Rapid
Incinerators

RAPID INCINERATOR
with digital display

can also reduce fluid substances to ashes

The operating temperature of 950°C is attained within the space of a few minutes. The upper level, with 8 recesses each with a diameter of 34 mm, can take porcelain crucibles of up to 50 mm diameter for drying, pre-heating and fuming-off. Can also reduce fluid substances to ashes. The uninhibited exposure to oxygen speeds up the incineration process. The tilted position of the crucibles in the lower incineration bed (which has a length of 400 mm) allows excellent observation and access to the contents during the operation. Modern design, easy to operate, reliable, space-saving and of very good value.

Equipment:
The upper part and the housing are made of high-grade, stainless steel. There is a built-in 120 minute time-switch fitted with a buzzer signal to indicate when the desired time has been reached, the switch is permanently in the ON position.
The Temperature Digital Instrument, which measures up to 999°C, is controlled by a NiCr-Ni. thermo-element. The pre-selected temperature digital display is shown with a constant ACTUAL temperature. The appliance is fitted with a safety switch that cuts of the power to the heater as soon as the service panel is opened. Power is restored automatically by closing the panel. The appliance is delivered ready-to-use with a 1,5 m cable and Schuko plug.