ELMAG™ - 600

Working Principle :

Electromagnetic Flow Meters are based on FARADAY'S LAW OF INDUCTION. A magnetic field is applied to the metering tube, which results in a potential difference proportional to the flow velocity perpendicular to the flux lines. The physical principle at work is electromagnetic induction. The meter features flanged construction and is available with choice of liner and electrode material. All meters consist of a sensor and a converter that may be mounted integral to the sensor or remotely either with a field mount kit.

These meters are also called as Magflow or, EM Flow Meters. These meters has no any moving parts in it, so these very well recommended for food grade applications like Juices, Milk, Municipal or Industrial Water etc. These meters are also suitable for liquids were there are lot of suspended particles like Sewage, Waste Water, Chemicals, Slurries etc.

Product Description :

Electronet series ELMAG™ - 600 are Battery Operated flow meters. It is ideal when connecting to a power is difficult or expensive. When no mains power is available, a battery powered ELMAG-600 gives the flexibility to install a reliable water flow meter virtually anywhere without distorting accuracy & performance.

ELMAG-600 is provided with 2 years continuous battery operation. It is extremely easy to use, simple to install into position & no regular maintenance required. The product is engineered for various industrial as well as domestic applications such as Munciple & Industrial water / Waste Water applications, Cooling tower water treatment, well usage reporting etc.

The graphical display provides indication for flow rate & totalised flow with key selection. The display will auto shut off after 10 min. Which saves the battery life. The display can be made ON by pressing the 'enter' key. The totalised value of a day is get stored in the data logger. Low battery indication is provided through software on display & flow meter will perform the expected results just replacing the battery.

Electronet Flowmeters are calibrated at our own Water Flow Calibration Facility. It uses the primary method of measuring the water flow by gravimetric principle. The facility is designed for calibration of flowmeters under controlled and stable flow conditions.