24 22 2 General Note BodyTel Europe GmbH markets MEDISCALE EICHFÄHIG under the name WeightTel Pro. Measuring accuracy WeightTel Pro measures with an accuracy of e=d=50/100g Compatibility and costs WeightTel Pro can only be used with compatible smartphones or the Home Gateway. A list of compatible devices can be found on For transmitting measured values to your patient file an internet connection via LAN, WLAN or mobile internet is required. Please refer to your mobile phone provider for related costs.

25 23 3 Introduction 3.1 General information on body weight The term body weight is used to describe the mass of a human body and is measured in kilograms throughout most countries of the world. There is no consent among health professionals as to what a person s ideal weight should be. The World Health Organization (WHO) considers a person with a body mass index (BMI) below 18.5 as underweight. Underweight may lead to an insufficient supply of important nutrients such as proteins, fats, carbohydrates, vitamins and minerals. Severely underweight individuals may have poor stamina and a weak immune system, making them prone to infectional diseases. The World Health Organization defines obesity as follows: pre-obese with a BMI of 25 to 29.9, obesity grade I with a BMI of 30 to 34.9, obesity grade II with a BMI of 35 to 39.9, obesity class III with a BMI of 40 or higher. A raised BMI is linked to an increased risk of cardiovascular diseases, diabetes, musculoskeletal disorders and some cancers. Regular physical activity and healthier food choices (e. g. a limited intake of fats and carbohydrates) can help reduce obesity. Regular weight measurement can help you control your weight and achieve a healthier lifestyle. If you are in care because of your weight share your online patient file with your physician or take a print-out of the file with you. This will give your physician a detailed overview of your weight changes to support your therapy. 3.2 Purpose WeightTel Pro is designed for weighing patients and intended for environments subject to calibration regulations. The product is also aimed on individuals requiring a high degree of measurement accuracy.

26 24 4 Using WeightTel Pro 4.1 Connect the power supply Before taking fi rst measurements with WeightTel Pro please connect the power supply to the back side of the scale. Fig. 1: Connect the power supply to the back side of the scale. After powering up the scale will show the calibration count for approx. 2 seconds. The calibration reading shows how often the scale has been calibrated. The value must match the calibration meter plate secured with an adhesive label. Fig. 2: The calibration meter shows how often the scale has been calibrated. Note: To immediately transmit measured values to your online patient fi le activate your smartphone s or Home Gateway s Bluetooth interface and start the BodyTel Mobile application. Please keep the phone close to the WeightTel Pro within a maximum range of 10 meters. Should you require assistance please contact the free support hotline: +800-BODYTEL ( ), Mo - Fry, 9am - 8pm.

27 25 4 Using WeightTel Pro 4.2 Set up your BodyTel user account To benefit from all the features of your new BodyTel telemedicine device please register for an user account at This will enable you to automatically transmit weight measures from your WeightTel Pro scale to your personal online patient file. Setting up is simple and free of charge. In order to register you will need: 1. A computer with an active internet connection 2. Your address Step by Step instructions: 1. Open your internet browser and enter the following URL into the address field: 2. Click on the Create account link, located in the upper right corner of the page. 3. In the following window enter your address into the corresponding field. Click on Register. 4. You will receive an confirmation to the specified address. This may take several minutes. 5. Open the sent from the address with the subject line -Bestätigungs-Anfrage von Attention! Please check whether your client has moved the confirmation to the spam or junk mail folder. Should you have not received a confirmation within 30 Minutes after registering please repeat steps 1 through 5. Please contact our free service hotline at +800-BODYTEL ( ), Mo - Fry, 9am - 8pm, if the problem persists. 6. Click on the link beginning with 7. A page will open in your browser reading Welcome to the first visit to your BodyTel center. Please read and accept the Terms of Service and the Privacy Policy by checking the appropriate boxes. Click on Next step. 8. Please fill out the required fields (sex, first name, last name) and choose a password. Click on Next step. 9. Fill out the remaining fields and click on Next step. 10. Choose your desired measurement units. The default settings should be: metric system, mg/dl, BE, 24 h, European. 11. Step 5 shows a summary of the data you have entered. Please check whether all details are correct and complete the registration procedure by clicking on Done. 12. Your data has been saved and you will reach the personal front page My BodyTel. This page is also displayed after you login at

28 26 4 Using WeightTel Pro 4.3 Installing BodyTel Mobile Activate Bluetooth on your mobile phone (smartphone) or Home Gateway and make sure visibility is set to visible to all. There are two different ways to download the BodyTel Mobile application: Option 1: 1. Open the internet browser on your mobile phone or Home Gateway and enter the following URL: 2. On the BodyTel download page please choose your language by clicking on the respective link. 3. Please read and accept BodyTel s Privacy Statement and the Business Terms by checking the boxes. Click on Send. 4. Please choose the version of the software compatible with your device. 5. The BodyTel Mobile software will now be downloaded. Please follow the instructions on your mobile phone or Home Gateway to install the software. Option 2: Download the software from the official Google Play Store. Please follow the instructions on your mobile phone or Home Gateway to install the software. For further information please visit the BodyTel knowledge base at: (Only german language)

29 27 4 Using WeightTel Pro 4.4 Pairing WeightTel Pro 1. Please start the BodyTel Mobile application on your mobile phone (smartphone) or Home Gateway. 2. Enter your BodyTel user account data. 3. When starting the software for the first time the application will ask you whether you would like to pair a device now. Please confirm with Yes. 4. Choose WeightTel Pro from the device list. Note: Your mobile phone or Home Gateway should be in close proximity to the WeightTel Pro scale because the scale will always attempt to pair with the device sending the strongest Bluetooth signal. Make sure that no other Bluetooth enabled devices (mobile phones, computers, etc.) are close to the WeightTel Pro while pairing. 5. Press the pairing key located at the bottom side of the scale s display (see picture) until the display shows the word COP. At the same time the Bluetooth status LED will start flashing. The scale will now try to pair with the device sending the strongest Bluetooth signal. The process is complete when the status LED is constantly on. Note: If you press the pairing key during the pairing process the process will be terminated. To start over please repeat step Check your mobile phone or Home Gateway and confirm all connection requests. 7. Enter the Bluetooth password On successful connection the status LED on the scale will flash twice and the display will show PAI.

30 28 4 Using WeightTel Pro 4.5 Positioning the WeightTel Pro scale Please note the following instructions regarding the correct positioning of the WeightTel Pro scale before you proceed with the first measurements: The scale has to be placed on a flat, firm (uncarpeted) and non-slippery surface (check Fig. 3). The scale cannot perform correct measurements if it is not completely level. Use the level indicator attached to the scale to align WeightTel Pro. 4.6 Weighing operation with WeightTel Pro Note: Please read the introductory remarks concerning body weight and reduction in Chapter 3 Introduction. Taring The function key on the bottom of the display resets the scale to zero. After pressing shortly the key the display shows 0.0 kg and the scale is ready to use. Fig. 4: The taring key (marked).

31 29 4 Using WeightTel Pro Weighing procedure Place both feet firmly on the weighing platform. Hold your position until a weight is shown in the display. If you leave the platform before the measurement is completed WeightTel Pro will automatically shut down. If you want to transmit measured data into your online patient file at please remain on the scale after the weight has been displayed. Please follow the instructions given in Chapter 4.8 Transmitting data. Should you not wish to transmit data leave the weighing platform. The scale will shut down automatically after approx. 15 seconds. 4.7 Transmitting data During transmission the measurements will be transmitted to your online patient file at If data is transmitted the blue Bluetooth status LED will start flashing. Upon completion, the LED will be activated constantly for 5 seconds. After that the scale will display 0,0 kg. In case of a transmission failure the LED will turn off and the scale will display an error code (see chapter 6 Troubleshooting ) for 10 seconds. Note: To transmit data the BodyTel Mobile software must be open and Bluetooth must be activated on your smartphone or Home Gateway.

33 31 5 Important instructions 5.1 Safety measures Please note: Before using the equipment for the first time please read the user manual carefully. It contains important information concerning the installation, correct use and maintenance of the equipment. The scale is not designed to be used in critical environments (CT, defibrillators). The scale must not be used in potentially explosive areas. Ensure that the scale can easily be disconnected from the mains (socket is accessible). The scale must not get wet from underneath. To move the scale lift the device. Do not drag the scale along the ground. After moving check leveling and adjust if necessary. The manufacturer shall not be held liable if the following instructions are not observed: The equipment is designed for use in buildings. The equipment complies with electromagnetic compatibility requirements. Do not exceed the maximum values specified in the standards. If you have any problems contact the BodyTel support. Only approved medical equipment may be connected to the interface. Warning: This device must not be changed without the manufacturer s permission. If the medical device is changed suitable tests and checks must be performed to ensure safe operation. Caution: The scale is subject to special precautionary measures regarding EMC and must be installed and put in operation according to the EMC instructions in the accompanying papers. Portable and mobile HF communications equipment can affect the scales if brought into close proximity.

34 32 6 Troubleshooting 6.1 Error Codes Error code Description Solution Err 5 The weighing process cannot Please check whether you have be completed. The error code applied to little (less than 1 kg) will be displayed as long as the or too much weight (more than malfunction persists. 250 kg) to the scale. Press the reset button with no load. Please make sure that the person on the weighing platform does not move. Please do not move the scale during the weighing procedure. Err 6 Taring cannot be performed. Please do not move the scale The error code will be displayed during the taring procedure. as long as the malfunction Press the reset button with no load. persists. Err 13 The converter is not working. Please contact your vendor. The error code will be displayed as long as the malfunction persists. Err 81 Error during Bluetooth Please check whether a transmission or pairing. After Bluetooth device is in reach. 5 seconds the scale will switch Start the BodyTel Mobile app back to the weight display. on your smartphone or Home Gateway and check whether Bluetooth connection is enabled. Retry the pairing procedure. Please check whether you have entered the correct Bluetooth password (9676) on your smartphone or Home Gateway. Your smartphone may be incompatible with WeightTel Pro.

35 33 7 Care and maintenance 7.1 Care The device may be cleaned with a damp cloth and regular cleaning products. Do not use any abrasive products. Disinfectants to be used: methylated spirit; isopropanol; 2% Kohrsolin; 1% aqueous Sokrena solution; 5% Sagrotan; 5% Gigasept. Do not spray directly on the device or the power adapter. 7.2 Interference In accordance with European directive 2004/108/EC this device is interference suppressed. However, the display reading can be affected by extreme electrostatic or electromagnetic interference, e.g. using a radio or mobile phone in the immediate vicinity of the equipment. Once the interference stops the product may revert to normal operation. It may be necessary to restart the scale. If the error persists contact your vendor or the BodyTel support. This device is a measuring instrument. Draft, vibrations, rapid temperature changes and direct exposure to sunlight may affect the reading. Avoid high air humidity levels, steam, abrasive fluids and contamination. Disposal of used electrical and electronic appliances According to our present knowledge the scale contains no environmentally harmful substances. This product is not to be treated as regular household waste, but should be handed in to an electrical/ electronic equipment recycling center. You can obtain further details from your local council, your municipal waste disposal company or the company from which you purchased the product. Maintenance A metrological check has been performed by the manufacturer. Re-calibration must be carried out in accordance with national regulations. Repairs must only be carried out by an authorized shop using original parts.

36 34 8 Warranty and repair 8.1 Warranty 1. BodyTel Europe GmbH grants a warranty of two years from date of sale. Proof of purchase must be supplied in the form of a correctly completed warranty card or a receipt. 2. Defects due to faulty material or workmanship will be repaired free of charge during the warranty period. 3. The warranty period is not extended as a result of warranty services being provided, either for the device as a whole or for any components. 4. The warranty excludes a. any damage caused by improper use, e.g. by failure to observe the operating instructions. b. damage caused by maintenance or interventions carried out by the purchaser or unauthorized third parties. c. damage resulting from transport, either from the manufacturer to the customer, or on the way to customer services. d. component parts subject to normal wear and tear. 5. Liability is excluded for direct or indirect damage caused by the device even if the damage to the device falls within the terms of the warranty. 6. Unauthorized modification of the warranty card invalidates it. 7. Further claims for whatever reason are excluded. BodyTel Europe GmbH does not warrant the batteries supplied. 8.2 Repair In the case of a warranted repair please complete the warranty card, giving precise details of the sender and the defect. Please contact our service department (see front cover) for shipping details.

39 37 10 EMV guidelines and Declaration of Conformity by Manufacturer The EMV guidelines and declaration of conformity for the MEDISCALE EICHFÄHIG scale (BodyTel product: WeightTel Pro) are provided by the manufacturer on request and are also available for download on the website.

Quick Guide Home Network Mode English > 1 German > 3 About the Home Network Mode EN Tivizen Nano & iplug normally work on their own created networks (whose SSID starts with tivizentv or iplug ) in which

p^db=`oj===pìééçêíáåñçêã~íáçå= Error: "Could not connect to the SQL Server Instance" or "Failed to open a connection to the database." When you attempt to launch ACT! by Sage or ACT by Sage Premium for

Version: 00; Status: E Seite: 1/6 This document is drawn to show the functions of the project portal developed by Ingenics AG. To use the portal enter the following URL in your Browser: https://projectportal.ingenics.de

p^db=`oj===pìééçêíáåñçêã~íáçå= How to Disable User Account Control (UAC) in Windows Vista You are attempting to install or uninstall ACT! when Windows does not allow you access to needed files or folders.

Newest Generation of the BS2 Corrosion/Warning and Measurement System BS2 System Description: BS2 CorroDec 2G is a cable and energyless system module range for detecting corrosion, humidity and prevailing

Ticketing Service by München Ticket https://www.muenchenticket.de/ 1 2 1 Enter here the name of the event Halls and Walls and press enter. 2 Click on TICKETS SICHERN to buy a ticket. How to get your tickets

CABLE TESTER Manual DN-14003 Note: Please read and learn safety instructions before use or maintain the equipment This cable tester can t test any electrified product. 9V reduplicated battery is used in

Mock Exam Behavioral Finance For the following 4 questions you have 60 minutes. You may receive up to 60 points, i.e. on average you should spend about 1 minute per point. Please note: You may use a pocket

CA_MESSAGES_ORS_HDTV_IRD_GUIDELINE 1/11 ORS NOTICE This document is property of Österreichische Rundfunksender GmbH & Co. KG, hereafter ORS, and may not be reproduced, modified and/or diffused in any way

Exercise (Part II) Notes: The exercise is based on Microsoft Dynamics CRM Online. For all screenshots: Copyright Microsoft Corporation. The sign ## is you personal number to be used in all exercises. All

Let me show you around 9 von 26 Where are we now? The administration building M 3 12 von 26 Let me show you around Presenting your company 2 I M 5 Prepositions of place and movement There are many prepositions

CA_MESSAGES_ORS_HDTV_IRD_GUIDELINE 1/8 ORS NOTICE This document is property of Österreichische Rundfunksender GmbH & Co. KG, hereafter ORS, and may not be reproduced, modified and/or diffused in any way

Filing system designer FileDirector Version 2.5 Novelties FileDirector offers an easy way to design the filing system in WinClient. The filing system provides an Explorer-like structure in WinClient. The

Benutzer- und Referenzhandbuch MobileTogether Client User & Reference Manual All rights reserved. No parts of this work may be reproduced in any form or by any means - graphic, electronic, or mechanical,