Dimensions

19″ rack version with mounting brackets and rubber fet

The DynamiQ-R has internal space for custom equipment such as pumps, electronics, filters and sensors.

Contains up to 4 GC ovens

The number and type of column ovens is targeted to the customer application.

Top Level Accurary500 ppb DL0.1% RSD0.01% CV

Designed for stability and robustness. Exact numbers depend on component and application.

Low consumptions

The DynamiQ-R can be configured for minimal carrier gas usage.

Typical 5 ml/minute carrier gas usage per channel during analysis

Typical 10 W power consumption per channel

Whole instrument weight < 20 kg

Easy maintenance

Each instrument is equipped with an easy replaceable cartridge, containing 1 to 4 GC channels with the analytical parts. There are no tubes or capillaries in the instruments which enhances the robustness.

Options

Sample Stream selector

3 inlet ports standard

optional integrated

Depending on the application, the DynamiQ can be equipped with a stream selector which allows to connect multiple sample gases and/or calibration gases. This integrated stream selector switches between the different inlets, so only 1 stream is let into the analyzer at a time. Note that in the case of applying a permanent calibration bottle to one of the inlets, for a standard 3-stream selector, only 2 sample gas stream inlets are possible.

Sample Pump

optional integrated

vacuum pump

In case of ambient pressure sample gas(es), 1 or 2 sample vacuum pumps can be integrated.

Multiple carrier Gases

1 standard

up to 4 possible

There is standard one carrier gas port, but the GC analyzer can be configured such that each channel has a different carrier gas.

Data / reporting

modbus over RS485 or TC/IP

RS232

The autonomous running instrument generates data according to a user defined format. This can be MODBUS, ASCII or customer specific over LAN, RS232 or RS485. Besides, result reports can be generated.