Search

Aeris™ Aeris Thermal Cycler

Aeris™ is the 3rd generation of thermal cyclers in the Esco PCR family over the last 5 years. It carries the proven reliability of Swift Maxi and MaxPro Thermal Cyclers. With the newly developed software functions, controlling the whole PCR protocol is now more user-friendly so that you can achieve optimized PCR result. Furthermore, the intuitive color touch screen gives you a new experience.

The Aeris™ Thermal Cycler offer 5 interchangeable blocks designed to meet critical requirements for different applications: it suits a variety of consumable PCR tubes, strips, plates, and slides. AeonStar™ peltier is qualified to deliver outstanding and precise performance and unique IsoHeat™ temperature control technology, which delivers high heating and cooling rates with excellent temperature accuracy and uniformity. SmartDrive™ automatic block recognition increases user convinience. Aerislink™ software enables the remote control of up to 30 individual units via one PC.

With the intuitive color touch screen, program setting becomes easier and faster with available functions to define different applications.

Self-testing provides more detailed information, like different sensors and heating and cooling status. This ensures reliable performance will be achieved before the actual run in 3 minutes.

Shortcuts can store regular programs, which can speed up the setting time.

Better Performance

Allows running of ‘Long PCR,’ a new technique that applies to amplification of PCR. It is also able to run ‘Nested PCR,’ which allows users to have two pairs of PCR primers to reduce the contamination in products due to the amplification of unexpected primer binding sites.

Temperature uniformity of ±0.1°C.

‘Pause’ function with timer can easily be set manually.

Suitable for many series of reagents and consumables, which can simplify experiments.

User-Friendly

Three language choices (English, Chinese and Spanish) on the control system for ease of use by operators worldwide.

Adjustable backlight for operator comfort, making it suitable for different working environments.

User log can store information for up to 20 experiments, which includes time and date, duration, program resuming time, power interruption, and program status. End users can make sure the test result is valid before doing the end point analysis.

Model Code

AERIS-MB

Block Code

AERIS-BG096

AERIS-B4830

AERIS-BG384

AERIS-BD048

AERIS-B4076

Sample Capacity

96 X 0.2 ml

48 x 0.2 ml
+ 30 x 0.5 ml

384 wells

48 x 0.2 ml
+ 48 x 0.2 ml

4 slides in situ

Application Consumables

0.2 ml tubes
96-well
microplates
12 x 8 strips
8 x 12 strips

0.2 ml tubes
0.5 ml tubes
4 x 12 strips

384-well microplates

0.2 ml tubes
6 x 8 strips

4 slides in situ

Max. Heating rate

4.0ºC/sec

2.8ºC/sec

2.8ºC/sec

4.0ºC/sec

1.8ºC/sec

Max. Cooling rate

4.0ºC/sec

2.8ºC/sec

2.8ºC/sec

4.0ºC/sec

1.8ºC/sec

Gradient Capability

Yes

-

Yes

-

-

Gradient Range

30-105°C

-

30-105°C

-

-

Maximum Gradient

1-30°Cmode

-

1-30°C

-

-

Temperature Control Mode

Tube or Block

Temperature Range

4-105ºC

Over-temperature Cut-Out

Yes

Number of Programs

Up to 250 programs, unlimited with USB flash drive

Maximum Hold Time

59 mins and 59 secs

Temperature Accuracy

≤±0.1ºC below 50ºC

Temperature Uniformity

≤±0.2ºC

PCR Volume Range

10-100 μl

Tm Calculator

Auto

Extensive Experiment Application

Option setting for time up/down is between 0-9m59s, which is suitable for long PCR Temperature when up/down is between 0.1ºC to 9.9ºC, it is suitable for Touchdown PCR

With the intuitive color touch screen, program setting becomes easier and faster with available functions to define different applications.

Self-testing provides more detailed information, like different sensors and heating and cooling status. This ensures reliable performance will be achieved before the actual run in 3 minutes.

Shortcuts can store regular programs, which can speed up the setting time.

Better Performance

Allows running of ‘Long PCR,’ a new technique that applies to amplification of PCR. It is also able to run ‘Nested PCR,’ which allows users to have two pairs of PCR primers to reduce the contamination in products due to the amplification of unexpected primer binding sites.

Temperature uniformity of ±0.1°C.

‘Pause’ function with timer can easily be set manually.

Suitable for many series of reagents and consumables, which can simplify experiments.

User-Friendly

Three language choices (English, Chinese and Spanish) on the control system for ease of use by operators worldwide.

Adjustable backlight for operator comfort, making it suitable for different working environments.

User log can store information for up to 20 experiments, which includes time and date, duration, program resuming time, power interruption, and program status. End users can make sure the test result is valid before doing the end point analysis.