Uncompromising weighing technology is the No. 1 priority. Incomparably fast response and display of weighing results and exceptional repeatability meet the very highest demands on accurate weighing. These results unconditionally measure up to the strictest requirements of both legal regulations and quality management systems.

Astonishingly fast response times with only 10-second stabilization time needed

Complete functionality for use as measuring and test equipment, with fully automatic calibration and adjustment function, isoCAL; the SQmin function for displaying the minimum allowable sample quantity according to the USP; and S.U.R.E. display of the uncertainty of measurement to comply with stringent accuracy standards, such as DKD (German Calibration Service)

14 built-in application programs as standard features, such as EUREKA program for air buoyancy correction, Differential weighing program for up to 999 samples, Statistical evaluation. All balance-generated data can be logged via the standard RS-232C interface port

Models ME5-F and SE2-F for weighing filters of up to 90 mm in diameter. The draft shield specifically constructed for this application is made completely of metal, thereby minimizing the interfering effects of static electricity

Flexible draft shield system with three independently motorized draft shield elements for fully automatic, intuitive operating at the press of a key.

Large, palm-activated keys with an intelligent learning function for user-friendly operation of the draft shield

Backlit, graphical display for consistently bright weight readouts.

Text prompts in plain English (and your choice of other languages) guide you quickly and confidently in configuring the balance.

The SQmin function displays the minimum allowable sample weight in accordance with the United States Pharmacopeia (can be activated by Sartorius Service)

S.U.R.E. function for dynamic display of the uncertainty of measurement and process accuracy in accordance with German Calibration Service (DKD) standards for continuous monitoring during all weighing routines (this function is activated by Sartorius Service)

14 built-in application programs as standard features, such as differential weighing program for up to 999 samples, statistical evaluation, calculation of weights using a definable factor or equation, air density determination.

This new ME36S microbalance features an exceptionally high weighing capacity of 31g and a readability of 1µg. The ME36S is designed for weighing the smallest sample quantities even in relatively heavy or bulky sample containers. The separate installation of weighing unit and display and/or electronic unit not only reduces thermal disturbances, but also delivers absolutely reliable weighing results. By virtue of its unique design, the ME36S display and weighing units are close enough for ideal operating convenience, with optimal spacing for ergonomic handling of microscopic sample quantities.

The YDP20-0CE data printer is designed to receive
weighing data from a connected weighing instrument
and print it on plain paper (roll width:
57.5 mm), with up to 24 characters per line.
The data printer is equipped with a real-time clock.
After the printer has been configured according to
the required settings, it is suitable for use as auxiliary
apparatus in legal metrology.

The following can be computed/printed out:

Number of weighing operations

Mean value

Standard deviation

Variation coefficient

Minimum value/weight

Maximum value/weight

Difference between the maximum and minimum value/weight

The YDP20-0CE data printer meets the quality standards
of the chemical and pharmaceutical industries, in
compliance with the requirements of GLP (Good Laboratory
Practice) and GMP (Good Manufacturing Practice).
In addition to recording individual weighing data, the
data printer can generate a statistical evaluation of
weighing data. All values transferred from an on-line
weighing instrument are stored. The following can be
printed out, depending on your printer configuration:
? Number of weighing operations
? Mean value
? Standard deviation
? Variation coefficient
? Lowest measured value
? Highest measured value
? Difference between lowest and highest measured values
? Date and time

Data is transferred either automatically or manually, at
the press of a key.