The instrument can measure double pressure (gauge pressure) at the same time, as well as dual temperature measurement (need test clip,not include), with automatic multi-unit pressure conversion, automatic conversion of temperature Celsius / Fahrenheit, to facilitate different needs. Built-in 89 kinds of refrigerant pressure-evaporation temperature database, also calculate the subcooling superheat, to facilitate direct reading of operating process data. Also it tests percentage of vacuum measurement; pressure leak measurement, leak time speed record. It is deserved to have this multi-functional, accurate and simply operated digital manifold let you do the job right.

EC/EN61010-1 safety standards for design and production.1) The pressure measured by the digtal manifold pressure tester is gauge pressure.2) Pressure testing ranges from -101Kpa to 6Mpa (-0.1bar to 60bar).3) The limit pressure is 10 Mpa (100 bar).4) The maximum operating pressure of standard hose is 600 PSI ( approximate 4.13 Mpa, 41.3 bar ). The limit pressure is 3000 PSI approximate 20.68 Mpa, 206.8 bar ).5) Please confirm the rated pressure value of the tested equipment before testing. Do not use it if it exceeds the range of the instrument. Also for DY517A, if the packedhoses does not match the pressure requirement, you can use suitable replacements for testing.6) Do not use and store the instrument in high temperature, high humidity, flammable, explosive and strong electromagnetic fields.7) Please do not change the instrument internal circuit, to avoid any damage of the instrument or danger occurring.8) Please wear qualified protective equipment to protect user during testing.9) Please use the instrument in a well-ventilated environment to prevent inhalation of toxic gases.