Whatever the task: material tests, ageing of computer chips, run complex experiments with highly sensitive loads, drying or tempering electronic components, hardening plastic resin or heating plasticine. Memmert Universal Ovens stand for unparalleled precise, even and gentle temperature control. With the SingleDisplay or TwinDisplay Generation 2012 technology that control has become easier than ever.

The Memmert Universal Ovens use corrosion-resistant and easily cleanable stainless steel for the working chamber and all but the rear of the housing.

In both the convection (UN) and fan ventilated (UF) appliances, supply air is heated in a pre-heating chamber, then through ventilation slits in the side panel of the working chamber the preheated air is introduced into the oven interior. The supply and exhaust air volume is controlled by an air flap on the rear panel of the oven.
From the wide selection of models, with both natural (UN) and forced air circulation (UF), and convenient, approved and tested options and accessories, it is possible to build a heating oven for every application. The two Generation 2012 model variants, with either SingleDisplay or TwinDisplay, cover all requirements for thermal safety, precision and quality assurance.

A wide range of options for programming and documentation using interfaces are available as part of the Memmert Generation 2012 class of products.
As much function as necessary, as much individuality as possible! Contact Rose Scientific to custom design the Memmert Universal Oven that does what it needs to do with the resources available.

The ordering number is assembled from the catalog numbers of the numerous options and accessories at the time the Incubator unit is ordered.

Memmert supplies a three-year guarantee for appliances, except for wear and tear parts

Chamber modification for the application of reinforced perforated stainless steel shelves
or stainless steel grids (bearing rails mounted in the working chamber), includes replacement of 2 standard grids by 2 reinforced grids, sizes 450 and 750 only

Inquire

R8

Fresh-air filter (filtration efficiency 80 %) mounted at the bottom, for UF or UFPlus, for model sizes 260 or smaller a subframe or subframe on castors is required

Interior nearly gastight with possibility for inlet/outlet through 2 tubes with ball valves

Inquire

FX

Entry port option, 23mm Ø, for introducing connections at the side, can be closed by flap, position F0 for left center center, F1 for left center top, F2 for right center center, F3 for right center top, F4 for special position left side, F5 for special position right side

Inquire

F6

Entry port option, 23mm Ø, for introducing connections at the back, can be closed by flap, special position back side

Inquire

D6

Entry port option, 14mm Ø, for introducing connections at the back, can be closed by flap, special position back side

Inquire

F7

Entry port option, 38mm Ø, for introducing connections at the back, can be closed by flap, special position back side

Inquire

F8

Entry port option, 57mm Ø, for introducing connections at the back, can be closed by flap, special position back side

Inquire

F9

Entry port option, 100mm Ø, for introducing connections at the back, can be closed by flap, special position back side

Inquire

D7

Entry port option, 120mm diameter, for introducing connections at the back, can be closed by flap, special position back side, for size 750 and smaller

Potential-free contact option (24 V/2 A) with socket to NAMUR NE 28 for signal generation, controlled by program segment, for a total of three selectable functions to be activated, including control of such elements as audible and visual signals, exhaust motors, fans or stirrers (only for units with TwinDisplay), maximum 2 contacts for 1–ph. units, Generation 2012

Inquire

H74

Potential-free contact option (24 V/2 A) with socket to NAMUR NE 28 for signal generation, controlled by program segment, for a total of three selectable functions to be activated, including control of such elements as audible and visual signals, exhaust motors, fans or stirrers (only for units with TwinDisplay), maximum 4 contacts for 3–ph. units, Generation 2012

Additional Pt100 temperature sensor option for positioning in chamber or load, for local temperature measurement (up to maximum of 3 additional sensors), measurements can be displayed, recorded in the integral ring store, documented with the AtmoCONTROL software or printed on an attached device, Generation 2012

Inquire

C3

MobileALERT option, notification by SMS in case of any error or alarm of the device, requires option H6 “floating contact for alarm”, Generation 2012

AtmoCONTROL Software, meets the requirements for the use of electronically stored data sets and electronic signatures as laid down in Regulation 21 CFR Part 11 of the US Food and Drug Administration (FDA), base license for the control of one unit (only for units with TwinDisplay), Generation 2012