Client Testimonials

Well done, KKInstruments!
We managed to monitor our production process effectively with your expertise! Will buy again!

Ahmad Azizul

IR Chief Engineerof Exxon Mobil

Our company prefer to source measuring instruments only from KKInstruments because they have a wider range of products and show a high standard of expertise. We are very happy with the deal. No regrets for choosing KKInstruments for the past 8 years,

The GSP-827 Spectrum Analyzer is a very handy instrument for RF measurement application, delivering a superior combination of easy-to-use design and exceptional price/performance value enclosed within a highly portable package. Covering 9kHz to 2.7GHz, the GSP-827 is designed to meet Production/Development test demands of RF products as well as on-site Maintenance/ Installation of wireless communication systems.

Portable and Easy-To-Use Spectrum Analyzer with Exceptional Price/Performance Covering a frequency range of 9kHz ~ 2.7GHz, the GSP-827 is designed to meet demands in production or development testing of RF products, maintenance or installation of on-site wireless communication systems. The exceptional price/performance, high portability, and easy-to-use design make GSP-827 a very accurate and convenient instrument for most RF measurement applications.

High Portability and Large Memory Size With the AC/DC/Battery power operation, a light 4.5kg weight, compact size, and coverage from 9kHz to 2.7GHz, the GSP-827 is a very convenient tool for field servicing RF systems. 100 sets of memory for saving 100 traces or setups make repetitive measurements and mass storage of test results possible. The large memory size of the GSP-827 is also a useful feature for characteristic verification of products and components. Repetitive measurement results can be saved internally, and then transferred to a PC for further analysis.

CH Power : By setting the channel center frequency and channel bandwidth, the GSP-827 automatically measures the total transmitted power within the bandwidth of the specified channel.

OCBW : By setting the channel center frequency and the total percentage of transmitted power, the GSP-827 automatically indicates the bandwidth that represents the percentage of the total amount of transmitted power.

ACPR :The ACPR measurement function allows users to check the interference or leakage power between the designated channel and its adjacent channels. Being able to apply to sophisticated modern communication systems, the GSP-827 has the capabilities to measure the power ratio between the specified channel and its adjacent channels either within the same communication system or among different communication systems.

10 MARKERS(PEAK TRACK, PEAK SEARCH)

The GSP-827 allows operating flexibility under both Delta-Marker mode and Multi-Marker mode. Up to 10 markers can be selected to show their absolute frequency and amplitude in a table on the screen. The relative frequency and amplitude of the selected markers-to-reference marker can be shown quite easily.

All Markers provide fast and comprehensive interpretations of measurements and full marking capabilities including, Peak Search, Marker to Center, and Peak Track.

100 TRACE/STATE MEMORIES

The GSP-827 provides up to 100 sets of memory to save or recall waveform and measurement set-ups. A specified file name and date/time stamp can be attached to each saved file for easy recognization of all saved information. With a large memory size, the GSP-827 facilitates measuring on site service and product characteristic verification.

LIMIT LINE

Upper and lower limit lines can be easily set to monitor abnormal signals under user defined conditions. Each Limit Line can be set by linking points positioned by frequency and amplitude data.

SPLIT-WINDOW

The Split-Window function allows you to view the frequency spectrum of an input signal on two separate screen areas under two different settings. The split window function provides the value of two spectrum analyzers in one instrument.

TRACKING GENERATOR

With the optional Tracking Generator, the GSP-827 becomes a convenient tool to verify characteristics of dual-port components such as filters and amplifiers. The normalization process of the output signal compensates the uneven signal loss caused by side effects from measurement cables and connectors to ensure an accurate measurement.

PC SOFTWARE

The GSP-827 is complimented with free PC software (EagleShot) to easily transfer acquired trace and the associated data from the GSP-827 to a PC through the RS-232C interface.

CARRYING CASE

With the carrying case, the GSP-827 makes on-site RF measurements more convenient than ever. The carrying case not only provides protection and portability for the GSP-827 but also carries test accessories such as RF cables, antennas, and a battery to facilitate measurements.