ZETLAB MASW

The multichannel analysis of surface waves is performed in two stages. At the first stage, measurements are performed with excitation in several points and saving the measured data to Seg-y files; at the second stage, the saved data are processed with the MASW method in the RadExPro software according to the manual //radexpro.ru/site/files/tutorials/RU_MASW_tutorial_rev120515.pdf.

When an excitation is caused by the impact hammer, the event is detected by ZET 7160 Logical Analyzer included in the mobile system; at that, the recorder registers the electric front of the digital signal and records this event in its internal log.

After the impact, ZETSENSOR mobile system is connected to a PC to transmit the recorded data for subsequent processing. Prior to the data uploading, files with the sampling rate over the recorded one (the data has been recorded with 200 Hz sampling rate, and those need to be re-sampled to 8 kHz rate, for instance) need conversion. Data conversion and transmission is performed using the ZETLAB software.

Transmitted data can be viewed using the historical data viewer (Figure 1).

Figure 1 “Historical data viewer” program

The event window displays the events corresponding to the surface wave excitation moments. Using zoom, mark a required fragment of the record (Figure 2).

Figure 2 “Historical data viewer” program

For subsequent processing, save the data in the Seg-Y format using File menu, then press Save… (Figure 3).