株式会社スリーエス | Safety, Speed, Service

3つのSで産業界に貢献する機械メーカー

AS200：Signal simulator

Designed for the easy calibration of I/P Positioners and 2-wire-type electrical transmitters.
Lightweight, portable and compact, allowing for a wide range of applications. At output currents other than 4～20mA, one machine can receive signals from a positioner’s output and 2-wire-type transmitter’s 4～20mA output.

◆Functions・Characteristic
◇SOURCE Mode
Emits a current signal of 4～20mA. The keypad allows seetings of 25％STEP or 0.01mA micro STEP settings.
Allows for selection of display of single units in mA and % terms. Can emit voltage signal of 1～5V (Optional)
◇RECEIVE」Mode
Self-powered (18V) specification enabling for easy receipt and display of positioner’s output and 2-wire-type transmitter’s output signal during recharging. Provision and wiring for DC power source and power measurement cuts down on cumbersome procedures.
◇READ Mode※1（Selection Mode）
Can measure and display 4～20mA current circuit.
◇「2WIRE」Mode※1（Selection Mode）
Available for loop testing of two-wire-system transmitters.
*1: Please select a either “READ Model” and “2WIRE Model” either.

◆LED light
LED light automatically on in response to ambient brightness.

◆Compact/Lightweight
Comes with a carry case to facilitate all round ease-of-use.

◆Heavy load response
Switch between up to 500Ω or up to 750Ω.

◆Power source
Dry-cell battery (9V1). Can also be operated using an AC Adaptor (sold separately).

*The rechargeable battery type went out of production at the end of May 2016.