How to Display Averaged Power Value During Integration on the WT1600

When the WT1600 is set into Integration mode, the averaged power (watt) values can be calculated and displayed. This is available only by using the User-Defined Function feature found in the MEASURE button menu.

The following equation must be entered in any one of the twenty Expression field:

WH(E1)/TI(E1)*3600

To display the calculated result, simply select and display the User Defined Function labeled F1 to F2 in the ITEM button menu.

The averaged power equation is pre-set on the WT500 and WT3000.

Related Products & Solutions

With 0.02% accuracy and 1MHz bandwidth, the WT3000 delivers where the highest precision measurements are required. It is the industry standard for R&D work on inverters, motor drives, lighting systems and electronic ballasts, UPS systems, aircraft power, transformer testing, and other power conversion devices.

The High-end WT1600 is designed for a wide range of applications, from energy-saving to large load applications. The WT1600 has a six element input capability to provide maximum flexibility, and Basic Power Accuracy of 0.1% with 1 MHz Bandwidth.

This site uses cookies to simplify and improve your usage and to provide you with the best experience of this website. Cookies are small text files stored on the device you are using to access this website. If you ignore this message and continue without changing your browser settings, we will assume that you are consenting to our use of cookies. For further information on our use of cookies, please see our Cookie Policy.