Abstract

Major advances characterize the Generation-V dual-Purkinje-image (DPI) eyetracker compared with the Generation-III version previously described. These advances include a large reduction in size, major improvements in frequency response and noise level, automatic alignment to a subject, and automatic adjustment for different separation between the visual and optic axes, which can vary considerably from subject to subject. In a number of applications described in the paper, the eyetracker is coupled with other highly specialized optical devices. These applications include accurately stabilizing an image on a subject's retina; accurately simulating a visually dead retinal region (i.e., a scotoma) of arbitrary shape, size, and position; and, for clinical purposes, stabilizing the position of a laser coagulator beam on a patient's retina so that the point of contact is unaffected by the patient's own eye movements.

References

You do not have subscription access to this journal. Citation lists with outbound citation links are available to subscribers only. You may subscribe either as an OSA member, or as an authorized user of your institution.