Monnit Wi-Fi Low Temperature Sensor

Also Available As:

Information

Price

$274.00

Format

Frequency

2.4 GHz

Part#

MNS-2-WF-TS-LT-L03-HG

Temperature Profile

Probe Length

Antenna

Monnit wireless low temperature sensors use an RTD to accurately measure temperatures from -200°C to +162°C. Perfect for low temperature critical applications such as freezers and coolers. User customization allows you to set the frequency of sensor readings and create notifications from the system when certain criteria have been met or exceeded.

Product Description

Monnit wireless low temperature sensors use an RTD to accurately measure temperatures from -200°C to +162°C. Perfect for low temperature critical applications such as freezers and coolers. User customization allows you to set the frequency of sensor readings and create notifications from the system when certain criteria have been met or exceeded.

In order to program the sensor to connect to your wif-fi network, you will need a Wi-Fi programming cable and the free MoWi setup utility. The MoWi setup utility is currently available for Windows® PC's only. Mac and Linux are not currently supported!

What does this wireless sensor's data look like in software?

What's in the box?

Wi-Fi Sensor

AA Batteries

SMA - Wi-Fi Antenna

Low Temp - RTD / Thermistor Specifications

Sensing Element

DIN 43760 Class B

Dissipation Constant @ 25°C

2mW/°C

Thermal Time Constant

15 sec max.

RTD Temperature Range

-200°C to +162°C (-328°F to +325°F)

RTD Accuracy at 0°C

+/- 3.3°C Standard (+/- 0.5°C Calibrated)

Wi-Fi Specifications

Battery Type

2 Replaceable AA 1.5 V Batteries

Max Operating Temperature Using AA Alkaline

-18°C to +55°C (0°F to +130°F)

Max Operating Temperature Using AA Lithium

-40°C to +85°C (-40°F to +185°F)

Battery Life

Up to 5 years depending on sensor type, Wi-Fi security, distance from Wi-Fi router, reporting frequency and other variables. (Testing surpassed 90,000 transmissions until battery depletion.)

Wi-Fi Security Standards

Open, WEP, WPA, WPA2 Security Standards

Wi-Fi Security Programming

Via USB cable (sold separately) and downloadable PC software.

Network Settings

Auto DHCP/DNS or Static

Data Logging (iMonnit Basic)

On Wi-Fi disruption, unit will log the most recent 50 readings and transmit when Wi-Fi connection is re-established.

Data Logging (iMonnit Premiere)

On Wi-Fi disruption, unit can log up to 50,000 readings and transmit when Wi-Fi connection is re-established.