It was reported that there was physical damage to the physician¿s programming tablet that occurred on (b)(6) 2014.
It was reported that the usb cable for the programming wand would not connect to the tablet usb port due to the port being bent.
It was reported that no patient's were affected because another programming system was available.
It was reported that the tablet was used the day prior without any difficulty and placed back in locked storage after use.
It is unknown if any mishandling occurred.
A new programming tablet was provided to the physician.
The tablet was received for analysis.
Analysis is underway, but has not been completed to date.

Event Description

Analysis of the programming tablet was completed on (b)(6) 2014.
During the analysis, it was identified that the usb port was damaged.
As a result, the tablet was unable to establish communication and generating port error messages.
No further anomalies were identified.

Manufacturer Narrative

Device failure occurred, but did not cause or contribute to a death or serious injury.

Manufacturer Narrative

Device manufacturing records were reviewed.
Review of manufacturing records of the programming tablet confirmed all quality tests were passed prior to distribution.