Product description

Thermofocus® is the first non contact (at-distance) clinical forehead thermometer in the world, result of a series of international patents.
Thermofocus is the no touch thermometer, designed to allow you to take your baby’s temperature without even touching him/her. With the Thermofocus thermometer, taking your baby’s temperature becomes a sweet gesture, just like a caress. It is enough to get Thermofocus close to your baby’s forehead to see his/her body temperature in a few seconds.
Thermofocus is a thermometer perfect for children and functional for adults, because the temperature measurement is quick, precise, hygienic, safe and can be done with your complete comfort.

Body temperature is measured by pointing the THERMOFOCUS® thermometer at the centre of the forehead. The right distance between the forehead and the thermometer for a correct measurement is easily determined thanks to a LED system emitting two light beams (normal red light which is totally safe). As the thermometer is moved closer to the forehead, at the right distance (approx. cm 3) the two beams converge to form a single red spot on the skin of the forehead. Release the button and keep still the thermometer for less than one second until the lights flash and the temperature appears in the LCD display of the thermometer. There is no need to concern if the aiming lights are pointed into the eyes: the beams are absolutely harmless and are not laser lights.

In case the subject, (i.e. a child) is moving, crying or has a perspiring forehead, it is possible to take the temperature aiming the thermometer on the navel or the neck, just below the ear. In case of old people with thick wrinkles on the forehead, a low supplying of blood may provoke a low temperature. In this case as well it is possible to proceed aiming the thermometer on the neck or on the navel (not recommended in case of not shaved areas).

«The use of Thermofocus® for the body temperature’s measurement is certainly positive, above all for its reliability, for the precision of the obtained values and also because this instrument is easy and functional to use, that is the reason why it has immediately “conquered” a stable place in my gown […] The child is convinced to be playing a game and not to be doing of a displeasing medical examination. Thermofocus’ characteristic of “familiarity” certainly makes the temperature data collection easier, more comfortable and more precise also in children […]»

Thermofocus and VisioFocus no touch thermometers are helping health authorities screen people coming from areas at risk for Ebola virus as well as Mers

Thanks to their exclusive aiming system Thermofocus and VisioFocus are the only no touch thermometers able to indicate clearly the correct distance from the skin and the correct point on which the temperature has to be taken. This is very important if you want to use against Ebola (as well as Mers) a non contact thermometer, able to detect on travellers possible fever due to Ebola Virus (as well as Mers).
Obviously, the presence of fever does not automatically mean that the subject has Ebola (as well as Mers), as fever may be caused by other factors. In the same way, the absence of fever does not automatically mean that subjects do not have Ebola (as well as Mers), for they may not show symptoms yet.
In any case, a thermometer must be able to detect the presence of fever on people. For this reason, it is important to use a thermometer really able to detect the presence of fever possibly caused by Ebola (as well as Mers).
Some no touch thermometers cannot find fever, unless it is very high.
Indeed, some of the non contact thermometers used against Ebola (as well as Mers) are laser industrial thermometers, which may be very dangerous if accidentally aimed at the eyes and cannot be adjusted to the room temperature as clinical thermometers should do. Accordingly, industrial thermometers used on the forehead may show a temperature from 1 to 3 °C (2 to 5°F) lower than the one measured in other conventional body sites (Oral, Axilla, Rectal, Core) according the environment situation. Therefore, these non contact thermometers may easily fail to detect fever due to Ebola virus (as well as Mers) or other reasons.
Other non contact thermometers are not provided with the algorithm necessary to obtain the actual body temperature from the surface temperature on the, so that they may fail to detect a passenger with fever caused by Ebola (as well as Mers), or they may detect the presence of fever on passengers having neither Ebola (as well as Mers) nor any other disease.
Finally, other no touch thermometers are not able to stabilize their temperature quickly when moved between environments with different temperatures, or are not able to keep a constant and correct temperature when handled for a long time (for example when many measurements are taken). Again, these faults may prevent a non contact thermometer from detecting fever due to Ebola (as well as Mers), or may cause them to detect fever on people having neither fever nor Ebola (as well as Mers).
Moreover, none of no touch thermometers used against Ebola (as well as Mers), except Thermofocus andVisioFocus, is provided with a system to indicate the correct point and the correct distance at which the temperature has to be taken, and, if this distance is not correct, the thermometer may fail to detect fever to a passenger with Ebola (as well as Mers), thus conveying to operators a false sense of security.

Thermofocus and VisioFocus use a special algorithm to process the surface temperature on the forehead in order to provide a body temperature comparable to the conventional ones (Axilla, Oral, Rectal Core). This algorithm was designed many years ago, was tested on many patients and by many engineers in many hospitals all over the world, and it is continuously verified and updated, especially after having been used in many international airports and hospitals during SARS in 2003 and Swine Flu in 2009.
Furthermore, Thermofocus and VisioFocus are the only thermometers provided with Automatic Quick Calibration System (AQCS) and Manual Quick Calibration System (MQCS) which keep the reference temperature of the device always constant and correct, no matter if the device is moving between rooms with different temperatures or if it is handled for a long time to take many measurements.

against SARS

Thermofocus® in asian airports

Thermofocus® in China

THERMOFOCUS® SUPPORTS CONTROL MEASURES AGAINST A/H1N1 FLU (swine flu)

Tecnimed takes part to the international cooperation network helping to face the spreading of A/H1N1 Flu (swine flu), by presenting its innovative Thermofocus®, the first and more advanced non-contact clinical thermometer in the world, the only one which guarantees absolute hygiene and an accurate, instantaneous, safe and repeatable temperature measurement, thanks to its exclusive patented optical aiming system, which allows to keep the thermometer always at the correct distance from the patient’s forehead.

In the case of A/H1N1 Flu, whose principal symptom is high fever, the body temperature measurement represents an extremely important action. That’s why the ‘new generation’ version of Thermofocus® thermometer, the 0800H series, is perfect to be used in airports, hospitals, schools, factories and all other at-risk situations and/or in all circumstances where it becomes necessary to make a mass, but accurate, check.

In fact, thanks to its exclusive at-distance measurement system, Thermofocus® thermometer always guarantees absolute hygiene, and reduces the risk of infection during the temperature monitoring, both for the patients and for the operator who is in charge of the measurements. A risk which becomes extremely high when the temperature monitoring is made at the same time/place on many people using traditional skin-contact thermometers (ear thermometer, digital, forehead, etc.).

In 2003, when the SARS pandemic spread out, Thermofocus® has been the most required thermometer by the sanitary authorities of many different Countries all over the world. Thermofocus® appeared to be the most hygienic, safe and accurate thermometer to detect body temperature in such at-risk situations. The exclusive MQCS system (Manual Quick Calibration System), developed after the SARS experience, makes the use of Thermofocus even more appropriate in case of mass controls and guarantees in every kind of situation accurate measurements and extreme safety.

Thermofocus® 01500A/H1N1 is the forehead non contact thermometer specifically studied for being used in hospitals, airports, schools, factories, etc. in case of emergency and/or pandemic situations. This thermometer is particularly quick, accurate and easy to use, allowing a fast and hygienic temperature reading for thousands of consecutive measurements. 01500 A/H1N1 is provided with the MQCS technology (Manual Quick Calibration System), thorough which the thermometer is always perfectly stabilized to the ambient temperature and ready to take series of measurements. Thermofocus 01500A/H1N1 comes with a special lanyard with additional brief instructions, which allows to safely carry the device in order to use it anytime it’s needed.

Thermofocus® is the first non contact (at-distance) clinical forehead thermometer in the world, result of a series of international patents.
Thermofocus is the no touch thermometer, designed to allow you to take your baby’s temperature without even touching him/her. With the Thermofocus thermometer, taking your baby’s temperature becomes a sweet gesture, just like a caress. It is enough to get Thermofocus close to your baby’s forehead to see his/her body temperature in a few seconds.
Thermofocus is a thermometer perfect for children and functional for adults, because the temperature measurement is quick, precise, hygienic, safe and can be done with your complete comfort.

Body temperature is measured by pointing the THERMOFOCUS® thermometer at the centre of the forehead. The right distance between the forehead and the thermometer for a correct measurement is easily determined thanks to a LED system emitting two light beams (normal red light which is totally safe). As the thermometer is moved closer to the forehead, at the right distance (approx. cm 3) the two beams converge to form a single red spot on the skin of the forehead. Release the button and keep still the thermometer for less than one second until the lights flash and the temperature appears in the LCD display of the thermometer. There is no need to concern if the aiming lights are pointed into the eyes: the beams are absolutely harmless and are not laser lights.

In case the subject, (i.e. a child) is moving, crying or has a perspiring forehead, it is possible to take the temperature aiming the thermometer on the navel or the neck, just below the ear. In case of old people with thick wrinkles on the forehead, a low supplying of blood may provoke a low temperature. In this case as well it is possible to proceed aiming the thermometer on the neck or on the navel (not recommended in case of not shaved areas).

«The use of Thermofocus® for the body temperature’s measurement is certainly positive, above all for its reliability, for the precision of the obtained values and also because this instrument is easy and functional to use, that is the reason why it has immediately “conquered” a stable place in my gown […] The child is convinced to be playing a game and not to be doing of a displeasing medical examination. Thermofocus’ characteristic of “familiarity” certainly makes the temperature data collection easier, more comfortable and more precise also in children […]»

Thermofocus and VisioFocus no touch thermometers are helping health authorities screen people coming from areas at risk for Ebola virus as well as Mers

Thanks to their exclusive aiming system Thermofocus and VisioFocus are the only no touch thermometers able to indicate clearly the correct distance from the skin and the correct point on which the temperature has to be taken. This is very important if you want to use against Ebola (as well as Mers) a non contact thermometer, able to detect on travellers possible fever due to Ebola Virus (as well as Mers).
Obviously, the presence of fever does not automatically mean that the subject has Ebola (as well as Mers), as fever may be caused by other factors. In the same way, the absence of fever does not automatically mean that subjects do not have Ebola (as well as Mers), for they may not show symptoms yet.
In any case, a thermometer must be able to detect the presence of fever on people. For this reason, it is important to use a thermometer really able to detect the presence of fever possibly caused by Ebola (as well as Mers).
Some no touch thermometers cannot find fever, unless it is very high.
Indeed, some of the non contact thermometers used against Ebola (as well as Mers) are laser industrial thermometers, which may be very dangerous if accidentally aimed at the eyes and cannot be adjusted to the room temperature as clinical thermometers should do. Accordingly, industrial thermometers used on the forehead may show a temperature from 1 to 3 °C (2 to 5°F) lower than the one measured in other conventional body sites (Oral, Axilla, Rectal, Core) according the environment situation. Therefore, these non contact thermometers may easily fail to detect fever due to Ebola virus (as well as Mers) or other reasons.
Other non contact thermometers are not provided with the algorithm necessary to obtain the actual body temperature from the surface temperature on the, so that they may fail to detect a passenger with fever caused by Ebola (as well as Mers), or they may detect the presence of fever on passengers having neither Ebola (as well as Mers) nor any other disease.
Finally, other no touch thermometers are not able to stabilize their temperature quickly when moved between environments with different temperatures, or are not able to keep a constant and correct temperature when handled for a long time (for example when many measurements are taken). Again, these faults may prevent a non contact thermometer from detecting fever due to Ebola (as well as Mers), or may cause them to detect fever on people having neither fever nor Ebola (as well as Mers).
Moreover, none of no touch thermometers used against Ebola (as well as Mers), except Thermofocus andVisioFocus, is provided with a system to indicate the correct point and the correct distance at which the temperature has to be taken, and, if this distance is not correct, the thermometer may fail to detect fever to a passenger with Ebola (as well as Mers), thus conveying to operators a false sense of security.

Thermofocus and VisioFocus use a special algorithm to process the surface temperature on the forehead in order to provide a body temperature comparable to the conventional ones (Axilla, Oral, Rectal Core). This algorithm was designed many years ago, was tested on many patients and by many engineers in many hospitals all over the world, and it is continuously verified and updated, especially after having been used in many international airports and hospitals during SARS in 2003 and Swine Flu in 2009.
Furthermore, Thermofocus and VisioFocus are the only thermometers provided with Automatic Quick Calibration System (AQCS) and Manual Quick Calibration System (MQCS) which keep the reference temperature of the device always constant and correct, no matter if the device is moving between rooms with different temperatures or if it is handled for a long time to take many measurements.

against SARS

Thermofocus® in asian airports

Thermofocus® in China

THERMOFOCUS® SUPPORTS CONTROL MEASURES AGAINST A/H1N1 FLU (swine flu)

Tecnimed takes part to the international cooperation network helping to face the spreading of A/H1N1 Flu (swine flu), by presenting its innovative Thermofocus®, the first and more advanced non-contact clinical thermometer in the world, the only one which guarantees absolute hygiene and an accurate, instantaneous, safe and repeatable temperature measurement, thanks to its exclusive patented optical aiming system, which allows to keep the thermometer always at the correct distance from the patient’s forehead.

In the case of A/H1N1 Flu, whose principal symptom is high fever, the body temperature measurement represents an extremely important action. That’s why the ‘new generation’ version of Thermofocus® thermometer, the 0800H series, is perfect to be used in airports, hospitals, schools, factories and all other at-risk situations and/or in all circumstances where it becomes necessary to make a mass, but accurate, check.

In fact, thanks to its exclusive at-distance measurement system, Thermofocus® thermometer always guarantees absolute hygiene, and reduces the risk of infection during the temperature monitoring, both for the patients and for the operator who is in charge of the measurements. A risk which becomes extremely high when the temperature monitoring is made at the same time/place on many people using traditional skin-contact thermometers (ear thermometer, digital, forehead, etc.).

In 2003, when the SARS pandemic spread out, Thermofocus® has been the most required thermometer by the sanitary authorities of many different Countries all over the world. Thermofocus® appeared to be the most hygienic, safe and accurate thermometer to detect body temperature in such at-risk situations. The exclusive MQCS system (Manual Quick Calibration System), developed after the SARS experience, makes the use of Thermofocus even more appropriate in case of mass controls and guarantees in every kind of situation accurate measurements and extreme safety.

Thermofocus® 01500A/H1N1 is the forehead non contact thermometer specifically studied for being used in hospitals, airports, schools, factories, etc. in case of emergency and/or pandemic situations. This thermometer is particularly quick, accurate and easy to use, allowing a fast and hygienic temperature reading for thousands of consecutive measurements. 01500 A/H1N1 is provided with the MQCS technology (Manual Quick Calibration System), thorough which the thermometer is always perfectly stabilized to the ambient temperature and ready to take series of measurements. Thermofocus 01500A/H1N1 comes with a special lanyard with additional brief instructions, which allows to safely carry the device in order to use it anytime it’s needed.

• Get the thermometer close to the forehead, you will notice two points, out of focus if the thermometer is too far; on the contrary they will be definite if it is too close. When Thermofocus is at the correct distance (about 3 cm from the forehead), the two lights will converge in only one luminous point.

• Release the button keeping the thermometer still until the lights flash.

• Read the value visualized on the display.

• If you wish, you can immediately take other measurement.

• Put the protective cap back on, rotating it lightly.

Do not be afraid if the tracking lights are unintentionally directed to the eyes: they are totally harmless!

It is possible to measure the temperature of any object that has a temperature between 1 and 55°C acting as it follows:

Mod. 01500A3 – use the button “house” (the value thus obtained corresponds to the object’s temperature).

Mod. 0700A2 – use the button “on” and subtract the value indicated on the tag; (the value thus obtained corresponds approximately to the object’s temperature, but the precision is lower if compared to the one of mod. 01500A3).

Congratulations! You have just bought the most modern and safest device to measure body temperature.

Thermofocus is convenient and easy to use. Indeed, it is the first “non-contact” medical thermometer. The device enables the temperature of adults and children to be taken without touching the skin, simply by moving the thermometer close to the forehead at the distance indicated by the device. With Thermofocus there is no need to place the thermometer in any part of your baby’s body. A sleeping baby will not be woken when
Thermofocus is used nor will it be caused discomfort by the device if awake.

WARNINGSPlease read all the instructions carefullybefore using the thermometer

Thermofocus is exceptionally easy to use to take a body temperature from a short distance. However, to avoid incorrect measurements, use Thermofocus in the same circumstances as conventional thermometers (fig. 1).

Use Thermofocus in a draft-free room at a constant temperature between 60.8°F (16°C) and 104°F (40°C).

If the thermometer is taken from a room (or drawer or cabinet) with a temperature different from that in which it is being used, wait for 10/30 minutes for the device to adjust to the room temperature before using it or make the calibration (MQCS) as explained at pag. 8.

Do not use the thermometer on a perspiring forehead, as this may affect the reading (fig. 2). Read the appropriate chapter (page 5).

Avoid taking the subject’s temperature in a drafty location and wait for several minutes before taking a measurement if the subject:

has exerted him/herself;

has been in a drafty room or a place where the temperature is significantly different from that in which the thermometer is being used (fig. 3);

has recently had a cold cloth or cold compress on the forehead;

has recently been wearing a hat or a scarf;

has been exposed to any activity which may affect the normal temperature of the head such as bathing, showering, shampooing, hair drying, etc.; also, holding a hand to the forehead for any length of time will affect the temperature reading.

In the cases outlined above, wait a few minutes for the temperature on the forehead to stabilize. Alternatively, in the case of babies, the temperature may be taken from the navel or the armpit or on the neck (please read the appropriate chapter – page 5).

The area from which the temperature is taken has a radius of about 3/4 inch (2 cm) around the point of light. It is very important to ensure that this area is not obstructed by hair or clothing and does not include eyes or eyebrows. If you change the point of measurement on the forehead you will get different results. For this reason, it is important that the two aiming lights are directed towards the middle of the forehead (between the nose and the hairline) and that the thermometer is held perpendicular to the forehead, taking care to hold aside any hair to clear an area of at least 3/4 inch (2 cm) around the light point.

Oils or cosmetics on the forehead may give a lower temperature reading than the actual one, while hair obstructing the forehead may cause a slightly higher reading.

The gilt guide wave (see fig. 8) of the thermometer is the most delicate part of the device. It consists of a small concave gold-plated mirror, which must be kept clean and undamaged. The accuracy of the reading may be affected if the guide wave is damaged or dirty.

Avoid handling the thermometer, particularly the tip, more than necessary before taking a reading.

Do not use the thermometer in contact with the ear or any other surface area of the body.

Do not use the device near large electromagnetic fields such as found on a person that is starting a calling with a cordless or cell phones.

Keep the device away from water and heat, including direct sunlight. Do not drop or knock the device, and do not use if damaged.

Failing to follow these guidelines may result in excessively high or low readings which are in no way attributable to a faulty device.

No harm can be caused should the aiming lights be accidentally pointed in the eyes: the beams are absolutely harmless!

The device is a sensitive measuring instrument and should not be used by children or as a toy.

Abrupt movement may not allow for enough data to give an accurate reading. Approaching a baby with the aiming lights already on may help, giving you enough time for a steady reading before the baby moves.

If you need to take your own temperature, you can either use a mirror, or easily show a child of six years or older to use Thermofocus.

Thermofocus never touches the body so there is no need for expensive disposable lens covers.

Thermofocus receives infrared emissions from the body, and in particular from the forehead. The forehead is the ideal part of the body from which to take a temperature because it is supplied by the temporal artery, which receives blood through the aorta and the carotid artery, guaranteeing a considerable flow of blood. Moreover, the forehead is the only part of the body close to the brain, which is not covered in hair.
The brain is the most important and delicate organ in the body, and the one most liable to damage from excessive temperature. Furthermore, the head is the first part of the body to modify its temperature in the case of a fever, whether rising or falling.

At every measurement, Thermofocus carries out every hundredth of a second, a sequence of 125 readings, which are amplified and computed by its sophisticated microprocessor, together with the value of the ambient room temperature, until the correct temperature is shown on the display.

It is important to be aware that no single “normal” temperature applies to everyone: each person has his/her own temperature, which changes not only during the day, but also according to certain physical activities undertaken by the subject (for example, if a baby has been crying). Moreover, an individual’s temperature may be different in different regions of the body, and may be influenced by the ambient temperature and other external factors, depending on how the temperature is being taken. Until now, all methods of measuring temperature involved a certain inconvenience to the subject, which can be stressful to a mother seeking to know her baby’s temperature but concerned about hurting or causing distress.
Because of the heat loss from parts of the body uncovered by clothing, the normal temperature measured on a person’s forehead is usually lower than that of covered parts of the body.
Therefore, when the “on” (0700 series) or “face” (01500 series) buttons are used, Thermofocus’ internal software automatically adjusts the temperature to provide a reading, which correlates approximately to the oral temperature. If you prefer you can change the setting from the oral temperature to the rectal. Note: when the device is on the Rectal setting the temperature displayed will be about 1°F (0,6°C) higher than the Oral setting. Please refer to the appropriate chapter (page 8).
In some version the device is showing the “core”temperature and you cannot choose neither Oral nor Rectal.

The normal temperature taken from the forehead of a healthy person, using Thermofocus, can vary between 95°F (35°C) and 100.4°F (37.5°C), but in an adult, it can also be below 95°F (35°C).

It is useful to know the normal temperature of each member of the family when in good health and at various times of day, in order to be able to detect a potential fever.

Fill in the following, simple chart and use it for reference. If you decide to consult a doctor, inform him/her of the type of measurement you have taken and the normal temperature under healthy conditions.

The thermometer will switch on and the last reading taken with the same button and the word “READY” will be shown, indicating that the thermometer is ready to take a measurement. After 2 seconds the display will show the room temperature, but the device is still ready to take a measurement.

Open the protective cap.

Press the same button again and hold it down: the two aiming lights will come on and the display will start to show the temperature, together with the symbol “face” (with the mod. 0700 this symbol is not shown).

Hold the thermometer perpendicular to the middle of the forehead and move Thermofocus closer to or further from the forehead until a single point of light appears. If the thermometer is too far away from the forehead, you will see two blurred points of light (fig. 4). If the thermometer is too close, you will see two separate points of light. When you see a single point of light on the subject’s forehead, the thermometer is at the correct distance for an accurate temperature reading to be taken (fig. 5).

Release the button and holding the device steady until the lights flash.

Read the value shown on the display. If necessary, you may take another reading immediately.

Close the protective cap.

After about 20 seconds of non-use the models 0700 switches off automatically, while the other models go on stand-by after 20 seconds and indicates the room temperature for 4 hours before turning itself off. If you want to make models other than the 0700 turn completely off, please read the appropriate chapter (page 8).

Sweating: taking the temperature from the neck

When the fever is going down the forehead may perspire making its temperature too low. In this case it’s possible to take the temperature from a distance at the neck about 1 inch (2,5 cm) below the outer ear (auricle) for adults and 0,4 inches (1 cm) for babies (fig. 6). Proceed as directed for forehead readings (use the “on” button for 0700 models or the “face” button for other models). The accuracy cannot be guaranteed, but it can be considered a reasonable approximation of body temperature and an indicator of fever. This measurement is particularly indicated for adults also where the forehead reading is affected by oil or cosmetics.

Taking the temperature from the navel or from the axilla (this feature is not available in the 0700 models)

If a baby is restless, crying or has a perspiring forehead, or if it proves difficult to take a temperature from the forehead, the reading can be taken from the belly, just on the navel, or from the axilla. It is very important to carry this out while the baby is normally dressed. Simply lift his/her clothes immediately before taking the temperature, in order to avoid heat loss from the belly. Measurement is not as accurate as the forehead measurement and should only be used as an alternative when a forehead measurement is not possible.

WARNING: taking the temperature from the navel or the axilla is recommended only for children or for adults when there is no hair to interfere with the reading.
Proceed as for a measurement on the forehead but use the “home” button and lift up, with your free hand, the baby’s clothes so as to have an area of at least 1-1/4” (3 cm) around the navel; when measuring on the navel, make sure the aiming lights converge directly on the navel, or a bit above it: the value which appears can be compared to the underarm temperature.
However, certain factors may affect this reading, such as the clothes being worn, the shape of the navel, digestive activity (length of time before or after a meal), or the thickness of the adipose tissue.
Therefore it is recommended that readings be taken when the baby is well and at different times during the day, noting the results on the apposite table and taking them as a reference when future readings are taken. Deviations from these reference values indicate a fever. The greater the deviation, the higher the fever. Remember to take the reading under the same conditions as the reference temperature: at the same time of day, at the same time before or after a meal, at the same ambient temperature.

Other measurements

Thermofocus can also be used to take other temperatures, using the same method as for the navel (use the “home” button). For the 0700 models, use the “on” button and subtract the value written on the silver sticker from the value shown on the display. For example:
1) the temperature of rooms, by pointing the device at a wall or piece of furniture;
2) the temperature of wounds, inflammations, scars (for medical professionals);
3) a baby’s bottle temperature (fig. 7) as well as baby food or bathwater temperature. In the last two cases, take the temperature quickly so that condensation does not form on the guide wave and on the sensor, and wait 30 minutes before taking any further readings.

MEMORY FUNCTION

In some models, there is a memory function, which allows the values of the last 9 readings to be recalled. To activate the memory when the thermometer is on, press the “Mem” button; the display will show the value of the most recent measurement followed by the number 1 and the RE (record) signal.
Press the button repeatedly and the display will show the second and third readings, together with the number 2, 3, etc, back from the most recent measurement.

CHANGING THE BATTERIES
1) Press inside the small button in the rear of the
device with a fingernail, as pictured in the fig. 9,
or with a pen. Lift the cover up.
2) Remove the old batteries and
discard safely.
3) Insert 4 new batteries (AAA alkaline)taking care to insert them
in the correct position as indicated.
4) Replace the cover.
5) After changing the batteries wait
for about 20 minutes for the
device to stabilize or recalibrate.
6) Remove batteries if the device is not
in use for any length of time.

CLEANING THE GILT GUIDE WAVE
It is important never to touch the gilt guide wave with your fingers or with any other object, as it is very delicate. For this reason, we recommend that you always protect the thermometer by closing its protective cap when not in use. If the guide wave need cleaning, lightly moisten a cotton swab with alcohol, and apply gently to the guide wave without pressing. Make sure that you clean away from the direction of the sensor, located at the bottom of the guide wave. Avoid cleaning with any other objects or liquids, which could scratch or in any way damage the guide wave. While cleaning, ensure that no liquid penetrates between the guide wave and the sensor.

CLEANING THE BODY OF THE THERMOMETER
Use a soft cloth slightly dampened with soapy water.

DO NOT USE the thermometer for at least 30 minutes after cleaning.

ROOM CALIBRATION (not available on the 0700 models)

If the device is handled for too long, or stored in a place where the temperature varies a lot, a countdown may appear on the display, depending on the model. This means that the unit will need time to stabilize and recalibrate itself. You have 3 different possibilities:

1. Automatic Quick Calibration System (AQCS): wait until the countdown comes to an end without touch it; if the room temperature continues to vary, or if the device is still being handled, the amount of time needed for the calibration will constantly be re-evaluated. As a result, the countdown will also reset itself. It is possible to take a temperature during the countdown, but accuracy is not guaranteed, especially if the temperature continues to rise. At the end of the countdown, the device is capable of measuring with enough accuracy. During the following stand-by the symbol “RE” will mean that the thermometer has been quickly stabilised.

2. Manual Quick Calibration System (MQCS): as an alternative you can quickly get an accurate room temperature when the thermometer is in stand-by and if the room temperature falls between 60.8 /104.0°F (16 to 40°C) by proceeding as follows:
1) press the “face” and “home” buttons at the same time. The word CAL will appear; open the protective cap.
2) Within 10 seconds, press the “home” button and aim the device at a wall or an object with a consistent temperature placed approximately 30 to 60 inches (80 – 150 cm) off the ground.
3) Release the button: the aiming lights will blink slowly and the display will then indicate the reading as a reference room temperature.
In order to be sure that this is the actual room temperature, aim the thermometer at a surface whose temperature remains stable and which is not influenced in any way by heat or cold (an inside wall, a piece of furniture; at any rate, an object which is neither in direct sunlight nor close to a lamp, a radiator, a window or in a draft). When using the thermometer in hospitals, do not hesitate to ask your supplier for the apposite Target Discs which you will then place in each room in order to mark the most appropriate area for the use of MQCS.
4) The thermometer is ready to take a temperature.

The MQCS may also be used when there is no countdown, should you need to change rooms which have different temperatures. This system also enables the thermometer to take good temperatures. Also in this case during the following stand-by the symbol “RE” will mean that the thermometer has been quickly stabilised.

3. Normal room calibration: in order to obtain the best possible measurement, let the thermometer stabilize itself completely. You have to wait 10/30 minutes according to the difference of temperature.

How to change the reading from CELSIUS (°C)to FAHRENHEIT (°F), from ORAL to RECTAL and from“ENERGY-SAVING” to “MAXIMUM PERFORMANCE” (or from “HOME” BUTTON ENABLED to “HOME” BUTTON DISABLED) or viceversa

Depending on the country of destination your thermometer has been set up to:
– display the temperature in Fahrenheit (°F) or Centigrade (°C) degrees;
– show “Oral” or “Rectal” reference;
– display off when in stand-by (“energy-saving” setting) or always on (“maximum performance” setting – not possible on the 0700 models);
– “home” button enabled or disabled.
However, you may change these settings to suit your own preference.
To change the setting, proceed as follows:

1. With the thermometer off (0700 models) or in standby mode (01500 models), press the “on” button (0700 models) or the “face” button (01500 models) and hold it down; after about 8 seconds, the visualization on the display changes showing in rotation the following combinations:
°C; °F; ORAL; RECTAL; “on” (home); “off” “home”;
(“on” and “off” – “home” means the display is on or off while in stand-by mode in the versions for consumer purpose, or “home” button enable or disabled in the version for professional purpose; these two possibilities are alternative each other according to the sector where the device is sold and cannot be present simultaneously in the same device).
2. When the preferred setting appears, release the button.

3. When the word “READY” appears the thermometer will be ready to take a measurement. You can change only one setting at a time.

POSSIBLE PROBLEMS AND THEIR SOLUTIONS

1) Display does not switch on:
– batteries are flat or low or have been incorrectly inserted; replace the batteries immediately or reinsert them correctly.
2) Two bright points of light are visible on the subject’s forehead:
– the distance from the device to the forehead is incorrect; move the thermometer closer or further away until you see a single bright point of light.
3) The point of light is hard to see:
– there is too much light in the room; position yourself to cast a shadow over the subject.
4) The gilt guide wave is damaged or water has penetrated the thermometer:
– contact the distributor for assistance.
5) The temperature shown on the thermometer appears too low:
– check that the device has been used correctly (see “WARNINGS” section);
– check that the thermometer is held correctly perpendicular to the forehead, as shown in figure 5;
-check that the gilt guide wave is not dirty or damaged; if so, clean as indicated in the appropriate section of this handbook or, if damaged, contact the distributor.
6) The temperature shown on the thermometer appears too high:
– check that the device has been used correctly (see “WARNINGS” section).
7) The aiming lights remain constantly on after releasing the button, or the thermometer appears to be stuck or frozen in process:
– reset the thermometer by removing and reinserting the batteries.

The display flashes,
alternating between showing the measurement and the figure “Hi.2″.

The subject has very high fever (over 40°C – 104°F).

Consult a doctor.

Measurement with “on” button (mod 0700): the display flashes, alternating between showing the measurement and the figures “Lo.3″.

If used on the forehead it means that its temperature
is below the working limits.

Check that the guide wave is not dirty or damaged, and that the subject has not just come from a location with a lower temperature.

The display shows “Hi.4″.

The ambient temperature is too high (over 40°C 104°F).

Move to a cooler location.

Measurement with “face” button: the display shows “Lo.5″. Measurement with “home” button or with “on” button (mod 0700): the display flashes, alternating between showing the measurements and the figures “Lo.5″.

accuracy from 34.0 to 35.9°C (93.2/96.7°F)
and from 39.1 to 42.5°C (102.3/108.5°F)

+/-0.5

+/-0.5

Other uses

1.0/55.0

1.0/55.0

Measuring range

33.8/131.0

33.8/131.0

* 16/40

* 16/40

Ambient temperature working range

* 60.8/104

* 60.8/104

0.1

0.1

Resolution

0.1

0.1

+/-0.2

+/-0.2

accuracy from 36 to 39 °C (96.8 to 102.2 °F)

+/-0.4

+/-0.4

+/-0.3

+/-0.3

accuracy from 34.0 to 35.9°C (93.2/96.7°F)
and from 39.1 to 42.5°C (102.3/108.5°F)

+/-0.5

+/-0.5

+/-1.0

+/-1.0

accuracy from 42.6 to 55°C (108.1/131 °F)

+/-1.8

+/-1.8

+2.0/-0.3

accuracy from 27.5 to 33,9°C (81.5/93.1°F)

+3.6/-0.5

+/-1.0

accuracy from 1.0 to 27.4°C ( 33.8/81.4°F )

+/-1.8

+/-0.3

accuracy from 20.0 to 33.9°C (68.0/93.1°F)

+/-0.5

+/-1.0

accuracy from 1.0 to 19.9°C (33.8/67.9°F )

+/-1.8

* The devices can work also in environments with a temperature between 5 and 16°C (41/60.8°F), but we do not guarantee the accuracy and the measuring range. Besides the readings are alternate to “Lo.5″

Store in a clean and dry place preferably at a temperature between +16 and +40°C (60.8/104°F)

Thermofocus is a IIa class medical device which has been tested in hospitals, clinics and consulting rooms.
Clinical accuracy, characteristics and procedures are available from the manufacturer on request.
This infrared thermometer meets requirements established in ASTM Standard (E1965-98).
Full responsibility for the conformance of this product to the standard is assumed by
Tecnimed srl, 12 p.le Cocchi, 21040 Vedano Olona (VA) – Italy
Classification according to regulation IEC 60601-1-2. Internally powered equipment for continuous operation.
Ordinary equipment not suitable for use in the presence of a flammable anaesthetic mixture with air or with oxygen or nitrous oxide.
The aiming leds of Thermofocus emit low luminous radiation largely under the level foreseen from IEC 60825-1 Standard for Class I lights.

MANUFACTURER GUARANTEE
Tecnimed srl guarantees this product against defect due to faulty materials or manufacture for the period of two years from purchase. In order to ensure that the product is covered by the guarantee it is necessary to return the product according to the distributors’ instructions. The guarantee does not cover damages caused by faulty or leaking batteries and damages caused by carelessness or improper use of the product.
The guarantee is void in the following additional cases:
– if the product has been damaged
– the label with the serial number (if any) was removed or damaged or rendered illegible
– if the product has been opened or repaired by non-authorized personnel
– if the product has been damaged due to a failure to follow the instructions set out in the instructions for use.
The product will be repaired or replaced at the manufacturer’s discretion. In no event can Tecnimed be liable for any damage inherent, direct or consequent to the purchase or the use of the product or for cost over the original cost of the product.
Returns: if you would like to return an item for any reason you must first contact the distributor to obtain a Return Authorization (RA) number and a Return Notification form which we will either email or mail to you. Please have your order/customer number and/or your proof of purchase when you call. The Return Notification form will have all details of where and how to ship your returned items.

Thermofocus collects at distance the infrared emissions that all the objects and the living beings diffuse in relation with the body’s superficial characteristics and in a bigger proportion when the body temperature is high.
Thermofocus takes the body temperature on the forehead. The forehead is the perfect place where to take the temperature because it is supplied by the temporal artery, that receives blood from the aorta and the carotid which ensure a significant blood flow in that area; moreover the forehead is the only part of the head in direct contact with the brain that is not covered with hair.
The brain is the most delicate organ of our body and it is the one that can be more damaged by temperature excesses.
At the same time the head is the first part of the body that changes its temperature when in a feverish state, both when fever is raising and when it is getting lower.

What is the normal temperature?

It is very important to know that it does not exist an only one “normal” temperature for everybody: everybody has his own temperature that varies during the day also on the basis of the person’s physical or mental activities (the cry in case of a baby). Moreover, the temperature changes from area to area and it can be influenced by the external temperature and by various factors, according to the kind of measurement. Also the temperature differences between the various areas (frontal, oral, axillary, rectal, tympanic) changes from person to person.

Following the dissipation of heat that the non-protected parts of the body undergo, the normal temperature that can be taken on a person’s forehead is usually lower than the one of other covered parts. For this reason, Thermofocus’ internal software automatically applies a correction to give a value that is approximately comparable to the oral temperature (that is usually lower than rectal one, with a variable value between 0,3 and 0,8°C). The temperature taken with Thermofocus on the forehead of a healthy person can usually vary from 35 to 38°C, but in some cases, above all in adults and in elderly people, it can be lower than 35°C.
A person who has a forehead temperature that is usually low, will have a lower feverish state’s threshold if compared to a person whose temperature is normally higher and vice versa.
For this reason, to better assess possible feverish states, it is convenient to know the temperature of every single family member, when in ideal health conditions and in the various moments of the day.

The instrument does not require additional costs and particular cares.
The only necessary maintenance, when the instrument is not used, is simply to avoid that dust or dirt enter in the golden wave guide placed inside the instrument’s prod, protecting it with its protective cap.

Distance from the subject during operation: about 3 cm (1.2 inch), set through an optical signal
*Thermofocus can work also in environments with a temperature between 5 and 16°C (41/60.8°F),but in this case the accuracy and the range temperature of working is not guaranteed.

Thermofocus is the non-contact thermometer, designed to allow you measuring your baby’s temperature without even touching him/her.
You know that sometimes the body temperature measurement is not easy for you and it is irritating for your child, also because when the babies are ill, they are often distrustful and irritable. And what should we say about the disappointment you feel when, having to check your baby’s temperature, you are obliged to wake him/her up during the night, maybe after hours of inconsolable cry?
Today, on the contrary, with Thermofocus measuring your baby’s temperature becomes a sweet gesture, just like a caress. If your baby is sleeping, he/she will not have to be woken up and if he/she is awake, will not be frightened, but he/she will be curious and eager to play with you and Thermofocus.

«In many cases, the child is convinced to take part in a game, […]. For me it has been common to see that the child tried to grab the instrument to play with it, since these little lights were not frightening for him/her at all.»

Thermofocus has been designed for your baby, but it can be used by all your family members. Since it does never get in contact with the forehead, it is by far the most hygienic thermometer and it does not need any disinfection action.

Moreover, Thermofocus can perform any measurement apart from the one on the forehead, on object and on other parts of the body.
It is a function designed just for you, mum; now you will be able to take very easily the temperature of the bath’s water, of the baby’s bottles, of the soups and also the temperature of any other object between 1 e 55°C.

Thermofocus is a vanguard thermometer, since it matches the advantages of its comfort, the speed in the measurement-taking and its hygienic quality, this latter being a fundamental factor above all if we think of consulting rooms or hospitals. Since it never gets in contact with the forehead, it can be used to take the temperature in all patients without having to intervene with any disinfection operation or without any kind of discomfort for the patient.

Besides the body temperature’s measurement, a doctor can use Thermofocus, using the button “home”, to scan the temperature in a certain area of the skin, comparing it to the surrounding area or to the symmetrically specular one (two legs, two feet etc.), it is then possible to find anomalies due to inflammations, circulatory problems, etc. If it is necessary, of course, it is possible to use it on open or healing wounds (even on internal organs during surgical operations since there is no contact between the thermometer and the part on which the temperature has to be taken).

Unlike other thermometers that require disinfection after each use or expensive hygienic disposable covers, Thermofocus does not require anything like this at all. And with savings that for a hospital becomes very important. Usually the costs of hygienic caps vary from 2 to 10 Euro cents (according to the kind of thermometer). Let’s suppose that they have an average cost of 5 cents each. Let’s also suppose that we have to take 4 measurements a day in a 25-patient ward with an auricular thermometer. We have a cost of 0,05 x 4 x 25 = 5 € a day. Let’s multiply by 365 and we get a cost of 1.825 € for each thermometer! Which is money that is literally wasted! Moreover the battery consumption is certainly lower with Thermofocus. Then, if we estimate the time saved by the nurses for each measurement (cap insertion, probe positioning in the auditory canal, cap removal) we can suppose that each Thermofocus can allow savings of more than 4.000 € per year!

Model Language Video

Sometime the temperature reading is too low. Why?

the lens may be soiled: the presence of dust or dirt filters the passage of the infrared emissions; as s consequence, Thermofocus will take a lower reading than the real body temperature.
It is necessary to clean the waveguide gently, using a cotton swab that has been slightly dampened with alcohol. Don’t use the thermometer for at least 30 minutes after cleaning.

The forehead can show signs of sweat or an oxygen mask is present or the subject is elderly. This can result in a low reading In all the above cases, you can take the temperature reading, again from a distance, on th eyelid (answer 10). – Oils or make-up con be present. This can results in a low reading. You can take the temperature reading, again from a distance, on the eyelid or on the neck (answer 10).

If I take several consecutive readings, my Thermofocus detects different temperatures (more than 0.2°C difference). Why?

Probably, the thermometer is not always aimed on the same spot and at the correct distance from the center of the forehead, or it has been moved before the lights flashed: make sure you keep the thermometer perpendicular to the center of the forehead and at the correct distance indicated by the aiming system, and keep it steady while the lights flash. Changing the reading point on the forehead will give incorrect results. – The thermometer was stored in another room (or even in a drawer, cabinet, etc.) at a different temperature than the room where the thermometer is used. Make the MQCS or wait until the countdown has run its course (AQCS) or for 10/30 minutes before taking any other measurements, so that the device can stabilize on its own.

Are there any external factors which can modify the measurement?

Usually not, provided that you consider as a normal situation a hill baby laying in bed at home. However, there may be some situations affected by external factors. In particular, Thermofocus should not be used on a person directly exposed to the sun, to sources of heat (a radiator or a chimney) or cold (outside during in winter, air conditioning) or again on a person that is training sport or it has just finished it (in fact, while walking, the forehead get colder because of the surrounding air).

Is it possible to take the temperature to a person coming from another room?

Before taking the temperature to a person coming from an ambient at a different temperature than the room where the thermometer is used (for example from outside), you must give them the time to stabilize themselves into the new room. To do this we suggest that the person stays in the room for some minutes, calm and still, possibly sitting down or laying in bed.

How can I make the MQCS (Manual Quick Calibration System – only 01500A3 model)?

The alternative is to promptly correct the thermometer temperature, adapting it to the real temperature of the room where the reading is to be taken. Proceed as follows (the room temperature must be in the 10-40°C (50-104°F) temperature range):1. with the thermometer in stand-by mode press and release the “face” and “home” button simultaneously: the symbol CAL will appear;
2.Within 10 seconds, open the cap and, focus the thermometer on an internal wall (not the inside of an external wall) or wardrobe with uniform temperature and at a point approximately 30 to 60 inches (80/150 cm) from the floor. Press the “home” button until the temperature reading is set squarely between the two arches
3. release the button: the lights flash slowly and the display shows the room temperature. To ensure a reliable temperature reading, do not focus the thermometer on an outside wall, window, source of heating or cooling (radiator, air conditioner, lamp, computer, surface in contact with the human body, etc.).
4. The thermometer is ready to take a reading.
Manual quick calibration (MQCS) can also be performed without the countdown if, for example, you need to move between rooms at different temperatures.
This system, too, enables the thermometer to take sufficiently accurate readings. In this case, when the thermometer is in stand-by mode, “MQCS” appears on the display indicating that manual quick calibration has been performed. It is also possible to just leave the thermometer in the room and wait 10/30 minutes, depending on the difference of temperature.

How does the AQCS (Automatic Quick Calibration system) work?

If the thermometer is handled at length or if there is a significant temperature difference with respect to the room temperature, a countdown will appear on the display indicating that you should wait until automatic temperature calibration has been completed. Without touching the thermometer, wait until the countdown has run its course and automatic quick calibration is completed. The countdown will continue updating as long as differences in temperature are detected (for example, because the thermometer has been handled).. You can take a reading before the countdown has run its course, although, in this case, accuracy cannot be guaranteed, particularly if the temperature is rising. At the end of the countdown, the thermometer can take sufficiently accurate readings. When the thermometer is in stand-by mode, “RE” appears on the display indicating that automatic quick calibration has been performed.

Is the temperature reading detected with Thermofocus the same as that taken with other thermometers and in other parts of the body?

The temperature reading detected with Thermofocus is normally very similar, but not necessary equal, to the temperature taken with other thermometers. When it is set in the oral or rectal mode, the temperature is very similar respectively to the oral and the internal rectal measurement taken with other thermometers. To obtain this result, the software of Thermofocus automatically applies a correction factor that compensates the heat dissipation from uncovered parts of the body (such as the forehead).
Moreover, the temperature differs from one area of the body to another (armpit, mouth, rectum, forehead) and that differences are not valid for everyone, but each person has his/her own body temperature:: for instance, the rectal and the oral temperature can, on some subjects, differ for less than 1°C (1.8°F), while be much more than 1°C (1.8°F) different in others.
For this reason, sometimes the forehead temperature can be different from the temperature measured in other parts of the body.
To correctly judged a fever, you need to know the normal temperature of your family members when they are in good health conditions and at various times of the day
For example, a person having a very low temperature in good health conditions will have a lower mark fever compared to others usually having a higher temperature. To help you with this, download the table in pdf format (available in English,French, and Spanish languages) fill it out and keep it as a reference.

Can I use Thermofocus even if it comes from a room having a different temperature?

All infrared thermometers must know the room temperature, and so all manufacturers suggest to wait for a certain time (usually from 10 to 30 minutes or more depending on the difference of the temperature) before using the thermometer when you come from another room at a different temperature then the room where the thermometer is used. With Thermofocus, you do not need for waiting, thanks to 2 exclusives patented systems for quick stabilization: the AQCS and the MQCS.

Can I change the setting from Celsius to Fahrenheit or vice versa?

Depending on where it is to be sold, your thermometer leaves the factory preset to: Celsius (°C) or Fahrenheit (°F). These settings can be modified as follows:
When the thermometer is in stand-by mode, press and hold the “mem” button (or “face” or “on” button in the older models); after about 8 seconds, the visualization on the display changes showing the settings in sequence: °C, °F, AXILLA, ORAL, RECTAL. When the preferred setting appears, release the button. After a few seconds, the thermometer will be ready for a measurement in the selected setting.

Can I change the setting from oral to rectal and axilla or vice versa?

Depending on where it is to be sold, your thermometer leaves the factory referred to AXILLARY, ORAL or RECTAL temperature readings These settings can be modified as follows:
When the thermometer is in stand-by, press and hold the “mem” button (or “face” o “on” button in the older models); after about 8 seconds, the visualization on the display changes, showing the settings in sequence: °C, °F, AXILLA, ORAL, RECTAL. When the preferred setting appears, release the button.
After a few seconds, the thermometer will be ready for a measurement in the selected setting. When the thermometer is set in RECTAL mode, the reading is 0.6°C (0.8°F) higher than an ORAL reading. When it is set in ORAL mode, the reading is 0.2°C (0.4°F) higher an AXILLARY reading.

Can I make my Thermofocus shut down immediately after the last use?

Depending on where it is to be sold, your Thermofocus (01500A3 model) leaves the factory preset with the display off when in stand-by mode (“Energy Savings”) or display on and showing the room temperature for 4 hours before shutting off (“Peak Performance”) , These settings can be modified as follows: When the thermometer is in stand-by mode, press and hold the “mem” button (or “face” button in the older models); after about 8 seconds, the visualization on the display changes, showing the settings in sequence: °C, °F, AXILLA, ORAL, RECTAL, “home” on, “home” off. When the desired setting appears, release the button (“home “ on means that the display will show the ambient temperature for 4 hours before shutting off, “home” off means that it will shut down immediately after the 20 second interval after the last reading.

Does the batteries’ consumption increase if Thermofocus displays the room temperature when it is in stand-by mode?

When the thermometer remains idle for 20 seconds, it reverts to stand-by mode and will display the room temperature for 4 hours before shutting off. When it is on stand-by, even if the display stays on visualizing the room temperature, it does not remain active, which means that the consumption stays lower than that of a cell phone when not in use. The device autonomy with new batteries is of about 3 years or 10.000 measurements, depending on use.

Why Thermofocus shows the ambient temperature? (only 01500A3 model)

Besides the comfort to have a clinical thermometer which works also as an ambient thermometer, the main purpose is to check that the temperature of the thermometer is constantly the same as the real ambient temperature in order to make measurements with the thermometer calibrated.

After the first measurement, how long do I have to wait before I can take another one?

With Thermofocus, you can immediately take another reading.

What happens if the red lights are unintentionally directed towards the eyes?

The lights are harmless. THEY ARE NOT LASER BEAMS!

Is there an area alternative to the center of the forehead?

We suggest to take the temperature reading again from a distance, by taking a close-up scan of the closed eyelid, using the “face ”button. No need to worry that your child could open its eyes while you are taking the reading: the lights are harmless. Performances are not guaranteed, but such reading can be considered a valid approximation of one’s body temperature and a fever’s indicator.

Is there any situations in which Thermofocus should not be used in the center of the forehead?

The forehead temperature reading can be affected by profuse sweating or if an oxygen mask is present or if the subject is elderly.
In case of sweating, dry the forehead is not enough, as the low readings do not depend on the sweat drops, but on the dissipation of heat which occurs when sweating.
The oxygen masks are not hermetic and so the oxygen can make the forehead cold: the temperature reading would not be correct.
A low vascularization of the forehead may occur in elderly people. In all the above cases, the temperature detected on the forehead will be lower than the real body temperature.

Why is the forehead the ideal site for taking a temperature reading?

The forehead is crossed by the temporal artery and is the only part of the body in direct contact with the brain and that is not covered by hair. Furthermore, the head is the first part of the body to change temperature as a fever, rises and falls. This characteristic is very useful, as it can give predictive information of how the fever is changing. This is why, sometimes, it is possible to find the fever on the forehead and not in other parts of the body or viceversa. This simply means that the fever is rising or falling.

Which is Thermofocus range of work?

The measurement range of the body temperature (“face” button) is between 34 and 42.5 °C (93.2/108.5 °F).
The measurement range of temperature of objects, liquids and surfaces (“home” button) is between 1 and 55°C (33.8/131.0°F).
Thermofocus can operate in rooms where the temperature is between 16 to 40 °C (60.8-104°F). It is possible to use the thermometer also in rooms where the temperature is lower than 16C (60.8°F), although accuracy and the operating range are not guaranteed.

Can Thermofocus be used to take the body temperature to people of any age?

For the comfort of its use, its non-invasive methods and its quick measurements, Thermofocus has been designed for children, but adults can also use it. However, it is important to bear in mind that sometimes in elderly people, the temperature taken with Thermofocus on the forehead could be lower than the correct body temperature.

Why is it so important for the thermometer to be at the correct reading distance?

The “field of view”, where every infrared sensor works, is cone-shaped: this means that the farer you go from the surface, the bigger the cone’s base – the area of which you take the reading – becomes.
If, for example, the distance from the center of the forehead is bigger than expected, the thermometer will include in the measurement area also the temperature of the eyebrows, of the eyes or of the hair. If the distance gets even bigger, it will include the temperature of the whole face.
In all these cases, the reading temperature would not be correct as it would be an average of all the parts of the face included in the “field of view”.

What is the correct reading distance?

The thermometer itself will tell you the correct reading distance, by means of its patented optical aiming system: if the thermometer is too close, or too far away from the skin, you will see two single or blurred points, respectively. When you see only one point, the thermometer is at the right distance for a correct reading.

How does Thermofocus work?

Thermofocus detects the infrared radiation naturally emitted by the body and, in particular, from the human forehead. In just a few hundredths of a second your Thermofocus takes a series of 125 readings, Its sophisticated microprocessor then amplifies and processes this information along with the room temperature, detected from a thermocouple probe incorporated within the sensor. The data are compared with the parameters contained in the memory, and the results of these processing is the temperature showed on the display.

What are Thermofocus’ advantages?

Thermofocus is able to take the body temperature without touching the skin: if your baby is sleeping, you will not wake him up; and, if he is awake, he will not be scared but, on the contrary, will be attracted and willing to play with you and Thermofocus.
Its optical aiming system indicates the correct reading distance, that is essential to obtain accurate measurements.
It is truly easy to use, reliable, safe and very fast. Moreover, there are no breaking risks, as in the case of mercury thermometers. Finally, unlike other thermometers, Thermofocus is hygienic as there is no need to disinfect it or to use hygienic protection. These features reduce in fact the risk of contaminations.