Search form

TS Series 5 kW to 45 kW

Features

High Frequency IGBT-based Switch-Mode Design

The TS Series utilizes the latest IGBT devices to drive down size and increase performance. The high frequency design delivers fast transient response and low output ripple. This performance coupled with Magna-Power's rugged current-fed power processing, enables confidence in even the most demanding applications.

Fully Programmable Power

All TS Series functions and features of are available and programmable from a variety of control methods, including:

Broad Voltage-Current Range, Consistent Controls

Magna-Power Electronics TS Series provides a wide voltage and current range while still maintaining among the highest power density rack-mount packaging. The TS Series IV covers voltages from 5 Vdc up to 4000 Vdc (floating) and current levels from 1.2 Adc up to 2700 Adc. Models 5 kW to 15 kW are available in a 3U chassis, models 20 kW to 30 kW are available in a 6U chassis, and 45 kW models are available in a 9U chassis.

Designed for Safety

TS Series power supplies have extensive diagnostic functions, including: phase loss, excessive thermal conditions, over voltage trip (programmable), over current trip (programmable), cleared fuse, and excessive program line voltage. When in standby or diagnostic fault, the AC mains are mechanically disconnected via an embedded AC contactor, providing confidence that the unit is only processing power when desired.

Scalable Master-Slave Operation

For maximum flexibility, all TS Series power supplies can be scaled a in master-slave configuration for higher voltage or current output. Using the plug and play Universal Interface Device (+UID47), units are tied together in via their 37-pin I/O, allowing an increase in system current when configured in parallel or increase in system voltage when configured in series—up to the power supply's output isolation rating.

Extensive Programming Support and Included Software

All TS Series power supplies come standard with a National Instruments LabVIEW and LabWindowsâ„¢/CVI drivers. In addition, an IVI driver is included supporting Visual C++, Visual C#, Visual Basic .NET, MATLAB, Measure Foundry, PAWS, Visual Basic 6.0, and Agilent VEE Pro. A Standard Commands for Programmable Instrumentation (SCPI) command set is also supported, allowing programming using raw ASCII commands. Finally, Magna-Power's RIS Panel software is provided allowing for a virtual control panel controlling the power supply via any of the available communication interfaces.

Models

There are hundreds of possible configurations for the TS Series product. Using the following ordering guide and models chart to define the best model for your application. Need less power? Visit our XR Series product line. Need more power? Visit our MS Series product line.

Model Ordering Guide

Models Chart

The following chart details the available standard TS Series models. The Current Maximum (Adc) column is separated by the available power levels. To determine the appropriate model, first select your output Voltage Maximum (Vdc) to find appropriate row. Next, select one desired Current Maximum from the row that contains your desired Voltage Maximum. Then, construct your model number according to the model ordering guide, above. Non-standard voltage and current configurations are available.

5 kW

10 kW

15 kW

20 kW

25 kW

30 kW

45 kW

Voltage Maximum

Current Maximum

Ripple (mVrms)

Eff. (%)

5

900

1800*

2700*

N/A

N/A

N/A

N/A

50

86

8

600

1200*

N/A

N/A

N/A

N/A

N/A

40

86

10

500

900

1800*

N/A

2700*

N/A

N/A

40

86

16

300

600

900

1200

1500

1800

2700

35

86

20

250

500

750

1000

1250

1500

2250

40

86

25

200

400

600

800

1000

1200

1800

40

86

32

150

300

450

600

750

900

1350

40

86

40

125

250

375

500

625

750

1125

40

87

50

100

200

300

400

500

600

900

50

87

60

83

166

249

332

415

498

747

60

87

80

62

124

186

248

310

372

558

60

87

100

50

100

150

200

250

300

450

60

87

125

40

80

120

160

200

240

360

100

87

160

31

62

93

124

155

186

279

120

87

200

25

50

75

100

125

150

225

125

87

250

20

40

60

80

100

120

180

130

88

300

16

32

48

64

80

96

144

160

88

375

13

26

39

52

65

78

117

170

88

400

12

24

36

48

60

72

108

180

88

500

10

20

30

40

50

60

90

220

88

600

8

16

24

32

40

48

72

250

88

800

6

12

18

24

30

36

54

300

88

1000

5

10

15

20

25

30

45

350

88

1250

4

8

12

16

20

24

36

375

88

1500

3.3

6.6

9.9

13.2

16.5

19.8

29.7

400

88

2000

2.5

5.0

7.5

10.0

12.5

15.0

22.5

450

88

3000

1.6

3.2

4.8

6.4

8.0

9.6

14.4

500

88

4000

1.2

2.4

3.6

4.8

6.0

7.2

10.8

550

88

Input Current Per Phase (Aac)

208/240 Vac, 1Φ

38

N/A

N/A

N/A

N/A

N/A

N/A

208/240 Vac, 3Φ

19

36

54

72

90

108

162

380/415 Vac, 3Φ

10

21

30

40

50

60

90

440/480 Vac, 3Φ

9

18

27

36

45

54

81

For models with the High Slew Rate Output Option (+HS), ripple will be higher. Models marked with an asterisk (*) are special low-voltage high-current models that may vary in size and input current from the standard TS Series models in the same power envelope. Contact your local sales partner for additional details.

User inputs and outputs: referenced to earth ground.Maximum input voltage to ground: ±2500 VacMaximum output voltage to ground:
±1000 Vdc for models less than or equal to 1000 Vdc,
±(2000 Vdc + Vo/2) for models greater than 1000 Vdc or with High Isolation Output Option (+ISO) where Vo is the maximum output voltage of the power supply.

Maximum Slew Rate

Standard Models:
100 ms for an output voltage change from 0 to 63%,
100 ms for an output current change from 0 to 63%.With High Slew Rate Option (+HS):
4 ms for an output voltage change from 0 to 63%,
8 ms for an output current change from 0 to 63%.

Diagrams

For maximum flexibility, all front panel function and controls have equivalent programming and monitoring functions on the included rear 37-pin analog-digital I/O connector and on the computer programming interface.

A Version Front Panel (TSA)

The A Version front panel provides a digital display with rotary front panel input, isolated 37-pin analog-digital I/O, and a RS232 computer interface.

D Version Front Panel (TSD)

In addition to the features of the A Version front panel, the D Version front panel provides digital 10-key entry, auto-sequencing with memory capability, and modulation for non-linear power profile emulation.

C Version Front Panel (TSC)

The C Version front panel is blank, providing only a switch to enable control power. All control must be performed by the provided isolated 37-pin analog-digital I/O or through a computer interface.

A

Engages and disengages main power

B

Meters display output voltage, output current, voltage set point, current set point, over voltage trip, and over current trip

C

Over voltage trip and over current trip setting potentiometers

D

Mode

POWER: Indicates power output

STANDBY: Indicates control power only

E

Configuration

REM SEN: Remote sense enabled

INT CTL: Front panel start/stop/clear enabled

EXT CTL: External start/stop/clear enabled

ROTARY: Front panel rotary knob input

KEYPAD: 10-digit numeric keypad control

EXT PGM: External analog voltage-current control

REMOTE: Computer control

F

Diagnostic Alarms

LOCK: Interlock

PGM LN: External input voltage beyond limits

OVT: Over voltage protection has tripped

OCT: Over current protection has tripped

THERM: Over temperature condition

PHASE: Under voltage AC input

FSE: Warns that a fuse has cleared

ARM: Indicates power supply is ready for or operating in auto-sequencing

G

Power switch energizes control circuits without engaging main power Diagnostic Alarms

H

Stepless rotary knob to set voltage-current

I

Function Keys

MENU: Selects function

ITEM: Selects item within function

V/I DIS: Displays voltage-current settings

TRIP DIS: Displays OVT and OCT setting

CLEAR: Clears settings or resets fault

ENTER: Select item

J

Memory location indicator, used for auto-sequencing applications

K

Digital input keypad

Front Panel Dimensions

Front panel dimensions diagram references to a, b and c correspond to values in following table:

Power

a

b

c

5 kW

2.250"

N/A

5.219"

10 kW

2.250"

N/A

5.219"

15 kW

2.250"

N/A

5.219"

20 kW

3.000"

2.250"

10.469"

25 kW

3.000"

2.250"

10.469"

30 kW

3.000"

2.250"

10.469"

45 kW

4.750"

4.000"

15.719"

Rear Panel Dimensions (3U) (Standard, Air Cooled)

Rear Panel Dimensions (3U) (Optional, Water Cooled +WC)

Rear Panel Dimensions (6U/9U) (Standard, Air Cooled)

Rear Panel Dimensions (6U/9U) (Optional, Water Cooled +WC)

Side Panel Dimensions (3U) (Standard, Air Cooled)

Side Panel Dimensions (6U/9U) (Standard, Air Cooled)

1

Front Panel Handles

2

Rear Air Exhaust

3

Output DC Connections - 3/8-16 Threaded Insert>

Qty (1) Insert Per Bus Bar, 5 kW to 15 kW Models

Qty (2) Inserts Per Bus Bar, 20 kW to 30 kW Models

Qty (4) Inserts Per Bus Bar, 45 kW Models

4

Computer and External Control Connections

5

Remote Sensing Connector

6

Input AC Connections
1/4-20 Bolt, Qty (4) PLC’S

7

Front Panel Handles

8

Side Air Intake (Air-Cooled Models Only)

9

Output DC Connections, 5 kW - 15 kW (Side View)

10

Output DC Connections, 20 kW - 45 kW (Side View)

Note: Metal Cover included for rear panel on 5 kW to 15 kW models. Stand-offs included for all other models.

Options

The following options are available for the TS Series programmable DC power supply:

High Isolation Output
Available for models rated for 250 Vdc to 1000 Vdc, the +ISO option extends the output isolation, used when the application demands floating or tying units in series beyond the standard 1000 Vdc output isolation rating. More

High Slew Rate Output
The high slew rate option is a control modification and replaces the standard output stage with one of low capacitance film and/or aluminum electrolytic capacitors, capable of high RMS currents. This stage provides higher bandwidth with faster output rise and fall times.More

IEEE-488 GPIB Interface
Full control of the product's features and diagnostics over a IEEE-488 GPIB interface. The GPIB interface allows up to 15 devices may be connected to one bus and is compatible with all Magna-Power provided software and drivers. More

The Integrated Blocking Diode Option (+BD) provides an internally heat-sunk protection diode on the positive output terminal of a MagnaDC programmable DC power supply to protect the product's output from reverse voltage up to 1200 Vdc.

LXI TCP/IP Ethernet Interface
Full control of all the product's features and diagnostics over a LXI Class C Certified TCP/IP Ethernet interface. Includes an embedded web server and compatibility with all Magna-Power provided software and drivers. More

The RS-232 to RS422 or RS-485 converter is an external industrial accessory to convert the power supply's standard RS-232. Unlike converters which require programming hardware handshaking signals to control RS-485 or RS-422 operation, the +RS485 option provides automatic control.

The UID47 accessory eases master-slave parallel or series configuration of Magna-Power DC power supplies, enabling near equal current or voltage sharing, depending on the configuration. The accessory includes the UID47 with integrated circuitry and (2) 6-ft DB37 cables.

Water Cooling
Water cooling is available instead of the standard air cooling. Water cooled units are fully enclosed and to not take in any external air, making them ideal for corrosive environments, densely packed cabinets or setups where water is readily available. More

How to Buy

For the fastest product and pricing information, please complete the form on the rightbelow and the Magna-Power team will address the request immediately.

Build-time

Magna-Power products are made-to-order with build-times that vary depending on the model's maximum power rating. Current build-time schedules are as follows:

Models 5 kW-15 kW, ≤ 1000 Vdc

4 weeks

All Models 20 kW-45 kW
All Models > 1000 Vdc

5.5 weeks

Sales Partners

In the USA and Canada, Magna-Power sells it products directly from the factory, with a nationwide network of technical representatives. Magna-Power products are available available worldwide through direct sales offices and authorized distributors. Requests from this form will either be quoted directly or automatically routed to the appropriate sales partner for quote.

Prefer to Talk?

Feel free use the Live Chat or give us a call: +1 (908) 237-2200

Please use the following form to request a MagnaDC price and lead-time quote. During business hours, quotes are generally emailed within 1 hour. Required Field*