TS 94.5 / Silver Bottle

Validation of the CO2-Testers

The Check Bottle TS 94.5 is used for monitoring the accuracy of the measuring heads of the CO2 tester TS 91.7. It consists chiefly of a polycarbonate container and an integrated digital precision instrument, combining manometer and thermometer.

Operation:

The polycarbonate container is filled up to the mark with carbonated beverage.

Then placed like any beverage bottle in the container of the CO2 tester TS 91.7 (or any other manometric tester), and shaken as during the standard measuring procedure for beverages.

When the measuring is finished, the pressure and temperature readings of the check bottle and the measuring CO2 head can be compared.

Maselli Guide & Brochures

Maselli Product Guide and specific product brochures and datasheets are available. Please contact us to get your free copy by email.

Video Lounge

You can experience Maselli devices in service along thousands of filling lines around the world - and here in our video lounge. Look at the films and see the performance of our checking, inspection, container transport, rejection and labelling.