Orbital shaking is controlled by the digital tachometer (rpm) and Digital time setting regardless of the temperature. The unit is equipped with the direct–drive system ensuring most reliable stable long–time operation (up to 30 day nights).

The ES-20 is extremely easy to operate, with very straightforward setup of temperature, speed and time, using the two line set–up and status display, which clearly indicates both set and actual values for each of the three parameters.

Specification

Temperature setting range

+25°C … + 42°C

Temperature control range

5°C above ambient … + 42°C

Setting resolution

0.1°C / 1 rpm

Temperature stability

±0.5°C

Speed control range

50–250 rpm

Orbit

10 mm

Display

LCD, 2 x 16 signs

Digital time setting

1 min–96 hrs / non-stop

Plexiglas walls thickness

7 mm

Load capacity

2.5 kg

Overall dimensions

340x340x435 mm

Internal chamber dimensions

305×260×250 mm

Weight

13.2 kg

Nominal operating voltage

230 V, 50/60 Hz or 120 V, 50/60 Hz

Power consumption (230V / 120 V)

160 W (0.7 A) / 170 W (1.6 A)

ES-20/60, Environmental Shaker-Incubator

Description

Orbital Shaker–Incubator ES-20/60 for biotechnological and pharmaceutical laboratories is a professional category equipment designed for cultivation of microorganisms and eukaryotic cells including animal, plant and insect cells. Shaker is equipped with a direct-drive mechanism for platform motion. It provides reliable and stable operation for the long term experiments needed for cell growth. Shaker–Incubator ES-20/60 provides smooth or intensive mixing in flasks installed on the platform.

Built-in noiseless thermoresistant brushless fan provides precise temperature distribution inside the chamber (adjustable for up to +80°C). The inner chamber is made of stainless steel. State–of–the–art motor, newest thermal insulation materials, soft-start of the platform motion and temperature PID–control decrease the energy consumption and make the Shaker–Incubator highly energy efficient despite its relatively large size.