Geräte Informationen:

R3753 series is the 500MHz vector network analyzer, which
has newly been designed based on a
concept ” an optimum tool for each application” .
We have fuily pursued high throughput such as 0.1 ms/points
high-speed measurement at a
resolution bandwidth (RBW) of 10kHz, 115dB wide
dynamic-range measurement, and two-device
simultaneous measurement with two-channel/four-trace
display. Also , we have added the
program sweeping function that can freely change the
resolution bandwidth (RBW), output level,
and input attenuator during sweep operation for each segment.
Since the analyzer employs semiconductor switches for
changing the output level and for
switching the input attenuator, it allows the optimum
high-speed level sweeping for the drive level
test of the oscillator.
With the built-in BASIC controller, a high-speed ATE system
can be easily configured with no
external controller for processes from adjustment to inspection.
Features
(1)
High throughput
• O.Tms/point high-speed frequency sweeping and 5ms short
blanking time
When two-channe! /four-trace (amplitude/phase) and RBW 10kHz
• 0.1 ms/point high-speed level sweeping
High speed and long life using semiconductor switches
(2)
Wide dynamic range
• With automatic switching of input attenuator, H5dB wide
dynamic range
High speed and long life using semiconductor switches for
switching input attenuator
(3)
Program sweeping function
• For each segment, allows setting of frequency, output
level, input attenuator, RBW, and
settling time.
(4)
MS-DOS format disk
• By using an MS-DOS personal computer, it is possible to
easily create programs and
analyze data because of the disk complied with MS-DOS format
standard.
• Three modes of storage capacity available (DD 720KB, HD
1.2MB, and HD 1.44MB)