The Multipiezo line is designed to treat the widest range of applications in the market. With an extremely wide ultrasound power range, innovative electronics, and unique ergonomics. The new Mectron Multipiezo and Multipiezo Pro devices have the power and versatility to solve a wide array of daily challenges: calculus, cavities, endodontic inflammation, periodontal inflammation, crown restorations, and beyond.

.

MultiPiezo Basic Touch Piezo Scaler by Mectron

reg $ 3,299Sale $ 1,859

.

Touch KeyboardEasy to clean touch keyboard to choose irrigation and power with the touch of a finger.

LED Handpiece

LED handpieces with new swivel-type light focus Light beam always on the insert tip, regardless of the insert shape Handpiece holder with security

Pulse Mode The Pulse Mode transforms the traditional sine wave formed ultrasound oscillation to a new profile (multipiezo pro, level 6 of the restorative function), characterized by specific power peaks, which results in an optimized performance for extractions and prosthetics.

Soft ModeWhenever needed mectron’s innovative Soft Mode reduces the amplitude of ultrasonic oscillations in order to provide a gentle and efficient insert movement. The result: sensitive patients are treated nearly painless and with maximum comfort.

Optional Water Connection Kit Secure and flexible option to connect to a fixed water line.

Flexible Irrigation System500 ml, no drip no leak bottles give you the option to simply use water or aggressive medical solutions for more involved procedures.

.

Multi Piezo Basic Touch Piezo Scaler Specifications:

Device compliant to Dir. 93/42/CEE: Class II aClassification as per EN 60601-1: IApplied part type B (handpiece, insert)IP 20 (device)IP 22 (foot pedal)Device for intermittent operation: 55 sec. ON – 30 sec. OFF with irrigation30 sec. ON – 120 sec. OFF without irrigationPower supply voltage: 100-240 Vac 50/60 HzMax. power absorbed: 90 VAFuses: Type 5 x 20 mm T 2AL, 250VOperating frequency: Automatic scanFrom 24 KHz to 36 KHzPower types: ENDOPERIO/SCALERRESTORATIVESOFT MODEPeristaltic pump capacity: Adjustable on the touch screen:7 flow levelsfrom 0 (0 ml/min) to 6 (28 ml/min)LED system of the handpiece: Light function on AUTO: LED system of the handpieces witches on as soon as the device starts to operate and switches off 3 seconds after the foot pedal is releasedLight function on ON: The LED of the handpiece is always on; after 100 seconds of foot pedal disuse it switches off by itself and the light function moves into position AUTOLight Function on OFF: The LED of the handpiece is always off.Protections of the APC circuit: No handpiece detectedCord interruptionInsert not tightened correctly or brokenOperating conditions: from +10°C to +35°CRelative humidity from 30% to 75%Pressure of air P: 800hPa/1060hPaTransport and storage conditions: from -10°C to +70°CRelative humidity from 10% to 90%Pressure of air P: 500hPa/1060hPaWeight 3.3 KgSize (L -W – H) 440 x 230 x 145 mmWarranty device 1 years; handpiece and handpiece cord 1 year