Entry / Access Port:
An entry port of Ø 20 mm diameter with silicon cork is centered on the left side for passing cables or probes.Interior Lighting:
The light is activated by a switch on the control panel.Insulation:
Double layered. The exterior is separated by a asbestos-free glass wool and another layer so that exterior and interior are completely separated without any thermal bridge.Door Sealing:
Silicon seal is used for thermal sealing and is easily replaceable.Heating:
By means of sheathed heaters, this eliminates any direct radiation on the specimen in the oven.Adjustable Air Changes:
Adjustable air outlet allows air exchange from ambient surroundings.

Controller:
France Etuves C3000 Temperature Controller is in-house developed and built with safety interlocks for product and user safety. Functions:
– Setpoint display and that measured.
– Delayed Start: 1min to 99h59min.
– Heat ramp.
– Heating timer from 0 to 99h59min.
– Repeat loop.
– Buzzer at the end of a cycle.
– Over temperature alarm.