Abstract

Purpose: Respiratory gated radiation therapy (RGRT) gives accurate results when a patient’s breathing is stable and regular. Thus, the patient should be fully aware during respiratory pattern training before undergoing the RGRT treatment. In order to bypass the process of respiratory pattern training, we propose a target location prediction system for RGRT that uses only natural respiratory volume, and confirm its application. Methods: In order to verify the proposed target location prediction system, an in-house phantom set was used. This set involves a chest phantom including target, external markers, and motion generator. Natural respiratory volume signals were generated using the random function in MATLAB code. In the chest phantom, the target takes a linear motion based on the respiratory signal. After a four-dimensional computed tomography (4DCT) scan of the in-house phantom, the motion trajectory was derived as a linear equation. The accuracy of the linear equation was compared with that of the motion algorithm used by the operating motion generator. In addition, we attempted target location prediction using random respiratory volume values. Results: The correspondence rate of the linear equation derived from the 4DCT images with the motion algorithm of the motion generator was 99.41%. In addition, the average errormore » rate of target location prediction was 1.23% for 26 cases. Conclusion: We confirmed the applicability of our proposed target location prediction system for RGRT using natural respiratory volume. If additional clinical studies can be conducted, a more accurate prediction system can be realized without requiring respiratory pattern training.« less

@article{osti_22649416,
title = {SU-G-JeP3-09: Tumor Location Prediction Using Natural Respiratory Volume for Respiratory Gated Radiation Therapy (RGRT): System Verification Study},
author = {Kim, M and Jung, J and Yoon, D and Shin, H and Kim, S and Suh, T},
abstractNote = {Purpose: Respiratory gated radiation therapy (RGRT) gives accurate results when a patient’s breathing is stable and regular. Thus, the patient should be fully aware during respiratory pattern training before undergoing the RGRT treatment. In order to bypass the process of respiratory pattern training, we propose a target location prediction system for RGRT that uses only natural respiratory volume, and confirm its application. Methods: In order to verify the proposed target location prediction system, an in-house phantom set was used. This set involves a chest phantom including target, external markers, and motion generator. Natural respiratory volume signals were generated using the random function in MATLAB code. In the chest phantom, the target takes a linear motion based on the respiratory signal. After a four-dimensional computed tomography (4DCT) scan of the in-house phantom, the motion trajectory was derived as a linear equation. The accuracy of the linear equation was compared with that of the motion algorithm used by the operating motion generator. In addition, we attempted target location prediction using random respiratory volume values. Results: The correspondence rate of the linear equation derived from the 4DCT images with the motion algorithm of the motion generator was 99.41%. In addition, the average error rate of target location prediction was 1.23% for 26 cases. Conclusion: We confirmed the applicability of our proposed target location prediction system for RGRT using natural respiratory volume. If additional clinical studies can be conducted, a more accurate prediction system can be realized without requiring respiratory pattern training.},
doi = {10.1118/1.4957074},
journal = {Medical Physics},
number = 6,
volume = 43,
place = {United States},
year = 2016,
month = 6
}

Purpose: AAPM TG-76 report advises lung patients experiencing tumor motion >5mm to use some form of motion management with even smaller limit for complex/special procedures like SBRT. Generally, either respiratory gating or abdominal compression is used for motion management. In this retrospective study, we are using an innovative index, Volumetric Indices (VI) = (GTVnn AND GTV{sub 50+}Xmm)/(GTVnn) to quantify how much of the tumor remains within 1, 2, and 3mm margins throughout the breathing cycle using GTV{sub 50+}Xmm margin on GTV{sub 50}[nn=0,10,20,…90]. Using appropriate limits, VI can provide tumor motion information and to check if RPM gates could have beenmore » used in conjunction with abdominal compression to better manage tumor motion. Methods: 64 SBRT patients with a total of 67 lung tumors were studied. 4DCT scans were taken, fully capturing tumor motion throughout the 10 phases of the breathing cycle. For each phase, Gross Tumor Volume (GTV) was segmented and appropriates structures were defined to determine VI values. For the 2mm margin, VI values less than 0.95 for peripheral lesions and 0.97 for central lesions indicate tumor movement greater than 4mm. VI values for 1mm and 3mm margins were also analyzed signifying tumor motion of 2mm & 6mm, respectively. Results: Of the 64 patients, 35 (55%) had motion greater than 4mm & could have benefited from respiratory gating. For 5/8 (63%) middle lobe lesions, 21/27 (78%) lower lobe lesions, and 10/32 (31%) upper lobe lesions, gating could have resulted in smaller ITV. 32/55 (58%) peripheral lesions and 4/12 (33%) central lesions could have had gating. Average ITV decreased by 1.25cc (11.43%) and average VI increased by 0.11. Conclusion: Out of 64 patients, 55% exhibited motion greater than 4mm even with abdominal compression. Even with abdominalcompression, lung tumors can move >4mm as the degree of pressure which a patient can tolerate, is patient specific.« less

Purpose: In spot-scanning proton therapy, the interplay effect between tumor motion and beam delivery leads to deterioration of the dose distribution. To mitigate the impact of tumor motion, gating in combination with repainting is one of the most promising methods that have been proposed. This study focused on a synchrotron-based spot-scanning proton therapy system integrated with real-time tumor monitoring. The authors investigated the effectiveness of gating in terms of both the delivered dose distribution and irradiation time by conducting simulations with patients' motion data. The clinically acceptable range of adjustable irradiation control parameters was explored. Also, the relation between themore » dose error and the characteristics of tumor motion was investigated.Methods: A simulation study was performed using a water phantom. A gated proton beam was irradiated to a clinical target volume (CTV) of 5 Multiplication-Sign 5 Multiplication-Sign 5 cm{sup 3}, in synchronization with lung cancer patients' tumor trajectory data. With varying parameters of gate width, spot spacing, and delivered dose per spot at one time, both dose uniformity and irradiation time were calculated for 397 tumor trajectory data from 78 patients. In addition, the authors placed an energy absorber upstream of the phantom and varied the thickness to examine the effect of changing the size of the Bragg peak and the number of required energy layers. The parameters with which 95% of the tumor trajectory data fulfill our defined criteria were accepted. Next, correlation coefficients were calculated between the maximum dose error and the tumor motion characteristics that were extracted from the tumor trajectory data.Results: With the assumed CTV, the largest percentage of the data fulfilled the criteria when the gate width was {+-}2 mm. Larger spot spacing was preferred because it increased the number of paintings. With a prescribed dose of 2 Gy, it was difficult to fulfill the criteria for the target with a very small effective depth (the sum of an assumed energy absorber's thickness and the target depth in the phantom) because of the sharpness of the Bragg peak. However, even shallow targets could be successfully irradiated by employing an adequate number of paintings and by placing an energy absorber of sufficient thickness to make the effective target depth more than 12 cm. The authors also observed that motion in the beam direction was the main cause of dose distortion, followed by motion in the lateral plane perpendicular to the scan direction.Conclusions: The results suggested that by properly adjusting irradiation control parameters, gated proton spot-scanning beam therapy can be robust to target motion. This is an important first step toward establishing treatment plans in real patient geometry.« less

Purpose: In respiratory-gated radiation therapy, a baseline shift decreases the accuracy of target coverage and organs at risk (OAR) sparing. The effectiveness of audio-feedback and audio-visual feedback in correcting the baseline shift in the breathing pattern of the patient has been demonstrated previously. However, the baseline shift derived from the intrafraction motion of the patient's body cannot be corrected by these methods. In the present study, the authors designed and developed a simple and flexible system. Methods: The system consisted of a web camera and a computer running our in-house software. The in-house software was adapted to template matching andmore » also to no preimage processing. The system was capable of monitoring the baseline shift in the intrafraction motion of the patient's body. Another marker box was used to monitor the baseline shift due to the flexible setups required of a marker box for gated signals. The system accuracy was evaluated by employing a respiratory motion phantom and was found to be within AAPM Task Group 142 tolerance (positional accuracy <2 mm and temporal accuracy <100 ms) for respiratory-gated radiation therapy. Additionally, the effectiveness of this flexible and independent system in gated treatment was investigated in healthy volunteers, in terms of the results from the differences in the baseline shift detectable between the marker positions, which the authors evaluated statistically. Results: The movement of the marker on the sternum [1.599 {+-} 0.622 mm (1 SD)] was substantially decreased as compared with the abdomen [6.547 {+-} 0.962 mm (1 SD)]. Additionally, in all of the volunteers, the baseline shifts for the sternum [-0.136 {+-} 0.868 (2 SD)] were in better agreement with the nominal baseline shifts than was the case for the abdomen [-0.722 {+-} 1.56 mm (2 SD)]. The baseline shifts could be accurately measured and detected using the monitoring system, which could acquire the movement of the marker on the sternum. The baseline shift-monitoring system with the displacement-based methods for highly accurate respiratory-gated treatments should be used to make most of the displacement-based gating methods. Conclusions: The advent of intensity modulated radiation therapy and volumetric modulated radiation therapy facilitates margin reduction for the planning target volumes and the OARs, but highly accurate irradiation is needed to achieve target coverage and OAR sparing with a small margin. The baseline shifts can affect treatment not only with the respiratory gating system but also without the system. Our system can manage the baseline shift and also enables treatment irradiation to be undertaken with high accuracy.« less

Purpose: To investigate the clinical utility of on-line verification of respiratory gated VMAT dosimetry during treatment. Methods: Portal dose images were acquired during treatment in integrated mode on a Varian TrueBeam (v. 1.6) linear accelerator for gated lung and liver patients that used flattening filtered beams. The source to imager distance (SID) was set to 160 cm to ensure imager clearance in case the isocenter was off midline. Note that acquisition of integrated images resulted in no extra dose to the patient. Fraction 1 was taken as baseline and all portal dose images were compared to that of the baseline,more » where the gamma comparison and dose difference were used to measure day-to-day exit dose variation. All images were analyzed in the Portal Dosimetry module of Aria (v. 10). The portal imager on the TrueBeam was calibrated by following the instructions for dosimetry calibration in service mode, where we define 1 calibrated unit (CU) equal to 1 Gy for 10×10 cm field size at 100 cm SID. This reference condition was measured frequently to verify imager calibration. Results: The gamma value (3%, 3 mm, 5% threshold) ranged between 92% and 100% for the lung and liver cases studied. The exit dose can vary by as much as 10% of the maximum dose for an individual fraction. The integrated images combined with the information given by the corresponding on-line soft tissue matched cone-beam computed tomography (CBCT) images were useful in explaining dose variation. For gated lung treatment, dose variation was mainly due to the diaphragm position. For gated liver treatment, the dose variation was due to both diaphragm position and weight loss. Conclusion: Integrated images can be useful in verifying dose delivery consistency during respiratory gated VMAT, although the CBCT information is needed to explain dose differences due to anatomical changes.« less