Com-Power CDN-AF4E is a part of the series of Coupling/Decoupling Networks designed specifically for testing product for conducted immunity per IEC / EN 61000-4-6.
The CDN-AF4E is designed for testing products that uses four wire unscreened cables for data communication. It has four 2 mm shrouded banana sockets for both EUT and AE power connection. The CDN-AF4E can handle up to 5 Amps of current.
The RF disturbance signal coupling port is female BNC. It can handle up to 40V of RF Input Voltage. The bottom surface of the CDN is not painted for easy and effective grounding.
CDNs are individually calibrated. The CDN-AF4E fully complies with the requirements contained in the IEC 61000-4-6 and CISPR 16-1-2.
All CDNs can be purchased at BRL Test separately or as part of the CIS series conducted immunity test system. This is a pre-packaged solution that includes an ACS series power amplifier and all accessories required for the test.
BRL Test is your EMC-EMI Headquarters. Call 866-275-8378 Today!

Product Name: Coupling Decoupling Network (CDN)

Compliant Test Standards: IEC / EN 61000-4-6

Application: Four wire unscreened, unbalanced Cables

Frequency Range: 150 kHz to 230 MHz

RF Input Voltage: 40 V (Max)

RF Input Connector: 50 Ω BNC (Female)

Voltage Rating: 310 V AC / 440 V DC (Line to Ground)

Current Rating: 5 Amp (Max)

AE and EUT Connections: 2 mm shrouded banana sockets

Common Mode Impedance:

150 kHz - 26 MHz: 150Ω ± 20Ω

26 MHz - 80 MHz: 150Ω + 60Ω / – 45Ω

80 MHz - 230 MHz: 150Ω + 60Ω / – 60Ω

Voltage Division Factor: 9.5 dB +3 / -1

Decoupling of Common Mode Disturbance: ≥ 40dB (EUT/AE)

Dimensions: 6 x 6 x 13 inches or 15.2 x 15.2 x 33 cm

Weight: 5 lbs. or 2.3 kg

Accessories Available from Com-Power for setting test levels and running the test: