(C)
A licensee shall use the dosimetry system
described in paragraph (A) of rule
3701:1-58-60 of the
Administrative Code to measure the output.

(D)
A licensee shall make full calibration
measurements required by paragraph (A) of this rule in accordance with
published protocols accepted by nationally recognized bodies.

(E)
In addition to the requirements for full
calibrations for low dose-rate remote afterloader units in paragraph (B) of
this rule, a licensee shall perform an autoradiograph of the source(s) to
verify inventory and source(s) arrangement at intervals not exceeding one
quarter.

(F)
For low dose-rate
remote afterloader units, a licensee may use measurements provided by the
source manufacturer that are made in accordance with paragraphs (A) to (E) of
this rule.

(G)
A licensee shall
mathematically correct the outputs determined in paragraph (B)(1) of this rule
for radioactive decay at intervals consistent with one
per cent
radioactive decay.

(H)
Full
calibration measurements required by paragraph (A) of this rule and radioactive
decay corrections required by paragraph (G) of this rule must be performed by
the authorized medical physicist.

(I)
A licensee shall retain a record of each
calibration in accordance with rule
3701:1-58-94 of the
Administrative Code.