Description

ONEday 4-section electric stretcher for long hospital stays with adjustment system on 2 couples of compass-leverages, ABS mattress sections, translating backrest, extendible mattress platform, electric trendelenburg and reverse-trendelenburg adjustments. Base frame in epoxy-coated steel pipes featuring wheels with breaks at its extremities.
200 mm diameter wheels with fifth wheel at the centre of the base frame, activated through the brake-unlock pedal.
Intermediate frame with six-strut high-containing folding side rails (H 38 cm) and metal accessory-holder bars (for drainage) on both sides, with 2 mobile sliding hooks in plastic material.
Equipped with proper housings at the head side corners for patient-lifting pole, I.V. pole and dedicated accessories. Height adjustment performed by 2 low-voltage electric actuators (24 V) for a maximum safe lifting load of 260 kg.
Electric controls on SATELLITE push-button control panel as a standard, on a flexible support enabling the adjustment of the mattress sections, of the height and of the trendelenburg and reverse trendelenburg positions.
When the push-button control panel remains unused for a few minutes, it goes automatically into auto switch-off mode, cutting down power consumption. Enhanced battery-pack, with day-long operating time, dedicated battery charger for a full and fast recharge. A sound alarm system warns in case of connection to the electrical grid and wheels not blocked. Backrest provided with roto-translating movement: when raised, it shifts backwards up to 11 cm.....

*Prices are pre-tax, exclude delivery charges and customs duties, and do not include additional charges for installation or activation options. Prices are indicative only and may vary by country, with changes in raw materials prices or exchange rates.