OilLab 690 - Tag

CE marked

Subject

Flash point on petroleum products having a flash point between ambient temperature and +93°C. Suitable for flash point detection on different substances and (NO) waste materials, solvents…

Measuring Tag Principle

The sample is warmed up according to the methods. When the sample reaches the selected test temperature, the shutter is opened and the ignition system introduces itself automatically. If the flash point is reached, the detection is done by an ionisation detector. If not, the shutter closes again and the sample continues to warm up until the next test temperature.

Measuring Tag Devices

Measurement of the Flash Point detected by an ionisation detector

Testing unit equipped with two ignition systems

Electrical system or flame exposure device

Built-in barometric sensor with automatic barometric correction of results executed by the software

Measuring Temperature Probe

Platinum resistance PT100 class A

Measuring Parameters

Temperatures: in °C

Measuring range: -50°C … +100°C

Testing range: +9°C … +93°C as per test methods

Resolution: 0.06 °C

Accuracy: ± 0.1 °C

Repeatability / Reproducibility: as per standards methods or better

Software Features

User friendly interface

All analytical parameters recorded

Customizable analysis parameters and methods

Customizable results report

Printable graphs and results

Self-identification of the typology of the analysers connected

The software includes:

Analysis Menu

Standard method as per ASTM / IP / ISO / EN / DIN… norms of reference

Unknow sample

Audible alarm and displayed messages at the end of the analysis and in case of errors and/or malfunctions

Diagnostic Menu

Direct access to all analog, digital, inputs and outputs

Selectable value displaying: °C / Volt

Calibration Menu

Automatic calibration of each temperature probe

Last calibration date referred to each single probe displayed and relative data printable

Display of calibration diagram

Insertion of offset values

Standard and advanced calibration modes

Data Utilities

Fields for introduction of operator and product name

Archive viewer for files recall

All analysis stored in Excel® compatible format

Storage capacity for more than 60’000 analysis

LIMS compatible

Integrated Touch Screen Panel PC

TFT/LCD 8"

Resolution 1024 × 768, 16.2 M colours

USB port for connection to an external printer and/or external PC

Storage capacity for more than 60'000 analysis

Test Cup

The cup is made of chromium plated brass provided with high temperature resistant handle