The Fluke 726 is a precise and powerful, yet easy-to-use field calibrator. It features broad workload coverage, calibration power, and unsurpassed accuracy needed by process professionals. It measures and sources almost all process parameters to calibrate almost anything. Use it to test sensors and valves and test and calibrate transmitters. The 726 interprets results without the help of a calculator and stores measurement data for later analysis.

If you want documenting and HART communication built into your calibrator, consider the Fluke 754 Documenting Process Calibrator with HART.

Measurement Accuracy

Voltage DC

30.000 V

0.01% + 2 mV (upper display)

20.000 V

0.01% + 2 mV (lower display)

90.00 mV

0.01% + 10 μV

-10.00 mV to 75.00 mV

0.025% + 1 count (via TC connector)

Current DC

24.000 mA

0.01%+ 2 mA

Resistance

0.0 to 4000.0 Ω

± 0.015%

Frequency

2.0 to 1000.0 CPM

0.05% + 0.1 cpm

1.0 to 1000.0 Hz

0.05% + 0.1 Hz

1.00 to 15.00 kHz

0.05% + 0.01 kHz

Sensitivity

1 V peak-to-peak-minimum

Pressure

Accuracy from 0.025% of range using any of 29 pressure modules (for detailed specifications refer to pressure modules in options and accessories). Modules available for differential, gauge, vacuum, absolute, dual and high pressure.