Description

The PD7000 is a member of the ProVu series of 1/8 DIN digital panel meters that accepts a direct temperature input from a wide range of temperature measurement devices (type J, K, T, E, R, S, B, N, and C thermocouples and 100 or 1000 Ω platinum, 10 Ω copper, 120 Ω nickel RTDs). It displays the measured temperature in either degrees Fahrenheit or degrees Celsius on a dual-line, 6-digit display that is available with optional Sunbright® sunlight readable LEDs. The PD7000 temperature indicator includes automatic cold junction compensation for thermocouples and the ability to average up to ten (10) RTD sensors. Like all ProVu meters, the PD7000 temperature meter includes UL / C-UL listing and CE marking, a NEMA 4X front, and AC or DC power options. ProVu meters can be equipped with up to four internal relays, a 4-20 mA output and Modbus® RTU serial communications. The PD7000 can be programmed via the front panel pushbuttons or free MeterView® Pro software.

Mounting Dimensions

Connections

MeterView® Pro Software

MeterView® Pro software is designed for use with ProVu, ProtEX-MAX, or Helios Series meters and allows users to remotely program, monitor, and datalog using a PC. Remote programming allows for all available meter settings to be programmed through an easy, user-friendly interface. The data acquisition feature allows the user to gather readings from a meter at user-selected intervals and generate charts using common tools like Microsoft® Excel. A linearization setup function is also included. With this utility the user can configure up to 32 linearization points and upload them to the meter. All configuration data can be saved to a file for future use. This software is accessible via the onboard USB connection on all Helios large display meters, ProVu panel meters, and ProVu-based ProtEX-MAX explosion-proof meters produced since 6 September 2016 (firmware version 4.0 or higher). In order for meters produced prior to 6 September 2016 (firmware version 3.1 or lower) to establish digital communications with a PC, a serial communications adapter is required. For an RS-232 connection, use a PDA1232adapter.To determine the software version of a meter:

Go to the Diagnostics menu () and press Enter button.

Press Up arrow button and scroll to Information menu (Info).

Press Enter to access the software number (), version (), and serial number () information. Write down the information as it is displayed. Continue pressing Enter until all the information is displayed.