LISST-100X

The LISST-100X is a submersible multi-parameter system for in-situ observations of particle size distribution and volume concentration. It also records the optical transmission, pressure and temperature. The instrument is fully self-contained with internal battery and data-logger.

The LISST-100X holds calibration for life.

In 2016, we introduced the LISST-200X as a generational advance on the LISST-100X. It is smaller, smarter, and goes deeper. The LISST-100X is being phased out.

Rentals: The LISST-100X can be rented. The minimum rental period is one (1) month. Please contact Sequoia for more…

Two size-ranges are available:1.25-250 microns (Type B) and 2.5-500 microns (Type C). Standard memory size is 128MB, suitable for storing approximately 1,600,000 size distributions, with option to expand to 1GB, or 12,800,000 size distributions. The depth range is 300 m, but a custom version (the LISST-DEEP) is available for deep-ocean environments (2500 m depth rating).

At the heart of the instrument is a collimated laser diode and a specially constructed annular ring detector (see Application Note L001). Scattering at 32 angles is the primary information that is recorded by the ring detector. This primary measurement is mathematically inverted to get the size distribution, and also scaled to obtain the volume scattering function (VSF). The size distribution is presented as concentration with units of micro-liters per liter (µl/l) in each of the 32 log-spaced size bins. Date and time, optical transmission, water depth and temperature are recorded as supporting measurements.

As an option, a modified endcap can be constructed, which makes it possible to connect either a Sea-Bird MicroCat CTD, a Campbell OBS 3+ or a Turner Designs Cyclops-7 fluorometer. Designed for extended field application, the LISST-100X is a self-contained battery-powered instrument that can be deployed on a tripod, used in a profiling mode, or moored. A range of accessories exists that makes it possible to use the LISST-100X as a laboratory particle sizer. A folded version and a short body version without internal battery also exists. The folded version is designed to fit in a Chelsea shuttle, while the short body version is designed for use with external power.

A large range of accessories has been developed for the LISST-100X, please view our accessories page for a detailed listing.

The LISST-100X and the LISST-Deep were both used intensively after the Deepwater Horizon accident in April 2010 to detect oil droplets and their size and concentration in the waters of the Gulf of Mexico. The EPA required the use of LISSTs for measurement of oil droplet size and concentration in the dispersed oil.

Accessories

BIOBLOCK – a mechanical anti-fouling accessory for the LISST-100X and LISST-25X. Sliding copper shutters in front of the windows limit fouling to a minimum. BIOBLOCK is controlled by the LISST-100X or LISST-25X to open before a measurement takes place and close again after the measurement.
LISST-100X/-STX/ Alkaline Battery Pack
LISST-100X/LISST-STX External AC Power Supply
Small Volume Flow Through Chamber for LISST-100X
LISST-100X Replacement Zscat Chamber
Replacement 2-meter Communications Cable
High Performance Conductivity and Temperature Sensor Modification
Custom Optical paths
Clamps
Replacement Instrument Stands
Custom LISST-100X with short pressure housing
Custom LISST-100X with folded pressure housing
Frames and Cages for LISST deployment
LISST-100X External Battery Pack