Kikusui PCR2000MA AC Power Supply / Frequency Converter.

The Kikusui PCR2000MA is a 2000VA compact variable frequency AC Power Supply. Weighing only 32kg's, the PCR2000MA is crammed full of features including AC, DC, AC+DC output and memory functions. The output voltage can be set anywhere between 0-310VAC and the frequency from 40-500Hz allowing the user to easily set different worldwide output voltages, including 120V 60Hz, 230V 50Hz and 115V 400Hz for aviation applications. Voltage and frequency can also be adjusted when the output is on for margin testing.
You can also measure the voltage, current and power of the AC and DC output and, using software, apparent power, reactive power, power factor, crest factor and current peak hold.

Compact AC power source employing the PWM inverter system

AC Output: 0V～155V / 0V～310V · 40Hz～500Hz

DC Output: ± 0V~219V / ± 0V~438V

Low distortion synthesised output waveform

Maximum peak current 3 times the rated RMS

Sensing function

The PCR2000MA is equipped with a USB and LAN interface and comes complete with web browser software control as standard. Instrument Drivers are also available.

PCR-MA Lineup

PCR500MA NEW Single phase 500 VA, compact AC power supply PCR1000MA NEW Single phase 1 kVA, compact AC power supplyPCR 2000MA NEW Single phase 2 kVA, compact AC power supplyPCR 4000MA NEW Single phase 4 kVA, compact AC power supply

Performance / Functions

Output characteristic : AC mode The output voltage variable range is 0 V to 155 V / 0 V to 310 V (2 ranges) and the frequency variable range is 40 Hz to 500 Hz, a wide range corresponding to the nominal voltage (single phase) of different countries. It can also be used for testing power supply equipment and actuators mounted on aircraft and ships.

Output characteristics : AC+DC mode AC+DC Mode is a function that superimposes a DC voltage on an AC voltage or an AC voltage on a DC voltage, it can be used through the LAN (including web browser) USB or the optional GPIB interface.

Measurement functionYou can measure voltage, current and power of both AC and DC output. You can display true rms value and average values (direct current) of output voltage and true rms value, peak value and average value (direct current) of output current. It is also possible to measure apparent power (VA), reactive power (VAR), power factor (PF), crest factor (CF), and peak hold current by using the communication interface.

Output ON phase setting functionOutput ON phase setting function In AC mode, you can set the OUTPUT ON phase between 0 and 359%.OUTPUT off The output turns off at the zero phase phase angle.

Memory functionThree sets of output voltage, frequency set value and limit value can be stored in the main body memory, so it is convenient for rapid changes in voltage/frequency. In addition, up to 11 pairs of memories can be used with communication commands only.

Maximum peak currentMaximum peak current up to three times the rated maximum current (effective value) can be output for a capacitor input filtered devices.