CNS2054 3479 JIS-L1018, P8112, ASTM-D2210, TAPPI T403ts-63 This tester is used to determine the bursting strength of paper and corrugated board. It is an essential piece of equipment for quality control.

Model

HT-8020A

HT-8020B

Capacity

10 Mpa

20/50, 50/100 kg/sq.cm(selectable)

Precision of display

0.1kg/sq.cm

0.1kg/sq.cm

Display

Digital indicator

3 pcs.of pressure gauge

Motor

1/4 or 1/3 HP, 50/60HZ

1/4 or 1/3 HP, 50/60HZ

Power source

110V or 220V

110V or 220V

Dimensions (approx)

53x36x63 cm

64x35x50 cm

Net weight (approx)

110 kg

100 kg

Features

Digital display

Clamping pressure gauge

Clamping pressure gauge

Speed of pressurization

High pressure: 170 ±20 ml/min

High pressure: 170±20 ml/min

Low pressure: 95±10 ml/min

Low pressure: 95±10 ml/min

Speed of depressurization

High pressure: 340±20 ml/min

High pressure: 340±20 ml/min

Low pressure: 190±10 ml/min

Low pressure: 190±10 ml/min

Diameter of clamping

High pressure

High pressure

Upper: 31.5±0.05 mm dia

Upper: 31.5±0.05 mm dia

Lower: 31.5±0.05 mm dia

Lower: 31.5±0.05 mm dia

Low pressure:

Low pressure:

Upper: 30.48±0.03 mm dia

Upper: 30.48±0.03 mm dia

Lower: 31.75±0.25 mm dia

Lower: 31.75±0.25 mm dia

Paper, Board & Packaging HT-8020A Bursting Strength Tester

CNS2054 3479 JIS-L1018, P8112, ASTM-D2210, TAPPI T403ts-63 This tester is used to determine the bursting strength of paper and corrugated board. It is an essential piece of equipment for quality control.

CNS2054 3479 JIS-L1004, L1018, L1031, K6328, P8131, P8112, ASTM-D2210, TAPPI T403ts-63 This tester is used to determine the bursting strength of paper and corrugated board. It is an essential piece of equipment for quality control.

Model

HT-8321

Features

Automatic

Capacity

2/5, 5/10 Mpa (selectable)

Speed of pressurization

High pressure: 170±20 ml/min

Low pressure: 95±10 ml/min

Precision of display

0.01 Mpa (0.1 kg/sq.cm)

Speed of depressurization

High pressure: 340±20 ml/min

Low pressure: 190±10 ml/min

Display

3 pcs. pressure gauge

Power source

110V or 220V

Dimensions (approx)

92x34x72 cm

Net weight (approx)

180 kg

Diameter of clamping

High pressure:

Upper: 31.5±0.05 mm dia.

Lower: 31.5±0.05 mm dia.

Low pressure:

Upper: 30.48±0.03 mm dia.

Lower: 31.75±0.25 mm dia.

Standard accessories

Glycerin

1 bottle

Rubber diaphragm

5 pcs

Standard aluminium foil

20 pcs

Driving wrench

1 pc

Paper, Board & Packaging HT-8047 Aging Oven Tester

JIS-K630, K-7212, ASTM D-2436, E-145, JTTC SPEC.-151 CNS 3556 The oven is used for degradation study of plastics & rubber when subjected to heat in a rotational condition for a specified time. One day test at 70°Cis theoretically equivalent to a six-month usage in normal environmental condition. The machine is equipped with a motor-driven rotary plate. Hot air circulation will ensure even temperature distribution.

Type

A

B

Inner dimension

40x40x45 cm

50x50x60 cm

Heat

1.8 KW

3.2 KW

Temperature

RT-200°C (or 300°C)

RT-200°C (or 300°C)

Rotating speed

8-10 rpm

8-10 rpm

Motor

1/4 HP

1/2 HP

Net Weight

80 kg

120 kg

Dimension

84x54x100 cm

95x65x118 cm

Paper, Board & Packaging HT-8056 Vibration Testing Machine

This machine is used for conducting the vibration test of mechanical, electronics, electrical parts, telecommunications, computer and auto parts. This application is available for checking the samples for resisting the effect while under transportation or durability testing.

Model

HT-8056

Vibration direction

up/down & left/right

Cycle

300-3300, 300-1500 cpm

Vibration speed

600-3300 cpm (adjustable)

Vibration capacity

0-7 mm adjustable, normally 1.5 or 2.0 mm

Load capacity

60 kgs

Testing table

50x60 cm

Timer

0-6 hours (adjustable, two sections)

Changing Direction

dual direction (up/down & left/right)

Motor

Two 2HP DC stepless motors

Power

AC220V 3ø, 220V/380V/415V

Dimensions (WxDxH)

105x95x55 cm

Net Weight

300 kgs

Paper, Board & Packaging HT-8056A Series Vibration Tester

This machine is used for conducting the vibration test of mechanical, electronics, electrical parts, telecommunications, computer and auto parts. This application is available for checking the samples for resisting the effect while under transportation or durability testing.

Model

HT-8056A

Vibration direction

up/down or left/right

Cycle

300-3300, 300-1500 cpm

Vibration speed

600-3300 cpm (adjustable)

Vibration capacity

0-7 mm adjustable, 1.5 or 2.0 mm normal

Load capacity

60 kgs

Testing table

50x60 cm

Timer

0-6 hours (adjustable, one section)

Changing Direction

up/down or left/right

Motor

One 2HP DC stepless motor

Power

AC220V 3ø, 220V/380V/415V

Dimensions (WxDxH)

70x80x55 cm

Net Weight

200 kgs

Paper, Board & Packaging HT-8136 Daylight Tester

This tester is made in accordance with C.I.E. and certified the minimum interferance of the ultra light.

Model

HT-8136

Test chamber (WxDxH)

70x53x35 cm

Net weight

26 kgs

Dimensions (WxDxH)

72x57x56 cm

Light source

D Light: North Sky Daylight (Standard Light Source)

A Light: Incandescent, Flame Light

UV Light: Ultra Violet Lamps

F Light: High Energy Fluorescent Lamp

Paper, Board & Packaging HT-8140 Package Drop Tester

CNS-2999 In the transportation process, packed products might be unavoidably dropped or accidentally impacted with hard substance. To prevent further damages in advance, this tester is designed to test and find out the ultimate solution for the package products with a purpose to reduce loss during transportation.

Model

HT-8140

Load weight

60 kgs

Test height

40-150 cm

Test space

60x60x60 cm

Length of single arm bracket

30 cm

Power

3ø, 220V/380V/415V

Dimension (WxDxH)

157x100x209 cm

Weight

200 kgs

Paper, Board & Packaging HT-8140A Dual-Arm Package Drop Tester

During the transportation process, packed products unavoidably being dropped or accidentally impacted with hard substance. To prevent further damages in advance, this tester is designed to test and find out the ultimate solution for the package products with a purpose to reduce loss during transportation. Compared with our previous version of single arm design, this twin-arm type is easier and more convenient to use.

Model

HT-8140A

Drop height

40 - 176 cm

Test load

60 kg

Test space ( W x D x H )

( 6 - 108 ) x 76 x ( 40 - 150 ) cm

Supporting area

30 x 76 cm

Size of base board

120 x 140 cm

Auto control up / down for drop height

adjustable

Motor control system

1 set

Positioning of drop height

setup 1 set

Height indication ruler

1 set

Support arm

auto control ( easier than manual control )

Power

3 ø, 380 / 415 V, 50 / 60 Hz

Dimensions (W x D x H )

Frame 228 x 140 x 225 cm

Control 30 x 40 x 120 cm

Base d imensions

120 x 140 cm

Weight

120 x 140 cm

Paper, Board & Packaging HT-8172 90° Adhesion-Tearing Strength Tester

This tester measures the adhesiveness of adhered materials such as copper foil, aluminum foil, tin foil or even paper labels. The lower layer is gripped firmly, while the upper layer is pulled laterally 90° from vertical. The force is read through a push-pull gauge.

Recently, for the convenience of most operators, we modify this machine to be speed adjustable, so that the operator could use different test speed in order to test his samples.

Micro-computer compress tester is applied to test the packing carton box or the container like commodities how heavy load it can afford. To determine the crack or collapse arose due to a positive, consistent pressure adding to the testing particles.

Model

8506A

8506B

8506C

Capacity

20, 50kN (2000,5000 kgs)

Sensing method

High precision load cell

Indicator

HT-9635A (16 bits)

Resolution

1/20000

Accuracy

within ±1%

Compression area

100x100x100cm

150x150x150cm

200x200x200cm

Test speed

13mm/min (5-100 mm/min)

Power

3 phase, 220V, 50/60Hz

Dimension (WxDxH)

186x100x203 cm

New weight

Apr. 480 kgs

Note: This tester can be equipped with computer system for data acquisition at extra charge.

Paper, Board & Packaging HT-8181 Elmendorf Tearing Tester

TAPPI-T414, JIS P-8116, CNS-B53 To test tearing strength of fabric, paper, PVC material etc. Specimen are being slightly splitted. One side is firmly gripped and the other is being gripped and moved forward to effect tearing action scale pointer indicates tearing strength.

Model

HT-8140A

Type

A

B

C

Capacity (gf)

0~100 or 0~200

0~1600 or 0~3200

0~6400

Sample size

63±0.15x76±2.0 mm

Dimensions (approx)

50x25x37 cm

60x40x50 cm

80x55x75 cm

Net Weight (approx)

15 kgs

35 kgs

60 kgs

Referent Stardards

TAPPI-T414,ISO-6383

JISP-8116,CNS-1355

* Optional: Micro-computer indicator is available upon request. (HT-8181D)

Paper, Board & Packaging HT-8380 Micro-Computer Ring Cursh Tester

Owing to the transportation needs, stacks of the carton boxes being piled upward, the endurance of such heavy load is determined by the composition of the paper board. To tell the difference of the burden strength, Micro-computer Ring Crush Tester is inevitable, useful and helpful.

To cut the paper board which is suitable for testing purposes ; otherwise irregular or misuse of the improper paper board may result in an incorrect test. There are two models : The parallel cutter and the corner cutter.

Model

HT-8380A

Adjustable space

30, 50, 60, 80, 100, 150 mm

Dimensions

48x62x16 cm (approx)

Net. Weight

25 kgs (approx)

With clear graduation, it cuts the paper board in precision and in a straight parallel line.

To cut the paper board which is suitable for testing purposes ; otherwise irregular or misuse of the improper paper board may result in an incorrect test. There are two models : The parallel cutter and the corner cutter.

Model

HT-8380B

Specimen size

4"x2.5"

Dimensions (approx.)

23x20x25 cm

Net. Weight

5 kgs

It can cut off one corner of paper boards to facilitate the operation of vertical compression testing.

Paper, Board & Packaging HT-8380C Column Compression Testing Grip

This grip is used for chucking the upper and lower ends of sample paper boards to make sure specimen stands vertically, thereby facilitating the vertical compression testing of the paper boards. The grip should be used together with ring crush tester.

Model

HT-8380C

Formula

Max. force/50 (kgf/mm)

Dimension (approx. WxDxH)

17x11x7 cm

Net weight (approx.)

2 Kgs

Paper, Board & Packaging HT-8380H Compression Tester For Paper Cones

This tester is for testing the transverse compression strength of paper tubes, bobbins, paper cans, composite cans, paper cones and so on. It can also be used for testing taper tubes.

This cutter is round in shape and is used for cutting corrugated paper boards bearing a plane surface pressure of 38.5 x 38.5c., thereby facilitating the testing of pressure resistance of the boards. It should be used together with ring crush tester. Size of sample : ( a ) 38.5 sq.cm ( b ) 100 sq.cm

Model

HT-8380D

Dia. after cutting

ø64 mm

ø112.8 mm

Area of Specimen

38.5 sq. cm

100 sq. cm

Dimensions

ø20cm, H20 cm

ø30cm, H20 cm

Weight

1.5 kgs

3.5 kgs

Paper, Board & Packaging HT-8380E Pin Adhesion Test / Join

This tester is applied to tear away the corrugated fiber boards with pins inserting into the flute of the fiber boards to test the adhesive strength between the flute and flat paper board of corrugated board.

Model

A

B

C

Specimen (mm)

50x150

Dia. Of pin (mm)

35±0.1

2.0±0.1

3.0±0.1

Pitch (mm)

17.6±0.1

12.0±0.1

15.0±0.1

Dimension (mm)

125x76x30

91x76x30

109x76x30

Weight (g)

500

350

400

Paper, Board & Packaging HT-8382 Spray Tester For Cardboard

CNS 3687, JIS P8137 In order to test the corrugated fiberboard under the precipitate conditions. Rain occasionally falls on paper when left outdoors. If the water is not immediately absorbed, water tracks are left behind. As a result standard test outcomes layout as 8 categories : R0,R2,R4,.....R10.

Model

HT-8382

Sample size

300x200 mm

Sample size

350x200 mm

Dropper capacity

50 c.c

Unit drop volume

0.1 ml each

Flow length

250 mm

Dimension

22x38x100 cm

Net weight

15 kgs

Distance between dropper and sample

10 mm

Slant degree of testing table

45 degrees

Paper, Board & Packaging HT-8383 Water Absorption Tester

Dropping 50 ~ 100ml. of distilled water directly on the specimen, after a certain period of time, the water is absorbed by the paper. This is a way to test a paper's texture and absorption rate.

Model

HT-8383

Internal Diameter

112.8±0.1 mm

Height

25 mm

Thickness

6 mm

Sample Size

130x130 mm

Dimension

150x150x60 mm

Net Weight

2 kgs

Paper, Board & Packaging HT-8507 Puncture Tester

CNS 11484, JIS P8134, TAPPI T803 In combination with the use of bursting strength tester, Puncture Tester turns to be an indispensable instrument for inspecting the structural strength of the finished corrugated paper board. In fact, Puncture Tester is a dynamic way to test the quality of the paper board whereas the bursting strength tester is a static method. Both of them are interrelated and neither of them can be neglected. It is suitable for testing general paper boards, corrugated paper boards, plywood board, plastic boards and fiber boards. The punch head is a triangle pyramid to break the specimen

For examining the durability of wheelchairs in order to test the overall construction, or to understand the weakness of the wheelchair as an important reference point for further improvement and development.

Model

HT-9363SP

HT-9363AM

HT-9363M

Frequency range

5-100 Hz, 1 Hz/step

5-100 Hz, 1 Hz/step

5-60 Hz, 1 Hz/step

Versatile testing mode

1.Manual

1.Manual

Manual

2.Auto-Linea

2.Auto-Linea

Log

Octave

Free random

Cycle random

Testing program

9 sets

7 sets

Total cycle time

10-9999 min 1 min/step

10-9999 min 1 min/step

1-15 min 1 min/step

Sub-cycle step

9 sets

7 sets

Sub-cycle time

10-9999 sec 1 sec/step

10-9999 sec 1 sec/step

Display

16x1 Dot matrix LCD, 2 set

7 segment red display, 13 sets

7 segment red display, 3 sets

Setting key

24 pcs

5 pcs

8 pcs

Power requirement

AC 220 V±10% single phase, 50/60 Hz

Vibration Table

Type

S type

L type

Dimension (WxDxH)

450x450x250 mm

550x700x300 mm

Amplitude p-p

0.35-2.5 mm ±1.5%

0.35-2.5 mm ±1.5%

Max. acceleration

100Hz: 14G 60Hz: 10G

100Hz: 12G 60Hz: 9G

Max.loading

100 kgw

200 kgw

Paper, Board & Packaging HT-9363M Vibration Testing System

For examining the durability of wheelchairs in order to test the overall construction, or to understand the weakness of the wheelchair as an important reference point for further improvement and development.

Model

HT-9363SP

HT-9363AM

HT-9363M

Frequency range

5-100 Hz, 1 Hz/step

5-100 Hz, 1 Hz/step

5-60 Hz, 1 Hz/step

Versatile testing mode

1.Manual

1.Manual

Manual

2.Auto-Linea

2.Auto-Linea

Log

Octave

Free random

Cycle random

Testing program

9 sets

7 sets

Total cycle time

10-9999 min 1 min/step

10-9999 min 1 min/step

1-15 min 1 min/step

Sub-cycle step

9 sets

7 sets

Sub-cycle time

10-9999 sec 1 sec/step

10-9999 sec 1 sec/step

Display

16x1 Dot matrix LCD, 2 set

7 segment red display, 13 sets

7 segment red display, 3 sets

Setting key

24 pcs

5 pcs

8 pcs

Power requirement

AC 220 V±10% single phase, 50/60 Hz

Vibration Table

Type

S type

L type

Dimension (WxDxH)

450x450x250 mm

550x700x300 mm

Amplitude p-p

0.35-2.5 mm ±1.5%

0.35-2.5 mm ±1.5%

Max. acceleration

100Hz: 14G 60Hz: 10G

100Hz: 12G 60Hz: 9G

Max.loading

100 kgw

200 kgw

Paper, Board & Packaging HT-8381 Die Cutter

The die cutter is used to cut a specimen into a long narrow stripe, and then place the stripe specimen into a round feature jig ( or ring crush tester ) for the operation of testing procedures.