4. Spinnerets with various holders.
4.1 Single needle with holder
4.2 Coaxial spinneret: Coaxial spinneret can produce Hollow nanofibers/microfibers, Core/Sheath nanofibers/microfibers. Using the coaxial electro-spinning method, different characteristics from each polymer can be combined into one fiber.
4.3 Side by side nozzle
4.3 Triaxial spinneret.
4.4 Tri-channel spinneret.
4.5 Multi-nozzle spinnerets:
5. Light source: Special designed light source, LED, cold light source, very useful for observing the spinning process by naked eyes. 6. 6. Exhaust fan.
This device is equipped to cool down the equipment in working process, and help the solvent vaporization. Press the ventilation button on the control panel to start working. This device is equipped to cool down the equipment in working process, and help the solvent vaporization. Press the ventilation button on the control panel to start working.

8. Heater
The syringe pump should be put outside when heater is using, otherwise, the syringe pump will be damaged. The best working temp. is 45~50 degree C. Please increase the temp. slowly when heating. In winter, heater is quite useful for solvent volatilization.

Designed for electrospinning process, Compatible to work under high voltage. The flat collector placed on the platform can move under software control. This X-Y platform is quite useful for Near-field Electrospinning, NFES.

13. Auto-clean device for needles

14. Other accessories.
15. Humidity control

As the solvent vapor should be removed from the chamber continuously, the inside humidity of the chamber can’t be controlled directly. While the following methods can be used to adjust the humidity of the room, as the air in the room will be withdrawn into the chamber continuously, the inside humidity of the chamber can be controlled accordingly.
1. Using an Industrial dehumidifier to reduce the level of humidity in the room.
2. Using an Industrial humidifier to increase the level of humidity in the room.
3. Insert dry air to adjust the humidity in the chamber.
Besides, when it is wet day, the heater in the chamber can be used with exhaust fan together to adjust the humidity within some extend.

15. Note: Grounding equipment should be prepared by the user. Specifications and design are subject to change without notice.