The versatility of the IDS 440PM allows you to perform a number of tasks.
These range from basic weighing operations to more complex tasks such as controlling
batches, check-weighing, or bulk -loading.

The IDS 440 includes, 16 programmable instructions (macros), and provides you with the capability of designing the printed record to satisfy your specific requirements. Compact in size and weight, the system is
housed in a rugged panel mount stainless steel enclosure which can be panel mounted or placed in a control console. Gaskets are provided to seal the panel to the panel or console in which it will be mounted.

One inch high, seven segment LED characters make it easier to read from longer distance
than most digital indicators. Brightness can be adjusted to suit conditions. The
keyboard features large keys which give you tactile and audible confirmation of input.

From gross, tare and net weighing to sophisticated trip-level event monitoring.
Applications are unlimited for the user who is capable of writing a basic program. For those who prefer
turnkey products, Industrial Data Systems offers a constantly growing number of applications at no additional charge.

PROGRAM DESCRIPTION

The basic weighing mode features push-button zero, push-button tare and keyboard tare. The
display indicates the weight and scale status. The weight data can be transmitted to a
printer, scoreboard, computer, etc. Additional functions allow the user or scale
technician to program his indicator for his specific application using Scale Basic. This
is done by entering instructions into the IDS 440 using codes for addition, subtraction,
activating events, etc. An Event Monitor is included to provide user-definable setpoints
that trigger relay outputs. The IDS 440 monitors the weight on the scale and compares it
with the setpoint level. When the weight on the scale reaches the setpoint, a relay output
is activated. The setpoints are defined in an event record which defines how the set-point
is calculated, what relay to activate, and what other function to initiate. EZ Link, a
windows-based software program, is optionally available to enable programming on a PC and
uploading to the IDS 440.

Analog Output: Provides4-20ma, 0-20ma, 0-24ma, 0-5v, 0-10v outputs with 16-bit resolution for process control.
Relay Module: Housed inside the IDS440, the relay module accommodates eight optically-isolated, solid-state
relays. Six output and two input relays are standard but can be modified for eight outputs
and no inputs.

EZ Link: A windows software program which allows you to perform all configuration and application programming
on your computer and upload it to the IDS440. The IDS440 can download information to the
EZ Link as well. With EZ Link, you can configure scale parameters I/O ports, print
formats, passwords, and set-points. You program with "event -driven
programming". EZ Link includes an editor to facilitate writing the program and a
compiler to check for errors.