Cl II ORA System with VerifEye

ORA System with VerifEye, Catalog Number 8065998300 For use during intraocular lens surgery
Recall Number Z-3050-2017

REASON
Some ORA Carts have the potential to return an incorrect IOL power measurement during cataract surgery. This issue appears to have been caused by a software coding error that results in the lens coefficients for an IOL model being downloaded from the Alcon server in an incorrect order..