The set up consist of permeability cell, water reservoir and pressure lines, with glass bottles.

Consists of a metal cylinder with a ledge at the bottom for retaining the specimen, a flange at the top, removable cover plate and a funnel. A rubber gasket is seated in matching grooves, between the cell and the cover plate, to render the joint water tight. Internal working size of the cell will be 100mm dia X 100mm hight.

Water reservoir consists of a length of metal pipe, 50 to 100 mm in diameter and about 500 mm long. The reservoir fitted with a graduated side arm gauge-glass for admitting water and compressed air and a cock for drainage, bleeding and connection to the permeability cell.