Citable URI

As published

DOI

Keyword

Abstract

A field evaluation of two new dissolved-oxygen sensing technologies, the Aanderaa Instruments AS optode model 3830 and the Sea-Bird Electronics, Inc., model SBE43, was carried out at about 32-m water depth in western Massachusetts Bay. The optode is an optical sensor that measures fluorescence quenching by oxygen molecules, while the SBE43 is a Clark polarographic membrane sensor. Optodes were continuously deployed on bottom tripod frames by exchanging sensors every 4 months over a 19-month period. A Sea-Bird SBE43 was added during one 4-month deployment. These moored observations compared well with oxygen measurements from profiles collected during monthly shipboard surveys conducted by the Massachusetts Water Resources Authority. The mean correlation coefficient between the moored measurements and shipboard survey data was >0.9, the mean difference was 0.06 mL L−1, and the standard deviation of the difference was 0.15 mL L−1. The correlation coefficient between the optode and the SBE43 was >0.9 and the mean difference was 0.07 mL L−1. Optode measurements degraded when fouling was severe enough to block oxygen molecules from entering the sensing foil over a significant portion of the sensing window. Drift observed in two optodes beginning at about 225 and 390 days of deployment is attributed to degradation of the sensing foil. Flushing is necessary to equilibrate the Sea-Bird sensor. Power consumption by the SBE43 and required pump was 19.2 mWh per sample, and the optode consumed 0.9 mWh per sample, both within expected values based on manufacturers’ specifications.

Many hypotheses have been advanced to explain the formation of mixed layers in the ocean; the salt finger type of double-diffusive convection, in particular, has received much attention. Because of their uniquely
ordered ...

This report is intended to illustrate and provide recommendations for how ocean observing
systems of the next decade could focus on coastal environments using combined
satellite and in situ measurements. Until recently, ...

A new relative humidity and air temperature sensor, the Sensirion Model SHTl, has been thoroughly tested
by the Upper Ocean Processes (UOP) group at the Woods Hole Oceanographic Institution. One-minute
averages from two ...

All Items in WHOAS are protected by original copyright, with all rights reserved, unless otherwise indicated. WHOAS also supports the use of the Creative Commons licenses for original content.
A service of the MBLWHOI Library |
About WHOAS