Chroma 62100H-450 DC Sources Power Supplies

New Chroma 62100H-450

Chroma 62100H-450 Details

Chroma 62100H-450

Programmable DC Power Supply 450V/23A/10KW

Chroma's new 62000H Series of programmable DC power supplies offer many unique advantages for telecom, automated test system & integration, industrial, battery charge & simulation for hybrid cars and solar panel simulation. These advantages include high power density of 15KW in 3U, precision readback of output current and voltage, output trigger signals as well as the ability to create complex DC transient waveforms to test device behavior for spikes, drops, and other voltage deviations.

The 62000H Series includes 14 different models ranging from 5KW to 15KW, with current range up to 375A and voltage range up to 1000V. The 62000H can easily parallel up to ten units capable of 150KW with current sharing for bulk power applications, for example, battery bank simulation of 450V/150A/67.5KW for electric vehicle and military use.

There are 100 user programmable input status on the front panel for automated test application and life cycle ON/OFF test. In addition, the 62000H has a 16 bit digital control with bright vacuum fluorescent display readout. The 62000H series DC power supplies are very easy to operate either from the front panel keypad or from the remote controller via USB / RS232 / RS485 / APG (Standard) and GPIB & Ethernet (optional). Its compact size with 3U only can be stacked on a bench in a standard rack without any difficulty.

Another unique capability of the 62000H supplies is their ability to create complex DC transient waveforms. This capability allows devices to be tested to DC voltage dropouts, spikes and other voltage variations making them an ideal choice for aerospace device testing, inverter testing and other devices which will experience voltage interrupts. Applications include DC/DC converter & inverter voltage drop test, engine start-up simulation, battery automated charging, electronic product life cycle test, etc.

Key Features

Power range: 5KW

Voltage range: 0 ~ 40V

Current range: 0 ~ 125A

High power density (15KW in 3U)

Easy master / slave parallel & series operation up to 150KW

Precision V&I Measurements

High-speed programming

Voltage & Current slew rate control

Digital encoder knobs, keypad and function keys

Current sharing operation

Voltage ramp function (time range: 10 ms ~ 99 hours)

Auto sequencing programming:10 programs / 100 sequences

OVP, current limit, thermal protection

Standard analog programming interface

Standard USB / RS232 / RS485 interface

Optional GPIB / Ethernet interface

Remote output ON / OFF (I / P)

Remote sense line drop compensation

LabView and Labwindows

Solar array simulation function

Shade I-V curve simulation

I-V curve programming: 10 program / 100 I-V files

CE certified

Chroma 62100H-450 datasheet & specifications

Fill out the form below for more information about the Chroma 62100H-450