YC-02 mini uht sterilizer is manufactured exclusively with advanced design & technologies to meet requirements from universities&comma; institutes&comma; enterprises' R&D Dept&period;&comma; and dairy industry for simulating industrial manufacture and research in laboratory&period; Rated capacity can be from 20L&sol;h to 25L&sol;h&comma; completely duplication the industrial production&comma; which is adaptive to wide ranges of viscidity&comma; and be same with preparation of fibers and granules&comma; precisely mimic prepare&comma; homogeniser&comma; aging&comma; pasteurism&comma; fast sterilization under ultra-temperature temperature and aseptic filling&period; The machine workstation system is integrated with the CIP function online&comma; which can be configured optionally a homogeniser and a aseptic filling cabinet on your needs&period;Product ParametersPRODUCT PARAMETERSThe technological flow

&lpar;1&rpar;Tube preheating & sterilizing&colon; to sterilize the tube by superheated water and last for some time&comma; then chill out to normal temperature and stand by&period;

&lpar;2&rpar; Material sterilizing&colon; to ensure the sterilizing temp&period; & time according to experiment requirement&lpar;In line homogeniser can be added either upstream or downstream&rpar;&period;&lpar;3&rpar; Aseptic filling&colon; hot filling or normal temp&period; filling based on laboratory needs&period;&lpar;4&rpar; CIP cleanup &period;Material&srarr; hopper&srarr;screw pump&srarr;preheating tube&srarr;&lpar;homo&period;&rpar; &srarr;sterilizing tube&lpar;137~150ºC&rpar;&srarr;water-cooling&srarr;&lpar;homo&period;&rpar;&srarr;&lpar;&srarr;ice-cold&rpar; &srarr;aseptic fillingCharacteristics1&period; Rated capacity 20L&sol;H~25L&sol;H&comma; lowest sample 3L~5L&comma; handled easily and minimum sample required&period;2&period;Completely duplicating the industrial production&comma; with high accuracy and good reproducibility&comma; the data and sample obtained through this machine indicate the industrial production and many terms and conditions can be adopted or referenced directly&period;3&period;YC-02 mini uht sterilizer workstation with selected system components &comma;all major4&period;components are housed within a stainless steel cabinet&comma; as standard two electrically5&period;heated hot water recirculators&comma;no steam required6&period;PLC controller&comma; LCD touch panel control fascia with flow path of system&period;7&period;Two modes of run&colon; Automatic-mode&comma; and Eye-monitored mode for the purpose of easily controlling experimental process &period;8&period;As standard a progressive screw pump&lpar;made in Germany&rpar; with CIP is used to avoid secondary pollution&period;9&period;YC-02 mini uht sterilizer is mobile&comma; and can be positioned in a wet or dry area&period;10&period;Non-material warning functions&period;11&period;Computer interface with real time temperature recording &comma;the trial data can be used for the paper directly with excel file&period;12&period;Each heating section is individually controlled to enable a wide setting of product temperatures&comma;the product is heated by pressurised hot water recirculators which are PID controlled for greater stability and accuracy&lpar;the precision of sterilizing temperature is ±0&period;5ºC&rpar; &period; The cooling is in 1 or 2 stages dependant on the final cooling temperature required&period;Technical parameter1&period;Rated capacity from 20 to 25 Ltr per hour &lpar;Highly available by your demand&rpar;&period;2&period;Heating to 148 deg C at the flow rates stated&period;3&period;Cooling to below 5 deg C at the flow rates stated&period;4&period;Temp&period; Precision&colon;±0&period;5ºC5&period;The times of sterilizing include 3S&comma;5S&comma;10S&comma;30S&comma;300S&lpar;optional&rpar;6&period;Product system pressures to 8 bar&period;7&period;SIP &lpar;Sterilise In Place&rpar; an option for Aseptic sampling&period;8&period;Built in true CIP &lpar;Clean In Place&rpar;&comma; flow for CIP more than 400 Ltr per hour9&period;Outlet temp&period; adjustable by request&period;10&period;380 volt 3 phase&comma; 50 or 60 Hz&period;11&period;Power&colon;10KWApplicationSuch as fruit juices&comma; milky products&comma; tea&comma; sauce dressings&comma; soups and so on&period;&bsol;

Model Number

YC-02

Item Name

Lab UHT Sterilizer

Machine Material

Stainless Steel

Capacity

20L&sol;H~25L&sol;H

Min Sample feed

3~5L

Air Inlet Temperature

105ºC-200ºC

Highest Sterilization Temperature

150ºC

Lowest Cooling Temperature

5ºC

Washing

CIP

Cabinet

Optional

Sterilizing Time

3s 5s 10s 30s 300s&lpar;optional&rpar;

Dimension

1650&ast;1150&ast;910mm

Power

10kw 380v

Temperature Precision

±2ºC

Product System Pressure

8Bar

Touch Screen

LCD

COMPANY PROFILEShanghai Pilotech Instrument Equipment co&period;&comma;LTD is a modern high-tech joint-stock enterprise&comma; a company specialized in the R&D&comma; production&comma; sale and service of laboratory drying equipment and small dairy beverage production lines&period; Over 10 years of development&comma; Shanghai Pilotech has formed a complete and strict enterprise management system and had a professional research and production team&comma; sales service team and after-sales service team&semi; the company meet CE standard of European Union&period;

OUR CUSTOMERS Our company's laboratory equipment is widely used by well-known universities in China such as Tsinghua University&comma; Peking University Fudan University&comma; Shanghai Jiaotong University&comma; Hong Kong University of Science and Technology&comma; Wuhan University&comma; Huazhong University of Science and Technology&comma; Sun Yat-Sen University&comma; Zhejiang University and Nanjing University&comma; and world-renowned companies like Coca-Cola and General Electric&period; It has more than 1&comma;500 users in mainland China&period;

PACKAGE TRANSPORTATIONSea shipping and air shipping all is available&comma; we packaged durable and careful&comma; to make sure the parcel will be fine&comma; when our customer receive it&comma; and more&comma; shippinh on time we are promise&period;