In regard to the guidelines of the Ministry of Health concerning the advertising of medical devices, in vitro medical devices and medical-surgical devices, we inform that the information contained on this website is intended exclusively for professional operators.

Contact Us

Give Us a Call

ECOLAB S.r.l.
Via Trento, 26
20871Vimercate - MB
Italy

Please fill out the contact us form with your details and we will aim to get back to you as soon as possible. A detailed query from your part will help us to be as accurate as possible in our response.

Instructions for use

The solution is produced using the provided dispensing spoon/beaker. After adding the appropriate powder quantity to water at about 25°C, stir the application solution frequently. After a few minutes the disinfectant solution is ready for use.

Application areas

Sekusept™ is actively suited for the manual cleaning and terminal disinfection of thermostable and thermolabile medical instruments including flexible endoscopes and for the manual pre-cleaning of endoscopes before mechanical reprocessing.

Not suitable for brass and copper instruments as well as for mechanically pre-damaged, chromium plated or nickel plated instruments. A light coloration of the powder has no effect on the product quality.

Manual reprocessing

Place the instruments in the solution immediately after use, ensuring that all surfaces are completely wetted; there must not be any cavities and air bubbles. Clean the instruments following the manufacturer’s instructions, including the mechanism (e.g. brushing, wiping). After the specified contact time, remove the objects from the liquid and rinse with water of at least drinking quality. For purely manual reprocessing, set up an instrument bath for cleaning and an instrument bath for the terminal disinfection.

Reprocessing in the ultrasonic bath

Sekusept™ Active can be used in all stainless steel ultrasonic baths. Set up ultrasonic bath with cold water, max. sonication time: 5 min. If needed, let it rest until the end of the contact time. Use disinfectants safely. Please follow the information provided by the instrument manufacturer.

Agent content of a 2% application solution: >1,000ppm peracetic acid.

Slight color variations in the powder do not mean any loss of quality.

Manual reprocessing of flexible endoscope

Disinfecting Pre-cleaning

Pre-cleaning at the light source

Wipe the outer jacket from top to bottom using a single use cloth soaked with the stated cleaning disinfectant solution

Use of the cleaning valve (manufacturer-specific)

Connection of the cleaning attachment to the instrument channel input

Suction an ample amount (200-500 ml) of cleaning disinfectant through distal end

Actuate the air water supply for a few seconds for every working sequence

Disconnecting the flexible endoscope from the light source

Cleaning at the reprocessing place

Connecting tightness tester

Inserting the flexible endoscope in the cleaning disinfectant solution

Perform tightness test according to the manufacturer's instructions

Cleaning of the channels and of the instrument channel inlet with special cleaning brush

Wipe endoscope from outside in the solution using a soft cloth

Rinsing the channels with disinfectant solution using special hand pump

Remove the endoscope from the solution and rinse with water from outside - flush channels with water by using hand pump

Rinsing

Thorough cleaning of the flexible endoscope from outside and of the channels using the hand pump

Water of at least drinking quality
sterile distilled water or fully demineralized water is better

Drying

Carefully dry from outside by using a clean, lint-free, single use cloth

Dry channels by means of the hand pump or compressed air of max. 0.5 bar

Suspended storage of the endoscope (without valves) e.g. in dry cabinet until the next use

Accessories, valves etc. are to be separately cleaned, disinfected or sterilized! The disinfectant solution is to be changed every working day according to RKI recommendations; if the stains are visible, it should be more frequent.

Composition

Sodium percarbonate, non-ionic surfactants, phosphonates.

Use (after commencement)

Shelf life

In container

until the printed expiration date

As a prepared application solution2

The same workday

1 Please note the following to ensure the correct storage of environmental hygiene and instrument disinfectants: The original containers (canisters, buckets or bottles) with the disinfectant concentrate (or powder) should be stored at temperatures below 25°C and kept dry and clean. A brief period below or above the given storage temperature (e.g. during transport) will not diminish the effectiveness.The containers are to be sealed tightly after each use. Please also take care that the concentrate remaining in the container is not contaminated when preparing the application solution.

2Assuming that the application solution that is produced is not contaminated by organic material, it is to be stored in a clean, closed container or a covered instrument basin.Please take note of the corresponding marking on the containers in which the disinfectant solutions are kept, e.g. contents, concentration, date, shelf life and batch number. Used and organically contaminated solutions are to be changed at least once per workday as per RKI recommendation.

Ecolab.com uses cookies to give you the best possible experience. By using Ecolab.com, you consent to our use of cookies. If you do not wish to receive our cookies, adjust your browser settings. Disabling cookies may impact your user experience while on Ecolab.com.