Area Monitors / Waste
Monitors

Gamma Radiation Detection

for monitoring trash, hospital waste, contaminated laundry and other applications.

Radiation Alert Area Monitor This is our newest area monitor for gamma and X-ray. It is similar to the Model 375, featuring internal and external detector options. GM detector and NaI scintillation detector configurations are available. Modern features include intelligent touch screen and free software.

Radwall-S Area MonitorsTheRadWall-S Area Monitor is a sensitve all-digital monitor, covering a wide range of Gamma and X-Ray radiation. This area monitor utilizes a high-performance YSO scintillation crystal combined with a state-of-the-art SiPM (YSO+SiPM) and multi-voltage threshold (MVT) algorithm to detect a wide range of radiation doses. This monitor features high detection accuracy, high sensitivity, and fast response, as well as other advantages. Dose range is 0.01 μGy / h ∼ 1 mGy / h. This device can be used for health physics, nuclear medicine, reasearch, and other fields that utilize radioactive sources.

Falcon Series Area Monitors This modern design of area monitors features large display, CsI scintillaiton detector, and a range from 1 uR/hr to 100 mR/hr. Other features include touch screen datalogging and MESH network capability. Different configurations are available that offer the user choice of enclosure color, display extended detection range to 50 R/hr and weatherproof enclosure. This unit can communicate to your network via RADConnect, Wi-Fi, Ethernet, USB, optional RADConnect-Link. Why settle for older technology? Please contact us for questions and pricing.

AM-806 Area / Waste MonitorThe Model AM-806 is a ruggedized, industrial grade area monitor that detects radioactive material in hospital laundry and waste carts. The system includes two large area plastic scintillators totaling 360 cubic inches, which detect gamma emitting isotopes passing through doorways and other passage ways. The system features a full color VGA touch screen for user interface. Counts and energy curve are graphically displayed on-screen. Alarm parameters can be easily changed by the touch of a finger, without recalibration. The AM-806 is durable and easy to operate. It requires minimal training. Installation of the system can be permanent or temporary.

AMP-50 Area Monitor The AMP-50 G-M monitor provides real-time remote radiological monitoring for gamma radiation. Local readout of hand-held meter allows for use as a portable survey instrument.The AMP-50 is designed specifically to be used in medium to low dose rate fields. The AMP-50 features linear response from 10 μR/h to4 R/h. The meter's sensitive electronics can be positioned far from the high field (25 to 350 feet away); they are not subject to destructive gamma exposure. The probe head may be placed directly into a filter compartment or against a resin tank or waste liner. Other features include: waterproof detector housing and cable, quick-disconnect connectors, built-in communication port for use with DDC-16, WRM transmitter, stabilized digital display, and user-selectable internal alarm threshold.

AMP-100 Area Monitor The AMP-100 would be the choice area monitor for medium range gamma radiation: 0.005 R/h to 1 000 R/h. The system includes an energy compensated G-M detector. The meter's sensitive electronics can be positioned far from the high field (25 to 350 feet away); they are not subject to destructive gamma exposure. The probe head may be placed directly into a filter compartment or against a resin tank or waste liner. Other features include: waterproof detector housing and cable, quick-disconnect connectors, built-in communication port for use with DDC-16, WRM transmitter, stabilized digital display, and user-selectable internal alarm threshold.

AMP-200 Area Monitor This unit is recommended for high range gamma: 5 R/hr to 10,000 R/hr. The system includes an energy compensated G-M detector. The meter's sensitive electronics can be positioned far from the high field (25 to 350 feet away); they are not subject to destructive gamma exposure. The probe head may be placed directly into a filter compartment or against a resin tank or waste liner. Other features include: waterproof detector housing and cable, quick-disconnect connectors, built-in communication port for use with DDC-16, WRM transmitter, stabilized digital display, and user-selectable internal alarm threshold.

Model 375P-336 Area / Trash Monitor The Ludlum 375P-336 Radiation Waste Monitor is a solution to hospital radioactive waste management. The plastic scintillation detectors are much more compact and sensitive than the larger, more awkward NaI probes. For outdoor use, the system can be modified to be weather-proof. If desired, the system can be modified to be connected with the facility computer network. The basic system includes the lightweight Model 375 Electronics and two 168 in³ (2753 cm³) Plastic Detectors with 200'(61m) cables. A 10 µCi Check Source is included to for periodic testing. Options include strobe lights, extra alarms, time/date printers, ethernet kit, software, and cameras.

Model 375-30 Area / Waste MonitorThe Model 375-30 is a Digital Model 375 Controller coupled to two shielded 7.6 cm (3 in.) diameter by 2.5 cm (1 in.) thick NaI(Tl) scintillation detectors. The detectors are encased in NEMA 4X rated weather tight enclosures suitable for the outdoor environment, while the 375 Controller is normally mounted to a wall indoors near an operator. This cost-effective system is simple, making it easy to operate and maintain. The controller supplies local alarms but can also be connected to external alarms or even put onto an Ethernet network if desired. It additionally has a 48-hour battery backup to keep the system operational in the event power is lost.