Bundle includes:

110V Appliance Adaptor

500 Pass Labels

200 Fail Labels

250 Passwrap Cable Labels

IEC Cloverleaf Adaptor

PAT Register Book

Description

Seaward Primetest 250 PAT Tester

The new Primetest 250 PAT tester will perform all tests required to comply with the IEE Code of Practice. These include RCD trip testing, protective conductor current and touch current testing. The Seaward Primetest 250 also carries out Insulation Testing at either 250V or 500V DC. The Seaward Prime test 250 is battery powered and can be plugged into the mains supply to run the protective conductor current and touch current tests.

As innovators within PAT, Seaward have also enabled the Seaward Primetest 250 to carry out full testing on 3-phase equipment found in factories, workshops, schools and colleges. The PT250 is the only tester on the current market that will test for Differential Leakage Current on 3-phase equipment using an optional Seaward 3-phase adaptor.

The battery powered Primetest 250 PAT tester is very easy to use with auto sequence test buttons and a large display with clear PASS/FAIL indication. This makes it an ideal choice of PAT tester for both novices to PAT testing or experienced test engineers who need to test a wide range of electrical equipment while fully complying the IEE guidelines. The Seaward Prmetest 250 is the next step in their already highly successful range of battery powered Primetest PAT testers, packing it with loads of features which will aid Electrical Safety within the workplace and help companies fully comply.

Seaward Primetest 250 Main Features:

Fully complies with the IEE Code of Practice

RCD Trip Time Testing

Battery driven for mains free testing

Tests IEC leads including polarity check

Automatic Test Sequences

Hand held and Lightweight Appliance Safety Tester

Selectable 250V or 500V Insulation Test.

3-Phase Equipment testing (using optional adaptor)

Carries out Protective Conductor/Touch Current (using mains supply)

Specification

Earth continuity

Output Current: 200mA

Test Voltage (o/c): 9V DC nominal

Test Duration (Class I Test): 2s nominal

Display Range: 0.01 ohm to 19.99 ohm

Pass Value (Class I Test): 0.2O

Insulation resistance

Test Voltage: 250/500VDC @1mA nominal, < 2mA (s/c)

Test Duration (Class I or II): 2s

Display Range: 0.01 Mohm to 19.99 Mohm

Pass Value: 1.0 MO Class I

2.0 MO Class II

Equivalent leakage

Test Voltage: 40V Nominal AC 50Hz Open Circuit

Test Duration: 2s – 30s nominal

Display Range: 0.1 mA to 19.99mA

Pass Value: 0.75mA CLI / 0.25mA CL II

IEC lead test

Test voltage: 5V DC nominal

Test Duration: Time for test (2s max)

Test: Live / Neutral Checks for o/c, s/c and crossed

Protective Conductor Current

Test Voltage: Supply Voltage, max current 13A

Test Duration: 30s maximum

Display Range: 0.15 mA to 9.99 mA

Pass Value: 0.75mA/3.5mA (user selectable with 0.75mA as default)

Touch current

Test Voltage: Supply Voltage, max current 13A

Test Duration: 30s maximum

Display Range: 0.10 mA to 5.00 mA

Pass Value: 0.25mA

3 phase differential leakage current

***Requires 3 phase adaptor accessory ***

Test Voltage: 3 phase supply voltage

Test Duration: continuous measurement

Display Range: 0.10 mA to 9.99 mA

Pass Value: 3.5mA

RCD trip time

Test Voltage: Mains supply voltage

Test Current: 10mA / 30mA rms sinusoidal

Test Current Phase: 0°/180° auto switched on alternate tests

Measuring Range: 0ms – 500ms

Pass Value: 200ms

Power socket check

Input voltage range: 225 – 254V AC max current 13A UK

Indicates configuration of voltage potential: Line potential phase to earth, Line potential phase to neutral, Line potential neutral to earth

Write Your Own Review

Delivery Information

At Test-Meter.co.uk we understand the importance of a reliable delivery service when shopping online. Especially when the goods are often essential to your working day. If your goods are required by a certain date please feel free to contact us for an honest and realistic indication whether this can be met.

We use DPD/Interlink for all larger items and Royal Mail for smaller items and accessories. With DPD/Interlink you will receive a 1 hour delivery window on the day of delivery via e-mail/text so you will not have to wait around all day. We also offer delivery options for goods required urgently on certain products, all which are sent by DPD/Interlink on a fully traceable and manageable service.

Our charges for UK deliveries:

FREE Delivery (on select products only): typically 3-5 working days*

Standard Delivery at £4.95: 2-3 working days

Next Working Day Delivery (on select products only) at £7.50

Pre-Noon Next Working Day Delivery (on select products only) at £9.50*

Pre-10.30am Next Working Day Delivery (on select products only) at £18.50*

Saturday Pre-Noon Delivery (on select products only) at £14.50*

Sunday Pre-Noon Delivery (on select products only) at £14.50*

*UK mainland only

NOTE! For next working day or Saturday/Sunday delivery options the order MUST be placed no later than 3pm on the working day before delivery is required. After 3pm, please arrange with our staff before selecting a priority delivery service. Priority services are available to UK mainland residents only.

Customers outside the UK:

Test-Meter.co.uk exports to many countries all over the world. Prices for shipping vary on location and weight/dimensions. Costs are calculated for certain countries at the checkout. If your country is not included in this list please e-mail admin@test-meter.co.uk with your name, address and the items you would like to order for an accurate shipping quote.