Application to high pressure measurement with its unnecessary opening hole to be made into high pressure instruments.

Complete tailored version sensors can be designed to different kinds of types according to users special needs. For instance, low temperature type, high pressure type, sanitary type, etc.

Insertion type can be made, that is applicable to large normal diameters measurement for its little pressure loss, low price, unnecessary stopping flow to take out it, and convenient installation and maintenance.

Description

The SRS series Turbine Flowmeter is a leading technology integrating with advanced design to produce the new generation of turbine flowmeter with the features of simple structure, light weight, high accuracy, good repeatability, flexible reaction a convenient installation / maintenance / application, etc.

It is widely applied to measuring the liquid of which kinematic viscosity is under 5*10-6m2 / s and have no impurity of fiber, grain etc., and no corrosive interaction with the stainless steel 1Cr18Ni9Ti, 2Cr13, and A12O3, and hard alloy in seal pipes. The liquid of kinematic above 5*10-6m2/s can be measured after real liquid calibration of flowmeter.

It can be used in value control, siren when excess, if co-ordination with special display instrument. So it is the ideal instrument of measuring flow value and saving energy.

Turbine Flow Meters measure the velocity of liquids, gases and vapors in pipes, such as hydrocarbons in fuel flow measurement, chemical flow metering, water flow metering, cryogenic liquid flow metering, air or gas flow metering, and general industrial flow metering

How Turbine Flowmeters Work

Turbine flowmeters use the mechanical energy of the fluid to rotate a “pinwheel” (rotor) in the flow stream. Blades on the rotor are angled to transform energy from the flow stream into rotational energy. The rotor shaft spins on bearings. When the fluid moves faster, the rotor spins proportionally faster. Turbine flowmeters now constitute 7% of the world market.

Shaft rotation can be sensed mechanically or by detecting the movement of the blades. Blade movement is often detected magnetically, with each blade or embedded piece of metal generating a pulse. Turbine flowmeter sensors are typically located external to the flowing stream to avoid material of construction constraints that would result if wetted sensors were used. When the fluid moves faster, more pulses are generated. The transmitter processes the pulse signal to determine the flow of the fluid. Transmitters and sensing systems are available to sense flow in both the forward and reverse flow directions.

Plus and Minus’s of Turbine Flow Meters

The cost is moderate. Very good at clean, low viscosity fluids of moderate velocity and a steady rate. Turndown is very good as it can read very low compared to the maximum flow. They are reliable if put in a clean fluid especially if it has some lubricity. AGA and API approved for custody transfers. They do cause some pressure drop where that may be a factor such as gravity flows. Not reliable for steam. Bearings wear out.

How to Use Turbine Flow Meters

Turbine flowmeters measure the velocity of liquids, gases and vapors in pipes, such as hydrocarbons, chemicals, water, cryogenic liquids, air, and industrial gases. High accuracy turbine flowmeters are available for custody transfer of hydrocarbons and natural gas. These flowmeters often incorporate the functionality of a flow computer to correct for pressure, temperature and fluid properties in order to achieve the desired accuracy for the application.

Be careful using turbine flowmeters on fluids that are non-lubricating because the flowmeter can become inaccurate and fail if its bearings prematurely wear. Some turbine flowmeters have grease fittings for use with non-lubricating fluids. In addition, turbine flowmeters that are designed for a specific purpose, such as for natural gas service, can often operate over a limited range of temperature (such as up to 60ºC) whereby operation at higher temperatures can damage the flowmeter.

This flowmeter can be applied to sanitary, relatively clean, and corrosive liquids in sizes up to approximately 24 inches. Smaller turbine flowmeters can be installed directly in the piping, but the size and weight of larger turbine flowmeters may require the installation of substantial concrete foundations and supports. The flow of corrosive liquids can be measured with proper attention to the materials of construction of all wetted parts, such as the body, rotor, bearings, and fittings.

Applications for turbine flowmeters are found in the water, petroleum, and chemical industries. Water applications include distribution systems within and between water districts. Petroleum applications include the custody transfer of hydrocarbons.

Industries Where Turbine Flow Meters are Used

Application Cautions for Turbine Flow Meters

Turbine flowmeters are less accurate at low flow rates due to rotor/bearing drag that slows the rotor. Make sure to operate these flowmeters above approximately 5 percent of maximum flow. Turbine flowmeters should not be operated at high velocity because premature bearing wear and/or damage can occur. Be careful when measuring fluids that are non-lubricating because bearing wear can cause the flowmeter become inaccurate and fail.

In some applications, bearing replacement may need to be performed routinely and increase maintenance costs. Application in dirty fluids should generally be avoided so as to reduce the possibility of flowmeter wear and bearing damage. In summary, turbine flowmeters have moving parts that are subject to degradation with time and use.

Avoid changing from gas flow to liquid flow should be avoided because they can mechanically stress the flowmeter, degrade accuracy, and/or damage the flowmeter. These conditions generally occur when filling the pipe and under slug flow conditions. Two-phase flow conditions can also cause turbine flowmeters to measure inaccurately.