Product Description

The testo 330i flue gas analyzer combines the proven measuring technology of the preceding model testo 330 LL with a revolutionary new method of operation: operation independent of measuring location via smartphone, a sturdy testoFix probe holder and paperless documentation. This instrument embodies the current trends in flue gas measurement.The centrepiece of the set is the testo 330i flue gas analyzer with H2-compensated CO sensor. The flue gas analyzer is therefore also approved for the official acceptance testing of heating systems.Operation of the flue gas analyzer and the display of readings are implemented independently of the measuring location via the testo 330i App on your smartphone or tablet. Here, you can make adjustments to the measuring instrument, analyze the flue gas values, enter customer data and create measurement protocols. You can then add comments or photos of the system to the protocols, and send them to the customer or your office via e-mail.

The testo 330i flue gas analyzer also has the following impressive features

User-replaceable long-life sensors with a service life of up to 6 years. This saves at least one sensor replacement over the flue gas analyzer's typical useful life

TÜV-tested according to 1st German Federal Immission Control Ordinance (BImSchV) and EN 50379, Parts 1-3

Integrated draught and gas zeroing without removing the probe. The probe can remain in the flue during zeroing

Fresh air dilution upon exceeding a CO threshold value up to 30,000 ppm

Fully enclosed, particularly robust plastic housing

Also included in the set: a modular flue gas sampling probe (180 mm) with a 65 cm long, extra-flexible hose and a 190 mm long combustion air temperature probe.The testoFix probe holder is included in the set, which enables the probes to be securely attached to the flue gas duct quickly and easily. This global innovation also serves as a fall-proof suspension system for the testo 330i flue gas analyzer. A mains unit and the practical system case, in which all set components can be stowed away compactly and safely, complete this set.

Flue gas analysis and flue gas measurement

At the core of the measuring technology of the testo 330i are the two long-life sensors for measuring O2 and CO. An NO sensor is optionally available. With the testo 330i, flue gas and draught are measured in parallel – therefore you always have an overview of all relevant readings on a screen in the testo 330i App.

Differential pressure measurement

You can also use the testo 330i flue gas analyzer to measure the gas supply on gas boilers. Measure the gas flow and gas resting pressure to ensure that a gas boiler can be commissioned properly. In conjunction with the manufacturer's specifications concerning the nozzle pressure, the results of this measurement will enable you to tailor the output of the boiler to the home's actual heat requirement.

O2 supply air measurement

In balanced flue systems, you can use the testo 330i to find out whether there is a leak in the flue gas system. For this, simply measure the O2 content of the combustion air in the dual wall clearance. This way, you can ensure that no flue gas is getting into the combustion air supply. This prevents damage due to condensate or impairment of the combustion process.

Product Brochures

Manuals

Software

Firmware testo 330i
(v2.10, BIN, 3.6 MB)
It is partially not possible to execute a firmware update with Windows 8.1 / Windows 10. This can be solved by installing a new Bootloader once.
A description and all necessary files can be found under Update-Kit / Bootloader

Update-Kit / Bootloader
(EXE, 380 KB)
It is partially not possible to execute a firmware update with Windows 8.1 / Windows 10. This can be solved by installing a new Bootloader once.
A description and all necessary files can be found under Update-Kit / Bootloader

Testo ZIV driver ZIV 2000 for testo 320 and testo 330
(ZIP, 8.8 MB)
Testo ZIV driver in the 2000 version. The Testo ZIV driver is used to connect the testo 320 and testo 330 measuring instruments to an application program (sweeping district administration program) according to the interface Version 2.0 defined by the Zentralverband des Schornsteinfegerhandwerks (Central Association of Chimney Sweeps, ZIV). Please check with the manufacturer of your application program as to whether this interface is supported. If Microsoft .NET Framework 4.0 has not been installed on the computer, it must be downloaded from the Microsoft website and installed on the system.

Testo ZIV driver for testo 300, testo 320 and testo 330
(v2.2 SP1, EXE, 64.193 KB)
The Testo ZIV driver is used to connect the testo 300, testo 320 and testo 330 measuring instruments to an application program (sweeping district administration program) according to the interface defined by the Zentralverband des Schornsteinfegerhandwerks (ZIV, Central Association of Chimney Sweeps) in version 1.0 of 01. August 2012 as well as in version 2.0 of 13. February 2017. Please check with the manufacturer of your application program as to whether this interface is supported.