Üveg

Laboratory melting furnace SC 8 with SiC rod heating for use with a customer crucible

Two-side heating with SiC rods

The laboratory melting furnace SC 8 was specially developed for melting glass in the laboratory. A customer crucible is entered into the furnace from the top. Glass is molten in the crucible. Heating is realized from two sides with powerful SiC rods. With this heating method a maximum furnace temperature of 1500 °C can be achieved. The very effective, multi-layer insulation with long-life lightweight refractory bricks inside the chamber guarantees low outside temperatures even if the furnace is in continuously used.

Model

Tmax

Inner dimensions in mm

Volume

Outer dimensions in mm

Heating power

Electrical

Weight

°C

w

d

h

in l

W

D

H

in kW1

connection*

in kg

SC 8

1500

200

150

250

8

840

715

730

30

3-phase

290

1Depending on furnace design connected load might be higher *Please see page 73 for more information about supply voltage