Mark-10 NIST Certifcate of Calibration with data for Series 3 gauges (ordered with a new gauge), add $71.25

Series TT03C digital cap torque gauges are quick and economical solutions for basic handheld cap torque testing. These portable instruments eliminate the need to secure closures in a fixture or more elaborate testing system. A dedicated remote torque sensor with Jacobs chuck is supplied with a universal grip for caps from 0.65 - 1.55 in (16.5 - 39.4 mm) diameter.

Series TT03C records both application and removal torque values (clockwise and counter-clockwise directions), and is available in three capacities from 12 to 100 lbFin (135 to 1150 Ncm).

Series TT03C features a sampling rate of 2,000 Hz, producing accurate results under a wide range of test conditions. A backlit graphic LCD displays large characters, and a menu is available to access the gauges’ features and configurable parameters. USB output is provided for data collection purposes.

Set point indicators are provided for pass-fail testing. The gauges are overload protected to 150% of capacity, and an analog load bar is shown on the display for graphical representation of applied torque.

The gauges include MESURTM Lite data acquisition software. Currently displayed readings may be sent to MESURTM Lite for tabulation and one-click export to Excel for further data manipulation.

Features

Ergonomic torque sensor with Jacobs chuck

Peak torque values saved and displayed

High-speed 2,000 Hz sampling rate

USB data output

Programmable set points with pass/fail indicators

Live load bar graphs with set point markers

Peak memory for clockwise and counter clockwise readings

Set points always displayed

Configurable audio alarms and key tones

MESURTM Lite data collection software included

Battery or AC operation

Safe overload: 150% of capacity

CE mark

Model

MTT03C-12

MTT03C-50

MTT03C-100

Price

$1,353.75

$1,353.75

$1,353.75

lbFin

12 x 0.01

50 x 0.05

100 x 0.1

kgFmm

140 x 0.1

580 x 0.5

1150 x 1

Ncm

135 x 0.1

570 x 0.5

1150 x 1

Accuracy:

±0.5% of full scale ±1 digit

Sampling Rate

2,000 Hz

Power

AC or rechargeable battery. Multi-step low battery indicator is displayed, gauge shuts off automatically when power is too low.

Battery Life

Backlight on: up to 7 hours of continuous useBacklight off: up to 24 hours of continuous use