Somat eDAQ Simultaneous High Level Layer

The Somat eDAQ Simultaneous High Level Layer (EHLS) offers 16 simultaneously sampled high-level differential analog inputs through independent connectors. Up to four EHLS layers can be added to a single eDAQ system enabling up to 64 channels. Networking multiple eDAQs make synchronous EHLS channel counts virtually limitless.

The EHLS can inherently handle any analog input from +/- 80V, and together with the Somat SMART conditioning modules, is a real multipurpose layer. Practically any input--thermocouples, strain gauges, accelerometers, microphones, amplified and unamplified transducers--can be utilized with this layer. Software-selectable sample rates, transducer power, and digital filtering simplify the set-up of any channel. There are also several calibration options, including defined value, external, and multipoint, as well as shunt calibrations with embedded software tools when using the Strain SMART Module.

The EHLS is available with an optional Analog Out. Calibration files, directly compatible with popular simulation software, scale the analog outputs to engineering units.

Includes

(16) Somat SAC-TRAN-MP-2 Transducer Cables (2 m)

Product Options

Somat eDAQ Simultaneous High Level Layer, Analog Out (EHLS-AO)

The Somat eDAQ Simultaneous High Level Layer, Analog Out (EHLS-AO) outputs are conditioned analog output signals that can be used in the creation of time-domain lab durability tests. When setting up the laboratory simulation, the Somat eDAQ system is brought into the lab with the component or vehicle. This practice is highly recommended, as it ensures that all of the transducer instrumentation and properties are exactly the same for the lab simulation as they were for the field data collection. Instead of being recorded, the analog-out signals are sent as time series data for the test rig to analyze. The controller can then develop drive files that are played into the test rig, reproducing exact field dynamics in the lab.

Benefits:

Consistency/Accuracy

The same signal conditioning, set-up file, and calibrations used in both the field and lab.

Ease of Set-up

Configure the eDAQ once, use it in both applications. The Analog Output Calibration File is directly compatible with popular simulation software.

Includes

(1) Somat SAC-TRAN-AO-2 Analog Output Cable

Part Number Code

The following number scheme is used to identify the layer configuration.

EHLS - XX

eDAQ Layer EHLS

Options

B = Base Layer. AO = Analog Out.

Cables and Accessories

Breakout Box, Analog Out

The SoMat Analog Output Box (EBB-AO) is a breakout box for the analog output option on the eDAQ EHLS and EBRG layers.

ICP-Type Signal Conditioning Module

The Somat ICP-Type Signal Conditioning Module (EICP) is a part of a family of signal conditioners for the eDAQ and eDAQlite HLS layers.

M8 Hex Nut Wrenches

The Somat M8 Hex Nut Wrenches (HDW-0034-00) work with the Somat M8 transducer and extension cables for both the eDAQ and eDAQlite.

Strain SMART Module

The Somat Strain SMART Module (SMSTRB4-xxx) is a part of a family of signal conditioners for the eDAQ and eDAQlite HLS layers. The Strain SMART Module provides excellent strain gauge signal conditioning with support for ¼-‚ ½- and full-bridges.

Thermocouple SMART Module

The Somat Thermocouple SMART Module (SMITC) is a part of a family of signal conditioners for the eDAQ and eDAQlite HLS layers.

Transducer Cable (2 m)

Transducer Cable (10 m)

The Somat Transducer Cable (SAC-TRAN-MP-10) is a high-quality 10 meter pre-molded cord set with tinned pigtail wires for transducer wiring.

Somat Pressure Sensor SPT-HLS

The Somat pressure sensor SPT-HLS is a compact, heavy-duty, pressure instrument designed for both OEM industrial and off-road applications where accuracy, repeatability, and stability can not be compromised.

Product Data Sheet & Literature

Somat EHLS

Somat EHLS

EN

DE

ES

FR

IT

PT

CN

JP

KR

RU

Data Sheet - SoMat EHLS

Brochure

EN

Candid data sheets: We keep our promises.

Precision in series: This also applies to the accuracy of the values specified in our data sheets. Ideal conditions can often be theoretically attained based on the values given, but our data sheets focus on realistic standard assumptions - such as those likely to be encountered in your application. Therefore, please consider this when comparing our data sheets with those of other manufacturers.

To help keep costs of on-track railroad tests under control, you need a reliable data acquisition system. For a recent series of tests on a new truck trailer design, SPE of Highland, IN chose SoMat eDAQ from HBM.

Making flying a pleasure again: This is the aim of the new, ultra-modern AKOYA light airplane. On-board safety and comfort is ensured through the use of strain gauges from HBM and the ultra-rugged SoMat eDAQlite data acquisition...

Trains in the Australian outback have to meet extremely demanding requirements. The renowned Institute for Rail Technology (IRT) decided to use HBM's particularly rugged SoMat eDAQ data acquisition system when designing a new...

The latest British military fast landing craft demonstrator has been developed and built by QinetiQ using HBM‘s eDAQ data acquisition system as part of the testing and analysis regime. The Partial Air Cushion Supported CATamaran...

Comparative tests of data acquisition equipment for use on high-speed trains undertaken by the Beijing Jiaotong University in China have demonstrated the effectiveness of HBM’s SoMat eDAQ equipment when operating under extremely...