The main board of TDS-100F is suitable for any kinds of transducers,including clamp-on type, insertion type, π in-line type, standard in-line type and same kinds of transducers from other manufacturers.the pipe parameters and calibration parameters of water meter and in-line pipe are input by manufacture, users do not need to input any parameter, only need to connect with flow meters to work.

Applied to on-line measure and system monitor for nearly all liquids from petrol chemical,metallurgy,electric power plant, irrigation, city water company,energy monitor fields, realize the functions of measuring and checking of flow velocity, flow rate, accumulation and heat quantity of different liquids, and flow rate on/off, liquids distinguish.

The user should tell the manufacture which kind of power supplying flow meter he need before ordering. Generally, the products provided by manufacture is 220 VAC power supplying flow meters. Special attention for operators : the flow meter will be badly damaged if the 220 VAC power supplying machine inserts into DC power resource, or the DC 24V power supplying machine inserts into AC 220V power resource. There are two power-supplying methods for TDS-100 fixed type machine (hang able type and plate type):

AC 220V (AC power suppl y in general)

DC 24V (DC power supply in general)

Technology features :

Operating power :AC 85—264V or isolation DC 8-36VDC

Repeatability:better than 0.2%.accuracy:better than 1%

Signal output :

One channel standard isolation RS485 output

One channel isolation 4-20mA or 0-20mA active output.

One channel OCT output(programmed between the pulse width(6-1000ms),default before leaving factory (200ms))

Automatic memory the positive,negative,net totalizer flow rate and heat quantity of the last 512 days,128 months,10years

Automatic memory the time of power on/off and flow rate of the last 30 times, realize to replenish by hand or automatically,read the data through Modbus communication protocol.

Protection level :

mainframe : IP65

IP68 (F4 mainframe)

Transducer : IP68

Ex-proof:EXd װ BT4 (TDS-100F2 type)

Transducer :

Clamp-on type

Insertion type

In-line type.

The new TDS-100F series ultrasonic flow meter can respectively or at same time use 16 key keyboard monitor. 16 key parallel or serial port key board, including 10 digit keys, 2 up/down arrow key, 1 Menu key(M), 1 enter key, 1 arithmetic point key, 1 backspace key. The keyboard can make users operate quickly and conveniently 4 key keyboard has 2 up/down arrow key, 1 menu key(M), 1 enter key (ENT), inputing digits, characters, and arithmetic point is by using up arrow key to input many times, the use of down arrow key is to move to next digit position.

SPECIFICATION

Item

Performance, Parameters

Accuracy

Better than ±1.0 %

Repeatability

0.2%

Velocity

±32m/s

Measurement Period

0.5S

MeasurementPrinciple

Transit-time measurement principle

Display

LCD with backlight. 2x20 letters

Keypad

4x4-key membrane keypad with tactile feedback

Input

Five channel 4-20mA current inputs for signals such as temperature, pressure, liquid level, and etc.. Accuracy 0.1%.Two of the five input channels are wired to terminal blocks. The remaining three channels are optional

Capable of offline compensation for flow totaliser, automatic / manual selectable. Self-diagnosis.v Automatically record the following information:The totaliser data of the last 64 days / 64 months / 5 years;The power-on time and corresponding flow rate of the last 64 power on and off events. Allow manual or automatic flow loss compensationThe instrument working status of the last 64 days

Strap(applied with installation of transducers on cast iron, GRP, PVC, etc. that can not be welded directly).

Installation of The new TDS-100F series is the easiest and convenient way in the installation of all flow meters, just choosing a suitable measurement point, input the pipe parameters of this pipe point to the flow meter, then fix the transducers on the pipe.

Author Details

Templatesyard is a blogger resources site is a provider of high quality blogger template with premium looking layout and robust design. The main mission of templatesyard is to provide the best quality blogger templates which are professionally designed and perfectlly seo optimized to deliver best result for your blog.