Avansee™ Preload1P

Avansee Preload1P is designed with the surgeon in mind

All Avansee lenses are manufactured from hydrophobic, highly cross‑linked, soft acrylic. There have been no reports of glistening since their launch in 2007.1,2 IOLs were incubated in saline at 45 ± 1°C for 24 hours and then at 37 ± 1°C for a further 2.5 hours to allow glistening to form.3

KOWAYP2.2+26.0D

ALCONSN60WF+20.0D

HOYANY-60+20.0D

Accelerated glistening formation3

Hoya Vivinex™ XY1

Average microvacuole –11.6 (MV)/mm2Glistening grade (MV) – 0

AMO Tecnis® ZCB00

Average microvacuole –6 (MV)/mm2Glistening grade (MV) – 0

Kowa Avansee™ PN6A

Average microvacuole –2.2 (MV)/mm2Glistening grade (MV) – 0

Alcon Acrysof® SN60WF

Average microvacuole –264.4 (MV)/mm2Glistening grade (MV) – 2,6

Zeiss CT LUCIA® 601P

Average microvacuole –71 (MV)/mm2Glistening grade (MV) – 1

Nidek Aktis SP NS-60YG

Average microvacuole –851.4 (MV)/mm2Glistening grade (MV) – 3+

Accelerated glistening formation14

Low PCO rate

The smooth lens surface and 360º square edge design, which extends around the lens including the optic-haptic junction6, are associated with reduced PCO formation.5,7,8

Excellent vision quality

The low asphericity of Avansee lenses means they provide a long depth of focus and are less influenced by IOL misalignment (decentration and tilt).10,11

IOLs that have little effect on the spherical aberration of the eye can potentially improve the quality of vision by reducing the risk of coma aberration associated with IOL misalignment (decentration and tilt).