TAR 120/220

The TAR series is a USP/ Pharm.Eur./ JP compliant ERWEKA unit for testing friability and/or abrasion.

Programming is easily performed by the symbol style keypad. The rotation speed can be programmed between 20 and 100 rpm and test run duration can be set in either time or total number of revolutions. During the test run the actual rotation speed and the test run duration is shown on bright LE displays.

For larger samples the complete unit can be set to the USP / Pharm.Eur. required 10° position by a standard switching leg.

To increase the ease of handling on the TAR series, the friability drums don’t have to be removed and opened for loading and emptying. Instead the friability drums are loaded through an opening. On completion of the test run the samples are automatically
removed into a container located under the drum.

According to the latest specifications the unit is equipped with one additional switching leg to bring the drum to the 10° position for testing of bigger sized samples.

The optional USB interface allows you either to download the test parameters (time, actual/ nominal speed) to a PC or receive a hardcopy printout when an optional printer is connected.

Friability and Abrasion drum compatible:

The TAR is compatible with our friability drum or can be used with our abrasion drum.