This is a 16-channel battery analyzer used for testing all kinds of battery including Li-ion battery, NiMH, NiCd, battery pack, and etc. Each channel is an independent constant-current (0.1 ~ 2A) and constant-voltage (0 ~ 10V) source which can be programmed and controlled by testing software. Brand new laptop with MS Window 8 and latest version of testing software are installed and calibrated for immediate use (Upto 8 sets of analyzers can be controlled simultaneously by one PC). Four-electrode testing method ensures the best measurement accuracy. Resume function after recovery from power failure to ensure data safety.

SPECIFICATIONS:

Electricity Requirement

AC110V±5% 50/60 Hz

Power Consumption

1000W

Current Range Per Channel

0.1-2A

Current Measurement Accuracy

±（0.2% FS + 0.3% RD)

Voltage Range Per Channel

0-10V

Voltage Measurement Accuracy

±（0.2% FS + 0.3% RD)

Max. Cycles

2000

Max. Testing Steps Per Channel

25

Total Channels

16

Internal Resistance Testing Accuracy

0.1mΩ

Internal Resistance Sampling

0~600mΩ

Current Sampling Rate

≤48S

Voltage Sampling Rate

≤3S

Programs & Software

The software (KTBS93en) with calibration function is included to set various working modes for measuring capacity and life cycle for all types of rechargeable batteries

With real-time monitoring interface and integrated graph/data windows, the testing process can be observed more directly and efficiently

Test Reports and Curves For Analysis

Different types of curves can be created by software base on user definition. (Voltage-time curve, current-time curve, capacity-time, energy-time, etc)

Data collected by software can be simply copied to Excel. User can easily compare the performance of the batteries tested in channels both visually and statistically

Protection And Auto-recover

If power failure occurs during the test, the system will shutdown all operating channels. Once power is recovered, the system will automatically resume those stopped channels and ensure that the test is normally conducted so that no cases will lose any data.

Laptop Computer

Brand new laptop with MS Windows 8, MS Excell 2013 in included and lastest version testing software are installed and calibrated for immediate use

Upto 20 sets of analyzers can be simultaneously controlled by one PC

Notice： If you use the laptop to run cycle life of battery for long time, must add a external hard driver ( > 3TB) to the laotop, and choose data collecting speed to the lowest to avoid data lost

New buyers must buy the software-installed laptop to avoid troubles from installation. The laptop bonded sale with the analyzer will cost extra $500 as a preferential price. If you like to run multiple battery analyzers on one Laptop, you can just buy multiple analyzers with one laptop which is offered at a discounted price. (The laptop will cost $799 if sold independently)

Customer can easily combine up to 8 battery analyzers (0.1-2A, 0-10V) by using normal Ethernet cable connector via one computer.