FLIR A315

FLIR A315

The FLIR A315 / A615 is a series of compact and affordable thermal imaging cameras, fully controlled by a PC. The FLIR A315 / A615 are Plug&Play with Viper Vision Process Control Software..

IR resolution: 320 x 240 pixels

Spatial resolution (IFOV): 1.36 mrad

Image frequency: 60Hz

Object temperature range: –20 to +120°C 0 to +350°C (Optional up to 1200C

Thermal sensitivity/NETD: < 0.05°C @ +30°C (+86°F) / 50 mK

Accuracy: ±2°C or ±2% of reading

Ethernet type: Gigabit Ethernet

Operating temperature range: 15°C to +50°C (+5°F to +122°F)

Excellent image quality

The FLIR A615 is equipped with an uncooled Vanadium Oxide (VoX) detector that produces crisp thermal images of 640 x 480 pixels. This allows more accuracy and shows more details at a longer distance. The FLIR A615 also has a high-speed infrared windowing option.

Users that do not need the high image quality of the FLIR A615 can choose the A315 that produces thermal images of 320 x 240 pixels. Both cameras make temperature differences as small as 50 mk clearly visible. They come with a built-in 25° lens with motorized focus and autofocus. Optional lenses are available.

GigE Vision™ standard compatibility

An industry first, GigE Vision is a camera interface standard developed using the Gigabit Ethernet communication interface. GigE Vision is the first standard to enable fast image transfer using low-cost standard cables even over long distances. With GigE Vision, hardware and software from different vendors can interoperate seamlessly over GigE connections.

GenICam™ protocol support

Another industry first - the goal of GenICam is to provide a generic programming interface for all kinds of cameras. The GenICam protocol also makes third-party software compatible with the camera.

16-bit temperature linear output

Allows you to do temperature measurements in a non-contact mode with any third-party software. A built-in Gigabit Ethernet connection allows real-time 16-bit image streaming to a computer.

Environmental housing (FLIR A315)

The FLIR A315 can be ordered with an environmental housing. The housing increases the environmental specifications of the FLIR A315 to IP66, protecting the cameras from dust and water without affecting the camera features. The housing is available for cameras that are equipped with a 25º, 45º or 90º lens, and can be ordered separately as an accessory.