Description

Features

The GX1838 provides eight output channels that can either be set as open or switched to any of the three voltage rails. Each of the three voltage rails can be programmed to output -10 VDC to +32 VDC or -20 VDC to +20 VDC with 14-bit resolution.

Voltage rails A through C are used as voltage sources which can be connected to a precision digital to analog converter (DAC) or to an external reference input. Rail A can be connected to DAC A or to external reference A. Rail B can be connected to DAC B or to external reference B. Rail C can be connected to DAC C, external reference C, or to ground. This architecture provides for maximum flexibility. If additional output channels are required, an external switch matrix (such as the GX6616) can be used. An on board EEPROM contains the DAC’s calibration parameters and provides for enhanced accuracy.

Programming and Software

The board is supplied with the GXPDO driver, a software package that includes a virtual instrument panel, and a Windows 32/64-bit DLL driver library and documentation. The virtual panel can be used to interactively program and control the instrument from a window that displays the instrument’s current settings and status. In addition, interface files are provided to support access to programming tools and languages such as ATEasy, LabView, LabView/Real-Time, C/C++, Microsoft Visual Basic®, C#, Delphi, and Pascal. An On-Line help file and PDF User's Guide provides documentation that includes instructions for installing, using and programming the board.

Applications

LRU/SRU avionics testing

Automotive ECU testing

Process control systems

Precision data acquisition

Automatic Test Equipment (ATE)

GX1838 Series Specifications

Number of Output Channels

8

Number of Voltage Rails

3

Number of DACs

3

Output Voltage Ranges

-10 V to +32 VDC (all rails)-20 V to +20 VDC (all rails)

Resolution

14-bit

Accuracy @ 1 kOhm Load

±14 mV

Slew Rate

6 V/µs

Maximum Load (Minimum Resistance)

50 Ω

Output Noise

< 4 mV rms, 1 KΩ load, 20 Hz - 300 KHz measurement BW

Output Protection

Short Circuit (Ground)

Power ON State

All Channels / Rails open

Maximum Current *

Per Channel

500 mA

Per Rail

500 mA

Per GX1838

500 mA

PXI Power Requirements

+5 V

2.9 A, max

+3.3 V

2.1 A max

+12 V

0.55 A, max

-12 V

0.55 A, max

Environmental

Operating Temperature

0 °C to +55 °C

Storage Temperature

-20 °C to +70 °C

Size

3U PXI

Weight

14 oz

Calibration

Calibration Interval

1 year

*The total maximum current is 500 mA (i.e., if one channel or rail uses the entire 500 mA, the other channels or rails cannot be used).
Note: Specifications are subject to change without notice.

Products similar to GX1838 Series

Order GX1838 Series and Related Products

To order call
1-888-837-8297
or order on-line via the form below.
Prices are specified for
USA
destination orders. Prices are noted in USD.
Import duties, taxes, and charges are not included in the item price.