THROUGH HOLE LOAD WASHER LOAD CELL COMPRESSION ONLY

LWO SERIES

CAPACITY RANGES:2,400 THROUGH 300,000 lb

The LW0 Series are washer shaped, strain gage based load cells that have been most commonly used in fastener testing and thru
hole load applications. Our stocked ranges from 2,400 through 300,000 lb compression are made from 17-4ph heat treated stainless steel and are matched to bolt diameters
and their specific load characteristics. Modifications of diameters, thicknesses, or load range to a specific diameter for OEM applications are welcomed. The
sensing element incorporates bonded foil strain gages of the highest quality and are sealed for protection against most industrial environments.

The Load Cells below come Calibrated in Compression Only

ModelOrder Code Model

CapacityCapacity Range

Calibration Description

Price

QtyQuantity

AddAdd to Cart

LW0-2

2,400 lb

Compression Calibration Only

$550.00

LW0-4

4,000 lb

Compression Calibration Only

$550.00

LW0-7

7,000 lb

Compression Calibration Only

$550.00

LW0-10

10,000 lb

Compression Calibration Only

$550.00

LW0-14

14,000 lb

Compression Calibration Only

$550.00

LW0-20

20,000 lb

Compression Calibration Only

$550.00

LW0-25

25,000 lb

Compression Calibration Only

$550.00

LW0-30

30,000 lb

Compression Calibration Only

$575.00

LW0-45

45,000 lb

Compression Calibration Only

$575.00

LW0-60

60,000 lb

Compression Calibration Only

$600.00

LW0-80

80,000 lb

Compression Calibration Only

$600.00

LWO-80A

80,000 lb

Compression Calibration Only

$650.00

LW0-125

125,000 lb

Compression Calibration Only

$775.00

LW0-190

190,000 lb

Compression Calibration Only

$775.00

LW0-260

260,000 lb

Compression Calibration Only

$950.00

LW0-300

300,000 lb

Compression Calibration Only

$950.00

Options

OPT-TEDS

N/A

Plug & Play Option with Connector

$75.00

Specifications

Rated Output (R.O.):

2 mV/V nominal

Nonlinearity:

2.0% of R.O.

Hysteresis:

2.0% of R.O

Nonrepeatability:

1.0% of R.O.

Zero Balance:

1.0% of R.O.

Compensated Temp. Range:

60° to 160°F

Safe Temp. Range:

-65° to 200°F

Temp. Effect on Output:

0.005% of Load/°F

Temp. Effect on Zero:

0.01% of R.O./°F

Terminal Resistance:

350 ohms nominal

Excitation Voltage:

10 VDC

Safe Overload:

150% of R.O.

Deflection Inches:

0.001 @ R.O.

Dimensions in Inches

Model

Capacity lb

O.D. DIA.

I.D. DIA.

D

H

LWO-2

2,400

.85

.192

.345

.350

LWO-4

4,000

.85

.262

.400

.350

LWO-7

7,000

.85

.326

.520

.350

LWO-10

10,000

.85

.390

.620

.350

LWO-14

14,000

1.00

.454

.725

.350

LWO-20

20,000

1.00

.517

.850

.350

LWO-25

25,000

1.25

.580

.960

.350

LWO-30

30,000

1.25

.644

1.040

.350

LWO-45

45,000

1.50

.770

1.200

.375

LWO-60

60,000

1.75

.897

1.490

.500

LWO-80

80,000

2.00

1.024

1.710

.500

LWO-80A

80,000

2.00

1.150

1.790

.500

LWO-125

125,000

2.37

1.279

2.140

.625

LWO-190

190,000

3.00

1.533

2.640

.875

LWO-260

260,000

3.50

1.787

3.000

1.250

LWO-300

300,000

3.75

2.041

3.300

1.370

The Load Cells below come Calibrated in Compression Only

ModelOrder Code Model

CapacityCapacity Range

Calibration Description

Price

QtyQuantity

AddAdd to Cart

LW0-2

2,400 lb

Compression Calibration Only

$550.00

LW0-4

4,000 lb

Compression Calibration Only

$550.00

LW0-7

7,000 lb

Compression Calibration Only

$550.00

LW0-10

10,000 lb

Compression Calibration Only

$550.00

LW0-14

14,000 lb

Compression Calibration Only

$550.00

LW0-20

20,000 lb

Compression Calibration Only

$550.00

LW0-25

25,000 lb

Compression Calibration Only

$550.00

LW0-30

30,000 lb

Compression Calibration Only

$575.00

LW0-45

45,000 lb

Compression Calibration Only

$575.00

LW0-60

60,000 lb

Compression Calibration Only

$600.00

LW0-80

80,000 lb

Compression Calibration Only

$600.00

LWO-80A

80,000 lb

Compression Calibration Only

$650.00

LW0-125

125,000 lb

Compression Calibration Only

$775.00

LW0-190

190,000 lb

Compression Calibration Only

$775.00

LW0-260

260,000 lb

Compression Calibration Only

$950.00

LW0-300

300,000 lb

Compression Calibration Only

$950.00

Options

OPT-TEDS

N/A

Plug & Play Option with Connector

$75.00

Wiring Color Code (WCC1)

4 Conductor

Internal Temperature Compensation and Balance Network Not Shown

OPT-TEDS Plug & Play Option

AD9 (9 PIN "D" Series) Connector attached to the end of a Load Cell or Torque sensor cable with a TEDS (Transducer Electronic Data Sheet) EEPROM.
Used with a Smart Plug & Play IEEE 1451.4 Compliant instrument, (shown on right), the Load Cell and Instrument will self calibrate. This option is a real time saver.
Read additional article...