US$369.88US$274.88

LCR-Reader-MP[LCRR-MP]

LCR-Reader-MP or Multipurpose offers users a vast range of features, a 0.1% basic accuracy and up to 100 kHz test frequency.

LCR-Reader-MP is well suited for measuring very large capacitances up to 680 mF. Electrolytic capacitors are measured at 120 Hertz with ESR measured at 100 kHz in accordance with the regular electrolytic capacitor test conditions. For Large Capacitance Mode, the test frequency is reduced to 25 Hertz, thus being able to measure very large capacitances with a reasonable accuracy.

The Oscilloscope Mode allows to analyze voltage waveforms with frequencies up to 100 kHz on an active circuit board. When used with the Kelvin Probe Connector the LCR-Reader-MP becomes a full probe station.

Note:LCR-Reader-MP has a maximum test signal of 0.65 Vrms that is not enough for reliable evaluation of some ceramic capacitors resulting in lower than actual capacitance values.
LCR-Reader MP has a high-test signal source resistance of 1 kOhm