Inductively Coupled Plasma (ICP-OES)

Reliability and high performance are the hallmarks of our multi-element detection ICP solutions. We have a long history of excellence and leadership in ICP-OES and ICP-AES technology, and our analytical platforms are engineered in response to real-world customer needs for accurate multi-elemental analysis.

We also provide a broad range of consumables and accessories -- including sample automation -- that are precision designed. And you can enjoy the peace of mind that comes with world-class training and support.

PerkinElmer’s new Avio® 500 is a truly simultaneous, dual view, and compact ICP-OES. It utilizes a vertical plasma and is engineered to handle even the most difficult, high-matrix samples without dilution, delivering productivity, performance, and faster return on investment.

The Avio 200 is a compact ICP-OES that combines a vertical plasma design with a host of unique hardware features to handle even the most difficult, high-matrix samples without dilution, delivering a whole new level of performance and flexibility to ICP.

Peristaltic pump tubing kit for MP2 Micro Peripump. Includes 34packages each containing 12 per pump tubes of various sizes and materials. The kit also includes 20 sets of per pump secure fittings. [Tubing included: 1 package each of 26 sizes of PVC

Secondary Drinking Water Metals, Matrix per Volume 2% HNO3 per 100mL, Concentration of Element: 100μg/mL Copper, 30μg/mL Iron, 5 µg/mL Manganese, and 500 μg/mL Zinc. Recommended for use in procedures for compliance monitoring of drinking water. Recommended for the analysis of ground and surface water where determination at the drinking water containment levels is required. Supplied in two bottles.

The PC3X for Avio is a compact, Peltier-heated or cooled inlet system which incorporates the ESI cyclonic spray chamber. In cooling mode the PC3X cools the outer walls of the cyclonic spray chamber that operates at -10°C to +80°C, reducing the water or solvent vapor loading on the plasma, resulting in enhanced stability and performance. In heating mode, sensitivity can be improved for low flow rate applications.

If both NaBH4 and SnCl2 are to be used as
reducing agents with FIAS systems configured for Hg/hydride
determinations. It is strongly recommended that separate mixing
manifolds, gas-liquid separators, quartz cells, and required tubing
and connectors be dedicated for use with each reducing agent. Can
be used with Flow Injection Mercury System.