Navigation

Changing from a UCClite-2 system to the UCCFusion controller

UCCFusion has the same functionality as UCClite-2; it supports 3-axis touch-trigger probing but NOT scanning or temperature compensation. The powerful integral amplifier can operate between 24 V and 60 V with an output up to 10 A. At only 1U high and 19 inches wide it can be rack mounted or stand alone on a flat surface. The unit has a built in touch-trigger interface and supports PH10 and TP200 via the standard PHC10-2 and PI 200 interfaces.

UCCFusion utilises an external power supply (PSU) which can be sourced locally or purchased from Renishaw. Use of an external PSU allows the most economical solution to match the power requirements of the CMM and motors.

Requirements

ESTOP bypass plug

If no external emergency stop is fitted to the CMM a bypass plug is required. Note that this has additional connections to previous versions of the servo power amplifiers.

If an external emergency stop is fitted please refer to the Fusion installation guide (link ?)

Blanking panel

Due to the size differences between the units blanking panels are available to fill the void.

Renishaw’s websites and Web shop uses cookies (including third party cookies). Some of these cookies are essential to make our site work properly and others are non-essential but help us to improve the site, including language and location settings and site analytics. You can read more about how we use cookies and how to configure or disable them on our site. By continuing to browse this site you agree to our use of cookies.