Fuel Consumption Meter :: VZO8

Available in two unique styles, either with a mechanical display or as a blind unit with pulse outputs

The standard meter is delivers a very low pressure drop and is suitable for flow ranging from 4 to 200 litres/hr.

A tamperproof mechanical counter indicates the total fuel consumed to the nearest 1/100th of a litre.

VZO oil meters are suitable for installation both indoors and outside. They can be installed in any position on vertical, horizontal and diagonal pipes except for completely upside-down and at temperatures up to 60°C

Specification

Flow Meter

VZO8 Standard

VZO8 OEM

Type

Rotary Pistion

Rotary Piston

Materials

Brass Body

Brass Body

Anodized Aluminium Piston

Anodized Aluminium Piston

Viton Seals

Viton Seals

Plastic Outer / Counters

Plastic Outer / Counters

Flow Range

4~200 litre/hr

4~200 litre/hr

Accuracy

±1%

±1%

Viscosity

1~50 mPa.s

1~50 mPa.s

Pressure (max)

25 Bar

25 Bar

Temp Range

-10°C to +60°C

-10°C to +60°C

Connections

¼" BSP(F)

¼" BSP(F)

10mm Compression Option

10mm Compression Option

Display

7x rolling couter 999999.9L

no display

1x dial indicating 0.09

Increments

0.01 litre

n/a

K-factor

n/a

0.0125 litre

Larger oil meters are also available for commercial and marine applications.

The cookie settings on this website are adjusted to allow all cookies so that you have the very best experience. If you continue without changing your cookie settings, we'll assume that you are happy to receive all cookies on our website. However, if you would like to, you can change your settings at any time using the Change cookie settings link in the Special menu.