**Accuracies above 1inch (25mm) are as good as setting to a
gage because the mechanical and electronic components are the same
on all ranges

With Standard Inch
Graduations on Shell and Thimble

Range

Resolution

Order No.

Inch

Approx. mm

Inch

mm

0-1

0-25.4

.00005

0.001

733XFL-1

0-1

0-25.4

.00005

0.001

733XFL-1W/SLC**

1-2

25.4-50.8

.00005

0.001

733XFLZ-2

2-3

50.8-76

.00005

0.001

733XFLZ-3

3-4

76-101

.00005

0.001

733XFLZ-4

4-5

101-127

.0001

0.001

733XFLZ-5

5-6

127-152

.0001

0.001

733XLFZ-6

6-7

152-178

.0001

0.001

733XFLZ-7

7-8

178-203

.0001

0.001

733XFLZ-8

8-9

203-228

.0001

0.001

733XFLZ-9

9-10

228-254

.0001

0.001

733XFLZ-10

10-11

254-279

.0001

0.001

733XFLZ-11

11-12

279-305

.0001

0.001

733XFLZ-12

12-13

305-330

.0001

0.001

733XFLZ-13

13-14

330-335

.0001

0.001

733XFLZ-14

14-15

355-381

.0001

0.001

733XFLZ-15

15-16

381-406

.0001

0.001

733XFLZ-16

16-17

406-432

.0001

0.001

733XFLZ-17

17-18

432-457

.0001

0.001

733XFLZ-18

18-19

457-482

.0001

0.001

733XFLZ-19

19-20

482-508

.0001

0.001

733XFLZ-20

20-21

508-533

.0001

0.001

733XFLZ-21

21-22

533-559

.0001

0.001

733XFLZ-22

22-23

559-584

.0001

0.001

733XFLZ-23

23-24

584-609

.0001

0.001

733XFLZ-24

Furnished with
Standards

**Includes
redemption card for Standard Letter of Certification
(SLC)

Micrometers are
furnished in protective cases.

Features:

This
is the Starrett No. 733 full function electronic outside
micrometer with all the features and controls you need to measure,
analyze and record the information you want. The anvil and spindle
diameter is .250" (6.35 mm).

Specifications
DescriptionInch Millimeter
Resolution through 4" (100 mm): .00005" 0.001 mmResolution over 4" (100 mm): .0001" 0.001 mmAccuracy**: ± .0001" ±0.002 mm
**Accuracies above 1" (25 mm) are as good as setting to a gage
because the mechanical and electronic components are the same
on all ranges.