The Thermo Scientific FD1530M Thermolyne Industrial Benchtop Muffle Furnace has a B1 single-setpoint digital controller with ramp and dwell, 2.2L (0.08 cubic foot) heating area capacity, and a temperature range of 100 to 1,200 degrees C for general laboratory use. Four heating elements embedded in refractory cement on the furnace top, bottom, and sides provide fast and even heating, with minimal temperature gradient. Thermal-efficient ceramic fiber insulation reduces heat loss and enhances energy efficiency. Heavy-duty firebrick insulation surrounds the chamber opening for increased durability. A 0.95 cm (0.38") diameter port at the rear of the furnace allows an independent temperature monitoring device to check internal temperature measurements. The furnace has an LED display that shows the chamber temperature as the unit heats, and the setpoint temperature can be checked. The counter-weighted front door swings up when opened, which vents heated air up and away from the operator. A ceramic hearth plate to protect the bottom heating element is included. This furnace is designed for laboratory-heating applications such as ashing, sintering, heat treating, quantitative analysis, and gravimetric analysis.

The Thermo Scientific FD1530M requires a 240V and 50/60 Hz electrical outlet and fuse (not included). The B1 single-setpoint digital controller with ramp and dwell allows the user to set a target temperature and control how quickly the furnace heats up and how long it holds the temperature before turning off. This controller also includes over-temperature protection (OTP), which sounds an alarm when the furnace heats beyond a specified limit. A mechanical OTP relay turns off power to the heating elements when the alarm sounds. Additional safety features include a door safety switch, which turns off the heating elements when the door is opened, and sensor break protection, which turns power off in the event of thermocouple failure. The furnace controller also includes a self-tuning feature. Self-tuning changes the control parameters to ensure temperature accuracy for a particular material type (load) and setpoint. A built-in vent port removes moisture and contaminants, which extends the life of the heating elements and the furnace. The furnace is Underwriters Laboratories (UL) and Canadian Underwriters Laboratories (cUL) Listed for safety. The power cord is included.

H is height, the vertical distance from the lowest to highest point; W is width, the horizontal distance from left to right; D is depth, the horizontal distance from front to back.

Laboratory furnaces are used for heating applications that require a range of high temperatures. They heat to high temperatures evenly and maintain consistent temperatures, while providing an environment free of undesirable contaminants and moisture. Factors such as object size, controllers, temperature range, and shape determine the uses of particular furnaces. Most furnaces have digital controllers that determine how fast they heat up (ramp), how long the temperature can be held (dwell), and how fast they cool down. Safety systems are included in most furnaces to prevent overheating the contents, as well as the furnace itself. Laboratory furnaces are used in applications such as petroleum analysis, chemical analysis of certain compounds, and heat-treating small steel parts.

Thermo Fisher Scientific manufactures analytical instruments, laboratory equipment, consumables, and software under the Thermo Scientific and Fisher Scientific brands, among others. The company was founded in 2006 and is headquartered in Waltham, MA.

Let customers speak for us

The La Crosse V10-TH Wireless Weather Station (Model No. C84343) is absolutely wonderful. The setup was easy with the Quick Setup Guide provided with the unit. The packaging contains additional support information with the website and phone number if there are questions. I especially liked the daily high low readings and the color on the unit! Tech Instrumentation provided great communication as to the availability of the product and shipping details. If fact, it arrived one day prior to the estimated delivery date. I would absolutely recommend this product and Tech Instrumentation for anyone looking for an indoor/outdoor weather station!

I can't way enough about the customer service at Tech Instrumentation. I was looking for a hard to find specialty part, which the manufacturer told me would need to be custom made. After calling and speaking with a representative at Tech Instrumentation, the part was located and order was filled, right over the phone. Delivery was prompt, and instrument is working great. Highly recommended company for lab supplies!