KONIK has been designing and continuously improving its HRGC, since 1978. The specifications and performance of the new model KONIK HRGC 5000 B are second to none. In the 5000B we have improved the pneumatics with a higher accuracy A/D converter. This once more, following KONIK’s commitment to upgradeability of its full range of instruments, is a KONIK module that can be retrofitted to models 4000B, 4000C and 5000A and is standard in model 5000B.

The KONIK HRGC 5000B has been designed for all conventional and special GC applications including Multivalve-Multicolumns Systems (RGA, NGA, PONA, PIANO, TOGA,…), as well asFast and Multidimensional GC-GC and for multidimensional HPLC-GC (a KONIK patent) and HPLC-GC-GC.All these options can be fitted into the KONIK 5000B at any time, providing full insurance for the fast evolving future as more and more applications and methods come into place.

The superb low inertia, fast heating and cooling, high precision oven (+/-0.1 ºC displayed) already introduced in the 5000A and the unique Cold Septum (purgeless) injection technology and exceptional flow controls of detector gases, along with the full range of detectors with their fast electrometers, remain unchanged. The state-of-the-art electronics, based on FPGA,s and micro embedded PC,s, that KONIK developed for its KONIK Q12 Mass Spectrometer, has now been also implemented in its range of HRGC’s allowing total diagnostics and remote control of the systems via internet and simple TCP/IP protocols.

Powerful and Flexible Sample handling. Multimodal and Multi-technique Sample Preparation and Sample Introduction is provided by the unique RoboKrom, the first and only Robotic Platform able to perform Static Head Space (HS), Dynamic Purge and Trap (P&T), Solid Phase Micro Extraction (SPME), Thermal Desorption, Sample Derivation and Micro-Concentration, besides standard injections in any GC and/or HPLC.

The KONIK HRGC 5000 B Series Gas Chromatograph is a High Resolution modular system integrated in one structural unit. Its unique design allows for constant upgrading and equipment modernization to any future analytical requirement, without modifying its external appearance or basic structure. The most common upgrades, as well as the necessary modifications for specific applications were considered while designing the HRGC 5000 B.

The optimal configuration for any specific application is obtained by selecting the basic version and customizing the unit by adding the required options for the analysis desired: injector options, detector options, valve switching devices, backflushing devices, autosampler, head space, control software, data processing, etc.

• High performance and low thermal inertia oven design with expanded temperature range from
-90ºC to 490ºC. Thermal inertia is dramatically minimized with perforated oven walls.
• Fastest heating and cooling oven: from 450ºC to 50ºC in less than 300s.
• PID Temperature control.
• Temperature resolution: 0.01ºC, 16bit ADC. 0.1ºC accuracy displayed.
• Two oven flaps, for maximum temperature stability even close to ambient temperature.
• Temperature gradients inside the oven have been reduced by forced air circulation.
• Oven is also pre-drilled to allow the installation of valves (sampling and switching), external interfaces (MS, FTIR) and other peripherals.
• Up to 25 positive and negative temperature ramps from 0.1ºC/min to 130ºC/min in 0.1ºC/min steps. Up to 26 isothermal periods.
• FAST GC option up to 500ºC/min using standard columns.
• Dual overheating protection, by hardware and software.

New electronics

The new electronics of the Konik HRGC 5000B Series based on FPGA technology of the latest generation has been developed to exceed the performance of all its predecessor microprocessors. Here are some of its crash throughs:

• Windows menu driven Touch PAD TFT Interface, 262566 colours/pixel
• Control of 8 isolated heated blocks at 16 BIT ADC (more than 0.01ºC precision) by PT-100: oven, 3 injectors, 3 detectors and one additional point of control
• Control of 2 points at 16BIT ADC by fast control PT-100
• Control of 2 points at 16BIT ADC by thermocouple with increased range up to 2000ºC (specially designed for Konik Quantitator®)
• Autoswitch on/off, Stand-by mode
• Self testing and autodiagnostics
• Power fail memory protection

Detectors

The KONIK Detectors are of modular design for quick interchangeability. Their unique design provides the best in flexibility and speed. KONIK offers a wide range of detectors such as: FID, TCD, ECD, NPD, FPD, Dual FID/FPD, OLFACTORY, PDHID/PDECD, MSD.

Provides full GC control through PC. It allows also the control of all other Konik instruments: HPLC, K2 (HPLC-GC), MS and Robokrom Multimode Autosampler.

KonikKrom Plus® is a windows based data acquisition and processing software with:
• Multiple instrument version
• Multidetector measurement
• Support tools for GLP/ 21 CFR Part 11 Compliance
• Add-on control modules and extensions
• Easy to install and easy to use

Engineered Solutions and Turn-Key Systems
The Konik HRGC 5000 B can be automated in the Lab or in the plant to become the "Autoanalyzer" of the products of your choice. Turn-key petrochemical solutions include, Natual Gas Analysis, Refinery Gas Analysis, LPG...