Identification and confirmation:The MAT95XP is a double focusing mass spectrometer featuring high performance and productivity. High mass resolution, accurate mass assignments and excellent sensitivity, together with high stability and sample throughput capabilities make the MAT95XP the most versatile mass spectrometer for solving routine analytical problems. Autotuning guarantees maximum sensitivity at all resolution settings and with all modes of ionization-routinely. The MAT95XP is used in numerous areas, like sports doping control, P.O.P. analysis, environmental analysis, drug discovery, and analytical service laboratories.

Productivity:Autotune in all ionization modes at low and high resolution One single mass calibration for all modes of operation Full automation including automated accurate mass determination and MS/MS experiments Turn-key solutions for target compound analysis (dioxins, steroids, etc.)

Versatility:Full data system control of the instrument including peripherals GC and LC interfaces EI, CI, ESI, Micro-ESI, APCI, FD, FI, FAB Rapid interchange of inlet systems and ionization modes Unmatched Ion Trap MSn in all ionization modes Autotuning at any resolution and any ionization modeOnly the ion source requires tuning Mass resolution is set by data system controlled adjustment of the slit widths without additional tuning

Once set, the mass resolution is valid over the full mass range with no mass dependent tuning, at all practical scan speeds.

The MAT 95 XP-Trap systemImagine you would be able to not only get high resolution data or accurate masses, but further analyze highly resolved ions using MS/MS and MSn - with high sensitivity, high collision efficiency and excellent mass resolution of the product ions.

The MAT 95 XP-Trap system offers all these capabilities by combining the high performance magnetic sector mass spectrometer with the unmatched Thermo Finnigan LCQ MSn ion trap system.

This seamless combination extends the power of the ion trap beyond its already outstanding capabilities by:

MSn with all modes of ionization.

MS/MS in the transfer octapole with no low mass cut-off.

MSn with high precursor ion resolution.

MSn of very small signals in the presence of intense interferences.

Plus the regular LCQ features:

Zoom scan for 10,000 (FWHM) mass resolution of product ions.

Resonant excitation for ultimate collision efficiency.

Exceedingly simple experiment setup and operation.

Specifications:

Full acceleration potential for all ionization modes

Resolution better than 60,000

Mass range 3,500 Da at full acceleration potential

Scan cycle time 0.2 s/decade

Multiple ion detection with 33 time windows per analysis and 31 masses per time window