New programmable controller for thickness and coplanarity measurement.

Micro-Epsilon
has launched a powerful, flexible programmable controller that
enables the real time processing (up to 100kHz) of up to six
input and output signals from the companys displacement
sensors.

The controller
is ideal for thickness, height, roundness, flatness or coplanarity
measurements, where signals from multiple displacement sensors
need to be synchronised and processed in real time. All processing
is carried out internally by Micro-Epsilons in-house developed
electronics unit.

The new CSP2008
universal controller is able to process up to six digital or
analogue input signals: two internal (via the front panel), plus
four external signals via plug-in Ethercat modules. Ethercat
can also be used as an external interface for connecting further
sensors and I/O modules.

The controllers
unique, simple to use, intuitive interface enables the user to
configure the unit via a PC or laptop using the Ethernet interface.
All user-selectable functions on the controller and measured
values can be viewed, displayed and stored in real time via the
users own web browser, without having to install any third
party software. Alternatively, simple configuration can be carried
out using ASCII commands via the RS422 interface.

Automatic sensor
recognition means that sensor connection is straightforward and
fast. Sensors can be configured using buttons via the front panel
and displayed on the controller or via a laptop. The CSP2008
is even upgradeable, with users able to use additional I/O modules
for specific applications.

Chris Jones,
Managing Director at Micro-Epsilon (UK) Ltd comments: The
CSP2008 is really a breakthrough in calculating differential
measurements very simply, without complex calibration or PLC
programming. Previously, the user would need a PLC or even a
PC to calculate these types of measurements. This added complexity
and necessitated having to write a user-specific programme. Using
the CSP2008 saves time, money and software development 
its all pre-configured and ready to go!

The controller
has a 52mm x 30mm, high luminance, colour graphics display, which
enables measured values to be read easily from a distance of
up to two metres. Mounting is also straightforward using DIN
rail TS 35.

The controller
is also compact and lightweight, measuring approximately 146
x 97 x 70mm at a weight of just 500g. The unit is protected to
IP40 and can operate from 0 to 50 deg C. ___________________________________________________