ROCKWELL TYPE HARDNESS TESTER

Obtained the supreme precision and overwhelming reliability by the adoption of entirely new designing and precision manufacturing with no compromise!

By the adoption of stealth type double information panel, improved the remarkable visibility and operation!

OUTSTANDING FEATURES

EMPHASIZED HIGH PRECISION

Reviewed the popular FR-e series and achieved the supreme precision.

STRENGTHENED BODY

Achieved the remarkable improvement of hardness measurement by the adoption of newly designed strong body.Prepared the maximum stroke of specimen of 255mm and long type of 405mm.
(from the horizontal anvil to X3 and X3L)

STABLE LOADING AXIS MECHANISM

Adopted highly sensitive load shaft to prevent the side shift by the introduction of ball guide at the top and bottom.
The stable measurement of hardness has become possible for the cylindrical and spherical surface.

STANDARD COMPENSATION FUNCTION

Low, middle, high compensation of multiple points to meet the each standard.
Obtain the test result which is comply with the different standard for all hardness range.

SHAPE COMPENSATION FUNCTION

Obtain the test result which is comply with the standard by entering the diameter of spherical surface & cylindrical specimen.

DISPLAY OF MAX.DEPTH OF INDENTER

Display the max.depth of indenter together with test result.
Be able to evaluate the elastic amount of different sample in hardness value.

STEALTH TYPE DOUBLE INFORMATION PANEL

Display the various information of OK / NO judgement and conversion and at the same time,display the measurement results.(Load and release of the test road, and hardness indication are automatic)

FUNCTION OF DATA JUDGEMENT

Display the judgement of measurement data (HI, OK, LO), if you set the maximum value and minimum one and the mean value, if you set measurement point on the sub-information panel.

LED LIGHT

Obtained the supreme visibility of specimen in the dark environment and achieved the minute positioning.

USB DATA OUTPUT

USB (B2.0) connector is equipped. USB data can be transferred to PC.
Capable in compliance with all ASTM (E-140), SAE (J-417b).