5. The Burkard Personal Spore Trap contains an internal pump that samples at 10LPM. In a routine environment, this pump is set for 9 minutes for a total volume of 90 L. In any analysis situation that is expected to be heavily contaminated with fungal spores the time of sampling should be decreased.

6. The Allergenco, which can provide multiple time-differentiated tests on the same slide, has an internal pump calibrated at 15LPM. The time can be set for anywhere from 6 min for a total volume of 90 L to 10 min for a total volume of 150 L.

7. The BioSIS 2000 is a slit impaction sampler that also impinges the air on a sticky glass slide

Notes:1. These pumps will collect both viable and non-viable particles but the laboratory will report only non-viable due to lack of the presence of the Fungal Fruiting structures.

The Air-O-Cell is a unique sampling cassette specifically designed for the rapid collection and quantitative analysis of a wide range of airborne aerosols. It collects both viable and non-viable particulates such as mold spores, pollen, insect parts, skin cell fragments, fibers, and inorganic particulates.

Advantages:

1. Useful for initial IAQ survey of the biological particulates in the air.
2. East to get quick turn around time.
3. Chance of sample contamination is low.
4. Easy to operate.

Disadvantages:

1. Cannot be cultured.
2. Fungi cannot be speciate to the species level.
3. Penicillium/Asperigllus cannot be differentiated due to similarity in size and color..

2. Remove the tape seals covering the inlet and outlet and placed them on the side of the cassette.

3. Connect the Air-O-Cell cassette to the sampling pump using flexible tubing.

4. Turn the sampling pump on for an appropriate sampling time range from 1 to 10 minutes

5. After sampling is complete, return the tape seals back.

LARO-100

The Laro 100 air-sampling cassette is an ideal method to analyze personal exposure to mold. Mold spores are collected on 0.8 micron mixed cellulose ester filter (like an asbestos PCM cassette). It is capable of capturing 100% of all airborne fungal particulate as well as any particulate or fibers 0.8 microns and greater in size. The sample can be analyzed immediately after collection by optical microscopy.. Overloading the sample can be a problem in dirty environments. Blank samples and outdoor control samples should be submitted with the personal samples. Laro 100 samples can also be cultured by dissolving the filter in water, extracting the water, and culturing the water on an agar plate.

Environmental situation

Flow rate/Sampling time

1

Inside/Outside-with no visible particulate in the air

4 L/min /10 min or longer

2

Special inside clean environments

4 L/min / 30 min or longer

4

Inside-8 hr /personnel monitoring

1.0 - 2.0 L/min / to obtain
300-480 L.

Notes:1. Laro 100 strip is the only one can be cultured.2. Differences between these pumps are: Flow Rate & Efficiency in collecting different sizes of the fungal spores.

The Micro5 Microcell airborne mold-monitoring cassette is another method to analyze personal exposure to mold. The Micro5 cassette contains a glass slide with adhesive that captures the mold spores when they impact the slide. The Micro5 has a 50% collection efficiency of 0.8 microns at 5 liters per minute. Micro5 samples can be analyzed immediately after collection by optical microscopy. The flow rate for the Micro5 must be set at 5 liters per minute to obtain a target sampling volume of 25 liters. Like the Laro 100, overloading the sample can be a problem in dirty environments. Blank samples and outdoor control samples should be submitted with the personal samples. The weakness of this method is the short time duration used to collect the sample. A five-minute sample may not give an accurate exposure concentration for an eight-hour workday. In order to collect 8 hours of sampling data, 96 samples would need to be collected.

Typical Environmental Conditions

Flow rate

1

Clean ``office" or outdoors (no visible dust)

5 Minutes

2

Indoor environment, high activity & personnel

3 Minutes

4

Outdoor environment

3 Minutes

Andersen Impactor

PURPOSE:Identification of culturable microorganisms and assessment of possible proliferation and dissemination of bacteria or fungi from building reservoirs.

1. Select at least three sites, one each to represent complaint area, a noncomplaint area (otherwise as similar as possible to complaint area), and outdoors.

2. In turn at each site, sample simultaneously for fungi, mesophilic bacteria, and thermophilic actinomycetes. Typical sampling time is ten minutes. Before moving to the next site, repeat twice to obtain triplicate, consecutive samples.

3. Load and immediately unload one set of sampling media in each sampler to serve as field blanks.

4. Collect another complete set of samples and blanks on the next day.

SAMPLING Procedure

1. Calibrate each sampling pump with a representative sampler in line.

2. Before each run, carefully and thoroughly wipe each sampler stage with rubbing alcohol. Allow to dry. Make sure air passages are not blocked.

3. Load sampling media into sampler, remove covers from media, and attach sampler to pump with flexible tubing.Note: Take special care to prevent contamination of media during loading and unloading. Do not touch agar surface.

4. Sample at known preset flow for an accurately known time, e.g., 5 min. (In heavily contaminated areas, a shorter sampling time may be necessary.)

5.Replace covers on sampling media, unload, and pack securely

Surface Sampling

Method of choice when there is visible mold growth. Done by using one of the following:

1.Choose the area of most concern for testing. We suggest testing any areas with visible mildew or water damage.

2.Wear a sterilized glove.

3. Take a three-inch piece of clear scotch tape, do not use frosty tapes. Fold one end over, sticky side to sticky side, to make a 1/2-inch tab handle (this will give you something to grasp). Leave the rest of the sticky side exposed but not for a long time.

4. Gently depress the sticky side of the tape against the area you are testing, and lift the tape off. You should try to get a sample about the size of a postage stamp. More is not necessarily better as it will be observed through a powerful microscope.

5. Adhere the sticky side of the tape to the inside of the zip lock bag.

6.Tape the piece of paper with your information and sample location inside the zip lock bag. Place the zip lock bag inside a mailing envelope and mail it to the lab.

Mycotoxin sampling guide

Air sampling

for mycotoxins has important limitations. Concentrations must be at least 100,000 spores or greater to ensure accurate detection of any mycotoxins. For all practical purposes, mycotoxin sampling should be done through bulk, surface or dust sampling.

Bulk samples

should be cut and aseptically removed from the source and placed in a clean, sterile container of pre-sealed zip lock bag. Proper identification of the sample should include date, time, location, temperature and humidity at the time of collection. A sample size of 25-50 grams is normally sufficient.

When possible, bulk samples should be collected that have visible fungi contamination. Bulk materials that can be used for the determination of mycotoxins include such items as wallpaper, cardboard, wood, plasterboard, paper covered gypsum board, mineral wool, plaster, sand, linoleum, polyurethane insulation, pipe insulation and paint chips. There is a wide possibility for a sample source.

Sterile swab samples

should be collected from an area approximately 4” x 4”. This will allow the laboratory to calculate a concentration based on a specific area. The sterile swab should be placed back in the sterile container so that the seal is a positive one. It is a good idea to seal the vial with a piece of tape to ensure that it doesn’t come apart during transport to the lab. Any cross contamination will render the sample useless. Surface sampling with sterile swabs is a good technique. It is non destructive. It is easily accomplished and the laboratory can get good results.

Dust samples

for mycotoxin identification are another good method of collection. With this technique it is important top collect a second sample from an area assumed to be non contaminated for comparison. A vacuum pump pulling approximately 15 liters per minute will do the task nicely. The sample should be collected from a location where obvious dust has accumulated.

It will be necessary to allow 2 to 5 minutes of sampling in order to collect enough material for proper identification. A clear filter is preferable because you can see if it starts to get overloaded with material. Too much material in the sampling cassette can actually hinder the process.

Multiple samples

should be collected fro comparison studies. At the current time there are no governmental or industrial regulations concerning allowable mycotoxins or toxigenic mold spores in indoor environments. Your best insurance against problems is to address the problem when the compliant come sup, identify any actual or potential fungal issues and address a remediation plan.