Horoscope

Sunday, 23 October 2011

Laboratory Instruments

Laboratory Instruments

Laminar Air Flow Horizontal

The
bench is made of thick ply board, and reinforced with teak wood where
every necessary. All visible surface are laminated with white decolam
Lamination. All interior surfaces are painted with air drying Epoxy
paint. The filter seat has rubber padding for a perfect rubber to rubber
sealing. The Front & side panels are made of Perspex sheet.

At
the delivery end, the unit is provided with Super Air H.E.P.A. Filters,
Comprising of non-woven microfibre glass pleated media. The pleats are
separated by corrugated aluminium foil separators. Initial pressure drop
is less then 25mm W.C. at rated flow D.O.P. efficiency is better than
99.97% down to 0.3% microns particle size. The HEPA Filters are
protected by a plastic/Aluminium grill.

FEATURES :-

•

Wooden/M.S Duley Power Coated /Stainless Steel Body.

•

HEPA Filter

•

Pre-Filter

•

Blower (Motor Blower Assembly)

•

U.V. Light

•

Fluorescent Lamp

•

Operating Switch

•

Pressure Gauge (Manometer)

•

Stainless Steel Table

•

High quality HEPA filter provides protection for the product (filtration efficiency of more than 99.97% at 0.3 micron.

•

Ultra-quiet noise level of not more than 65 dBA.

•

Durable and easy to clean stainless steel work.

•

Cock for gase or vacuum line.

•

OPTIONAL ACCESSORIES :-

•

Particle Dust Count-Validation at site.

•

Timer for U.V. Lights.

•

Speed Regulator

•

Caster Wheels.

•

Magnahelic Manometer.

Laminar Air Flow Horizontal With Stand

FEATURES :-

•

Wooden/M.S Duley Power Coated /Stainless Steel Body.

•

HEPA Filter

•

Pre-Filter

•

Blower (Motor Blower Assembly)

•

U.V. Light

•

Fluorescent Lamp

•

Operating Switch

•

Pressure Gauge (Manometer)

•

Stainless Steel Table

•

High quality HEPA filter provides protection for the product (filtration efficiency of more than 99.97% at 0.3 micron.

•

Ultra-quiet noise level of not more than 65 dBA.

•

Durable and easy to clean stainless steel work.

•

Cock for gase or vacuum line.

•

Work area properly illuminated by diffused glare free fluorescent light.

Laminar Air Flow Vertical

Laminar Air Flow Vertical

1

The bench is made of thick ply board, and reinforced with teak wood
where every necessary. All visible surface are laminated with white
decolam Lamination. All interior surfaces are painted with air drying
Epoxy paint. The filter seat has rubber padding for a perfect rubber to
rubber sealing. The Front & side panels are made of Perspex sheet.

At the delivery end, the unit is provided with Super Air H.E.P.A.
Filters, Comprising of non-woven microfibre glass pleated media. The
pleats are separated by corrugated aluminium foil separators. Initial
pressure drop is less then 25mm W.C. at rated flow D.O.P. efficiency is
better than 99.97% down to 0.3% microns particle size. The HEPA Filters
are protected by a plastic/Aluminium grill.

The unit is
illuminated by fluorescent tube lights and is designed to provide a work
area confirming to Class 100 of revised US Federal Standard 209B.

Laminar Flow Cabinets are self-contained workstation units available in
either horizontal or vertical laminar flow models. They can be placed
within a clean room to create an ultra-clean environment or used to
provide an individual clean air environment for small items not
requiring a full-size Clean room.

Laminar Air Flow

For
some applications it may not be ideal to use either a horizontal
laminar flow hood or a vertical. For example, where an operator has to
work on Celling / Filling machine . This simple and portable unit
produces a unidirectional flow. A bacteria free air space at an
inclination over the work area. It is specially designed to minimize the
risk of transferred contamination. the unit is self supported from the
floor and totally isolated from the Celling / Filling machine. Rest of
the features are same as of LAMINAR .

FEATURES

•

Wooden/M.S Duley Power Coated /Stainless Steel Body.

•

HEPA Filter

•

Pre-Filter

•

Blower (Motor Blower Assembly)

•

U.V. Light

•

Fluorescent Lamp

•

Operating Switch

•

Pressure Gauge (Manometer)

•

Stainless Steel Table

•

High quality HEPA filter provides protection for the product (filtration efficiency of more than 99.97% at 0.3 micron.

OPTIONAL ACCESSORIES:-

•

Particle Dust Count-Validation at site.

•

Timer for U.V. Lights.

•

Speed Regulator

•

Caster Wheels.

•

Magnahelic Manometer.

Clean Air Tent

These are overhead clean air modules mounted on roof or with steel
frame on the ground. Units can be built with / without casters so the
clean enclosure can be moved over equipment or used to isolate a
controlled process. It is basically used for filling, refilling weighing
and sampling/Dispensing of Raw Material, Basic Compounds, Active
Compounds, works on the principal of laminar air flow with HEPA Filters
& Pre-Filters.

It provides:- This will act as a Class 100
Room. Protection of the personnel in the cabinet from the processed
substances by vertical flow of clean air protection of the raw and
intermediate products to be processed. Protection of the room where the
cabinet is built up and environment from highly active substance which
otherwise would disperse in the room and escape in the atmosphere
unfiltered. Safe collection and deposition of substance released during
work.

•

Wooden/M.S Powder Coated /Stainless Body.

•

HEPA Filter

•

Pre-Filter

•

Blower (Motor Blower Assembly)

•

U.V. Lamp

•

Fluorescent Lamp

•

Operating Switch

•

Static Pressure Gauge (Manometer)

•

Stainless Steel Table

•

High quality HEPA filter provides protection for the product (filtration efficiency of more than 99.97% at 0.3 micron.

•

Ultra-quiet noise level of not more than 65 dBA.

•

Durable and easy to clean stainless steel work.

•

Work area properly illuminated by diffused glare free fluorescent light.

•

Stainless steel work.

OPTIONAL ACCESSORIES:-

•

Cold rolled M.S. sheet body duly powder coated.

•

Stainless Steel Body.

•

Particle Dust Count-Validation at site.

•

Timer for U.V. Lights.

•

Speed Regulator

•

Stainless Steel Body duly powder coated.

•

Caster Wheels.

•

Magnahelic Manometer.

SAMPLING / DISPENSING BOOTHS

These
are overhead clean air modules mounted with Stainless steel frame on
the ground. Units can be built with / without casters so the clean
enclosure can be moved over equipment or used to isolate a controlled
process. It is basically used for filling, refilling weighing and
sampling / Dispensing of Raw Material, Basic Compounds, Active
Compounds. It works on the principal of Laminar Air Flow with HEPA
Filters & Pre-Filters.

This will act as a 100 Room.
Protection of the personnel in the cabinet from the processed substances
by vertical flow with Reverse Flow of clean air protection of the raw
and intermediate products to be processed. Protection of the room where
the cabinet is built up and environment from highly active substance
which otherwise would disperese in the room and escape in the atmosphere
unfiltered. Safe collection and deposition of substance released during
work.

FEATURES

•

Wooden/M.S Powder Coated / STAINLESS STEEL BODY (304)

•

HEPA Filter (0.3 micron)

•

Fine Filter (10-5 micron)

•

Pre-Filter (10 micron)

•

Motor Blower Assembly

•

U.V. Light

•

Fluorescent Lamp

•

Operatating Switch

•

Static Pressure Manometer

•

Stainless Steel Standing Base

•

High quality HEPA filter provides protection for the product (filtration efficiency of more than 99.97% at 0.3 micron.

•

Ultra-quiet noise level of not more than 65 dBA.

•

Durable and easy to clean stainless steel body.

•

Work area properly illuminated by diffused glare free fluorescent light.

•

PVC Flexible Curtains on all the three sides.

OPTIONAL ACCESSORIES:-

•

Particle Dust Count-Validation at site.

•

Timer for U.V. Light.

•

Speed Regulator

•

Caster Wheels.

Magnahelic Manometer.

Inclined Laminar Air Flow For Filling & Ceiling Machine

For
some applications it may not be ideal to use either a horizontal
laminar flow hood or a vertical. For example, where an operator has to
work on Celling / Filling Machine. This simple and portable unit
produces a unidirectional flow. A bacteria free air space at an
inclination over the work area. It is specially designed to minimize the
risk of transferred contamination. the unit is self supported from the
floor and totally isolated from the Celling / Filling machine. Rest of
the features are same as of LAMINAR .

FEATURES

•

Wooden/M.S Duley Power Coated /Stainless Steel Body.

•

HEPA Filter

•

Pre-Filter

•

Blower (Motor Blower Assembly)

•

U.V. Light

•

Fluorescent Lamp

•

Operating Switch

•

Pressure Gauge (Manometer)

•

Stainless Steel Table

•

High quality HEPA filter provides protection for the product (filtration efficiency of more than 99.97% at 0.3 micron.

•

Ultra-quiet noise level of not more than 65 dBA.

•

Durable and easy to clean stainless steel work.

•

Cock for gase or vacuum line.

•

Work area properly illuminated by diffused glare free fluorescent light.

OPTIONAL ACCESSORIES:-

•

Particle Dust Count-Validation at site.

•

Timer for U.V. Lights.

•

Speed Regulator

•

Caster Wheels.

•

Magnahelic Manometer.

Laminar Air Flow Table Top Model

The
bench is made of thick ply board, and reinforced with teak wood where
every necessary. All visible surface are laminated with white decolam
Lamination. All interior surfaces are painted with air drying Epoxy
paint. The filter seat has rubber padding for a perfect rubber to rubber
sealing. The Front & side panels are made of Perspex sheet.

At the delivery end, the unit is provided with Super Air H.E.P.A.
Filters, Comprising of non-woven microfibre glass pleated media. The
pleats are separated by corrugated aluminium foil separators. Initial
pressure drop is less then 25mm W.C. at rated flow D.O.P. efficiency is
better than 99.97% down to 0.3% microns particle size. The HEPA Filters
are protected by a plastic/Aluminium grill.

The unit is
illuminated by fluorescent tube lights and is designed to provide a work
area confirming to Class 100 of revised US Federal Standard 209B.

Laminar Flow Cabinets are self-contained workstation units available in
either horizontal or vertical laminar flow models. They can be placed
within a clean room to create an ultra-clean environment or used to
provide an individual clean air environment for small items not
requiring a full-size Clean room.

Biological Safety Cabinets (biohood) Class Ii

Applications
include isolation and identification of plant/human pathogens, work in
virology, tissue culture, sterility testing and recombinant DNA
research. This Biohazard Cabinet offers operator protection from low to
moderate risk biological agents (or chemical carcinogens(, while
providing a sterile work environment for the procedure. The cabinet is
kept under negative pressure with partial quantity of recirculated air
exhausted through a HEPA filter. External exhaust from the laboratory is
optional.

The Biological Safety Cabinet is similar in performance and specification to the Vertical Series, described on reserse.

Integral
powerful exhaust system. Built-in-positive Air Curtain effect to
prevent inside air from going out and room air from entering in,
providing total process and operator safety.

•

Exhaust ducting to carry the air outside.

•

Front sliding door, U.V. light, Manometer, Gas cock etc. optional.

AIR
FLOW:- 70% of the air is recirculated back into the work area and 30%
of the air is HEPA filtered and exhausted out to the environmental via a
dedicated duct without mixing with the recirculated air.

STANDARD SIZE:-

2' x 2'

3' x 2'

4' x 2'

6' x 2'

8' x 2'

FACE VALUE:- a cafe velocity of 90±20 cfm is maintained.

PRESSURE MODULE

Positive
pressure modules are designed to provide a clean area within a small
enclosure or cell. These pressure module adopts the same Laminar Air
Flow Principle. It generate positive pressure where the Room is totally
Closed.It provides a source of Fresh Air in the Room.

FEATURES

•

Wooden/M.S Duley Power Coated /Stainless Steel Body.

•

HEPA Filter

•

Pre-Filter

•

Blower (Motor Blower Assembly)

•

Pressure Gauge (Manometer)

•

High quality HEPA filter provides protection for the product (filtration efficiency of more than 99.97% at 0.3 micron.

•

Ultra-quiet noise level of not more than 65 dBA.

•

Durable and easy to clean stainless steel work.

•

Work area properly illuminated by diffused glare free fluorescent light.

OPTIONAL ACCESSORIES:-

•

Particle Dust Count-Validation at site.

•

Timer for U.V. Lights.

•

Speed Regulator

•

Caster Wheels.

•

Magnahelic Manometer.

A.c. Module

It
is useful in Front of Air Conditioner to provide a sterile, bacteria
free & dust free Air of 0.3 microns in a particular temperature . It
works on the Laminar Air Flow Principle .

SPECIFICATIONS:- Sizes Available asper the size of A.C. (Window/split type)

Incinerator Lab Model

Incinerator

The
Incinerator is our recent development; it was initially designed to
meet the requirement of medical colleges and small hospitals to dispose
off their bio-medical waste in an environmental friendly manner. The
growing awareness of cleaner environment the need for incinerators is
growing in other sectors as well.
Our recent addition in product
list is bird and small animal incinerators that can be fuelled by
electric power, oil or gas and fully automatic. Our design team can
design and build even larger incinerators to meet the requirement of
municipal wastes, kitchen waste etc.
Our incinerators are built
to the need of our customers to improve the environment to control
hazardous solid waste. The incinerators reduces most combustible solid
wastes to nearly 2 -3 percent by volume and 5 percent by weight.

Key features of our incinerators are :

Robust construction.

Virtually smoke and smell free with secondary burners.

High temperature refractory lining.

Quick heat up time and lesser operational time.

Easy installation.

Trouble free operation and maintenance, our mechanics are available on call.

Designed to meet standards of Central Pollution Control Board (CPCB) Delhi.