2537 Paddlewheel Flow Sensor

The GF Signet 2537 Paddlewheel Flow Sensor represents an enhancement to a recognised industry standard with the added functionality of various output options, including flow switch, pulse divider, digital (S3L) or 4 to 20mA. Additionally, it offers low flow, low power and high resolution options and can be configured on-site directly through the built-in user interface. Installation is simple because the Signet 2537 utilizes the same fittings as the popular Signet 515 and 2536 Paddlewheel Sensors, and fits into pipe sizes ranging from DN15 to DN200. Available in a variety of wetted materials, such as Polypropylene and PVDF, it is ideal for a variety of applications including chemical processing, water and wastewater monitoring and scrubber control.

Product Specification:

The Signet 2537 Paddlewheel Flow Sensor represents an enhancement to a recognised industry standard with the added functionality of various output options, including flow switch, pulse divider, digital (S3L) or 4 to 20mA.​

Specification Detail of the Signet 2537 Paddlewheel Flow Sensor:

Pipe SizeDN15 – DN200

Material BodyPolypropylene or PVDF

Flow Range1.2 l/min – 51,951 m3/hr

Linearity1%

Operating Temp Range-18 to 85ᵒC

Pressure Range1.7 – 14 bar

Outputs & CommsDigital (S³L), 4 to 20 mA

Insertion / InlineInsertion

Fluid TypeClean water, dirty water, chemicals

Applicable LiquidsWater, waste water, chemicals

Power12-24 VDC

ApprovalsCE, FCC, UL

Connection TypeScrewed (custom fittings)

Top Companion Products

Looking for help with flow rate, pipe diameter or flow velocity? Our flow rate calculator will provide the answer.

This is the background of the Bottom Area shown below. It'll display correctly on published site.
If you don't want background for the bottom area, simply hover over the bottom of this image to delete it. The background of the bottom area will be a solid dark color.