Cookies on the OMEGA websitesWe use cookies on this website, these cookies are essential for the website to work correctly.If you continue without changing your settings, we'll assume that you are happy to receive all cookies on this website.To find out more information about these cookies please click here.

The circular char recorder CTXL Series retains the proven CT485
design and are state-of-
the-art circular chart recorders,
available in four models CTXL-TRH,
CTXL-DTC, CTXL-DPR, and CTXL-PH.

The CTXL-TRH is a temperature/relative humidity chart recorder with
a temperature range of -17 to 49ºC, and a humidity range
of 2 to 98%. The CTXL-DTC is a
dual thermocouple input chart
recorder that accepts J, K, or T
thermocouple probes. The
CTXL-DPR is a dual-process input
chart recorder that accepts 0 to 1 Vdc,
0 to 5 Vdc, 0 to 10 Vdc, 0 to 20 mA,
and 4 to 20 mA inputs. The CTXL-PH model accepts a pH electrode with a BNC connector and a Pt100 sensor for automatic temperature compensation (pH electrode sold seperately).

Models have dual backlit LCD's that
display two inputs in real-time. The
charted data is stored in the nonvolatile
memory that can be
downloaded to a PC at a later time.
This provides an electronic copy of
the charted data. In addition, the
CTXL Series provides many
powerful functions such as audible
and visual alarms with built-in relay
contacts; real time clock to monitor
and store data vs. time; monitoring
of the Minimum, Maximum, and
Average values; 1, 7, or 32-day
recording; chart re-scaling, and more.

The high and low alarm set-points
for the two channels can be
programmed and enabled using the
front panel keypad and comes with
two built-in relays for high alarms,
and two additional voltage outputs
to drive external mechanical relays
for the low alarm conditions.

The models have unique features including double-sided
chart paper and the magnetic hub.
The linear pen and chart drives
make recorded values easy to read and interpret. All models
come with a stabilising arm for
bench top operation, and a decorative
foot cover for wall mount applications.

The model has a hidden key (White
Box) to enable/disable unauthorised
access to the front keypad. This will
keep away any unauthorised personnel
from changing the recorder settings.

Place Order(Specify Model Number)

What Other People Bought: When you see this icon, click on it to expand a list of products that other people have bought when they purchased this model.

Part Number/ Desc.

Qty

RoHS

Description

RoHS

The part number is RoHS Compliant.

The part number is NOT RoHS Compliant.

RoHS Compliancy of the part number is undetermined.

May be ordered as RoHS compliant, however, that requirement must be specified at the time of ordering by adding "-rohs" to the end of the part number.