Cyclic Compression Porometer - China

The instrument applies cyclic stress on a sample and measures the pore structure characteristics after a desired number of cycles. The stress cycles and testing after any desired number of cycles are automatically continued. Operation Principle The instrument applies cyclic stress on a sample and measures the pore structure characteristics after a desired number of cycles. The sample is loaded in the sample chamber and is subjected to stress cycles in the specified stress limits. At the end of the desired number of stress cycles, the pressure of a non-reacting gas on one side of the sample is increased to initiate gas flow through pores. The gas pressure and flow rates are measured. The pores in the sample are spontaneously filled by a wetting liquid. The gas pressure and flow rates are measured through the wet sample. After acquisition of data, the sample is re-wetted and again subjected to cyclic compression. Pressure and flow rates are measured after the desired number of cycles. The test is continued to acquire data as a function of number of stress cycle. Application The Cyclic Compression Porometer is used to test the sample and measure the maximum pore diameter, mean flow pore diameter and pore distribution. Theses data represents the actual characteristics of pore structure.
Applied industries：
Automobile manufacturing Fuel battery pack Paper Battery separator Geofabrics
Filtration Nonwoven Wool felt Features · acquires data, continues to cyclically stress the sample
· Very little operator involvement
· Operator adjustable: stress limits in a cycle; number of cycles after which data is to be acquired ; stress free duration for data acquisition ; total number of cycles.
· Concurrent measurement of compressive stress in the sample as a function of stress cycle. Technical Parameters Items Parameters
Pore Size Range 0.013 - 500 microns
Permeability Range 1 x 10-3 - 50 darcies
Sample size 1.75” - 2.5” diameter
Pressure range 0~500psi
Pressurizing gas Clean, dry, and compressed air or nonflammable and noncorrosive gas
Pressure transducer range 0~500psi
Resolution 1 in 60, 000
Accuracy 0.15% of reading
Mass flow controller range 10cc/minute~100cc/minute
Power supply 110/220VAC, 50/60Hz (Others Available)
Dimension D30” H x 19” W x 18.5” D
Weight 100lbs

Related listings

With test rigs of series DHP 145 it is possible to test tightness against over and low-pressure of components of different dimensions and shapes. The leak rate is determined depending on pressure. Overpressure is accomplished with compressed air (fro...

This air filter test system is designed to meet requirements of EN1822-4 Appendix E “efficiency leakage test for particle sizes from 0.3 μm to 0.5 μm”. This part of the standard is related to filter elements which cannot be scanned due to its geometr...

The Gravimetric Measurement System allows the comfortable detection of oil content in blow-by after oil mist separators with a filter blank sheet or filter cartridge directly on engines or engine test stands. Decrease of oil consumption as well as re...

Safe Trade Tips

Ask for business references and check them

Use a payment method that offers better protection for all such as Letters of credit and verified professional escrow services. Avoid payments in advance such as money transfers

Verify their business via their local Chamber of Commerce

Search the internet using their website address, their business name, their phone and fax numbers, and their email addresses to see if you can find any feedback about them.