Thanks to a reply from Angus the 53131A is sending data every sample
now!!!
"Something to check - in the 'Utility' menu, if the HP-IB is set to
'Talk' rather than a numeric address, it does just what you describe
and only sends data every few seconds.
Angus.
Thanks!!!!
Corby