USB TC-08

The TC-08 thermocouple data logger is designed to measure a wide range of temperatures using any thermocouple that has a miniature size thermocouple connector. Additionally, the TC-08 can also measure other sensors using a 70 mV range.

Featuring built-in cold junction compensation (CJC), the TC-08 has an effective temperature range of -270 to +1820 °C (The actual temperature range depends on the thermocouple being used.)

All the benefits of USB

The TC-08 connects to the USB port of a Windows based PC and enables the host PC to automatically detect the TC-08 — avoiding the need for any complex setup procedures. The USB connection also allows the TC-08 to be powered directly by the USB bus — eliminating the need for an external power supply and making the TC-08 ideal for measuring temperatures both in the lab and in the field.

The TC-08’s USB connectivity allows up to 20 TC-08 USB data loggers to be used simultaneously on one PC — making the TC-08 ideal for use as a large multi-channel temperature acquisition system with a cost per channel significantly lower than other systems of this size.

Fast and accurate temperature data acquisition

With the TC-08 thermocouple data logger your measurements can be made both fast and accurately.

The low conversion time of the TC-08 means up to 10 temperature measurements can be taken every second. Whilst the high (20-bit) resolution ensures that the TC-08 can detect minute changes in temperature.

Single Channel Terminal Board

The USB TC-08 Single-Channel Terminal Board is an accessory for the USB TC-08 Thermocouple Data Logger. The screw terminals allow wires to be attached to the data logger without soldering and enable the USB TC-08 to measure voltages from 0 to +5 V, or 4-20 mA loop current.