Www.pro-lite.co.uk www.photometrictesting.co.uk The Top Ten Errors (and Misconceptions) in LED Photometry LED Specifications: the Truth, the Whole Truth.

Similar presentations

Presentation on theme: "Www.pro-lite.co.uk www.photometrictesting.co.uk The Top Ten Errors (and Misconceptions) in LED Photometry LED Specifications: the Truth, the Whole Truth."— Presentation transcript:

2
www.pro-lite.co.uk www.photometrictesting.co.uk Can We Trust LED Specifications? Many people distrust LED and SSL specifications. Why? Well consider how LED specifications are determined The importance of absolute photometry The common errors made in photometry (some are LARGE!) Understanding why CCT is a bad metric

3
www.pro-lite.co.uk www.photometrictesting.co.uk We tested an off-the-shelf GU10 LED replacement: ClaimedActual Luminous Flux (lumens) 320252 CCT (Kelvin)3,0002,900 CRI (R A )N/S72 Power (Watts)55 Beam Angle45° The luminous flux was tested in a 1m integrating sphere at 25.5°C after a 20 minute stabilisation time Call me an Old Cynic.....

4
www.pro-lite.co.uk www.photometrictesting.co.uk In the Good Old Days..... Relative Photometry ruled: Fluorescent lamp output is constant regardless of what fitting it is put into Multiply lamp flux by fitting LOR to calculate flux of luminaire Photometric data formatted in units of candelas per 1000 lumens

5
www.pro-lite.co.uk www.photometrictesting.co.uk Relative photometry is no longer applicable to SSL: LED output depends upon operating temperature In turn it depends upon the thermal design of the fitting For most SSL we have to perform absolute photometry However with SSL...

6
www.pro-lite.co.uk www.photometrictesting.co.uk How Are LED Specifications Tested? LED specifications – fact or fiction? LED specifications are determined under idealised laboratory conditions: short flash of current, junction temperature of 25°C An LED will typically operate at >60°C in a luminaire If you use ten 100 lumen LEDs in a luminaire, you must not expect to generate 1,000 lumens from the luminaire, you will probably get 750-800 lm in practice As LEDs get hotter: Luminous flux DECREASES CCT INCREASES CRI changes

10
www.pro-lite.co.uk www.photometrictesting.co.uk Measuring Total Luminous Flux Luminous flux – in lumens – is usually measured in an integrating sphere. However..... Not all spheres are created equal Beware SHADOWS! Size IS important! You might be suffering from self absorption

11
www.pro-lite.co.uk www.photometrictesting.co.uk Shadowing Effects in Spheres The perfect sphere is perfectly spherical, has no holes and is painted with a 100% reflective, perfectly Lambertian coating. The aim is to achieve a perfectly uniform luminance distribution over the surface of the sphere

12
www.pro-lite.co.uk www.photometrictesting.co.uk Low Reflectance = High Errors Recent studies have shown that low reflectance coatings introduce a significant directionality to integrating sphere readings: The lower the reflectance, the more directionally sensitive Significant errors with spotlights +15% to -22% in a sphere painted with an 80% coating

13
www.pro-lite.co.uk www.photometrictesting.co.uk Sphere Size Does Matter The perfect sphere is perfectly impossible! There is no one-size-fits- all policy when it comes to integrating spheres. If the light source is too big, the sphere wont integrate properly, giving large errors. Rules of thumb: For 4 measurements of 2D/3D luminaires, sphere diameter should be 10x source size or 1.5x for linear lamps (3% surface area per IES LM79-08) For external 2 measurements, sphere diameter should be 3x source size Credit: DW Windsor Ltd

14
www.pro-lite.co.uk www.photometrictesting.co.uk Are You Self Absorbed? The light source placed inside a sphere both emits light and absorbs some of that light. This results in the test lamp flux being measured LOW (usually by some tens of percent – a significant source of error). Also effects CCT & CRI. These errors are simply corrected using an auxiliary lamp.

15
www.pro-lite.co.uk www.photometrictesting.co.uk #4: Use a high quality integrating sphere – a wooden box or other geodesic shape painted with Dulux white vinyl matt will NEVER give good results! #5: Make sure your sphere is large enough – if you can barely squeeze the fitting inside youll get shadowing errors #6: Correct for self absorption – if you dont, youll probably sell yourself very short! Dr Johns Top Tips

18
www.pro-lite.co.uk www.photometrictesting.co.uk #7: Use the old 5x fitting size rule at your peril – placing the photometer at a distance of 5x the light sources luminous aperture (e.g. 7.6m for a 5 foot fitting) was OK for extended, diffused sources but for narrow beam angle sources you could be measuring in the near-field and seeing very low candela readings #8: Think about your scan resolution – its no good scanning a 15° spot with a 5° increment – youll only be taking 2-3 slices through the beam (beam shape will be inaccurate) Dr Johns Top Tips

19
www.pro-lite.co.uk www.photometrictesting.co.uk CCT is a Really Bad Metric Correlated colour temperature (CCT, Kelvin) is a simplified metric used to indicate the colour of white light produced by a lamp. The issue with CCT is that two light sources can have the same CCT but look completely different colours. The problem is down to the mathematical definition of CCT and our high sensitivity to colour difference.

20
www.pro-lite.co.uk www.photometrictesting.co.uk The Definition of CCT In 1960 CIE UCS, lines of iso-CCT fall within ± 0.02 duv Two LEDs with the same CCT can have distinctly different colours. The difference can be as much as ± 0.02 duv which is about 20 times greater than the minimum perceivable colour difference

21
www.pro-lite.co.uk www.photometrictesting.co.uk #9: As a luminaire designer, under no circumstances should you use CCT as a purchasing metric – to stand any chance of colour matching LEDs or LED modules you have to specify colour in terms of the CIE chromaticity coordinates, Cx & Cy. And then negotiate binning tolerances with your vendor Dr Johns Top Tips

22
www.pro-lite.co.uk www.photometrictesting.co.uk White Light from Blue - Fluorescence Differences in the thickness or optical path length through an LED phosphor can lead to significant differences in the correlated colour temperature (CCT) at different angles from the LED. When the phosphor is thinner, less blue light is converted and the CCT is higher; when it is thicker, more blue light is converted and the CCT is lower

23
www.pro-lite.co.uk www.photometrictesting.co.uk Colour Shift Through an LED Phosphor This is the colour variation from a remote phosphor LED spotlight. In the centre of the beam, the CCT is 7,500 Kelvin. At 30°, the CCT has dropped to 4,500 Kelvin. An integrating sphere measurement would only yield the average CCT.

24
www.pro-lite.co.uk www.photometrictesting.co.uk #10: Beware the effect of colour shift from white LEDs – the rated CCT is probably based upon a directional measurement in the centre of the beam or an average value if performed using an integrating sphere. Ask to see evidence of the colour shift with angle from the LED or LED module. Dr Johns Top Tips

25
www.pro-lite.co.uk www.photometrictesting.co.uk Photometric Measurements Dont Have to be Testing

About project

Feedback

To ensure the functioning of the site, we use cookies. We share information about your activities on the site with our partners and Google partners: social networks and companies engaged in advertising and web analytics. For more information, see the Privacy Policy and Google Privacy &amp Terms.
Your consent to our cookies if you continue to use this website.