Ideal for use in both new systems as well as retrofit rooms. The lightweight cassette sized DR wireless panel uses state of the art acquisition software to capture images within 10 seconds from exposure.

Panel Specifications:

Scintillator: CsI Direct Deposit

Active Area: 434 x 355mm

Pixel Pitch – 154um

Pixel Array – 2816 x 2304

Limiting Resolution 3.3 lp/mm

Exposure Control: Full-Field Automatic Exposure Detection (F2AED)

Software Features:

Generator Integration

Customizable study list

Standard tools on mouse

Multi-Scale Contrast Enhancement & Dynamic Range

Optimized Image Processing Parameters

CareView® Series Panels

CareView® series x-ray flat panel detectors incorporate the proven technology of amorphous silicon. The amorphous silicon photodiode array and TFT underneath convert invisible X-ray photons into visible photons, and then into electric charges. The electron signals are measured and amplified in the data line, and then converted to digital signals by the Analog/Digital converter. Finally, the digital signals are transmitted to a PC for processing to form an image.