Discussion Agenda :

Front Panel and Basic Functionalities :

Front Panel and Basic Functionalities

Display Interface and Basic Functionalities :

Display Interface and Basic Functionalities

User’s Key points :

User’s Key points We must want to know some important key points before using the equipment under consideration.
General check
Functional check
Probe compensation
Automatic settings for waveform display
Getting to know the vertical system
Getting to know the horizontal system
Getting to know the trigger system

General Check :

General Check Check the unit for possible shipping damages
Check the accessories
Thorough inspection of the entire unit

Functional Check :

Functional Check Power on the unit and default setup
Accessing signals

Probe Compensation :

Probe Compensation Adjust the probe on 1x or 10x according to the signals.
If you see a “Undercompensation” or “Overcompensation” waveform display, adjust the variable capacitor on the probe with a screwdriver with non-metal handle, until a “Correct Compensation” waveform

Automatic settings for waveform display :

Automatic settings for waveform display Connect the signal to be measured to the signal input channel.
Press [AUTO]. The oscilloscope will automatically set the vertical deflection factor, scanning time base and trigger mode.

Vertical System :

Vertical System Turn the vertical position knob to display the signal in the centre of the window.
Change the vertical setups and check changes of status information.

Horizontal System :

Horizontal System Use the horizontal scale knob to change the horizontal time base setup and check any changes in status information.
Range of horizontal scanning rate is 5ns~50s, in steps of 1-2-5.
Use the horizontal position knob to adjust the
horizontal position of the waveform window
Press [MENU] to display the ZOOM menu.
Shortcut Key Return to Zero!

Trigger System :

Trigger System Use the trigger level knob to change the trigger level.
Open the [TRIGGER MENU] to change trigger setups.
Press [F1] and select EDGE TRIGGER
Press [F2] and set TRIGGER SOURCE to CH1
Press [F3] and set EDGE TYPE as RISING
Press [F4] and set TRIGGER MODE as AUTO
Press [F5] and set TRIGGER COUPLING as DC
Press [50%] and set the trigger level at the vertical
centre point of the trigger signal amplitude.