The MCR20 probe module changing rack is designed to securely store stylus modules ready for rapid automatic changing, whilst protecting mating surfaces from any airborne contaminants within the working environment.

Probe maintenance - CK200 (A-1085-0016) is a specialised cleaning material supplied for the removal of contamination from the location faces of the magnetically retained kinematic couplings of the TP20, TP200 and SP25M probe systems. The frequency of cleaning should be determined by the conditions of use.

The standard force stylus modules, identified by black caps, are suited to the majority of applications.

300 mm (11.81 in)

300 mm (11.81 in)*

300 mm (11.81 in)*

22 g (0.78 oz)

28 g (0.99 oz)

30 g (1.06 oz)

MF

The medium force stylus module, identified by a grey cap, is for use where a higher trigger force than standard is required.

300 mm (11.81 in)

22 g (0.78 oz)

EF

The extended force stylus module is identified by a brown cap. Typically, this stylus module will only be required with large stylus assemblies, and where spurious ‘air' triggers caused by machine vibration and acceleration, preclude the use of either SF, LF or MF modules.

300 mm (11.81 in)

22 g (0.78 oz)

6W

The 6-way stylus module, identified by a blue cap, has been designed for applications requiring measurement in the –Z direction, for example when measuring the width of undercuts.

Renishaw’s websites and Web shop uses cookies (including third party cookies). Some of these cookies are essential to make our site work properly and others are non-essential but help us to improve the site, including language and location settings and site analytics. You can read more about how we use cookies and how to configure or disable them on our site. By continuing to browse this site you agree to our use of cookies.