We use cookies to ensure that we give you the best experience on our website. If you click 'Continue' we'll assume that you are happy to receive all cookies and you won't see this message again. Click 'Find out more' for information on how to change your cookie settings.

Introduction of a new mobile-bearing total knee prosthesis: minimum three year follow-up of an RCT comparing it with a fixed-bearing device.

As part of the step-wise validation of a new prosthesis (TMK), we previously published the 1 year results of a randomised controlled trial in patients undergoing bilateral knee replacement [Price A., Rees J., Beard D., Juszczak E. et al. A mobile-bearing total knee prosthesis compared with a fixed-bearing prosthesis. JBJS B 2003;85-B-1:62-7.]. Forty patients had the new mobile-bearing prosthesis implanted in one knee and an established fixed-bearing device in the other (AGC). We now report the 3 year status of these patients and, in addition, review a separate multi-centre cohort of 172 patients who had undergone unilateral arthroplasty with the TMK. No significant differences were found in outcome (American Knee Society Score and Oxford Knee Score) between the two prostheses. The greater incidence of "clicking" in the mobile-bearing knee, reported in the previous review, persisted (TMK=48%, AGC=30%). The presence of this mechanical noise was found to have no relationship with outcome in either of the prostheses. The unilateral cohort study showed an acceptable complication rate for the new prosthesis, although some patients reported subjective instability. The method of controlled introduction of the TMK, of which this constitutes a further step, has allowed us to assess the significance of a reported problem (clicking) and to provide scientific data from which other surgeons can decide about use of the implant.