LIS Resources

This assay is performed on an inductively coupled plasma-mass spectrometer. Calibrating standards and blanks are diluted with an aqueous basic diluent containing internal standards. Quality control specimens and patient samples are diluted in an identical manner. In turn, all diluted blanks, calibrating standards, quality control specimens and patient specimens are aspirated into a pneumatic nebulizer and the resulting aerosol directed to the hot plasma discharge by a flow of argon. In the annular plasma the aerosol is vaporized, atomized, and then ionized. The ionized gases plus neutral species formed in the annular plasma space are aspirated from the plasma through an orifice into a quadrupole mass spectrometer. The mass range from 1 amu to 263 amu is rapidly scanned multiple times and ion counts tabulated for each mass of interest. Instrument response is defined by the linear relationship of analyte concentration vs. ion count ratio (analyte ion count/internal standard ion count). Analyte concentrations are derived by reading the ion count ratio for each mass of interest and determining the concentration from the response line.(Unpublished Mayo method)

Creatinine is measured using an enzymatic method based on the determination of sarcosine from creatinine with the aid of creatininase, creatinase, and sarcosine oxidase. The liberated hydrogen peroxide is measured via a modified Trinder reaction using a colorimetric indicator.(Package insert: Roche Diagnostics, Indianapolis IN, 2004)