Satlook Digital
- NIT
B/W Spectrum Analyzer

The Satlook Digital is a Swedish made Sat-TV Instrument and Spectrum-Analyzer. The unit is made for exact alignment and adjustment of Satellite Dishes. It is intended for professional use when high accuracy and precise information are needed.

It's easily operated without a lot of unnecessary buttons and knobs. The basic functions are easy to get a hold on and takes only minutes to learn. This instrument is provided with a 4.5" B/W-monitor which either shows normal Sat-TV channel, the frequency spectrum 920-2150 MHz (or parts thereof) alt. Digital data. Menus/help-displays are shown on the LCD-screen (64x128) beside the monitor.

Analog Sat-TV channels can be tuned in and viewed upon (Multistandard Video PAL, NTSC and SECAM). Audio frequencies between 5.5-8.5 MHz can be listened to. The Spectrum can be extended (zoom in) for correct adjustment for the polarization ("cross-polarization"). The instrument has very high resolution and accuracy. It presents measured data _+2 dB (at around 20 C.)

SATLOOK Digital presents, under one of the sub menus. digital information like BER (bit error rate), constellation diagram (QPSK) and S/N (Signal/Noise Ratio). Satlook Digital can be handle a lot of memory-positions and both spectrum-pictures and certain frequencies can be saved (both analog and digital transponders).

The polarization of the LNB is switchable 13V/18V and Hi-Lo Band with 22 KHz-tone. The DISEQC-function controls all DISEQC-accessories (like switches, LNB's and actuators). The power of the instrument is supplies by a built in and rechargeable battery. Even though the instrument has lot of functions it is still very light and flexible.

Specifications:

Input Frequency: 920-2150MHz (easily reduced to 250MHz max zoom in)

Frequency Display: L.O. can be setup. Yes, IF default. All standard LNB

Memory: 100 spectrum pictures can be stored with name. Stored spectrum can be mixed for easy identification of satellite.-"Max hold" function.

Favorite channels: Both analogue and digital frequencies can be stored (with name). Simultaneous measuring of up to 10 channels at the same time. Each channel with individual setup of 13/18v and 22 kHz on/off.

PC-connection: Yes, RS232-output

Power out: Yes, 13-18V

22 kHz tone: Yes, on/off.

DiSEqC: Yes, all 1.0 and 1.1. Also Tone burst on/off.

DiSEqC actuator: Built in positioner for DiSEqC 1.2, SatScan and SatSelect.