PG-350E Portable Gas Analyzer

Обзор

The PG-350E achieves measurement performance equal to laboratory equipment in a highly portable package. As part of the PG-300 series, the PG-350E is a compact stack gas analyzer that can simultaneously measure up to five separate gas components. The approximately 20% lighter body from the previous PG-250-model is optionally protected by rugged side guards, which prevent the analyzer from shocks and damages. PG-350E covers NOX, SO2, CO, CO2 and O2 - measurement and is operating according to: DIN EN 15267 - 3, DIN EN 14181 andapproved as Standard Reference Method (SRM) for:CO (DIN EN 15058)O2 (DIN EN 14789)NOX (DIN EN 14792)

The field concious design supports whatever the measurement scene is. An optional electronic cooler unit is available for long-term measurement under tough environments such as gas turbines, boilers, and incinerators facilities.

Cross-flow modulation method requires no optical adjustments as dual optical path measurement does, because sample gas and reference gas flow into a single measurement cell switching one by one. Since clean air is fed into the sample cell in between each batch of sample gas, the cell remains clean which reduces the span drift and keeps long-time stability.

Ease of operation ensured by high visibility color LCD touch screen. The useful functions such as screen capture, trend graph, or operation guide, etc. can be executed on the screen.

Reducing warm-up time in half, the PG-350E is ready to measure within 30 minutes. Moreover, the timer function sets up an automated warming-up according to Your time settings.

95% efficiency of NOX-convertor enables more precision measurement.

Energy efficiency rises up to 35% by reduced power consumption compared to our previous models.

A SDTM memory card slot is installed in the front panel and quick data saving is available. Ethernet interface for connection to a LAN environment enables real-time data import over the network.