Each unit comes with an integral centimeter ruler on the front lip of the tabletop, magnetic instrument bar, and towel dispenser

Standard Type 304 or optional Type 316 stainless-steel construction

Choosing the Exhaust System to Fit Your Lab:

Recirculating Exhaust: For existing labs or labs without facility exhaust, it may be necessary to filter and recirculate the air. Specify a unit with an integral blower and filters.

External Exhaust: For distances that do not exceed 20ft. (6.1m), you may choose a unit with integral blower and filters that may be connected to your facilityÃ¢â‚¬â„¢s exhaust duct (Facility duct required by others; remote blower required by others for distances exceeding 20ft.).

Remote Exhaust: If your facility can supply an exhaust duct and remote blower system that provides an airflow of 400cfm (0.189 cu. m/s), a ducted unit will best suit your needs (Facility duct and remote blower by others; no filters).

Validated Exhaust Specifications:

Recommended Exhaust Airflow Volume:

Recirculating: N/A

External: N/A

Remote: 400cfm (0.189 cu. m/s)

Pressure Loss through Unit:

Recirculating: N/A

External: N/A

Remote: 1.15 in. w.g. (2.87 mb)

Average Capture Velocity at Vented Area:

Recirculating: 119-121fpm (0.604-0.615m/s)

External: 122-123fpm (0.620-0.625m/s)

Remote: 148fpm (0.752m/s)

Average Capture Velocity at Front of Unit:

Recirculating: 10fpm (0.051m/s)

External: 10-13fpm (0.051-0.066m/s)

Remote: 15fpm (0.076m/s)

Filters

Recirculating Exhaust: Use of filters is required

External Exhaust: Use of filters if required for application

Remote Exhaust: No filters are required

Replacement Filter Set

Includes two formaldehyde-neutralizing (potassium permanganate) filters and two fine-particle filters (No.100619)

Includes:

Work surface height of 36 in. (91.4cm)

Louvered backdraft ventilation

Sink

Cold/hot water faucet with manual control

Disposal, 1/2hp with manual control

Hand spray combined with cold/hot water faucet

Two duplex GFCI electrical receptacles (110-120V units)

Two specified point-of-use electrical receptacles (220-240V units; see table at the end of this section)

Centimeter ruler on front lip of tabletop

Fixed lower shelf

Fixed upper shelf

Fluorescent light

Type 304 stainless-steel construction

Programmable microprocessor for date, time, and filter

Magnetic instrument bar (instruments not included)

Towel dispenser for C-fold towels

C-fold towels

Electrical Specifications:

110-120V and 220-240V models require facility receptacles with 15 amp circuits for each power cord

Related Products

Product Content Feedback

We continue to work to improve your shopping experience and your feedback regarding this content is very important to us. Please use the form below to provide feedback related to the content on this product.

Product Title

Full Name:

Email Address:*

User Name:

Telephone Number:

What type of issue did you experience with the content on the page?*

Please provide a detailed description of the issue or request*

By clicking Submit, you acknowledge that you may be contacted by Fisher Scientific in regards to the feedback you have provided in this form. We will not share your information for any other purposes. All contact information provided shall also be maintained in accordance with our
Privacy Policy.