Products

Products

SM1105 Medical Galvo Scanner

Release time:2015-12-24 15:51

SM1105 is the latest R&D
of high performance medical beauty galvanometer scanner. The scanner hand piece
designing was adopted the unique vertically into the light structure. It has
the characteristic of small volume, light weight, high speed, easy operate etc.
A full range of security protection function, custom software development which
makes the stability and reliability of the scanner reached the world leading level.
It also has independent intellectual property rights.

Motor Specification

Working Temperature: 0℃-45℃

Linearity: 99.9%

Setting Time: ≤0.35ms

Scale Drift: <40PPM/℃

Zero Drift: <15μRad./℃

Long-term Drift Over 8
Hours: <0.5mRad

RMS Current: 2.0A

Peak Current: 10A

Maximum Scan Angle: ±15°

Storage Temperature: -10 to
+60℃

Resolution: 12μrad

Repeatability: 8μrad

Input Aperture: 5.0-7.0mm

Beam Displacement: 10.7mm

Weight: 13g

Frequency: ≤1000Hz

Servo Drive Board
Specification

Input Voltage: ±15VDC

Interface Signals Analog: ±15V

Analog Signal Input
Resistance: 200KΩ±1%

Position Signal Input
Resistance: 1KΩ±1%

Position Signal Input Scale Factor:
0.33V/°

Position Signal Output Scale
Factor: 0.33V/°

Working Temperature: 0 degrees-45 degrees

Dimension (length*width*height): 75x58x33mm

The Characteristics of SM1105
Scanner:

1. The
scanner hand piece designing was adopted the unique vertically into the light
structure, only weights 300g, the operation is flexible and convenient.

2. Adopted the world’s smallest scanning motors, the photoelectric
sensors is imported from America, only weights 13g, the marking speed is two
times faster than common scanners, the setting time reaches 0.35ms.

3. The Driving system adopted the control method of Germany glavo
scanners, two motors are centralized controlled by only one drive board, the
performance is stable, reliable and high accuracy. All precision electronic
components are imported from aboard.

4. The control board are designed according to the industrial
computer motherboard standard, comprehensive configuration isolation measures,
with good electromagnetic compatibility, stronger anti-interference ability,
can effectively prevent the motherboard damage caused by external input error
signals or power supply.

5. With powerful software function, convenient parameter settings,
a variety of scanning shapes and scanning mode for options, the scanning shapes
can be two-way adjusted.

6. This model has the self-inspection and protection signal output
function. It will output the protection signal within 0.1ms when the scanner
working abnormal, to shut down the laser power supply thereby effectively to
prevent medicals accidents.