HOBO Bluetooth Low Energy Carbon Dioxide - Temp - RH Data Logger

This data logger requires either the HOBOmobile App and a compatible mobile device or Onset's HOBOware software. The system requirements for the HOBOmobile app can be found at the bottom of our HOBOmobile software page.

Overview

Onset’s HOBO MX1102 CO2 logger makes it more convenient than ever to measure and record CO2 in buildings and other non-condensing environments. It measures CO2 from 0 – 5,000 parts per million (ppm) – and our free HOBOmobile app* allows you to access data right from your mobile phone or tablet within a 100-foot range.

The MX1102 also features a USB port so it can be used with a computer running HOBOware graphing and analysis software.

What measurements does this data logger support?

Onset’s HOBO MX1102 CO2 logger makes it more convenient than ever to measure and record CO2 in buildings and other non-condensing environments. It measures CO2 from 0 – 5,000 parts per million (ppm) – and our free HOBOmobile app* allows you to access data right from your mobile phone or tablet within a 100-foot range.

The MX1102 also features a USB port so it can be used with a computer running HOBOware graphing and analysis software.

6 months, typical with logging and sampling intervals of 5 minutes or slower; 6 months or less with logging and sampling intervals faster than 5 minutes while logging CO2. Entering burst logging mode will impact battery life. With HOBOmobile use, battery life can be reduced by remaining connected, excessive readouts, checking of Full Status Details, audible alarms, and paging. Visual/audible alarms and other events can have a marginal impact on battery life.

Memory

128 KB (84,650 measurements, maximum)

Download Type

USB 2.0 interface or via Bluetooth Smart

Full Memory Download Time

20 seconds via USB; approximately 60 seconds via Bluetooth Smart, may take longer the further the device is from the logger

LCD

LCD is visible from 0° to 50°C (32° to 122°F); the LCD may react slowly or go blank in temperatures outside this range

Size

7.62 x 12.95 x 4.78 cm (3.0 x 5.1 x 1.88 inches)

Weight

267.4 g (9.43 oz)

Environmental Rating

IP50

The CE Marking identifies this product as complying with all relevant directives in the European Union (EU).

The following are associated resources for this product. In general users of this product can gain insight into its possible use by utilizing the resources below. Our user forum can also be a valuable resource for questions, use cases and general feedback. Share with us today, here!

Ideal for a range of indoor monitoring applications, this self-contained wireless data logger lets you access temperature data anytime from your mobile device over a 100-foot range. And, it requires no dedicated equipment beyond a mobile device for configuring the logger or reading out data.

The self-contained wireless data logger, which works with Onset’s free HOBOmobile® app for logger setup and data management, enables you to access data anytime from mobile device for iOS or Android over a 100-foot range, and it requires no dedicated equipment beyond a mobile device for configuring the logger or reading out data.

As Onset’s highest-accuracy data logger, it provides twice the accuracy of previous models, a deployment-friendly LCD, and flexible support up to four external sensors for measuring temperature, current, CO2, voltage, and more.

The data logger is ideal for energy audits, building commissioning studies, equipment scheduling, and more.

6 months, typical with logging and sampling intervals of 5 minutes or slower; 6 months or less with logging and sampling intervals faster than 5 minutes while logging CO2. Entering burst logging mode will impact battery life. With HOBOmobile use, battery life can be reduced by remaining connected, excessive readouts, checking of Full Status Details, audible alarms, and paging. Visual/audible alarms and other events can have a marginal impact on battery life.

Memory

128 KB (84,650 measurements, maximum)

Download Type

USB 2.0 interface or via Bluetooth Smart

Full Memory Download Time

20 seconds via USB; approximately 60 seconds via Bluetooth Smart, may take longer the further the device is from the logger

LCD

LCD is visible from 0° to 50°C (32° to 122°F); the LCD may react slowly or go blank in temperatures outside this range

Size

7.62 x 12.95 x 4.78 cm (3.0 x 5.1 x 1.88 inches)

Weight

267.4 g (9.43 oz)

Environmental Rating

IP50

The CE Marking identifies this product as complying with all relevant directives in the European Union (EU).