Supply, installation and testing of medical waste incinerator with a capacity of 75 kg / hr gas (LPG) complete with all accessories and consequently with the chimney in accordance with the technical specifications and conditions of public health
make a room for the incinerator that accommodate three incinerators as in item 3.1.1 and attached to the incinerator room waste store that accommodate waste for a period of seven days and all the constructions of reinforced concrete with preparingg incinerator building and store with all the work of wiring, electrical and lighting systems, water supply and sanitation
supply incinerator building and the waste warehouse with a fire extinguish sprinkler system depends on the source of the water from network of teh general site , as well as a fire alarm system contains sensors for heat, smoke and this system is linked with the hospital fire alarm systems
Supply, installation and testing of the ventilation system of the incinerator room and the warehouse with fans ROOF TOP EXHAUET FANS providing them with filters with high efficiency (HEPA FILTER ride to pull fans with taking the air change rate of 5 times per hour with streams and air required grilles
Supply, installation, testing separate conditioning unit for waste repository DUCTED SPLIT UNITES and capacity sufficient to maintain the internal temperature at 14 degrees Celsius in summer and winter
Supply, installation, testing of the cooling store of insulated prefabricated walls inside the waste room with a capacity of 35 cubic meters to save the medical waste at a temperature of zero Celsius till burning including the cooling unit and all accessories
HAZARDOUS wood
play wood
petroleum residual waste
hydraulic hose system
calcium hydroxide bags
bags or wooden boxes containing cyanide
medical waste
filters
pharmaceuticals
NO HAZARDOUS sawdust
plastic
plastic and textil awnings
wood
plastic tubs
rubber
sanitary waste
EPPs
plastic bags
construction residual
contaminated rags or towels
plastic or packing of food
polyestirene dishes, glasses, spoon, fork, etc.Capacity incinerator medical waste to high combustion temperature
With a smokeless device to protect the environment and with two combustion chambers equipped with gas burners or oil
-incinerator to burn large volume pieces of 70 cm
deviation: Automatic Control System
-destructive capacity from 20 to 30 kgs
First-room / Temperature = 900 ° C combustion
2nd bedroom / Temperature = 1200 ° c afterburner