Helpful "How-to's"

Articles

Product Group Specifications

Instruction Manual

Print

MW11 Handy Photometer

The revolutionary new range of Milwaukee handy photometers are compact and easy to use, have a modern design, and measure Free Chlorine (MW10); Total Chlorine (MW11); Phosphate (MW12); Iodine (MW13) and Iron (MW14). Their main advantages are the small size, and the real simplicity when doing measurements on the spot. Within a few minutes you can get the required result on the LCD and the instrument directly displays the concentration of the specific component in ppm. The meter automatically turns off after 2 minutes in order to save the battery.They work exactly the same way as the other Milwaukee photometers:

When specific reagents are added, the water sample takes on a degree of coloration that is proportional to the concentration of the parameter being measured. The photometer measures this coloration. When a light beam passes through the coloured sample, energy with a specific wavelength is absorbed by the test substance. The photometer determines the coloration of the sample by measuring the transmission or absorption of light of this wavelength. The photometer then uses a microprocessor to calculate the concentration and displays the result.

The field of application is really wide including agriculture, drinking water analyses, aquariums, swimming pools or fish farms. As a result of this versatility and the affordable price it is undoubtedly worth using these products.

SPECIFICATIONS

Range

0.00 to 3.50 ppm

Resolution

0.01 ppm

Accuracy

±0.03 ppm ±3% of reading @ 25 °C

Typical EMC Dev.

±0.01 ppm

Light Source

Light Emitting Diode @ 525 nm

Light Detector

Silicon Photocell

Method

Adaptation of USEPA method 330.5. The reaction between free chlorine and the DPD reagent causes a pink tint in the sample.

Environment

0 to 50°C (32 to 122 °F) max. 95% RH non-condensing

Battery Type

1 x 1.5V AAA

Auto-shut off

After 2 minutes of non-use.

Dimensions

81.5 x 61 x 37.5 mm (3.2 x 2.4 x 1.5”)

Weight

64 g (2.25 oz.)

ACCESSORIES

2720216

Total Chlorine powder reagent, 25 pcs.

Mi0011

10 mL small glass cuvets (2 pcs)

Mi0012

Caps for Mi0011 glass cuvets (2 pcs)

Mi0013

Stoppers for Mi0011 glass cuvets (2 pcs)

ORDERING INFORMATION

All brand new handy photometers are supplied in carton box including 2 cuvets, reagents and instruction manual.