INMEL 52

Calibrator INMEL 52 integrate generation and measurement function of electric signals, used in automatic industry.
The main features of this devices is high accuracy and resolution of generating or measuring signals, friendly operation and large display.

Checking and calibration:

measuring transducers,

temperature meters,

measuring and control lines.

BASIC PARAMETERS

The measurement and generation

tabela1

Range

Resolution Measurement/simulation

Accuracy

4W ± Ω

Accuracy

± Ω

Measuring Current

mA

0.00Ω ~ 400.00Ω

0.01 / 0.1 Ω

0.1

0.15

0.1

0.1 ~ 0.5

0.5 ~ 3.0

400.0Ω ~ 1500.0Ω

0.1 / 0.1 Ω

0.5

0.5

0.05 ~ 0.8

1500.0Ω ~ 3200.0Ω

0.1 / 0.1 Ω

1

2

1

0.05 ~ 0.4

Admit excitation current only apply on simulate mode. The admit excitation current could be marked on the OHM meter or RTD meter which was connected to the calibrator. admit excitation current: 0.2mA. MAX input voltage: 30V. 2W/3W/4W – two wire/tree wire/four wire the resistance measuring set

Measure (input)/Simulate (output) RTD specification

The resolution of measurement and simulationi 0.1°C

tabela1

Mode

Range

Accuracy °C

Measuring current mA

Measurement 4W

Measurement
2W/3W

Simulation

Pt10 385

-200~800°C / -328~1472°F

3.5

4.5

1.5

0.1~3.0

Pt50 385

-200~800°C / -328~1472°F

0.7

1.0

0.7

0.1~3.0

Pt100 385

-200~800°C / -328 ~ 1472°F

0.33

0.5

0.33

0.1~3.0

Pt200 385

-200~250°C / -328~482°F

0.2

0.3

0.2

0.1~3.0

250~630°C / 482~1166°F

0.8

1.6

0.8

Pt500 385

-200~500°C / -328~932°F

0.3

0.6

0.3

0.05~0.8

500~630°C / 932~1166°F

0.4

0.9

0.4

Pt1000 385

-200~100°C /-328~212°F

0.2

0.4

0.2

0.05~0.8

100~630°C /212~1166°F

0.2

0.5

0.2

Pt100 JIS

-200~630°C /-328~1166°F

0.3

0.5

0.3

0.1~3.0

Admit excitation current only apply on simulate mode.
The admit excitation current could be marked on the OHM meter or RTD meter which was connected to the calibrator.
admit excitation current: 0.2mA.
MAX input voltage: 30V.
2W/3W/4W – two wire/tree wire/four wire the resistance measuring set

General Specifications:

Maximum voltage applied between any jack and earth ground or between any tow jack 30V