Hooked up the he leak detector and he leak checked chamber
(top and bottom) picked up a very small leak at the baratron
before the shutoff valve (1/2 scale) tighten fitting and he leak checked
again (good). Pumped down chamber for 10mins and ran manual
leak check , l/r is 9mt per mins.