A Peltier deivice (thermoelectric module, thermoelectric device) is a plate type element, inside which P-type semiconductors and N-Type semiconductors are located alter-nately. If direct current is supplied to the Peltier device , heat is transferred inside the device , and one face generates heat and increases temperature while the other face sucked heat and decreases temperature . Therefore, changing the direction of the current supplied to the Peltier device and achieve heating and cooling operation . This method has a fast response and can shift quickly between heating and cooling, so temperature can be controlled very precisely .

Prevention of electric shock and fire

​Do not operate the switch with wet hands . Also , do not operate the Thermoelectric Bath with water or fluid left on it .

Action in the case of error

​If any error such as abnormal sounds , smoke , or bad smell occurs , cut off the power at once , and stop supplying facility water . Please contact Adcol or a sales distributor to repaire the Themoelectric Bath .

​Regular inspection

The inspection must be done by an operator who has sufficient knowledge and experience.
​a , Check of displayed contents

b , Check of temperature , vibration and abnormal sounds in the body of the Thermoelectric Bath .

c , Check of the voltage and current of the power supply system .

d , Check for leakage and contamination of the recirculating fluid and instrusion of foreign objects to it .
​e , Check radiation air flow condition and temperature .
​f , Check for leakage , quality change , flow rate and temperature of facility water .