Downloads

Further Information

CH-5 Chiller and PC-5 Pump

The CH-5/PC-5 Chiller/Pump Unit is designed for use in conjunction with the TECAL 140 H & S Calibrators to achieve calibration temperatures down to -40°C. This unit provides a chilled water supply which is pumped through the Heatsinks on the Tecal 140 H & S units allowing the Tecal to reach -40°C.

The unit is designed to run on distilled or deionised water.

The CH-5/PC-5 can also be used to pass chilled water through the cooling coil on the Liquid Calibration Baths (LCB) and Cooling Probe (FDB00CP) giving rapid cooling or a minimum temperature of +10°C.

Technical Specification

Specifications

CH-5 and PC-5

Temperature range

4°C to 15°C

Cooling capacity

400W

Temperature Control

Thermostat onoff

Electrical control

Manual

Bath Capacity

5 litres

Pump capacity

20 litresminute

Pump Head height

6 metres with water

Bath Dimensions L x W x H (mm)

300 x 140 x 150

Overall Dimensions L x W x H (mm)

430 x 235 x 524

Weight

26.5kg

Ordering Information

Product Code

Description

FCH5D

CH-5 chiller, 230V 50Hz

FP52D

PC-5 pump, 230V 50Hz

FCH5P

CH-5 chiller,120V 60Hz

FP52P

PC-5 pump, 120V 60Hz

Please note:

Chillers & pumps used to be ordered as a kit (FCH5H 240V, FCH5P 120V) but are now ordered separately (as above). Price is unchanged