Full Stainless Steel Steam Sterilization Model:HA-BD Sterilizing Volume:150L,200L,280L Features: 1. The steam can't go through to chamber when the door haven't close, and the pilot off inform. You can processing dry operate. 2. When the pressure inner over 0, 027MPa. The door be locked and can't be open. 3. If you not choose the sterile time and start the machine, buzzer sound. 4. If you mistake open the door in the course of start, the jacket value auto off and open the chamber value. 5. Lack water the pilot on, heating and sterile pilot off, the buzzer sound and auto release steam. 6. Overload and leakage protection, when leakage or over current occur, it auto off the power. 7. Temperature controlled not work and the inner pressure over 0.165MPa, the safety value auto open.

Cylindrical Sterilizer Machine Model:HA-BC Sterilizing Volume:150L,200L,280L,400L,500L Features: 1.Sterilizing course automatic sterilization controlled,easy to perate. 2.With drying function,suitabledressing drying. 3.With over-temperature,over-pressure auto-protect device. 4.The door opening mechanism can't be perated until the pressure in the chanber is reduced to 0.027Mpa.It can't be start on if the door doesn't close well. 5.The safety valve will be open automatically when the inner pressure over0.24Mpa and the steam be exhausting to the water tank. 6.The power will be cut-off automatically,cut-off the water and alarm when the machine lacing water. 7.The chamber of the sterilizer is made of stainless steel.

Steam Sterilization Model:HA-BA Steam Sterilization Volume:150L,200L,280L Features: 1. The steam can't go through to chamber when the door haven't close, and the pilot off inform. You can processing dry operat. 2. When the pressure inner over 0, 027MPa. The door be locked and can't be open. 3. If you not choose the sterile time and star the machine, buzzer sound. 4. If you mistake open the door in the course of start, the jacker value auto off and open the chamber value. 5. Lack water the pilot on, heating and sterile pilot off, the buzzer sound and auto release steam. 6. Overload and leakage protection, when leakage or over current occur, it auto off the power. 7. Temperature controlled not work and the inner pressure over 0.165MPa, the safety value auto open.

Steam Pressure Sterilizer Model:HA-BB Steam Pressure Sterilizer Volume:150L,200L,280L Features: 1. Sterilizing course automatic sterilization controlled, easy to operate. 2. Autoclave horizontal With drying function, suitable dressing drying. 3. Horizontal Cylindrical Pressure Steam Sterilizer With over-temperature, over-pressure auto-protect device. 4. The door opening mechanism cannot be operated until the pressure in the chamber isreduced to 0.027 Mpa. It can't be start on if the door doesn't close well. 5. The safety valve will be open automatically when the inner pressure over 0.24 MPa andthe steam be exhausting to the water link. 6. The power will be cut-off automatically, cut-off the water and alarm when the machinelacking water. 7. Fully stainless steel structure. 8. With printer.

LCD Pulse Vacuum Class B Autoclave Model: LX-BV Sterilizing Volume: 35L, 50L, 75L, 100L Features: 1. Micro-computer control system, Sterilization temperature, time dynamic LCD displayed. Fault automatic displayed and alarm. 2. Automatic control system for water feeding, temperature rise, sterilizing, exhausting air, dryness. 3. Quick door open structure, safety and interlock device, horizontal sliding door. 4. Glass reinforced plastics cover to prevents the operator to be injured. 5. Adjustable temperature from 105-134'c 6. Have protector for exceed temperature, exceed pressure and electric leakage, Safety valve of auto release against over-pressure. It opens when the designed pressure is surpassed. 7. Material made by SUS304 stainless steel. 8. Built-in water tank. Steam and water inner cycle system, no external water source, not exhaust steam outside, environment is clean and dryness. 9. Timer range: 0-999min 10. Safety and interlock device: The door can't be opened until there is no pressure in the chamber and power for heating can't be switched on until the door is firmly closed. 11. Alarm for sterilizing and fault state, The buzzer alarms when a cycle of sterilization is completed. 12. Silicone rubble for seal.

Steam sterilizers Model: HA-BA150L, HA-BA200L,HA-BA280L Steam sterilizers Volume: 150 liters, 200 liters, 280 liters USE: Large horizontal steam sterilizer which is designed for the Lab, hospital, clinic, etc. and suitable for sterilizing of surgical, dental instruments, glassware, culture medium and biological dressing, food and goods, etc. Steam sterilizers Characteristics: 1. Sterilizing course automatic sterilization controlled, easy to operate. 2. With drying function, suitable dressing drying. 3. With over-temperature, over-pressureauto-protect device. 4. The door opening mechanism cannot be operated until the pressure in the chamber is reduced to 0.027 Mpa. It can't be start on if the door doesn't close well. 5. The safety valve will be open automatically when the inner pressure over 0.24 MPa and the steam be exhausting to the water link. 6. The power will be cut-off automatically, cut-off the water and alarm when the machine lacking water. 7. The chamber is made of SUS304. 8. With printer.

Table Top Steam Sterilizer Application Range: The sterilizer is a table type in safety and auto-controlled, which is designed for the unit of medical health, pharmacy, science and technology etc. It is suitable for sterilizing of surgical, dental and eye instruments, glassware medicine, culture medium and biological dressing, food and other goods. Technical Features: 1.Indicator light indicates working state; 2.For 4-6 minutes rapidly sterilizing; 3.Sterilizing temperature and time can be preset; 4.Steam-water inner circulation system, no steam discharge and the environment for sterilizing will be clean and dry; 5.Over temperature & over pressure auto-protection; 6.Safe protection of water lacking; 7.Exhaust the cool air automatically; 8.Door safety lock system; 9.With three stainless steel sterilizing plates; 10.Fully stainless steel structure; 11.Automatically shut off with beep reminding after sterilization;

Table Top Steam Sterilizer Application Range: The sterilizer is a table type in safety and auto-controlled, which is designed for the unit of medical health, pharmacy, science and technology etc. It is suitable for sterilizing of surgical, dental and eye instruments, glassware medicine, culture medium and biological dressing, food and other goods. Technical Features: 1.Indicator light indicates working state; 2.For 4-6 minutes rapidly sterilizing; 3.Sterilizing temperature and time can be preset; 4.Steam-water inner circulation system, no steam discharge and the environment for sterilizing will be clean and dry; 5.Over temperature & over pressure auto-protection; 6.Safe protection of water lacking; 7.Exhaust the cool air automatically; 8.Door safety lock system; 9.With three stainless steel sterilizing plates; 10.Fully stainless steel structure; 11.Automatically shut off with beep reminding after sterilization;

Portable Pressure Steam Sterilizer Application Range: The product uses saturated steam to rapidly and efficiently sterilize articles. It can be used in hospitals, public health centers, health stations, clinics in factories and mines, scientific research institutes, etc to sterilize medical equipment, surgical dressings, glass utensils, solutions, substrata. It can also be used as cooker in plateau areas and be used to make high quality drinking water in industrial and mining enterprises. Technical Features: 1.Fully stainless steel structure; 2.Double scale indication pressure gauge; 3.Electric heated; 4.Easy to operate, safe and reliable;