Teledyne Leeman Labs Hydra II AF Gold

Share:

Hydra II AF Gold

Hydra IIAF Gold is a fully automated cold vapor atomic fluorescence Hg analyzer designed to meet the requirements of all current low-level Hg analysis methods including: EPA methods 1631 and 245.7 as well as EN methods 13506 and 12338. This system delivers a detection limit well below 0.05 ppt when gold amalgamation is used and 0.1 ppt without it. It also provides a dynamic range that extends to 250 ppb and productivity tools that are unmatched by other CVAF instruments. Tools like advanced high concentration protection and an autosampler that holds up to 270 samples.

Productivity Run more samples with Hydra's high capacity autosampler -Up to 180 samples when amalgamation is used and 270 samples when amalgamation is not needed -Huge cups for continuing QCs (e.g. CCB and CCV) Protect yourself from those samples with unexpectedly high concentrations of Hg -Hydra II's advanced over-range protection system will prevent productivity loses by preventing system contamination (and reruns) from samples that contain unexpectedly high concentrations of Hg Run your samples faster for improved productivity and lower cost of analysis -Analysis times of about 3 minutes per sample with gold amalgamation and under 2 minutes per sample without it

Photos

Hydra IIAF Gold

Manuals and Resources

Note: We appreciate the support of individuals and companies making their documents and resources available to our community. If you find a document or resource that does not belong please let us know by emailing info@labwrench.com