After December 2018, we will be moving
elements from the doctorportal newsletter to MJA InSight
newsletter and rebranding it to Insight+. If you’d like to
continue to receive a newsletter covering the latest on research
and perspectives in the medical industry, please subscribe to the Insight+
newsletter here.

As of January 2019, we will no longer be
sending out the doctorportal email newsletter. The final issue
of this newsletter will be distributed on 13 December 2018.
Articles from this issue will be available to view online until
31 December 2018.

Manufacturing a human heel in titanium via 3D printing

February 24, 2015

Med J Aust 2015; 202 (3): 118.

doi:10.5694/mja15.00052

Author: John E Barnes

3D printing continues to find new uses as more people get familiar with the capabilities of this relatively new manufacturing tool. Professor Peter Choong at Melbourne’s St Vincent’s Hospital proved the adage “chance favours the prepared”. His patient faced losing a foot due to a cancerous heel, so he came up with the idea of a 3D-printed alternative. He engaged medical device company Anatomics to design the implant. The CSIRO’s Lab 22 was contacted due to its capability in 3D printing and, within a week, a foot-saving replacement was used in surgery — a world first.

Using engineering principles and computer-assisted design, Anatomics manipulated the computed tomography data of the patient’s healthy heel to design a device to include muscle attachment points to support the new heel once implanted, internal structures to handle the forces of an adult heel, and porosity to reduce implant weight and facilitate tissue integration.

The CSIRO manufactured the design in its Arcam A1 printer. Using titanium Ti6Al4V powder, the printer produced the device overnight. The CSIRO and Anatomics cleaned and prepared the device for surgery. The collaborative team made several prototypes, refining the designs for form, fit and function. The final version was implanted about 24 hours after manufacturing.

There are a number of learning outcomes from this endeavour. The patient was able to keep his foot, which otherwise…