Related products

Main products

The Metrohm Instant Raman Analyzer (MIRA) DS is a ruggedized, high-performance, handheld Raman spectrometer used for rapid, nondestructive determination of illicit materials, e.g. drugs, explosives, starting materials, and hazardous agents. Despite the small size of the instrument, Mira DS is extremely rugged and features a high-efficiency spectrograph equipped with our unique Orbital-Raster-Scan (ORS) technology.The basic package includes the Library of illicit materials, calibration standard, universal attachment for analyses in bottles or bags or for direct analysis, and the right-angle attachment, ideal for running samples on a surface and/or in a bag. Class 3B operation.

Taking high-precision Metrohm ion chromatography to an even higher level of autonomy, the Process IC can connect up to 20 sample streams and provide reliable multicomponent results for superior process control.

Promotions

The all-new OMNIS platform makes your work in the lab faster, safer, and easier: Up to four parallel titrations, automated titration and sample processing, contact-free reagent exchange, and an intuitive, easy-to-use titration software.

Metrosep C 3 - 250/4.0

Order number: 6.1010.430

The innovative substrate on a polyvinyl alcohol base increases selectivity for monovalent and divalent cations significantly. A characteristic of this is the number of "theoretical plates per meter". On the Metrosep C 3 - 250/4.0, for example, 42,000 plates are achieved for sodium, 51,000 for ammonium, and 31,000 for barium with its delayed elution. The peak forms on this cation column are highly symmetrical.

The selectivity of the Metrosep C 3 - 250/4.0 also permits the separation of transition metals. Because Metrohm ion chromatographs generally determine the cations without chemical suppression, the transition metals can be analyzed on the Metrosep C 3 - 250/4.0 together with the alkaline and earth alkaline metals.

6.1010.450

Metrosep C 3 Guard/4.0

The Metrosep C 3 S-Guard/4.0 contains the C-3 column material and is used to protect Metrosep C-3 cation columns. Particles and contaminations are reliably retained, allowing the lifetime of the analytical separation column to be prolonged considerably. The Metrosep C 3 Guard/4.0 also functions according to the "On Column Guard System" and is screwed directly onto the separation column with nearly no dead volume.

6.1010.460

Metrosep C 3 S-Guard/4.0

The Metrosep C 3 S-Guard/4.0 contains the C-3 column material and is used to protect Metrosep C-3 cation columns. Particles and contaminations are reliably retained, allowing the lifetime of the analytical separation column to be prolonged considerably.