C.A 1510

Description

In
indoor environments, dense occupation and insufficient air renewal lead to high
CO2 concentrations. The level of CO2 is therefore an excellent indicator of air
quality, alongside pollution analysis performed by inspection organizations and
laboratories, as well as showing the effectiveness of the air renewal systems
for environmental engineering professionals.

In addition to its portable function, the simple, user-friendly C.A 1510 is
capable of recording the parameters measured and monitoring air quality
criteria based either on the level of CO2 or on a combination of the three
physical quantities measured.

CO2: 0
to 5,000 ppm

Accuracy: ± (50 ppm + 3% of the
value measured)

Temperature: -10 °C to +60 °C

Accuracy: ± 0.5 °C

Humidity: 5 to 95 %RH

Accuracy: ± 2%RH

Recording: 1 million values
stored on the product in the form of several measurement campaigns

USB interface (product
recognized as a USB key) or Bluetooth for wireless use

Software supplied
as standard and 1510 application available with Android

Locking of the instrument
possible during recording (display & keyboard)

Data Logger Transfer

Data Logger Transfer

Data Logger Transfer software allows you to pilot your loggers.

With the software Data Logger Transfer, you will be able to :

Setup the instruments,Download and display the trends,Display real time values,Display data in curves or table,Export automatic reports in Word (.docx format), Export data into Dataview (.dvb files) or into Excel

Compatible with Windows 7, 8, 8.1 and 10.

Attention : DataView is not provided with the software Data Logger Transfer