- The critical working pressure gauges easy for monitoring the working pressure for both interior and exterior chambers.

- Sterilization timer starts working when the set temperature rises to the set point.

- Temperature is adjustable from 121 to 134 at 1.2kg/cm2 and 2.2kg/cm2.

OPTIONAL FEATURES:

- Automatic Water filing – when the water level in house of steam generator arrives the minimal level below, the built-in system will fill the water to the safe and working water level automatically.

- Automatic Drying & Exhaust – Set drying time and when the sterilization cycle is complete, the drying function will process automatically.

- Safety door lock activated by pressure – in order to prevent the personal injury by negligence accidentally, the door can’t be opened when pressure in interior chamber exceeds 0.2kgf/cm2

- Pre-vacuum and drying vacuum – pre-vacuum to speed up the sterilization cycle. Drying Vacuum can exhaust the residue steam and water in the chamber, which can reach the requirement for Bowie & Dick Vacuum testing at 134°C for minutes continuously.