Features • 1. The handpiece is cast from titanium alloy to ensure sound mechanic function and durability. • 2. Assembling with excellent piezo-ceramic, the handpiece has high and long lasting power and generates little heat. • 3. Detachable handpiece can be autoclaved under the high temperature of 135 and pressure of 0.22Mpa avoiding cross-infection effectively.• 4. The material of handpiece made in America is appointed material of aero plastic components in Europe and America, which is FDA and NSF certificated.• 5. The cable of the Woodpecker UDS-K ultrasonic scaler is made from silica gel tube, soft and durable.

Item has been added to your cart.

Features

2. Automatic frequency tracking ensures that the machine always works on the best frequency and performs more steadily.

3. Digitally controlled, easy operation and more efficient for scaling.

4. The handpiece is detachable and can be autoclaved to the high temperature of 135℃ and pressure of 0.22MPa

5. The scaler is composed of electrocircuit, water way and ultrasonic transducer

6. UDS-K is used for the dental calculus elimination

Specifications

Power source input: 220 - 240V, 50Hz/60Hz, 150mA

Main unit input: 24V - 1.3A

Output primary tip vibration excursion: ≤100um

Output half-excursion force: ＜2N

Output tip vibration frequency: 28kHz ± 3kHz

Output power: 3W to 20W

Main unit fuse: T 1.6AL 250V

Power source fuse: T0.5AL 250V

Water pressure: 0.1bar to 5bar (0.01MPa to 0.5MPa)

Weight of main unit: 0.75kg

Weight of power source: 1.2kg

Dimension: 170mm x 111mm x 65mm

Operating mode: Continuous operation

Type of protection against electric shock: Class II

Degree of protection against electric shock: Type BF applied part

Applied part of the equipment: handpiece and tip

Degree of protection against harmful ingress of water: Ordinary equipment (IPX0), Protection degree against water (used on the foot switch): IPX1

Degree of safety of application in the presence of a Flammable Anesthetic Mixture with air or Oxygen or Nitrous Oxide: Equipment not suitable for being used in the presence of a flammable anesthetic mixture with air or with oxygen or nitrous oxide

Net weight: 2.5kg

Gross weight: 2.7kg

Packing size: 380*120*300mm

Storage and Maintence

1. The equipment should be handled carefully and lightly. Be sure that it is far from the vibration, and is installed or kept in a cool, dry and ventilated place

2. Don't store the machine together with the articles that are combustible, poisonous, caustic, or explosive

3. This equipment should be stored in a room where the relativce humidity is ≤80%, atmospheric pressure is 50kPa to 106kPa, and the temperature is -10℃ to +50℃

4. If the machine is not used for a long time, please make it get through to the power and water once per month for five minutes