Оборудование информацию:

The 9310A family forms a range of ideal general-purpose
digital oscilloscopes. They capture single-shot events
at up to 100 MSamples/s, and repetitive signals at
10 GSamples/s. Record lengths up to 1M points provide
high-quality horizontal resolution, and allow fast
digitizing of long-duration events. Memories can be
segmented, for minimum dead time between acquisitions.
Live waveforms may be viewed simultaneously with up
to 3 expansions, showing all of the signal detail.
Expansions are shown as highlights on the main trace.
The LeCroy ProBus™ intelligent probe system allows
automatic sensing of the probe type. For LeCroy's
active FET probes it also provides variable offset at
the probe tip; offset and coupling are controlled
from the scope front panel.
SMART Trigger modes like Glitch, Window and Dropout
allow you to capture precisely the events of interest.
A comprehensive range of signal processing functions,
on live or stored waveforms, allows waveform manipulation
without destroying the underlying data.
The 9310A family features the proven user-interface
of LeCroy's portable scope family. A bright, high-resolution
9" CRT allows optimum waveform viewing under any conditions.
Menus and text are arranged around the graticules -they
never overwrite the waveforms. Each of the main control
functions has a dedicated single knob, to keep the scope’s
performance at your fingertips.
DOS compatible floppy disk and memory card options
store waveforms and test setups, and make transferring
data to a PC easier than ever before. Hardcopies may
be made on GPIB, RS-232-C or Centronics printers or
plotters. An optional internal printer is also available.
Optional packages provide extensive Waveform Processing
including FFT and Enhanced Resolution to 11 bits.