The DT547x is a new CAEN Power Supply Family providing a Single High Voltage channel in a small size Desktop form factor. The unit is powered and controlled by any Laptop/PC USB 2.0 port and this make it a really portable and practical solution for testing and educational needs.

A dedicated LabVIEW based Control Software allows for an easy set and monitor of the High Voltage parameters. The High Voltage output is delivered through SHV connector

Three versions are available spanning from 500 V to 5 kV and from 200 µA to 1 mA, (1 W max) meeting the needs of a wide range of applications. Each unit is provided with Imon Zoom. This feature enhances the Current monitoring, increasing the resolution of a x10 factor
according to the specific model. The HV output Ramp-Up and Ramp-Down rates can be set independently in the range 1÷500 V/s in 1 V/s steps (1÷100 V/s for DT5472).

Channel can be enabled or disabled through the front panel manual switch or via Interlock logic. Overvoltage and Undervoltage warning when the output voltage differs from the programmed value. Overcurrent detection: when a channel attempts to exceed the programmed current limit, it signaled to be in "overcurrent" and enter in a TRIP status. The output voltage is varied to keep the current below the programmed limit for a programmable TRIP time, then the channel is switched off. If TRIP is set to "constant current mode", the channel behaves like a current generator. Hardware VMAX: maximum output voltage can be set via front panel potentiometer. VMAX value can be read out via software.