The OMEGA®
iSeries is a family of
microprocessor-based instruments
offered in three true DIN sizes with
NEMA 4 (IP65) rated front bezels.
All of the instruments share the same
set-up and configuration menu and
method of operation, a tremendous
time saver for integration of a large
system. The iSeries family includes
extremely accurate digital panel
meters "DPi" and single loop PID
controllers "CNi" that are simple to
configure and use, while providing
tremendous versatility and a wealth
of powerful features.

The DPi Series covers a broad
selection of transducer and
transmitter inputs with 2 input
models.

The Universal temperature and
process instrument (DPi models)
handles 10 common types of
thermocouples, multiple RTDs and
several process (DC) voltage and
current ranges. This model also
features built-in excitation, 24 Vdc @
25 mA. With its wide choice of signal
inputs, this model is an excellent
choice for measuring or controlling
temperature with a thermocouple,
RTD, or 4 to 20 mA transmitter.
The strain and process instruments
(DPiS models) measure inputs from
load cells, pressure transducers,
and most any strain gage sensor as
well as process voltage and current
ranges. The DPiS has built-in
5 or 10 Vdc excitation for bridge
transducers, 5 Vdc @ 40 mA or
10 Vdc @ 60 mA (any excitation
voltage between 5 and 24 Vdc is
available by special order). This
DPiS model supports 4- and 6-wire
bridge communications, ratiometric
measurements. The DPiS features
fast and easy "in process" calibration/
scaling of the signal inputs to any
engineering units. This model also
features 10-point linearization which
allows the user to linearize the signal
input from extremely nonlinear
transducers of all kinds.
Programmable
Color Display
The DPi Series are 1/8, 1/16 and 1/32 DIN
digital panel meter featuring the big
iSeries color-changing display. The
digits are twice the size of typical 1/8 DIN panel meters. The iSeries
meters feature the only LED displays
that can be programmed to change
color between
GREEN, AMBER, and RED. .
Embedded internet and serial
communications featuring optional
"embedded Internet" (specify "-EIT"
option) the iSeries are the first
instruments of their kind that connect
directly to an Ethernet network and
transmit data in standard TCP/
IP packets, or even serve Web
pages over a LAN or the Internet.
The iSeries are also available with
serial communications. With the "-C24" option, the user can select
from the pushbutton menu between
RS232, RS422, and RS485, with
straightforward ASCII commands or
MODBUS.

Note: "-DC", "-C24" and "-C4EIT" not available with excitation. Models "-EIT" and "-C4EIT" are only offered on DPi8 and DPiS8 models. * 20 to 36 Vdc for DPi8A, DPi16A, -C4EIT, or -EIT. Ordering Example: (1) DPi8-C24 1/8 DIN Meter with isolated RS-232 and RS-485 300 to 19.2 k baud, $260.00

Part Number Builder

(1)

(2)

(3)

(4)

(5)

D

D

P

P

i

i

Qty:

Option Descriptions

(1)

Input Type

SelectNothing(leave field blank) for Temperature/Process MonitorS for Strain

(2)

Case Size

Select32 for 1/32 DIN16 for 1/16 DIN16A for 1/16 DIN and isolated analog output8 for 1/8 DIN8A for 1/8 DIN and isolated analog output8C for 1/8 DIN with compact case

(3)

Network Options

SelectNothing(leave field blank) for No Network Options-C24 for Isolated RS232 and RS485 300 to 19.2 k baud-EIT for Ethernet with Embedded Internet (only for DPi/DPiS8)-C4EIT for Ethernet with Embedded Web Server + Isolated RS485/422 hub for up to 31 devices (only for DPi/DPiS8)

(4)

Power Supply

SelectNothing(leave field blank) for Standard Power (90 to 240 Vac/dc @ 50 to 400 Hz)-DC for Low Voltage Power (12 to 36 Vdc/24 Vac for single display. 20 to 36 Vdc/24Vac for dual display ethernet or isolated analog output)

(5)

Factory Setup

SelectNothing(leave field blank) for Standard Unit-FS for Factory Setup and Configuration

NOTE: All combinations may not be valid, check spec sheet for valid part numbers.