Advanced search

Advanced search is divided into two main parts, and one or more groups in each of the main parts. The main parts are the "Search for" (including) and the "Remove from search" (excluding) part. (The excluding part might not be visible until you hit "NOT" for the first time.) You can add new groups to both the including and the excluding part by using the buttons "OR" or "NOT" respectively, and you can add more search options to all groups through the drop down menu on the last row (in each group).

For a result to be included in the search result, is it required to fit all added including parameters (in at least one group) and not fit all parameters in one of the excluding groups. This system with the two main parts and their groups makes it possible to combine two (or more) distinct searches into one search result, while being flexible in removing results from the final list.

Limit the search further

Here you can limit your search further, your result list will only contain those who match all of the criteria that you fill out in this part (combined with the advanced search from above)

Frostell, Claes

Karlsson, Sigbritt

Abstract [en]

OBJECTIVES: The objectives were to compare the duration of use of polymeric tracheostomy tubes, i.e., silicone (Si), polyvinyl chloride (PVC), and polyurethane (PU), and to determine whether surface changes in the materials could be observed after 30 days of patient use. METHODS: Data were collected from patient and technical records for all tracheostomized patients attending the National Respiratory Center in Sweden. In the surface study, 19 patients with long-term tracheostomy were included: six with Bivona TTS Si tubes, eight with Shiley PVC tubes, and five with Trachoe Twist PU tubes. All tubes were exposed in the trachea for 30 days before being analyzed by scanning electron microscopy (SEM) and attenuated total reflectance Fourier transform infrared spectroscopy (ATR-FTIR). New tubes and tubes exposed in phosphate-buffered saline were used as reference. RESULTS: Si tubes are used for longer periods of time than those made of PVC (P < .0001) and PU (P = .021). In general, all polymeric tubes were used longer than the recommended 30-day period. Eighteen of the 19 tubes exposed in patients demonstrated, in one or more areas of the tube, evident surface changes. The morphologic changes identified by SEM correlate well with the results obtained by ATR-FTIR. CONCLUSIONS: Si tracheostomy tubes are in general used longer than those made of PVC and PU. Most of the tubes exposed in the trachea for 30 days suffered evident surface changes, with degradation of the polymeric chains as a result.