Laboratórium

These compact melting furnaces for the melting of non-ferrous metals and alloys are one of a kind and have a number of technical advantages. Designed as tabletop models, they can be used for many laboratory applications. The practical counter balanced hinge with shock absorbers and the spout (not for KC) on the front of the furnace make exact dosing easy when pouring the melt. The melting furnaces are available for furnace chamber temperatures of 1000, 1300, or 1500 °C. This corresponds to melt temperatures of about 80 °C - 110 °C lower.

Crucible with integrated pouring spout of iso-graphite included with delivery

Additional spout (not for KC), mounted at the furnace for exact pouring

Compact bench-top design, simple emptying of crucible by tiltiing system with gas damper

Crucible for heating up of melting furnace insulated with a hinged lid, lid opened when pouring

Defined application within the constraints of the operating instructions

Additional equipment

Other crucible types available, e.g. steel

Design as bale-out furnace without tilting device, e.g. for lead melting

Over-temperature limiter for the furnace chamber with automatic reset to protect against overtemperature. The limit controller switches off the heating when the pre-set limit temperature has been reached and does not switch it on again until the temperature falls below the setting again.

Observation hole for melt

Model

Tmax

Crucible

Volume

Outer dimensions in mm

Connected

Electrical

Weight

°C

in l

W

D

H

load kW

connection*

in kg

K 1/10

1000

A 6

0.75

520

680

660

3.0

1-phase

85

K 2/10

1000

A10

1.50

520

680

660

3.0

1-phase

90

K 4/10

1000

A25

3.00

570

755

705

3.6

1-phase

110

K 1/132

1300

A 6

0.75

520

680

660

3.0

1-phase

120

K 2/132

1300

A10

1.50

520

680

660

3.0

1-phase

125

K 4/132

1300

A25

3.00

570

755

705

5.5

3-phase1

170

KC 1/153

1500

A6

0.75

580

630

580

10.5

3-phase

170

KC 2/153

1500

A10

1.50

580

630

580

10.5

3-phase

170

1Heating only between two phases *Please see page 60 for more information about supply voltage2Outer dimensions of furnace, transformer in separate housing (500 x 570 x 300 mm)3Switchgear and controller mounted in a floor standing cabinet