Temperature sensor

NRT 100/1000

“High reliability and best accuracy”This is the indoor temperature sensor that uses platinum measurement resistance Pt 100 or Pt 1000 and this product is to measure the indoor temperature by using the principle of the platinum measurement resistance that has the resistance increased in proportion to the temperature.

System configuration

Major characteristics

Broad range of temperature for -50℃ ~ +150 ℃

Measurement accuracy for 0 ℃ ±0.3 ℃ (Tolerance)

Available for separate order for 3-type product in addition to 2-type (temperature calibration possible following the cable length)

Available to connect to M3323, M323 and B321

Available to connect to various devices/modules to have the input of PT 100/1000

Providing the support for collection on the walls (non-reclaimed type)

Refer to the product catalog for detailed specification

NPT 100/1000, NDT 100/1000

“High reliability and best accuracy”The platinum measurement resistance Pt 100 or Pt 1000 is used to sense the through of water and air flowing on the duct or pipe. There are duct acceptance NDT and pipe acceptance NPT module and, for the NPT product, the temperature sensor part is SS material with 100% seal processing.

Internal wiring drawing

Major characteristics

Broad range of temperature for -50℃ ~ +150 ℃

Measurement accuracy for 0 ℃ ±0.3 ℃ (Tolerance)

Connection of 3-type (temperature calibration possible following the cable length)

Measure the temperature by connecting M3323, M323 and B321

Available to connect to devices to have the input of PT 100/1000

Refer to the product catalog for detailed specification

Temperature and humidity transmitter

NT-300P/300P2, NH-300P, NHT-300P/300P2

“High reliability and best accuracy”The platinum measurement resistance Pt 100 and semiconductor-type precision humidity sensor is used to measure the indoor temperature and humidity and is the transmitter to convert and output in proportional 4-20mA current. In the event that the distance between the sensor location and controller distance are separated for 4-20mA output with the industrial standard of the current loop method, the temperature and humidity information can be transmitted accurately.

25 ℃ based, up to ±2% (10%~90%RH zone), in other zones, ±4%RH of accuracy (Tolerance)

Scope of broad voltage input (9-30VDC)

Measure the temperature by connecting to M325, M3325 and B325

Available to connect with all types of controller that may receive the 4-20mA current output of industrial standard

For temperature (NT), for humidity (NH) and for temperature and humidity (NHT)

Refer to the product catalog for detailed specification

NTD-300P1/300P2, NHD-300P, NHTD-300P1/300P2

“High reliability and best accuracy”The platinum measurement resistance Pt 100 and semiconductor-type precision humidity sensor is used to measure the temperature and humidity of the air in the duct and is the transmitter to convert and output in semiconductor-type proportional 4-20mA current. In the event that the distance between the sensor location and controller distance are separated for 4-20mA output with the industrial standard of the current loop method, the temperature and humidity information can be transmitted accurately.

System drawing

Major characteristics

Range of temperature measurement for 0℃ ~ +100 ℃ (P1 type)

Range of temperature measurement for -30℃ ~ +70 ℃ (P2 type)

Accuracy of measurement for 0℃ ±0.25 ℃ for accuracy (Tolerance)

Scope of humidity measurement of 0~100%RH (Not to be condensed)

25 ℃ based, up to ±2% (10%~90%RH zone), in other zones, ±4%RH of accuracy (Tolerance)

Scope of broad voltage input (9-30VDC)

Measure the temperature by connecting to M325, M3325 and B325

Available to connect with all types of controller that may receive the 4-20mA current output of industrial standard

For temperature (NT), for humidity (NH) and for temperature and humidity (NHT)

Refer to the product catalog for detailed specification

NTO-300P, NHO-300P, NHTO-300P

“High reliability and best accuracy”The platinum measurement resistance Pt 100 and semiconductor-type precision humidity sensor is used to measure the indoor temperature and humidity of external air and is the transmitter to convert and output in proportional 4-20mA current. In the event that the distance between the sensor location and controller distance are separated for 4-20mA output with the industrial standard of the current loop method, the temperature and humidity information can be transmitted accurately.

System drawing

Major characteristics

Range of temperature measurement for -30℃ ~ +70 ℃

Accuracy of measurement for 0℃ ±0.25 ℃ for accuracy (Tolerance)

Scope of humidity measurement of 0~100%RH (Not to be condensed)

25 ℃ based, up to ±2% (10%~90%RH zone), in other zones, ±4%RH of accuracy (Tolerance)

Scope of broad voltage input (9-30VDC)

Measure the temperature by connecting to M325, M3325 and B325

Available to connect with all types of controller that may receive the 4-20mA current output of industrial standard

For temperature (NT), for humidity (NH) and for temperature and humidity (NHT)