Hello all,
The 6 months pm is done, the wec has been rebuild and the power supply has been tested and the uniformity test has been checked.
We ran 1 um 3612 in hard contact and resolved 1.5 um resolution , exposure 1 second.
The same test done in vacuum contact and 1 um resolved nicely on all 5 sites with 1 second of exposure.
Please use the system and let us know if there are any issues.