This kit is designed to help electricians and HVAC technicians solve problems quicker. Test first for overheated electrical devices using an IR thermometer; then use the clamp meter and voltage detector to find out more about the problem

Fluke 62 MAX+ Infrared Thermometer

±1.0°C or ±1.0% of reading, whichever is greater -10°C to 0°C: ±2.0 -30°C to -10°C: ±3.0

Response time (95%)

< 300 ms (95% of reading)

Spectral response

8 to 14 microns

Emissivity

0.10 to 1.00

Optical resolution

12:1 (calculated at 90% energy)

Display resolution

0.1°C (0.2°F)

Repeatability of readings

±0.5% of reading or < ±0.5°C (1°F), whichever is greater

Power

AA battery

Battery life

8 hours with laser and backlight on

Physical Specifications

Weight

255 g (8.99 oz)

Size

175 x 85 x 75 mm (6.88 x 3.34 x 2.95 in)

Operating temperature

0°C to 50°C (32°F to 122°F)

Storage temperature

-20°C to 60°C (-4°F to 140°F), (without battery)

Operating humidity

10% to 90% RH non-condensing at 30°C (86°F)

Operating altitude

2000 meters above mean sea level

Storage altitude

12,000 meters above mean sea level

IP rating

IP 54 per IEC 60529

Drop test

3 meters

Vibration and shock

IEC 68-2-6 2.5 g, 10 to 200 Hz, IEC 68-2-27, 50 g, 11 ms

EMC

EN 61326-1:2006 EN 61326-2:2006

Standards and agency approval

Compliance

EN/IEC 61010-1: 2001

Laser safety

FDA and EN 60825-1 Class II

Fluke 323 True RMS Clamp Meter

Specifications

AC current

Range

400.0 A

Accuracy

2% ±5 digits (45 Hz to 65 Hz) 2.5% ±5 digits (65 Hz to 400 Hz)

AC voltage

Range

600.0 V

Accuracy

1.5% ±5 digits

DC voltage

Range

600.0 V

Accuracy

1.0% ±5 digits

Resistance

Range

400.0 Ω/ 4000 Ω

Accuracy

1.0% ±5 digits

Continuity

≤ 70 Ω

AC response

True-rms

Data hold

Yes

Size

H x W x D (mm)

207 x 75 x 34

Max wire diameter

30 mm (600 MCM)

Weight

265 g

Category rating

CAT III 600 V CAT IV 300 V

Warranty

Two-year

Fluke 1AC-II VoltAlert

Specifications

Operating principle

Senses the steady state electrostatic ﬁeld produced by ac voltage through insulation without requiring contact to the bare conductor. A red glow at the tip and a beeping noise (if not switched OFF) indicates the presence of voltage.

Voltage sensing ranges

Nominally, 90 VAC to 1000 VAC or 200 VAC to 1000 VAC depending on model, 45 Hz to 405 Hz; also a 20 V to 90 V control circuit model