Scope of the ArtThe R&S®RTE oscilloscopes are available with two or four channels and a bandwidth of 200 MHz, 350 MHz, 500 MHz, 1 GHz, 1 GHz, or 2 GHz. With a sampling rate of 5 Gsample/s and a memory depth of 10 Msample per channel (can be optionally expanded to 50 Msample per channel), the instruments provide excellent performance parameters.

An acquisition rate of more than one million waveforms per second ensures that signal faults are found quickly. A highly accurate digital trigger system and the excellent dynamic range of the A/D converter with its more than seven effective bits provide precise results.

Besides the cursor functions and automated measurements, the R&S®RTE offers even more analysis tools that help users to complete tasks quickly. At the push of a button, for example, the QuickMeas function simultaneously displays the results of multiple measurement functions for the signal that is currently active. The FFT analysis is also unique. It gives the impression of a live spectrum on the screen and reliably detects sporadic signals.

A broad range of dedicated application solutions is also available, including trigger and decode options for serial buses such as I2C, SPI and CAN, and a power analysis option. The logic analysis capability offered by the R&S®RTE is essential for analyzing digital components of embedded designs. The R&S®RTE-B1 mixed signal option can be added to any base unit and offers 16 additional digital channels. It is possible to decode up to four parallel buses simultaneous

The wide variety of measurement and analysis functions can be easily operated via the high-resolution 10.4" XGA touchscreen. Transparent dialog boxes ensure that the measurement diagrams always maintain their original size. Signal flow diagrams in the dialog boxes simplify navigation. Signal icons with realtime preview on the edge of the screen clearly show what is currently happen.

More reliable measurements, more tools and fast results, more fun to use — that's the R&S®RTE oscilloscope. From embedded design development to power electronics analysis to general debugging, the R&S®RTE offers quick solutions for everyday T&M tasks.

R&S RTE Series Oscilloscopes

Model

Bandwidth

Ch.

Memory Depth*

Unique New Features

RTE Series w/ HZ-15 near-field probe set

R&S®RTE1204

2 GHz

4

40 Msample/Channel

FingertipZoom: Move the finger along the signal to get a quick overview of signal details

FFT based spectrum analysis: powerful & user-friendly

SaveSet tool: Fast access to instrument configurations

QuickMeas: Key results at the push of a button

R&S®RTE1154

1.5 GHz

4

40 Msample/Channel

R&S®RTE1104

1 GHz

4

40 Msample/Channel

R&S®RTE1102

1 GHz

2

20 Msample/Channel

R&S®RTE1054

500 MHz

4

40 Msample/Channel

R&S®RTE1052

500 MHz

2

20 Msample/Channel

R&S®RTE1034

350 MHz

4

40 Msample/Channel

R&S®RTE1032

350 MHz

2

20 Msample/Channel

R&S®RTE1024

200 MHz

4

40 Msample/Channel

R&S®RTE1022

200 MHz

2

20 Msample/Channel

* On 1-channel, standard. Memory Depth can be increased with B101 (x2) and B102 (x5) options.