Atomlab Wipe Test Counter - Biodex

BACK TO CATALOG

Atomlab Wipe Test Counter

Atomlab Wipe Test Counter - Biodex

Easy to use, easy to understand.

The Atomlab® Wipe Test Counter is easy to use, easy to understand, fast and dependable. A touchscreen display utilizing intuitive software eliminates the tedium of wipe testing. Simply perform a daily calibration and background count, then count the wipe for each predetermined location. Trigger levels can be set for any isotope at any location including 200 dpm for iodine. In seconds the system will determine if the location is above or below the user defined trigger level.

When performing a wipe test, the full spectrum is displayed. A wide window that includes the isotope energies expected in a particular department is set by the user. The efficiencies of the isotopes selected for the window can be either factory defaults or user determined using an integrated detector efficiency program. Individual isotope ROIs along with the wide window can be set. This feature helps identify the isotope(s) causing contamination.

The user may define an unlimited number of “Area” names and wipe “Locations” and designate them as restricted area, unrestricted area, sealed source or package. The Wipe Test Program automatically calculates net contamination levels and reports results in dpm, uCi and Bq.

The system consists of a lead shielded 2" x 2" sodium iodide (NaI) well detector and a 64 channel multichannel analyzer. The displayed energy range (spectrum) is 0-800 KeV, which is typically found in nuclear medicine departments.

The wipe counter is designed to meet or exceed all NRC (10 CFR 35.70, 10 CFR 20.1906 and 10 CFR 35.2067) and state wipe test requirements. There are automated programs for the quality assurance functions: calibration, FWHM, chi-square and minimum detectable activity (MDA).

Wipe test results and QA test data can be stored in memory and printed at any time.

Manual MCA Program

For performing sample counts and analysis beyond the scope of the standardized programs.

Expandable library of 34 commonly used isotopes

Unlimited user-defined isotopes

Two counting methods: preset time and preset ROI

In addition to providing “standardized” programs for routine test and survey procedures, the Atomlab 500 now offers a “manual” program to accommodate other counting and spectrum analysis tasks. The user can select from three counting methods: preset time, preset ROI counts, and continuous counting. After any count, the Spectrum Analysis option can be used to produce a Spectrum Analysis Report. The monitor shows when a test is in progress, and graphically displays the spectrum, time, counts and cpm.

Eliminate the tedium of wipe testing with the Atomlab™ Wipe Test Counter

Windows® 10 Operating System

Microsoft SQL Database

Communicates with most commercially available NM management systems.

USB connectivity for external monitor, printer, and software upgrades

64 Channel MCA

NEW Manual MCA Program

Adjustable wide window and individual isotopes

2x2 NaI drilled-well detector

Remote shielded well

Energy spectrums with individual ROI

Ability to help identify isotopes causing contamination

User-specific wipe locations and trigger levels

Wipes that exceed trigger levels are immediately recognized

Detailed wipe reports including cpm and dpm

Wipe testing results stored

Upgradable at any time to a dose calibrator by adding an ionization chamber

Report Printer available

Two-year warranty

Enquire about this product

Do you have a question or would like further information? We are here to help!.

Product:Name:Phone:Email:Your Message:

InMed is a leading supplier of medical and scientific products and services.

We combine over 20 years of experience with an uncompromising commitment to our customers – to ensure we provide comprehensive services and top quality brands at competitive prices.