Sunbeam Blood Pressure Monitor 7652 User's Manual

INSTRUCTION MANUAL
Automatic Digital Blood Pressure Monitor
Model 7652
Sunbeam-Oster Household Products
Before operating this unit please read these instructions completely.
P.N.99944
1. DO Read instructions carefully before using Monitor.
2. DO Rest arm on a flat surface so that upper arm is at heart level when
taking measurements.
3. DO Remain calm, quiet and still when measurements are being taken.
4. DO make sure that the cuff is in the proper position according to your
instruction manual.
5 . DO Insure that the exhaust velocity is 2-5 mmHg per second at the systolic
point each time a measurement is taken.
6 . DO Wait at least IO minutes between taking measurements.
7. DO Pump unit pressure 30 to 60 mmHg above normal systolic before
beginning measurement.
8. DO Consult your physician before adjusting your medication based on
readings from this Monitor.
9. DO Use fresh alkaline batteries.
10. DO Use a dry, soft cloth to clean the instrument.
1. DO NOT move arms or hands when measurement is being taken.
2. DO NOT talk or chew when measurement is being taken.
3. DO NOT take a series of measurements without waiting at least 10 minutes
between measurements.
4. DO NOT adjust medication based on readings from this Monitor.
5. DO NOT make any adjustments to unit, there are no user serviceable parts.
Call l-800-597-5978 for further instructions.
6. DO NOT smoke during or at least 30 minutes prior to taking blood pressure
measurements.
7. DO NOT drop or otherwise mechanically shock the Monitor. This is a delicate
precision device.
8. DO NOT use thinner, alcohol, benzene or wet cloth to clean unit.
9. DO NOT subject to extremes in temperature, humidity, direct sunlight or dust.
10. 00 NOT fold the Arm Cuff or Rubber Cuff Hose tightly during storage, as
such treatment may shorten the life of the components.
Manufactured to meet specifications set by the Association for the
Advancement of Medical Instrumentation (AAMI)
What is Blood Pressure?
Blood pressure is a measurement of the force of blood flowing against the walls of the
arteries. Arterial blood pressure is constantly changing during the course of the cardiac
cycle. The highest pressure in the cycle is called the SYSTOLIC BLOOD PRESSURE;
the lowest is the DIASTOLIC BLOOD PRESSURE. Both pressure readings, the SYSTOLIC and DIASTOLIC, are necessary to enable a physician to evaluate the status of a
patient’s blood pressure. Many factors such as physical activity, anxiety, or the time of day,
can influence your blood pressure. Blood pressure is typically low in the morning and
increases from afternoon to evening. It is lower in the summer and higher in the winter.
Why Is It a Good Thing to Measure Blood Pressure at Home?
Having one’s blood pressure measured by a doctor in a hospital or a clinic, and group
health checks, tend to stimulate nervousness in the subject and may even create high blood
pressure. Also blood pressure varies in accordance with a variety of conditions, and so
judgement is not possible on the basis of a single measurement. (See Fig. 2)
The blood pressure measured first thing in the morning after getting up, before taking any
food, and with the subject still, is known as the fundamental blood pressure. In practice it
is rather difficult to record the fundamental blood pressure, but to come as near as possible
to measuring the blood pressure in an environment that is close to this, is why it is useful
to take the measurement at home.
Systolic: The highest point of blood pressure; level of blood pressure produced
as your heart beats.
Diastolic: The lowest point of blood pressure; level of blood pressure produced
as your heart rests between beats.
Millimeters of Mercury (mmHg): Blood pressure is measured in terms of the
height of mercury in a column. Measurements are written one over the other.
For example, a systolic presssure of 120 mmHg and diastolic pressure of 80
mmHg is expressed as “120 over 80.”
IMPORTANT: Only a physician is qualified to interpret your blood pressure
measurements, and no device can replace regular medical examinations by
your physician. It is recommended that your physician review your procedure
for using this blood pressure monitor. Your physician should verify blood
pressure measurements before making adjustments to medication.
Standards for assessment of high or low blood pressure, without regard to age,
have been established by the World Health Organization (WHO), as shown in this
chart.
(Fig. 1)
160
140
mmHg
Reference Material: Investigation into Adult Diseases Report by the Ministry of Health
and Social Security, 1971.
Individual blood pressures vary greatly both on a daily and a seasonal basis.
These variations are even more pronounced in hypertense patients. Normally the
blood pressure rises while at work and is at its lowest during the sleeping
period. The graph below illustrates the variations in blood pressure over a whole
day with measurement taken every five minutes.
53
E
160
[
140 *
120,
.c
P
z
100.
&
'
60.
-g
40.
0
m
60 *
.
20'...'."'.'
1 2 1314 1 5 1 6 1 7 1 8 1 9 2021 2 2 2 3 2 4
1
2
.
3
4
5
6
.
7
8
9
.
1011
Time
AM
Shown is data for mea&!ments taken every 5 minutes. The thick line represents sleep.
The rise in blood pressure at 4 PM (A in the graph) and 12 PM (B in the graph) correspond to an attack of pain and sexual intercourse. (Beven, Honour & Stott: Clin. Sci.
36:329, 1969.)
DIAGRAM A
Systolic blood pressure
display section
Diastolic blood pressure,
pulse display section
Pressure
setting switch
Memory switch
Start
switch
Power switch
Hose
connector
Main kame
Hose Plug
- Cuff hose
Measurement in
progress
Mark appears in the measurement condition and
flashes when pulse is
detected.
Measurement in progress remain quiet.
Mark flashes when power is
applied and there is air
remaining in the cuff.
Automatic exhaust is performed.
Measurement has begun
but the pressure was
insufficient.
Raise the cuff
pressure 30-40 mmHg
more andmeasure once
again.
Appears when the battery
voltage is excessively low.
Replace all four batteries
with new ones.
Exhaust
A. ,
Insui’ficient
pressure
Replace
batteries
Measurement error
PU ERR
Pulse error
Appears when the blood
pressure could not be
obtained accurately.
Appears when the pulse
rate could not be obtained
accurately.
Exhaust the air from the
cuff, and re-apply the cuff
properly and re-measure.
(Wait 10 minutes before
re-measure)
1. Battery Installation
Remove the battery cover and insert batteries into the battery compartment as shown, taking
extreme care that the polarities+ and - are observed. (Fig. 3)
(Fig. 3)
2. Attaching the Arm Cuff
Wind the arm cuff around the left upper arm tightly (for best results) so that the bottom
edge of the cuff is positioned one inch above the elbow joint.
(Fig. 4)
.Do not roll up a shirt or jacket.
It may cause constriction of the upper
arm and proper measurement results
will not be obtained. (Fig. 5)
3. Measuring Posture
Remain in a seated or reclining position
Do not move your arm while measuring.
(Fig. 5)
1. Turn Power ON
l When power is turned ON, all display marks appear for approximately 1 second.
l When the “0” is displayed the unit is ready for measurement. If the ‘7” mark appears, release air
in the arm cuff by squeezing it or removing the hose plug.
Approx. 1 second
2. Attach the arm cuff to the left arm for best results.
3. Set the pressure value approximately 40 mmHg above your normal
Systolic blood pressure by pressing the PRESSURE SWITCH. There are 9
settings: 120,140,160,160,200,220,240,260,
and 280 mmHg. See “Important
Note” on following page.
4. Press the START SWITCH.
l The arm cuff pressurizes when the Start Switch is pressed.
l When the Start Switch is subsequenHy pressed during pressurization, air in the arm
cuff is released.
5. When Pressurization is completed, the automatic exhaust mechanism will
gradually reduce arm cuff pressure, and the “v“ mark will appear indicating that
measurement is in progress.
1) Pressure value appears on the left, and exhaust speed on the right side of the display.
Current Pressure Value
(gradually being reduced)
Exhausl
Speed
Measure
Mark
2) When a pulse is detected, the pressure value moves to the right side of the display. At
this time the ‘I e “ mark flashes at the same rate as pulse rate, and the buzzer
sounds when th6 pulsation is sufficient.
urrent Pressure
Flashing
alue
IMPORTANT NOTE
If insufficient pressure is obtained, the unit will inflate the cuff again to the next
pre-set pressure. The unit will continue to reinflate until it is able to obtain an
accurate pressure, each time inflating to the next preset pressure. At that point, if the
unit is unable to obtain sufficient pressure the “Err” mark will appear in the display.
6. Turn Power OFF
Press the Power switch to turn power OFF. (See Diagram A)
7. Previous Measurement Display
When the Memon/ switch is pressed, the previous measurement value is shown for
about 5 seconds.8. Automatic Power OFF Function
If this unit is left on after measurement an Automatic Power-OFF function turns the
power OFF after approximately 2 min. 30 seconds. To continue measurement press,
the power switch turning the unit ON.
l
Name and model number
Display system
Measuring method
Power source
Measuring range
Digital Blood Pressure Meter 7652
Digital display system/LCD 6 digits
Oscillometric method
Type AA batteries x 4
20-280 mmHg (Pressure)
40-200 Pulses/minute (Pulse)
Accuracy
- 3 mmH or 2% whichever greater (Pressure)
-5% (Pu9 se)
Pressurization
Depressurization
Deflation
Memory
Automatic pressurization
Constant-Air-Release Valve System
Manual air release valve
Built-in memory enabling display of the previously measured value and
currents measurements.
To be automatically cut off after 2 minutes 30 seconds of nonuse to
save energy.
Approx. 12 months with 3 min. usage per day
50” to 14O”F, less than 85% RH
14” to 122”F, less than 85% RH
5”(W) x 3”(D) x i&(H)
13.0 oz (including 4 batteries)
Automatic Power OFF
Battery life
Operating environment
Storage environment
Dimensions
Weight
MODEL 7662
REPLACEMENT PARTS
AND ACCESSORIES
-
Health@ Home
ParIt
47834-042
47834-033
47834-012
47834-036
- 47834-044
47834-037
99935
47834-006
*
47834-038
Description
Reaular Adult Cuff
Battery Cover
Replacement Hose
Exhaust Valve Assembly
Replacement Bulb
Storage Case
Instruction Book
Blood Pressure Record
Sheet (25/pack)
L Connector
Above prices include shipping and handling.
Please print your name and address below.
Name
Street Address
City
State
Price
12.95
2.00
4.95
4.95
2.95
4.50
1 .oo
3.95
4.95
Order 1
Quantity
Total
Ii: $ I
Zip
Send to: Sunbeam-Oster Household Products
Purvis, Mississippi 39475
or call l-800-394-2326
For Mastercard or VISA: Circle one
Mastercard
Card X
Exp. Date
Authorization
Signature
VISA
LIMITED ONE YEAR WARRANTY
This product is warranted for one year from the date of purchase against defects in
material and workmanship. If the product fails to operate during this period, return
it, prepaid, to Appliance Service Station, 117 Industrial Central Row,
Purvis, MS 39475 for repair or replacement without charge at the manufacturers option or call l-800-597-5978. This warranty gives you specific legal
rights, and you may also have other rights which vary from state to state.
Warrantor: Sunbeam-Oster Household Products.
’
@Sunbeam TM Health at Home, 01992 Sunbeam-Oster Household Products.
Distributed by Sunbeam-Oster Household Products, Schaumburg, IL 60173
e
MODEL 7652
REPLACEMENT PARTS
AND ACCESSORIES
Health #Home
__________________------------------------------------Please print your name and address below.
City
State
Zip
Include check or money order made payable to Sunbeam-Oster Household Products
and mail to:
Sunbeam-Oster Household Products
Purvis, Mississippi 39475
MODEL 7652
REPLACEMENT PARTS
AND ACCESSORIES
HealthmHome
PartX
47834-042
47834-033
47834-012
47834-036
47834-044
47834-037
99944
47834-006
Description
Reaular Adult Cuff
Battery Cover
Replacement Hose
Exhaust Valve Assembly
Replacement Bulb
Storage Case
Instruction Book
Blood Pressure Record
Sheet (25/pack)
L Connector
47834-036
Above
Please
N a
Street
City
prices include shipping and handling.
print your name and address below.
m e
Address
State
Price
12.95
2.00
4.95
4.95
2.95
4.50
1 .oo
3.95
4.95
Order 1
Quantity 1 Total
I
I:$
-
Zip
Send to: Sunbeam-Oster Household Products
Service & Return Center
117 Industrial Central Row Purvis, Mississippi 39475
or call l-800-394-2326
For Mastercard or VISA: Circle one
Mastercard
VISA
Card #
Exp. Date
Authorization
Signature