100kV, 100W X-Ray Source

Spellman’s XRB100 Monoblock® X-Ray source is designed for OEM applications powering its internal X-Ray tube up to 100kV at 100W. Features like universal input, small package size and a standard analog and RS-232 digital interface simplify integrating the XRB100 into your X-Ray system. Proprietary emission control circuitry pro­vides excellent regulation of X-Ray tube current, along with outstanding stability performance.

Typical applications:

X-Ray Scanning: Food Inspection

Fill Level Confirmation

Security Applications

사양

(Ref. 128072-001 REV. H)

X-Ray Characteristics:

Tube Type: Stationary anode, tungsten target

Focal Spot: 0.5mm (IEC 336)

Beam Filter: Lexan: 3.2mm

Oil: 10mm ±0.1mm

Glass: 1.8mm max

Beam Geometry: Symmetrical fan 74° x 10° ±1%.

Input Voltage:

Power factor corrected input 0.98, 90-264Vac, 47-63Hz, 2A, maximum

X-Ray Tube Voltage:

Nominal X-Ray tube voltage is adjustable between 40kV to 100kV

X-Ray Tube Current:

100uA to 1mA over specified tube voltage range

X-Ray Tube Power:

100W, maximum continuous

Voltage Regulation:

Line: ±0.1% of maximum output voltage over a ±10% change of nominal input line voltage

Load: ±0.1% of maximum rated voltage for 100uA to 1mA load change

Voltage Accuracy:

Voltage measured across the X-Ray tube is within ±2% of the programmed value

Voltage Risetime:

Ramp time shall be 1 second from 10% to 90% of maximum rated output voltage

Voltage Overshoot:

5% of maximum voltage, to return within 2.5% of maximum voltage in less than 50ms

Voltage Ripple:

0.5% peak to peak of maximum voltage for frequencies ≤ 1kHz

Emission Current Parameters Current Regulation:

Line: ±0.5% of rated output current over a ±10% change of nominal input line voltage

Load: ±0.5% of rated output current for a change from 50% to 100% of rated output voltage

Current Accuracy:

Current measured through the X-Ray tube is within ±1% of the programmed value