Abstract. Reliable and reproducible measurements of atmospheric aerosol particle number size distributions below 10nm require optimized classification instruments with high particle transmission efficiency. Almost all differential mobility analyzers (DMAs) have an unfavorable potential gradient at the outlet (e.g., long column, Vienna type) or at the inlet (nano-radial DMA), preventing them from achieving a good transmission efficiency for the smallest nanoparticles. We developed a new high-transmission inlet for the Caltech nano-radial DMA (nRDMA) that increases the transmission efficiency to 12% for ions as small as 1.3nm in Millikan–Fuchs mobility equivalent diameter, Dp (corresponding to 1.2 × 10−4m2V−1s−1 in electrical mobility). We successfully deployed the nRDMA, equipped with the new inlet, in chamber measurements, using a particle size magnifier (PSM) and as a booster a condensation particle counter (CPC). With this setup, we were able to measure size distributions of ions within a mobility range from 1.2 × 10−4 to 5.8 × 10−6m2V−1s−1. The system was modeled, tested in the laboratory and used to measure negative ions at ambient concentrations in the CLOUD (Cosmics Leaving Outdoor Droplets) 7 measurement campaign at CERN. We achieved a higher size resolution (R = 5.5 at Dp = 1.47nm) than techniques currently used in field measurements (e.g., Neutral cluster and Air Ion Spectrometer (NAIS), which has a R ∼ 2 at largest sizes, and R ∼ 1.8 at Dp = 1.5nm) and maintained a good total transmission efficiency (6.3% at Dp = 1.5nm) at moderate inlet and sheath airflows (2.5 and 30Lmin−1, respectively). In this paper, by measuring size distributions at high size resolution down to 1.3nm, we extend the limit of the current technology. The current setup is limited to ion measurements. However, we envision that future research focused on the charging mechanisms could extend the technique to measure neutral aerosol particles as well, so that it will be possible to measure size distributions of ambient aerosols from 1nm to 1µm.

High transmission efficiency is key for classifying and counting atmospheric aerosol below 10 nm. We developed a new high­-transmission inlet for the Caltech nano-­radial DMA (nRDMA) and successfully deployed the nRDMA, equipped with the new inlet, in chamber measurements, using a particle size magnifier (PSM) and a booster CPC as a counter. With this setup, we were able to measure size distributions of ions between 1.3 and 6 nm in mobility diameter.

High transmission efficiency is key for classifying and counting atmospheric aerosol below...