4D B&W ultrasound is the most advanced black and white ultrasonic equipment in the world. 4D ultrasonic technology is based on the 3D ultrasonic image and with the time dimension parameters, which can obtain the 3D image in real time...

Quick Details

System Overview

The Clover series system is a high performance multipurpose, portable, simple to use, and durable system with best-in-class image quality. The Wisonic Clover has the patented unique features, such as HoloTM PW, Auto Doppler positioning, and our best in class 20 MHz probe. With our industry leading technologies, Clover is able to bring exceptional operating experience like never before.

The Clover is designed for cardiac, abdominal, obstetrics, gynecology, vascular, musculoskeletal, small parts, pediatric, neonatal, nerve block (assisting needle guidance in the administration of peripheral nerve block), and urology applications.

Packaging & Delivery

Specifications

Static 3D/4D doppler ultrasound imaging MSLCU56

1. System Overview

The Clover series system is a high performance multipurpose, portable, simple to use, and durable system with best-in-class image quality. The Wisonic Clover has the patented unique features, such as HoloTM PW, Auto Doppler positioning, and our best in class 20 MHz probe. With our industry leading technologies, Clover is able to bring exceptional operating experience like never before.

The Clover is designed for cardiac, abdominal, obstetrics, gynecology, vascular, musculoskeletal, small parts, pediatric, neonatal, nerve block (assisting needle guidance in the administration of peripheral nerve block), and urology applications.

Static 3D/4D doppler ultrasound imaging MSLCU56

1.1Application

Abdomen

Obstetrics

Gynecology

Cardiology

Small Parts

Urology

Vascular

Pediatrics

Emergency Medicine

Anesthesia

Others

1.2Transducer Type

Convex Array

Linear Array

Phased Array

1.3Imaging Modes

B Mode

Harmonic Imaging

M Mode

Color Doppler Imaging

Directional Power Doppler Imaging

PW Doppler

CW Doppler

TDI

3D/4D

Anatomic M

HoloTM PW Imaging

1.4Standard Features

B Mode

Harmonic Imaging

M Mode

Color Doppler Imaging

Directional Power Doppler Imaging

PW Doppler

HoloTM PW Imaging

Spatial compounding imaging (SCI)

Time-spatial speckle suppression imaging (imaging enhancement)

One key auto B imaging optimization

Auto Doppler

FZoom (Full Screen Zoom)

ExFOV(Extended FOV)

Raw Data

Clinical Application Packages

240GB SSD

Battery

2 USB 3.0 Ports

HDMI

Ethernet Port

WIFI Adaptor

1.5Optional Features

CW

Anatomic M

TDI

IMT

WiNeedle

3G/4G Network Adaptor

Video Convetor Box

Extra Battery

ECG module

Probe Extend Module

1.6Language

Software: English, Chinese

Keyboard Input: English, Chinese

User Manual: English, Chinese

Static 3D/4D doppler ultrasound imaging MSLCU56

2. Physical Spec

2.1System Dimension and Weight

Height: 65mm

Width: 352mm

Depth: 366mm

Weight: Around 4.9kg (Include Battery)

2.2Trolley Dimesion and Weight

Height: 820-1020mm

Width: 510mm

Depth: 662mm

Weight: Around 22kg

2.3Monitor

15-Inch LED

Resolution: 1024×768

Brightness Adjustable

Max Opening Angle:135°

2.4Probe Extend Module(option)

One port extend three ports

Height: 155mm

Width: 148mm

Depth: 36mm

Weight: Around 920g

2.5ECG module(option)

3 Leads

AHA or IEA standards

Height: 90mm

Width: 108mm

Depth: 31mm

2.6Audio Output

High Quality Speaker

2.7Ergonomic Trolley

Height Range: 200mm

Wheel Diameter: 125mm

Wheel (4): Break and Lock

2.8Transducer Holder and Port

Connection Ports: 2 Fully Active

Holder: 4 on Trolley

2.9 AC Adapter

Model:MANGO120-19AD-WIS

Voltage: 100-240V~

Frequency: 50/60 Hz

Currency: 2.0-1.0A

Output: 19V,6.3A

2.10Battery

Model: LI14I06A

Lithium-Ion, 14.4V, 5800mAH

Charge to Full: 3 hours

Max Duration: 70 minutes realtime scan

2.11Operating Evironment

Temperature: 0-40 °C

Humidity: 30%-85% (non-condensing)

Pressure: 700hPa-1060hPa

2.12Storage and Transport

Temperature: -20-55 °C

Humidity: 30%-95%(non-condensing)

Pressure: 700hPa-1060hPa

3. User Interface

3.1Control Panel

User-Centric control panel, efficient and ergonomic.

Backlit control panel

Multiple programmable key

8-Segment TGC

Standard keyboard, with back light

Adjustable key volume and trackball sensitivity

Dedicated palm rest area, reduce wrist fatigue

3.2Boot-up and Power off

Hard boot-up in 23s

Power off in12s

Quick recover from sleep in 5s

3.3Comment

Text and arrow available

Size of text and arrow adjustable

Set home position for comment initiation

Full packages for all application

User-defined text

3.4Bodymark

Over 144 bodymark patterns for all application

Support transducer mark

3.5Screen Info*

Displayed Info:

-Wisonic Logo

-Hospital Name

-Exam Date

-Exam Time

-Acoustic Power

-MI

-Thermal Index

-ID, Name

-Probe

-TGC Curve

-Focus Position

-Thumbnail

-Image Parameters

*Not all items are listed in here, please refer to user manual for detailed info.