Bacharach 10-5046 Fyrite Classic Gas Oxygen O2 Analyzers

The Fyrite 10-5046 performs oxygen testing in connection with oxygen & gas anesthesiology in the 0 to 60% range. Accuracy is within +/- 0.5 % O2.

The number of possible tests with a single fluid charge is dependent on the concentration level of samples being tested. At midpoint scale reading the O2 fluid is good for 100 tests. Fluid replacements can easily be determined with an easy test & replacements are a simple procedure. These test procedures, as well as other good information, are provided in the Fyrite manual 11-9026.

Fyrite Indicators have a wide range and may be exposed to ambient temperatures from -30 to 150 degrees F. Gases to 850 degrees F may be tested using standard aspirator type sampling equipment. Special equipment for high gas temperatures or dry gases are available. For temperatures over 1400 degrees F, ceramic sampling tubes are available.