Research Stereo Microscope System SZX10

Product Details:

Product Description

Description of Research Stereomicroscope System SZX10

The SZX10 is specially designed for routine research. With a maximum numerical aperture (NA) of 0.2, producing a resolution of 600 line pairs per millimeter and a zoom ratio of 10:1, more information can be gained efficiently from samples. This suitable for oblique and bright field illumination, and compatibility with different imaging devices enables use with fluorescence and flexible imaging stations. The range of objectives with outstanding working distances offers versatility, while the Galilean optical system provides a natural view and distortion-free observation. The innovative design ensures optimal visual comfort and ergonomic use.

Specification of Research Stereomicroscope System SZX10

Observation Method

Fluorescence Blue/Green Excitations

✓

Fluorescence Ultraviolet Excitations

✓

Simple Polarized Light

✓

Brightfield

✓

Darkfield

✓

Zoom

Zoom Ratio

10

Magnification Indication

0.63, 0.8, 1, 1.25, 1.6, 2, 2.5, 3.2, 4, 5, 6.3

Motorized

✓

Features

Zoom variable magnification system with parallel optical axis

Zoom drive system: Horizontal handle

Click-stop for various zoom positions incorporated(Manual)

AS

Built In

Illuminator

Fluorescence Illuminator

Hg Lamp

✓

Xenon Lamp

✓

Light Guide Illumination

✓

Revolving Nosepiece

Manual

Standard Type

✓

Optics

Galilean Optical System

✓

Focus

Focusing Mechanism

Coarse/Fine Focus

✓

Motorized

✓

Load Capacity

Manual: 0-10.0kg/ 2.7-15.0kg/ 8.0-25.0kg

Motorized: 0.0-23.0kg

Coarse Handle Stroke

Manual: 80 mm/ 120 mm

Motorized: 78 mm

Coarse Handle Stroke per Rotation

Manual: 36.8 mm

Motorized: 2.7 mm

Fine Handle Stroke per Rotation

Manual: 0.77 mm

Motorized: 0.27 mm

DIDAC INTERNATIONAL

Describe Your BUYING Requirement *

* Tips on getting accurate quotes. Please include product name, order quantity, usage, special requests if any in your inquiry.