SparX Pro Digital X-ray System

SparX Pro Portable Digital X-ray System

The SparX Pro is the affordable option for the NDT operator that does not want to compromise on quality. It is a leading, cost effective DR system which is compatible with 1.6 km WiFi option and with our new rugged tablet. The entire system was designed to meet the real needs of Vidisco’s system operators who use our DR in the field on a regular basis.

SparX Pro Technical Configurations*

Rugged Case (STD)

Backpack

Chaser***

Casing

Small rugged case

Small backpack

Chaser case and bag

Default Computer

Standard laptop (rugged laptop optional)

Ruggedized laptop

-

CDU to ICU Cables

50m CDU to SparX Pro cable on removable reel

35m CDU to SparX Pro cable on built-in reel

-

Remarks

Additional imager to an existing system.

SparX Pro Technical Specifications**

Specification

SparX Pro

Imaging Area (mm):

320 x 255 mm (~12.8 X 10.2")

Imaging Area Edges Distance:

Right & Bottom Borders: 8.7mm

Imager Size:

385 X 320 X 18 mm (h,w,d) (~15.2 X 12.6 X 0.7 ”)

Imager Weight:

3.3 Kg (~7.2 lbs)

Dynamic Range:

16 Bit

Resolution:

127 μm (3.9 lp/mm)

Pixel Pitch:

127 μm

Robust, Field Design:

YES

Battery Operation:

Over 4 hours

Battery Power Level Indication:

LED Indicator

Charge During Nonstop AC Operation:

YES

Support Vehicle DC/AC Inverter:

YES

Accessories:

High Power Wireless (up to 1.5 km line of sight) WiFiX (300m line of sight between operator and down-range) WXR for Wireless Communication between X-Ray and Panel Dual Energy Image with Vidisco’s XR-DE Plus Tripod Mount SparX Slider - Automatic Multiplication of the Imaging Area