flox[on] J

FCTechnik Mass Flow Controller

flox[on] J

flox[on] J for huge gas flow up to 7000 Nl/min

The measurement of the flox[on]J mass flow controller for hugh gas flow is temperature and pressure compensated with a very high control accuracy in the lower and upper flow ranges. The adjustment control of the mass flow controller uses a pressure compensated proportional valve of the most recent design, which makes it unsusceptible to pressure fluctuations in the in- and outlet. The flox[on] mass flow controller is equipped with a high-performance micro-controller. This guarantees an unusually quick and dynamic regulation.

Application

Especially the flox[on]J mass flow controller was developed for controlling hugh gas flows up to 7000 Nl/min.It is often used in the steel and NE-heavy industry.
It is suitable for all noncorrosive gases like, air, nitrogen, argon or hydrogen.

Measurement Principle

The measurement is based on the principle of thermal anemometry. The measuring cell is located in the bypass. A gas flow of max. ca. 1 Nl min.-1 undergoes a constant energy feed by means of an electric heating element. The thermal feedback is converted to an electrical signal and then linearized through a calibration curve to give the effective flow rate.

Instead of the trademark flox[on] you can of course call our mass flow controller floxon too.

Data operation flox[on] J

Maximum flow

Nlmin-1

7000

Minimum flow

Nlmin-1

1500

Maximum operating pressure

bar

16

Regualtion ratio

1 : 40

Accuracy

± 3%

Step response (10% - 90%)

s

1 t0 5

Operating temperature

°C

from -10 to +60

flox[on] J Electrical Equipment

Connection voltage

V DC

24 ± 15 %

Capacity max.

W

20

Connection analog

mA

Thread 1" BSP

Signal Set/Real

4 - 20

Bypass Signal

V DC

24

dimensions flox[on] J

flox[on] J

Application:

For noncorrosive gases like air, nitrogen, argon or hydrogen

Materials:

Housing and flange connector - aluminium

Valve and sinter filter - brass

Any further questions?

If you have any additional questions, please do not hesitate to contact us.