FL Series Flush Industrial Pressure Transmitter

Designed for the highest system integrity

Dylix’s FL Series Pressure Transmitter is designed for the OEM and end user where ruggedness, high performance, and reliability are required at a competitive price. Key feature of the FL is its non oil –filled sensing element. The sensing diaphragm is machined from a solid piece of steel, and can withstand the most abrasive / cyclical applications.

Manufactured for extended life cycles

Dylix’s engineers developed this product to last in high cyclical environments with or without the presence of waterhammeringor spiking. Advanced manufacturing techniques, extreme environmental burn-in and thorough residual stress relieving procedures ensure the FL Series maintains its performance over time.

Dylix’s Customer Service

Every FL Series is shipped with a NIST traceable calibration certificate

Standard Features:

±≤ 0.5% FSO Static Accuracy

NEMA 4X Packaging

±≤ 0.25% FSO/YR stability

Available Options:

±≤ 0.25% FSO Static Accuracy

Hastelloy Wetted Materials

Zero & Span Controls / Shunt Cal

Production Specialist

Electrical

Excitation

FL1

2-15 Vdc

FL2 & FL3

8-38 Vdc*

Output

FL1

1-2 mV/V*

FL2

0-5 Vdc*

FL3

4-20 mAdc

Zero Balance

≤± 1% FSO

FSO Setting

≤± 1% FSO

Resolution

Infinite (± 0.001% FSO usable)

Response Time

FL1

< 0.5 mS

FL2 & FL3

< 3 mS

Insulation Resistance

1,000 M Ω @ 50 Vdc

Reverse Polarity

Protected

Warm-up

< 10 mS

Power Supply Effect

≤± 0.002% FSO per V input
(FL2 & FL3)

EMY/RFI

Internal Filtering (FL2 & FL3)

Short Circuit Protected

Up to 40 Vdc (FL2 & FL3)

* Options Available

Mechanical

Pressure Ranges

0-50 through 0-7,500 psi

(Customer may specify any range/eng. unit)

(Absolute, vacuum, compound are available options)

Proof Pressure

2X Full Scale*

Burst Pressure

5X Full Scale* (15k psi max)

Materials

Wetted Parts

316 & 17-4 SST*

Non-wetted Parts

316 SST plus electrical connector*

Pressure Port

¾-16 UNF Flush with O-Ring**

Electrical Connector

NEMA 4X cable exit (24” cable)*

Dimensions

Per outline below

Weight

Nominal 10 oz.

Environmental

Compensated Temp Range

0 to 170º F

Operating Temp Range

-40 to 200º F

Storage Temp Range

-40 to 250º F

Performance

Static Accuracy

≤± 0.5% FSO* (BFSL, RSS)*

(Combined effects of non-linearity, hysteresis & repeatability)

Repeatability

≤± 0.01% FSO

Temperature Effects

≤± 1.5% FSO over comp range*

(Combined effects of Zero & FSO with reference at 70º F)

Long Term Stability

≤± 0.25% FSO per year

Standard Wiring

Model

Output

+ Power

- Power

+ Signal

- Signal

FL1

mV/V dc 4 wire

Red/Pin 1/Pin A

Black/Pin 2/Pin B

Green/Pin 3/Pin C

White/Pin 4/Pin D

FL2

0-5 (10) Vdc 3 wire

Red/Pin 1/Pin A

Black/Pin 2/Pin B

Green/Pin 3/Pin C

FL3

4-20 mAdc 2 wire

Red/Pin 1/Pin A

Black/Pin 2/Pin B

**Dylix Corporation reserves the right to change specifications without prior notification. Please contact the factory for the latest revision.

Dylix Corporation is a U.S. manufacturer of pressure transmitters and transducers