The combined distillers for volatile acidity, sulphur and alcohol in wines

The VA/SO2 Kombo Glasschem stills consist of a distillation unit for volatile acidity combined with 2 or 4 units to determine the sulphur dioxide, according to the Kombo version. These instruments are certainly the ideal solution for wine industries and third-parties laboratories which must perform distillations of more parameters, simultaneously: the VA/SO2 Kombo Glasschem distillers allow the additional analysis of the alcoholic content, with an accuracy better than 0,2%, thanks to the ebulliometric technique and to the special supplied software. The new design, the auto-draining system for eliminating the exhausted wine sample at the end of the distillation for volatile acidity, the flow-rate control system of the air stream for SO2, the high efficiency of the wine heating and the absence of gas leaks in the environment get these new instruments really exclusive.

Advantages:

"All-in-one" instruments for the analysis of volatile acidity, SO2 and alcoholic content

2 available versions to satisfy all needs: Kombo 1+2 and Kombo 1+4

Appliable to all wines, mosts, grape juice and spirits

25% faster distillation time for volatile acidity than the traditional distillers

Easy separation and draining of exhausted wine samples at the end of the procedure

Several distillations of volatile acidity can be run in a row, without adding cold water

2 or 4 distillation heads for the total or free or bound sulphur dioxide

Alcoholic content analysis based on the ebulliometric method (accuracy: better than 0,2%)

Optional "Alcohol Determination Kit" for the determination of alcoholic content

High-efficiency and energy-saving electrical heating

Constant heat input: no differences from distillation to distillation

Space ergonomical design

Very compact sizes

Easy to install and use

Technical features:

Distillation time: 12 min/sample for volatile acidity, 10 minutes for free SO2 and 10 minutes for bound or total SO2, 5 minutes for alcohol distillation and 5 minutes for the boiling point determination