A major component of the Tek-CARE®300II Nurse Call System, the NC351A and NC351A/2 central equipment supply the central point for system component integration. The NC351A supports up to 4 master consoles and 256 patient/staff stations, while the NC351A/2 supports up to 8 master consoles and 512 patient/staff stations.

The NC351A and NC351A/2 house the necessary firmware and control circuits to operate all the Tek-CARE300II system functions. From a single durable metal enclosure, the central equipment supports 2 isolated serial ports, field station ports, master console ports, and power requirements for up to 256 patient, staff and duty stations (512 with external PK305A Supply).

SPECIFICATIONS

Dimensions: 15.25" × 13" × 6.25" (387 mm × 330 mm × 159 mm)

Capacity: up to 4 (NC351A) or 8 (NC351A/2) masters, up to 256 (NC351A) or 512 (NC351/2A) stations

Power Input: 115 VAC, 60Hz, 140W

Environments: 25°C; maximum 80% relative humidity

Terminations: Plug-on connectors

System Supervision: Continuous polling of all stations. Stations or wiring faults displayed at masters.

System Protection: All outputs are protected by self-resetting current-limiting devices.