Teseq CDN 163 EFT / Burst Coupling Network

Tech Support
(800) 404-2832

With Every
Rental

Worldwide
Delivery

High frequency pulse signals do result in cross-talk between conductors running in parallel even over short distances. This behavior is taken into account in the latest and currently valid edition of the burst standard EN/IEC 61000-4-4 (2004) and results in the fact that coupling of burst pulses is now only specified in pure common mode. The CDN 163 uses this coupling method.

The CDN 163 is a 690 VAC three-phase or 1000 VDC burst pulse coupling network for large EUT currents. Although designed for continuous duty at up to 100 A per phase, the unit can be subjected to considerably higher currents for short durations. As a result, high inrush currents or pulse-shaped peak currents can be handled with ease. The current limitation is largely down to a matter of thermal loading. A built-in thermometer enables the heating effect to be monitored.