Agilent 8757A-001 Specs:

Agilent HP 8757A-001. Scalar Network Analyzer, 60GHz

Scalar Network Analyzer, 60GHz

The HP 8757A features 76 dB of dynamic range (-60 dBm to +16 dBm) when used with the HP 11664 A/E detectors. With square wave modulation and detection (AC), the HP 11664 detectors enable reliable, drift-free measurements from 10 MHz to 40 GHz. With the HP 85025 and 8506 AC/DC detectors, make scalar measurements with or without modulation to 100 GHz. In DC mode (no modulation), use the HP 85025 A/B to make accurate swept-frequency measurements of power (dBm).

High directivity bridges (>40 dB) covering RF and microwave frequencies help produce excellent measurement results. Using the HP 85020 series and 85027 series directional bridges, make accurate measurements of reflection and transmission parameters simultaneously.