Transformer Oil Insulating Tester

JY6613 Transformer oil breakdown tester

JY6613 transformer oil breakdown tester adopts completely new electromagnetic compatibility concept(EMC) which designed to avoid tester crash accident during testing, and also can still work well in high magnetic field.

JY6601 transformer oil breakdown tester use advanced voltage breakdown control technology which will keep breakdown energy in a very low level,that will avoid pollution to oil sample during testing and make sure the test result are accurate and reliable.

FUNCTION INTRODUCTION1. Never crash:JY6613 adopts electromagnetism compatibility(EMC) to avoid tester crash in test process. The instrument can work well in high-intensity electromagnetic environment

2. High precision: JY6613 adopts completely new breakdown voltage control technology to make the breakdown discharge energy to the lowest,which avoid oil sample pollution during test process and make test results reliable and accurate.

3. New type oil cup: JY6613 oil cup is made by new material with high strength and long service life to avoid fragile and leakage problem during testing.

4. Complete protection system: JY6613 has a variety of protection devices which will make sure the safety of operator and tester itself in accident

like inferior oil beakdown and empty cup breakdown.

High performance of MPU (microprocessor) can provide high precise data and fast calculation.

Oil test cup made from advanced material which is high intensity with long working life.

LCD screen provide clear display for all tested value.

Thermosensitive printer provide fast and clear tested value printing.

JY6613 supply RS485 and USB interface for data transmission.

We supply a infrared temperature measuring gun with tester for free!

TECHNOLOGY SPECIFICATION:

Output voltage

0~80kV/100kV

Minimum resolution

0.1 kV

Accuracy

2％×read±0.2kV

Voltage raise speed

1.0/2.0/3.0kV／s

Breakdown disconnecting time

≤1ms

Experimental times

1～6

Volume of the oil cup

400ml: 200ml

Distance between the poles

2.5mm(Electrode gap is adjustable)

Temperature in operation

0~40℃

KRI will supply fast delivery work to make sure your can receive tester in time!