Active Humidity Module

STAGE TOP CHAMBER

The Stage Top Chambers fits in the XY stage or PIezo insert of the microscope, connects to Okolab Temperature, Gas and Humidity Controllers and creates the proper environment for live cell imaging right on the microscope stage.

TEMPERATURE CONTROLLER

H301-T-UNIT-BL-PLUS is the most advanced solution for Temperature Control. It is compatible with Digital Gas Controllers and with Active Humidity Controller to ensure best environmental control for live cell imaging.

ACTIVE HUMIDITY CONTROLLERS

Active Humidity Controllers control the Relative Humidity inside the chamber in the range 50-95% with a resolution of 1%.

Features

Connects to any Bold Line Gas controller

Controlled via OKO-TOUCH

Humidity sensor resolution 1%

A heated tube connects the Active Humidity Controller to the Chamber

Stand alone version (HM-ACTIVE-STANDALONE) is available to run the humidifier independently of the gas controller.

The dry gas is humidified by flowing over warm water. The Humidity sensor regulates water temperature in order to achieve the desired relative humidity in the chamber.

To prevent water condensation, the tube connecting the humidity controller to the chamber is heated.

Evaporation rate as a function of relative humidity inside the chamber

OKO-TOUCH

OKO-TOUCH is an intuitive and user-friendly touch screen interface. It connects to any one of the Bold Line Controllers, detects all equipment in the system and displays all functions supported by the equipment detected.

It allows to:

Turn all Bold Line Controllers on/off

Define set point values

Set flow rates gas controller

Define gas cycles of gas controller

Define temperature cycles of temperature controller

Select sample or chamber feedback mode

Select chamber model and plate adapter

Display all set points and measured parameters

Run self calibration procedures

OKO-TOUCH

Top Incubator Overview

Home Page: T, CO2 and O2 real time values

T Unit: Sample / Chamber Feedback operation mode

T Unit: Chamber selection

CO2 - O2 Unit: flow rate visualization

CO2 O2 Unit: Total output flow rate selection

Objective Heater Enable / Disable

Objective Heater Self Calibrartion

Objective Heater Self Calibrartion

SMART BOX

Smart Box connects to the Bold Line Controller and provides the following features: