Catalog no.W-6814-A2

General Purpose Flexure Fixtures

Center pivot supports for flexural testing of brittle material specimens up to 300 mm (12.0) width in accordance with ASTM C78, ASTM C293, ASTM C348, ASTM D1635, and ASTM C1609. Four supports are supplied for either 3 point or 4 point (sometimes referred to as third point) flexural testing. Upper and lower supports are mounted to a t-slot base for adjustment of upper and lower spans.

Lower support span: 76.2 mm to 920 mm (3.0 to 36.3 in)

Upper support span: 76.2 mm to 420 mm (3.0 to 16.6 in)

Support diameter - 76.2 mm (3.0 in)

Specimen width - up to 300 mm (12.0 in)

Support height - 66.7 mm (2.6 in)

Maximum Capacity of lower support fixture (applicable to 3 point flexure testing where upper span is zero):

Overall maximum: 200 kN (45,000 lbf)

If installed with overhang, capacity is rated per below

Capacity = 200kN (22,500 lbf) where x < 63 mm (2.5 in)

Capacity (kN) = 12,600/x when x > 63 mm

Capacity (lbf) = 111,900/x when x > 2.5 in

Where x = distance from end of machine loading table
to lower flexure support