Abstract

Purpose:

To test the radiation delivery robustness of the first MR-IGRT system using a commercial cylindrical diode array detector (ArcCHECK) and an ionization thimble chamber (Exradin A18).

Methods:

The MR-IGRT system is composed of three evenly spaced Co-60 sources on a rotating gantry located between two magnet halves. The collimator for each source consists of 30 doubly-focused leaf pairs that allow the system to deliver both conformal and intensity modulated (IMRT) treatment plans. The system's delivery robustness was tested over a span of 6 months from September 2013 through February 2014. This was achieved by repeatedly delivering 10 patient plans. These plans consisted of 2 conformal prostates, 2 IMRT prostates, 2 IMRT head and neck, 2 IMRT breast, 1 IMRT pancreas, and 1 IMRT bladder. The plans were generated with the system's treatment planning software. Once the plans were generated, quality assurance plans were created on a digital ArcCHECK dataset. The ArcCHECK used for testing was specially designed to be MR-compatible by moving the power supply outside of the magnetic field. The A18 ionization chamber was placed in a custom plastic plug insert in the center of the ArcCHECK. Gamma analysis was used with the ArcCHECK for relative dose evaluating both 3%/3mm and 2%/2mm. Absolute point dose was compared between ion chamber measurement and treatment plan.

Results:

The ArcCHECK passing rate remained constant over the 6 month period. The average passing rate for 3%/3mm and 2%/2mm analysis was 98.6% ± 0.7 and 88.8% ± 2.9, respectively. The ion chamber measurements showed little variation with an average percent difference between planned dose verses measured dose of 0.9% ± 0.7.

Conclusion:

Minimal differences were noted in the delivery of the 10 patient plans. Over a period that included acceptance testing, commissioning, and clinical deliveries, the MR-IGRT system remained consistent in radiation delivery.