X-ray Sources and Sync Cables

Vidisco produces special adapter cables that enable its systems to control most X-ray sources, which are on the market today. Full control of kV, mA and exposure time means having a fully synchronized solution. Vidisco's portable digital radiography systems are compatible with various constant potential X-ray sources as well as high frequency X-ray sources. Vidisco’s systems are also compatible with Isotopes (e.g Se-75, Ir-192, Co-60).

XRS-4 Pulsed X-ray Source

The newest and largest pulsed X-ray source from the house of Golden Engineering. The XRS-4 produces pulses of 370 KVP, and weighs 10 kg (22 lbs) including battery. It is powered by an 18V battery and produces a maximum of 199 pulses in 22 seconds.

XRS-3 Pulsed X-ray Source

XRS-3 produces maximum photon energy of 270KVP and canemit consecutively 99 pulses in 6.6 seconds. It weighs 5.4 Kg (approx. 12lbs) and can fit into a Vidisco system configuration casing. It is powered by a 14.4V battery and is supplied with 2 batteries and a charger. The XRS-3 dimensions are 35.6 cm (14”)X 11.5 cm (4.5”)X 19 cm (7.5”). It is compatible with Vidisco’s XR-DE Plus capabilities.

XR-200 Pulsed X-ray Source

The XR-200 produces maximum photon energy of 150KVP with a maximum duty cycle of 3000 pulses per hour (200 pulses every 4 minutes). Its dimensions are smaller than the XRS-3 (12.5” (31.75 cm) x 4.5” (11.5 cm ) x 7.5 “ (19 cm)) and it can also fit into a Vidisco portable System casing configuration and it is compatible with XR-DE Plus capability.

XR-150 Pulsed X-ray Source

The smallest member of the Golden Engineering family with dimensions of 10.5” (26.5 cm) x 3” (8 cm) x 4” (10 cm) and weighing only 2Kg (approx. 4.5lbs). It can produce 99 pulses in 9.9 seconds, with maximum photon energy of 150KVP. Its maximum duty cycle is 200 pulses counts in 8 minutes. It is powered by a 7.2 Volts Nickel-Cadmium sub C cells (6) battery and is supplied also with a charger.