UHP-T-SR LED

USA/Canada: +1 (248) 436-8085

Ultra-High Brightness LED Light Source

Introduction

Click to enlarge

The UHP-T-SR is a Ultra-High Brightness LEDs series
featuring unique LED chips. The UHP-T-SR proprietary
thermal management enables us to drive the LED die
at high current density, making the UHP-T-SR series a
top-brightness system. The UHP-T-SR
collimated LED head features built-in adjustable aspheric collimator
to provide collimated LED beam. Applications such as
fluorescence microscopy, confocal microscopy,
chemical reaction activation and numerous others can
benefit from this design.

The shielded UHP-T-SR LED head contains the high
current driver, while the LED controller box contains
control functions. This arrangement eliminating much
of RFI / EMI interference common in high current light
sources. The controller features Optically Isolated TTL and Analog Inputs. These features
make this product especially suited for electrophysiology rig applications.
UHP-T-SR can be used with a full range of other optical OptiBlocks such as Fiber Coupler,
Liquid Light Guide coupler, Beam Combiner, Filter Wheel, Beam Switcher and others See
Optional Accessories section below.

The Power Collim. is the total beam power emitted from the LED head after passing the internal collimating aspheric lens.

The Power LLG3 is total power emitted from 3 mm core liquid light-guide (Prizmatix P/N: LLG-3) connected to the LED head by suitable coupler (Prizmatix P/N: LLG-A3)
(Please see list of Accessories below - parts #9 and #10)

Benchtop LED Current Controller Specifications

Constant current or chopping modes

Precise LED current setting by 10 turn dial

TTL external trigger input

Analog input for external LED power control (0-5Vdc)

Optically isolated TTL and Analog inputs

Compact and robust enclosure

Click to enlarge

Electrical Specifications

Connectors for TTL and Analog Input

Optically isolated BNC

Digital modulation frequency

Hz

DC-30000

Rise / Fall time

μs

<5

Analog input voltage range

V

0-5

Analog modulation frequency

Hz

DC-~100

Current controller supply voltage

V

12

Power adaptor input

85-264 VAC, 47-63Hz, 1.5A

General Specifications

Operation temperature range

0C

10-35

Storage Temperature range

°C

-10-55

Operating relative humidityNon condensing

%

<90

Head dimensions

See drawing below

Head weight

g

350

Controller dimensions (L x W x H)

mm

197 x 174 x 80

Controller Weight

g

400

Power adaptor dimensions(Lx W x H)

mm

175 x 72 x 35

Power adaptor weight

g

650

Power adaptor safety

LED Head fan noise

dBA

38

Mechanical drawings

Standard UHP-T LED head:

* Specifications subject to changes without notice

Optional Accessories

Beam Combiner [2]

Multiple LED beams can be combined into one output beam. For example UV LED can be combined
with White LED to create Mercury lamp like configuration. For more details please see:
http://www.youtube.com/watch?v=iv7dlwLHaUE

Fiber Coupler Adaptor [3]

All Prizmatix LED heads can be easily changed from collimated LED source to fiber coupled LED
configuration by means of Fiber Coupler Adaptor (SMA, FC or ST connector). .
For more details please see
video clip

Liquid Light Guide Adaptor [9]

The Microscope-LED can be easily changed from direct microscope coupling to Liquid Light Gide
coupled LED configuration by means of LLGA Adaptor
For more details please see
video clip

Fiber Optics Collimator [13]

The output from optical fiber is divergent according to fiber NA. In order to reduce the divergence
angle a collimator module can be used. Prizmatix collimator was especially designed to fit thick core
high NA Polymer Optical Fibers. See more details at
http://www.prizmatix.com/optics/collimator.htm