The Fiber Meter (850/1300) – Optical Power Meter is primarily used for checking the signal output power of the optical communication equipment in fiber optical networks. It measures the average power of a continuous light beam which emits from the equipment or other optic sources. It measures power in both 850nm/1300nm (multi–mode) and 1310nm/1550nm (single–mode) respectively.

The Fiber Meter (850/1300) – Optical Power Meter consists of a solid state InGas photo diode, signal power measurement circuitry, and a 2 digit LED display. Users just connect a fiber cable between the transmission port of the equipment and the universal interface on the Fiber Meter (850/1300) – Optical Power Meter, or they can connect to other light source. The product will show the exact amount of the received power value (in–dBm). The test ranges are 30dBm to –15dBm in wavelength 850nm/1300nm and –30dBm to –6Bm in wavelength 1310nm/1550nm.

The universal connector can be used for all the most common fiber interfaces such as ST/SC/FC/FDDA and it also suitable to test both single–mode and multi–mode cables. With an optional 2.5mm to 1.25mm adapter, users can even test for 1.25 fiber cables.

The Fiber Meter (850/1300) – Optical Power Meter can operate in coordination with Hobbes’s opti source or most other fiber LD light source to be a complete fiber measurement kit.

Model No.: 257135/257136

Wavelength: 850nm/1300nm for 257135; 1310nm/1550nm for 257136

Sensor type: universal

Dynamic range: 257135: -30dBm~-15dBm, 257136: -30dBm~-6dBm

Accuracy: ±1 db

Resolution: 1dbm

Unit: dbm

Fiber connector: universal

2.5mm to 1.25mm adapter spec.: 2.5mm to 1.25mm

Operating temp.: 0ºC to 50ºC

Storage temp.: 0ºC to 70ºC

Battery: AAA 1.5V x2

Dimensions: 141 x 29.5 x 22mm

Weight: 45g

Fiber Meter (850/1300)Optical Power Meter

The Fiber Meter (850/1300) – Optical Power Meter consists of a solid state InGas photo diode, signal power measurement circuitry, and a 2 digit LED display. Users just connect a fiber cable between the transmission port of the equipment and the universal interface on the Fiber Meter (850/1300) – Optical Power Meter, or they can connect to other light source. The product will show the exact amount of the received power value (in–dBm). The test ranges are 30dBm to –15dBm in wavelength 850nm/1300nm and –30dBm to –6Bm in wavelength 1310nm/1550nm.