Specification:
Load capacity: Various different capacities can be offered up to 5000 kg.
No. of load cell: Optional One or Two.
Cross Travel: up to 1000 mm
Facility of full scale elongation as well as extensometer gauge length elongation (OPTIONAL)
Horizontal day light: 450mm
Speed: Variable between 1.33mm/minute to 500mm/minute.
Paint: Powder coated.
Power: 230 Volts, 50Hz, single phase.

THIS SYSTEM HAS A MULTI-STEP SPEED FACILITY WHERE BY YOU CAN SET THREE VARIABLE SETTABLE SPEEDS FORWARD TEST SPEED, REVERSE FAST SPEED AND REVERSE SLOW APPROACH SPEED.

GRAPHIC DISPLAY AND PRINTOUT THROUGH PC:
In addition to the above reports, further values like Young’s modulus, Yield stress, Yield point, Flexural stress, Flexural modulus, Standard deviation, Intermediate result and Load Vs Elongation graph could
be obtained by attaching a printer to the PC.

THROUGH THE PC THIS WILL BE A MULTI UNIT DISPLAY AS WELL AS PRINTOUT WHERE BY THE OPERATIONAL UNITS WOULD BE KG, CM, POUND, INCH, NEWTON & MM FOR CONVERSIONS IN ANY COMBINATION.