Comment AG4108 SNF 2009-01-23 07:39:02: Field Service 1/26/09

Field service found no cooling problems on the AG4108, but did find that several of the lamps were not properly seated in their holders. When there is time, another qual should be run to see if this addresses the non-univormity issues following the rebuild.
Note: a pryometer calibration was also performed at this time.