It mainly applies in treating the patient who suffers . We should adopt acute hemoferfusion if the patient takes too much drug , after , infusion, , forced , but the condition grows worse, especially accompanied with brain function barrier or stupor.

Adopting ascites reinfusion to reduce the losing of albumen, if the patient suffers hepatocirrhosis with ascites.

Main Function:

Hemoperfusion , reinfusion of ascites(hepatocirrhosis)

Technical parameter:

Power supply: AC 220V 50Hz

Power input: ≤60W

Ambient Temperature: 50 --40(on working ); -40--70(on storage )

Relative humidity: ≤80%

Working noise: ≤60dB Alarm: ≥62dB

Venous pressure inspection: 0KPa--40Kpa(0-300mmHg)

Indication precision:&61617;1.3Kpa(&61617;10mmHg)

Adjustable Range of vennous pressure:1.3KPa-40Kpa(10-300mmHg)

Adjustable range of blood flow:0--400ml/min

Allowable error of the blood pump: &61617;1%

Bubble monitor method : ultrasonic theory, when the fluid drop to the guard line, there will be a sound and light alarm