Please choose how you would like to display pricing for the store. If the shipping address for an order is outside of the USA, Canada, Mexico or territories of the USA, pricing for that order is marked up by 20% of list price. This markup covers the expenses of an International Warranty, preparation of export paperwork and additional processing time.

Note that all orders are reviewed by our sales staff and pricing will be corrected if it does not align with our policies.You may change this setting at any time by clicking on the Pricing button on the top right of your screen.

The ESU-2300 is an Electrosurgical Analyzer. The “measurement engine” of the ESU-2300 is a specially-engineered true-RMS RF ammeter and wattmeter, designed to measure the complex RF waveforms produced by electrosurgical generators. The ESU-2300 has been designed to comply with electrosurgical manufacturer industry standard current sensing technology and utilizes an internal precision RF Current Transformer. The analyzer utilizes a precision internal test load resistor network. The ESU-2300 is intended to be routinely used to measure the various parameters relating to the testing and performance validation of electrosurgical generators. DFA® Technology allows the ESU-2300 to aggressively digitize the complex electrosurgical generator waveform, analyze its components and provide accurate and stable results. It may be used as a stand-alone meter or in conjunction with a PC utilizing the optional software.