The miniDOT® logger is a completely submersible instrument that logs dissolved oxygen and temperature measurements. The oxygen sensor is an optode that measures dissolved oxygen concentration in water through a fluorescence method. Data are recorded to an internal SD card. Operation of the miniDOT® logger such as setting the time and sample interval can be accomplished via the USB cable.

Commonly Used for Measuring

Device Applications

More about the miniDOT® Logger

User replaceable batteries

100 Meter depth rating

Shipped calibrated

Portable and easy to use

The miniDOT logger measures dissolved oxygen in water and is completely self contained. It includes an optical DO sensor, a temperature sensor, batteries and a micro SD memory card. The logger collects measurements of dissolved oxygen with an accuracy of +/- 5% and temperature to +/- 0.1 degrees C. All measurements are stored internally. Data can be offloaded to a computer via the miniDOT USB cable. Temperature range is 0 to 35 degrees C with an accuracy of +/- 0.1 degrees C. Oxygen range is 0 to 150% saturation with an accuracy of +/- 10 umole/L.

PME also has an anti-fouling wiper compatible with the miniDOT logger. Increase your data longevity by using the PME wiper. More about the miniWIPER.

Product Design

The miniDOT logger is constructed from a delrin plastic, which is a hard material that does not easily crack or break. All internal sensor components have been specifically designed to fit on the logger spine. The black sensor cover allows the user to easily grip and unscrew the top.

Logger Information & Sensor Specifications

The miniDOT sensor is an optode that measures lifetime-based luminescence quenching of a thin membrane. The sensing foil contains a coating that has a variable fluorescence that depends upon oxygen concentration in the surrounding water. The sensor is able to compute oxygen concentration in mg/L and record these measurements internally. Every miniDOT logger is calibrated at PME headquarters. PME calibrates each logger at 12 oxygen concentrations each at 8 temperatures for a total of 96 calibration points. The miniDOT logger can sample any where between once every minute to once every hour. Contact PME for sample intervals outside this range.

Sample Interval Minutes

Days of Sampling

Number of Samples

1

365

500K

10

365 +

52,000 +

60

365 +

8,000 +

Product Case Studies

Device Software

Software is provided with every miniDOT logger and can be found on the micro SD card once connected to a computer. The software displays the miniDOT logger files and creates visual plots to easily read oxygen measurements. The software also allows the user to set the internal clock and determine a sample rate.