The purpose of this Calibration and Measurement Guide is to introduce the basic calibration and operation of the VectorStar MS464xB Vector Network Analyzer (VNA) System and reduce the time required to become
proficient at performing basic measurements. The procedures in this manual assume a working knowledge of, and a familiarity with, vector network analyzers. P/N: 10410-00318

This manual provides a reference for the VectorStar Series VNA User Interface (UI) menus and dialog boxes. In addition, this manual provides the context-sensitive portion of the instrument Help system. P/N: 10410-00319

This manual provides information for remote operation of the VectorStar MS4640B Series VNAs using commands sent from an external controller via the IEEE 488 General Purpose Interface Bus (GPIB), USB, and Ethernet. P/N: 10410-00322

The MU100040A CPRI RF module for the MT1000A Network Master Pro adds CPRI RF measurements to Anritsu's transport and fiber test platform. The modular design of the MT1000A means that it can be configured just for CPRI measurements or combined with the 10G transport module and OTDR module. P/N: 11410-00944

This simple guide is intended for test engineers with little experience in 5G base station Tx testing and for operators requiring efficient measurement; it explains an easy procedure using the MS2850A/MS269xA to measure the EVM of 5G NR downlink Test Model signals.

These compact handheld Spectrum Analyzers are designed to meet the needs for portability. Whether it is for spectrum monitoring, broadcast proofing, interference analysis, RF and microwave measurements, or Wi-Fi and
wireless network measurements, the Spectrum Master is the ideal instrument for making fast and reliable measurements. P/N: 11410-00511

The Site Master S331E and S361E is a handheld cable and antenna analyzer designed to make
Return Loss, VSWR, Cable Loss, and Distance-T
o-Fault (DTF) measurements in the field.
The S332E and S362E is an integrated multi-functional test instrument that eliminates the need to carry and learn multiple test sets. In addition to the cable and antenna measurements, the S332E and S362E can be configured to include a Spectrum Analyzer, 2-port Transmission Measurement, Interference Analyzer with Interference Mapping capabilities, Coverage Mapping, Channel Scanner, CW Signal Generator, AM/FM/PM Analyzer, PIM Analyzer, Power Meter, and High Accuracy Power Meter. P/N: 11410-00484

The Site Master S331L is a one-port, 2MHz to 4GHz hand held cable and antenna analyzer designed to make Return Loss, VSWR, Cable
Loss, Distance-To-Fault (DTF), and power measurements in the field. P/N: 11410-00616

This version of software (and any other software version 2.1.0 and later, including those of format V201x.x.x) requires installation of Bittware 8.33 on the VNA prior to installation of the VectorStar application. Installation of Bittware is only needed once and does not need to be reinstalled at every
VectorStar application installation. Please refer to Appendix A in this document for details about Bittware versions and installation of this driver.

With the OSL calibration, you have to calibrate each time you change the frequency range. With the FlexCal, you can change the frequency range and you don't have to perform another calibration. FlexCal is great for troubleshooting purposes. The accuracy can be affected if your span is really narrow so for optimum accuracy, it is recommended to use the OSL calibration.

The Fast CW is an ideal mode of operation to attain rapid data over GPIB. To achieve a fast measurement rate, the display is not updated and only the raw S-parameter or user-defined parameter of the active channel is measured.