Fluid Components International

FS10A Analyzer Flow Switch/Monitor

FS10A Analyzer Flow Switch/Monitor

Water and wastewater operationsand plant engineers in search ofa flow switch/monitor to ensure flow to their water or gas sampling systems or analyzers will find their solution in the Model FS10A Analyzer Flow Switch/Monitor fromFluid Components International (FCI). No matter how sophisticated a fluid analyzer system may be, it will be ineffective if a sample flow fails to reach the analyzer sensor. The most advanced systems in the world cannot provide accurate results without a valid fluid sample. That’s why FCI developed the advanced FS10 Analyzer Flow/Switch Monitor.

The FS10A Flow Switch/Monitor is designed for use with tubing from 1/8 inch to ½ inch and the electronics may be either integrally mounted or remote mounted from the sensor element. It is extremely sensitive to low flow rate detection, there are no moving parts to foul or clog, and there are no cavities, orifices, or dead-legs to trap or contaminate samples.

FS10A is suitable for application in liquid or gas analyzer sampling systems. It is small, lightweight, and features a choice of electronic outputs including open collector, relay, and

4-20mA outputs. Limit alarms are user programmable for time delay and/or hysteresis. An on-board 10-LED array provides at-a-glance visual indication to the user of flow rate and whether the alarm limit has been tripped. All FS10A set-up parameters can be performed via its two front panel push buttons or via a PC over its RS232C I/O. Its exclusive “learn” mode facilitates easy user set-up of flow range zero and span settings.

Water and wastewater end-users and system integrators will find FCI’s FS10A provides an overall solution for sampling system and analyzer flow assurance. With its advanced electronics and reliable thermal dispersion flow sensing technology, it will ensure the highest integrity process analysis without interruption.