OverviewThe BA75 has a maximum output of 75kV, more than
sufficient to meet almost all testing standards
worldwide. This rugged, portable instrument is the
smallest and lightest unit on the market with an
integrated battery and carrying handle that makes it
ideally suited for the field environment.

With a
color, ultra-bright sunlight viewable Liquid Crystal
Display (LCD), this unit is equally at home in the
testing laboratory as is in the field. The intuitive,
menu driven display, with built-in international testing
sequences, allows the user to simply start a test cycle
and the unit takes over and completes the test. Results
are sent to an integrated printer for immediate
hardcopy. In addition a PDF file can be created directly
by the unit and stored onto a standard memory stick.

The
supplied BA Control center software allows for the
control of the instrument and for downloading of the
test results via wireless link to a PC.

Features

Output Voltage: 0 to 75kVrms.

Easy to use, intuitive, menu driven display.

Ultra bright Liquid Crystal Display (LCD).

Scope display of voltage breakdown.

Direct, precise voltage and current measurement.

Ultrafast, low energy switch-off (<5Ás).

Viewable, screened HV test
chamber.

Automatic PDF generation of
results.

Integrated printer and
temperature measurement.

Compatible with mineral,
silicon, and ester based oils.

Pre-installed fully automatic standards.

User-programmable test sequences.

No nuisance error message.

Adjustable gap spacing mechanism on optional
test cells.

Moisture impervious test cell
construction.

Low volume test cell available where applicable.

EMC shielding.

Multiple safety interlocks.

ATA approved transport / cases and test cells
available _see options.

Application
Electrical insulation oils often form the critical
insulation medium for many types of electrical
apparatus, such as transformers, circuit breakers, taps
changers, cables, medical apparatus, bucket trucks etc.
When this liquid insulation becomes compromised due to
aging and or contamination, the electrical insulation
properties can be negatively affected causing a
potential insulation flashover, often resulting in
catastrophic consequences.