Laboratorium

TR 240

TR 60 with observation window

TR 450

Electrical rotating device as additional equipment

TR 1050 with double door

TR 60 - TR 1050

With their maximum working temperature of up to 300 °C and forced air circulation, the ovens achieve a perfect temperature uniformity which is much better than in ovens of most competitors. They can be used for various applications such as e.g. drying, sterilizing or warm storing. Ample warehousing of standard models provides for short delivery times.

These chamber furnaces with air circulation are characterized by their extremely high temperature uniformity. Hence, they are especially suitable for processes such as cooling, crystalizing, preheating, curing, but also for numerous processes in tool making. Due to the modular concept, the air circulation furnaces can be adjusted to the process requirements by adding suitable equipment.

Tmax 450 °C, 650 °C, or 850 °C

Horizontal air circulation

Swing door hinged on the right

Temperature uniformity up to +/- 5 °C according to DIN 17052-1 (model N 15/65 HA up to +/- 7 °C)

Optimum air flow and temperature uniformity through high circulation rates

One shelf and rails for two additional shelves included (N 15/65 HA without removable tray)

Stainless steel air-baffles in the furnace for optimum air circulation

Base frame included in the delivery, N 15/65 HA designed as table-top model

Air inlet and exhaust air flaps as additional equipment for using as drying oven

Defined application within the constraints of the operating instructions

Model

Tmax

Inner dimensions in mm

Volume

Outer dimensions in mm

Connected

Electrical

Weight

°C

w

d

h

in l

W

D

H

load kW

connection*

in kg

NA 30/45

450

290

420

260

30

1040

1290

1385

3.6

1-phase

195

NA 60/45

450

350

500

350

60

1100

1370

1475

6.6

3-phase

240

NA 120/45

450

450

600

450

120

1200

1470

1575

9.8

3-phase

310

NA 250/45

450

600

750

600

250

1350

1650

1725

12.8

3-phase

610

NA 500/45

450

750

1000

750

500

1500

1850

1800

18.8

3-phase

1030

N 15/65 HA1

650

295

340

170

15

470

845

460

2.7

1-phase

55

N 30/65 HA

650

290

420

260

30

607 + 255

1175

1315

6.0

3-phase2

195

N 60/65 HA

650

350

500

350

60

667 + 255

1250

1400

9.6

3-phase

240

N 120/65 HA

650

450

600

450

120

767 + 255

1350

1500

13.6

3-phase

310

N 250/65 HA

650

600

750

600

250

1002 + 255

1636

1860

21.0

3-phase

610

N 500/65 HA

650

750

1000

750

500

1152 + 255

1886

2010

31.0

3-phase

1030

N 30/85 HA

850

290

420

260

30

607 + 255

1175

1315

6.0

3-phase2

195

N 60/85 HA

850

350

500

350

60

667 + 255

1250

1400

9.6

3-phase

240

N 120/85 HA

850

450

600

450

120

767 + 255

1350

1500

13.6

3-phase

310

N 250/85 HA

850

600

750

600

250

1002 + 255

1636

1860

21.0

3-phase

610

N 500/85 HA

850

750

1000

750

500

1152 + 255

1886

2010

31.0

3-phase

1030

1Table-top model*Please see page 60 for more information about supply voltage2Heating only beetween two phases