Overview

The Mod. N8031 provides 8 independent 100 V/ 10 mA channels in a double width NIM mechanics. Each channel can provide a 100V max voltage and a 10 mA max current. The unit is available with positive, negative and “mixed” (4 positive and 4 negative) channels.

Channels outputs are delivered through BNC connectors. The HV output RAMP-UP and RAMP-DOWN rates may be selected independently for each channel in the range 1÷50 V/s in 1 V/s steps. The module features 50 nA Iset resolution.

Module control can take place either locally thanks to a 2.8″ Touchscreen Graphic color LCD display with a completely redesigned user interface or remotely, via USB 2.0 or Ethernet.

A complete set of Software Tools is available to control these units; the user can freely download low level libraries, LabVIEW driver and Graphical application software.

Safety features include:

OVERVOLTAGE and UNDERVOLTAGE warning when the output voltage differs from the programmed value by more than 2% of set value (minimum 10V).

Programmable VMAX protection limit

OVERCURRENT detection: if a channel tries to draw a current larger than its programmed limit, it enters TRIP status, keeping the maximum allowed value for a programmable time (TRIP), before being switched off

Call

Support

Sales network

This site uses own technical and third parties cookies. By closing this banner or continuing the navigation, you accept the use of cookies. If you want to know more or opt-out of some cookies see our cookies policy. Accept