Biological Safety Cabinets

Biological Safety Cabinets are safety equipments to be used for handling Hazardous Pathogens. These cabinets, apart from creating a Clean & Sterile environment, to protect the product/sample also protects the Lab personnel and the Atmosphere.

The vertical downward airflow protects the product, the inflow along the front edge protects the user and the HEPA filtered exhaust air protects the atmosphere. We are looking for inqueries from South India & Overseas Countries.

Pathogens have been classified into four risk levels from 1 to 4. Based on the level of risk of the pathogen and the procedure, NSF-49 of USA has classified the Biosafety cabinets into three Classes viz. Class I, Class II & Class III.

BioMARK-II-B2 is a Class II Type B2 Biohazard Safety Cabinet or Total Exhaust Cabinet. These cabinets provide personnel, product and environment protection. Type B1 cabinets have 0% re-circulation arrangement and 100% exhaust through HEPA filter. Vertical downward airflow to the work zone creates parallel streams of ultra-clean air that carries with it all the contaminants leaving the work area at a high degree of cleanliness of the order of Class 100.

Standard Features :

The filtration arrangement for this Type B1 cabinet is designed as per NSF Standard 49. All contaminated plenums are surrounded by negative pressure, virtually eliminating leakage from the cabinet. Provides maximum protection. Blowers are fully balanced and are made for continuous operation. Motor capacity is optimized to deliver the required air with low noise.The cabinet work zone has minimum welded joints to avoid collection of contaminants or rusting Cabinet work area is made of fully SS304 construction with minimum joints.Maximum front sash opening is 8" which can be increased for introducing larger instruments or samples and for cleaning. An alarm is provided to prevent opening of front door beyond 8" UV germicidal lamp (15W or 30W) is provided for preliminary sterilization.. Built-in white, electronically ballasted, fluorescent lighting provides excellent illumination in the work zone and reduces operator fatigue. Powder coated GI construction in 2 colour combination provides a pleasing look and enhances the aesthetic appeal of the equipment, Fully SS304 construction is also available. Exhaust Module with HEPA filter (has to be located near the Cabinet above the false ceiling or outdoors on sunshade etc. with access, for service.) Every Cabinet is tested in the factory for HEPA filter integrity, Cleanliness and Air flow. Additionally these are repeated at the Site again as per requirement. All Test instruments are Calibrated with traceability to NABL or International Standards. The cabinets is warranted for 12months excluding consumable parts and accessories. Additional IQ/OQ documentation is available upon request.

Note 1: When air is exhaust from the Biosafety Cabinet, not only pathogens but air conditioned air is also thrown out. To compensate for the air lost, air has to be introduced into the room using Positive Pressure / Fresh Air Modules, to avoid leakage into the room through door opeingins / gaps. If fresh air is not fed from an air conditioned room, the lab is heated up to cool this additional Air conditioners have to be installed. As a thumb rule for every 200cfm of fresh air added one ton of AC should be added. For eg. for a 3' Type B2 cabinet, minimum of 2TR AC should be added.

Note 2: For a 3ft Biosafety Cabinet the exhaust air quantity is 450cfm, to heat this air even to to 80degC, 15kW of heater is required, which is too high considering power situation in India. To avoid this we have developed an alternative system that is being used in USA and Europe, which is an additional stage of HEPA filtration in exhaust. An additional HEPA filter Module with a blower and HEPA filter will be connected to the exhaust system. It may be noted that all the HEPA filters in the Biosafety cabinet have to be tested for Integrity using Aerosol Photometer after Installation at Site to ensure safety of the environment.

Yes! I am Interested

View PDF

Request CallBack

Get Best Quote

Biological Safety Cabinets

Approx. Price: Rs 80,000 / SetGet Latest Price

Product Details:

Minimum Order Quantity

1 Set

Cabinet Material

Stainless Steel, Mild Steel

Main Filter

HEPA Filter

Sterilization

UV Germicidal Tube

Electric Supply

Single Phase,230V (less than 1kW)

Noise Level

Less than 65 db

Brand

bioMARK-Advanced

Size

91cm X 62cm X 66cm

Working area size

3ft,4ft & 6ft

Open-fronted equipment with atmost protection for User, Product and the Environment. Ideal for handling Pathogens. HEPA filters / ULPA Filters Tested at factory and onsite* See all variants and choose the most appropriate for your application and risk level. Popular Reliable model, 100s of installations in India.Warranty and AMC available.

Bio-Safety Cabinets Class II Type A2

Approx. Price: Rs 72,000 / Piece(s) OnwardsGet Latest Price

bioMARK-II-A2 is a Class II Type A2 or A/B3 Biohazard Safety Cabinet. These cabinets provide personnel, product and environment protection. Type A2 cabinets have 70% re-circulation arrangement and 30% exhaust, both through HEPA filters. Vertical downward airflow to the work zone creates parallel streams of ultra-clean air that carries with it all the contaminants leaving the work area at a high degree of cleanliness of the order of Class 100.

Standard Features :

The filtration arrangement for this Type A2 cabinet is designed as per NSF Standard 49. All contaminated plenums are surrounded by negative pressure, virtually eliminating leakage from the cabinet. Provides maximum protection. Blowers are fully balanced and are made for continuous operation. Motor capacity is optimized to deliver the required air with low noise.The cabinet work zone has no welded joints to avoid collection of contaminants or rusting Cabinet work area is made of fully SS304 construction with minimum Maximum front sash opening is 8" which can be increased for introducing larger instruments or samples and for cleaning. An alarm is provided to prevent opening of front door beyond 8" UV light timer is provided to know the usage of UV lamp. Built-in warm white, electronically ballasted, fluorescent lighting provides excellent illumination in the work zone and reduces operator fatigue. Powder coated GI construction in 2 colour combination provides a pleasing look and enhances the aesthetic appeal of the equipment. Every Cabinet is tested in the factory for HEPA filter integrity, Cleanliness and Air flow. These are repeated at the Site again on request. All Test instruments are Calibrated with traceability to NABL or International Standards. The cabinets is warranted for 12months excluding consumable parts and accessories. Additional IQ/OQ documentation is available upon request.

Yes! I am Interested

Save time! Get the Best Deal

*Your information is safe with us.

Request CallBack

Get Best Quote

Bio-Safety Cabinets Class II Type A1

Approx. Price: Rs 72,000 / Piece(s) OnwardsGet Latest Price

bioMARK-II-A2 is a Class II Type A2 or A/B3 Biohazard Safety Cabinet. These cabinets provide personnel, product and environment protection. Type A2 cabinets have 70% re-circulation arrangement and 30% exhaust, both through HEPA filters. Vertical downward airflow to the work zone creates parallel streams of ultra-clean air that carries with it all the contaminants leaving the work area at a high degree of cleanliness of the order of Class 100.

Standard Features :

The filtration arrangement for this Type A2 cabinet is designed as per NSF Standard 49. All contaminated plenums are surrounded by negative pressure, virtually eliminating leakage from the cabinet. Provides maximum protection. Blowers are fully balanced and are made for continuous operation. Motor capacity is optimized to deliver the required air with low noise.The cabinet work zone has no welded joints to avoid collection of contaminants or rusting Cabinet work area is made of fully SS304 construction with minimum Maximum front sash opening is 8" which can be increased for introducing larger instruments or samples and for cleaning. An alarm is provided to prevent opening of front door beyond 8" UV light timer is provided to know the usage of UV lamp. Built-in warm white, electronically ballasted, fluorescent lighting provides excellent illumination in the work zone and reduces operator fatigue. Powder coated GI construction in 2 colour combination provides a pleasing look and enhances the aesthetic appeal of the equipment. Every Cabinet is tested in the factory for HEPA filter integrity, Cleanliness and Air flow. These are repeated at the Site again on request. All Test instruments are Calibrated with traceability to NABL or International Standards. The cabinets is warranted for 12months excluding consumable parts and accessories. Additional IQ/OQ documentation is available upon request.

Yes! I am Interested

Request CallBack

Get Best Quote

Class III Bio-Safety Cabinets

Approx. Price: Rs 1 Lakh / Piece(s) OnwardsGet Latest Price

Recommended for Product, Personnel & Environment protection only Air Balancing: 100% exhaust with incineration & working zone under negative pressure. As the power consumption for incineration is high (around 10kW), we suggest 2-stage exhaust filtration through HEPA/ULPA.

bioMARK-III, Class III Biological Safety Cabinet is designed for work with Microbiological agents assigned to biosafety level 4, and provides maximum protection to the environment and the worker. Access for passage of materials into the cabinet is through a double-door pass-through box that can be decontaminated between uses. Both supply and exhaust air are HEPA filtered. The exhaust air passes through two stage HEPA Filters, or a HEPA filtered and an air incinerator, before discharge to the outdoors. Airflow is maintained by a dedicated independent exhaust blower outside the cabinet, which keeps the cabinet safe, i.e. under negative pressure.Arm-length, heavy-duty rubber gloves are attached in a gas-tight manner to ports in the cabinet and allow for manipulation of the materials isolated inside. Although these gloves restrict movement, they prevent the users direct contact with the hazardous materials.

Standard Features :

Exhaust air is double-filtered through high-quality HEPA/ULPA filters (per IEST-RP-CC-001.3) with typical efficiency of >99.999% for 0.1 to 0.3 micron particles

Available in GI powdercoated or SS304 construction with leak free welded-sealed joints.

An angled cabinet front ensures an ergonomic working posture.

Cabinet operates at negative pressure relative to the laboratory in order to prevent migration of pathogenic materials out of the work area.

All cabinets individually tested for important paramenters as per NSF-49 with capability to Calibrate on Site after installation or periodically.

Calibrated Test Instruments used with traceability to NABL or (Inter)National Standards.

Yes! I am Interested

Request CallBack

Get Best Quote

Bio Safety Cabinets

Approx. Price: Rs 72,000 / Piece(s) OnwardsGet Latest Price

Our Biological Safety Cabinets, bioMARK are made for handling Pathogens falling under risk level 1 to 4 and offer protection to the person, product and the environment. The cabinets are Fabricated and tested as per the guidelines of NSF Standard 49. These cabinets are stringently checked as per NSF49 requirements for design, construction and testing requirements.

The different types of Bio Safety Cabinets made by us are:

• Class I cabinet- offers protection only to the user • ClassII & III cabinets- offer protection to the user, product and the environment • Class II Biosafety Cabinets are further classified as Type A, B1, B2 & B3 • TypeA/B3 cabinets exhaust 30% of the air handled and recirculates 70% of the air. Type A exhausts air into the room itself, whereas Type B3 exhausts air into the atmosphere. TypeA/B3 cabinets are suitable for work with low to moderate risk biological agents in the absence of volatile toxic chemicals and volatile radionuclides. • Type B1 cabinets exhaust 70% and recirculate 30% of the air. TypeB1 cabinets are suitable for work with agents treated with no more than a few micrograms of toxic chemicals and trace amounts of radionuclides required as an adjunct to microbiological studies. • Type B2 cabinets exhaust the entire air handled in the cabinet. Type B2 cabinets may be used with biological agents treated with toxic chemicals and radionuclides required as an adjunct to microbiological studies • Class III Cabinet is defined as a totally enclosed ventilated unit of gas tight construction with glove ports for access. The workspace is maintained at -ve pressure of >0.5" water gauge. Class III Cabinets are suitable for work with agents that require biosafety level 1, 2, 3 or 4 containment

Bio-Safety Cabinets Class II Type B1

Approx. Price: Rs 80,000 / Piece(s) OnwardsGet Latest Price

bioMARK-II-B1 is a Class II Type B1 Biohazard Safety Cabinet. These cabinets provide personnel, product and environment protection. Type B1 cabinets have 30% re-circulation arrangement and 70% exhaust, both through HEPA filters. Vertical downward airflow to the work zone creates parallel streams of ultra-clean air that carries with it all the contaminants leaving the work area at a high degree of cleanliness of the order of Class 100.

Standard Features :

The filtration arrangement for this Type B1 cabinet is designed as per NSF Standard 49. All contaminated plenums are surrounded by negative pressure, virtually eliminating leakage from the cabinet. Provides maximum protection. Blowers are fully balanced and are made for continuous operation. Motor capacity is optimized to deliver the required air with low noise.The cabinet work zone has no welded joints to avoid collection of contaminants or rusting Cabinet work area is made of fully SS304 construction with minimum Maximum front sash opening is 8" which can be increased for introducing larger instruments or samples and for cleaning. An alarm is provided to prevent opening of front door beyond 8" UV light timer is provided to know the usage of UV lamp. Built-in warm white, electronically ballasted, fluorescent lighting provides excellent illumination in the work zone and reduces operator fatigue. Powder coated GI construction in 2 colour combination provides a pleasing look and enhances the aesthetic appeal of the equipment. Every Cabinet is tested in the factory for HEPA filter integrity, Cleanliness and Air flow. These are repeated at the Site again on request. All Test instruments are Calibrated with traceability to NABL or International Standards. The cabinets is warranted for 12months excluding consumable parts and accessories. Additional IQ/OQ documentation is available upon request.

Yes! I am Interested

Request CallBack

Ask For Price

Class I Bio-Safety Cabinets

Product Price : Get Latest Price

Mark Air Class I biohazard safety cabinet is maintained under negative pressure, ventilated cabinet usually operated with an open front. All of the air from the cabinet is exhausted through a HEPA either into the laboratory or to the out side. The Class I BSC is designed for general microbiological research with low-and moderate-risk agents, and is useful for containment of mixers, blenders, and other equipment. These cabinets are not appropriate for handling research materials that are vulnerable to airborne contamination, since the inward flow of unfiltered air from the laboratory can carry microbial contaminants into the cabinet.

Yes! I am Interested

Request CallBack

Ask For Price

Virus Burn Out Unit

Product Price : Get Latest Price

Fabricated by using IS 304 stainless steel, equipped with high resistance electric heater with EPS / glass wool insulation, automatic temperature controlling thermostat to raise the exhaust air temperature to 80-degree Celsius (Adjustable), to incinerate / destroy hazardous pathogens in the exhaust air to protect the environment.

Note 1: When air is exhaust from the Biosafety Cabinet, not only pathogens but air conditioned air is also thrown out. To compensate for the air lost, air has to be introduced into the room using Positive Pressure / Fresh Air Modules, to avoid leakage into the room through door opeingins / gaps. If fresh air is not fed from an air conditioned room, the lab is heated up to cool this additional Air conditioners have to be installed. As a thumb rule for every 200cfm of fresh air added one ton of AC should be added. For eg. for a 3' Type B2 cabinet, minimum of 2TR AC should be added.

Note 2: For the 3ft Biosafety Cabinet the exhaust air quantity is 450cfm, to heat this air to 80degC, 15kW of heater is required, which is too high considering power situation in India. To avoid this we have developed an alternative system that is being used in USA and Europe, which is an additional stage of HEPA filtration in exhaust. An additional HEPA filter Module with a blower and HEPA filter will be connected to the exhaust system. It may be noted that all the HEPA filters in the Biosafety cabinet have to be tested for Integrity using Aerosol Photometer after Installation at Site to ensure safety of the environment.