The Cope-Elga Vision Deioniser provides quality water as and when it is required. The need for elaborate and costly storage is eliminated. Just turn on the mains water tap and take off the water instantly either through the top dispensing spout or the accessory hand held dispensing gun. The resin changes colour when the cartridge is exhausted.

Model

Vision 125

Max. flow rate

l/h

60

Max. input pressure

mm

90 (6 bar)

Mains water connection

0.5" BSP hosetail

Typical water quality

µS/cm

0.1 to 10

Typical output, water areas

Hard (400-500ppm)

litres

80 to 100

Mid (100-300ppm)

litres

150 to 400

Soft (40-80ppm)

litres

500 to 900

Overall dimensions H x W x D

mm

385 x 92 x 100

Weight

kg

2.5

With colour change cartridge and dispensing spout.

Ultra-pure water, as produced by the Cope-Elga Vision, is almost entirely free of dissolved mineral salts, including silica, and will have a neutral pH since the dissolved CO2 will have been removed. From an inorganic standpoint, ultra- pure water will conform to International Pharmacopoeia and laboratory grade specifications for purified water. For less critical applications the Cope-Elga Vision is capable of producing pure water at the end of each cartridge run, extending its life by approximately 25%. Pure water is similar to ultra-pure water although CO2 and silica will be present.

Ultra-Pure Water

Pure Water

Conductivity

1 to 0.1 µS/cm

10 to 1 µS/cm

Resistivity

1 to 10Ω -cm

0.1 to 1Ω -cm

Silica

<0.05 ppm (mg/l)

not removed

Carbon dioxide

<0.5 ppm (mg/l)

not removed

Trace dissolved metals

<0.001 ppm (mg/l)

<0.005 ppm (mg/l)

Residual Solids

<0.5 ppm (mg/l)

<5 ppm (mg/l)

Average pH

Neutral

4 to 7

Subscribe To Our Newsletter

By subscribing to our newsletter, you’ll receive promotions, offers and discounts from The Lab Warehouse. We’ll never use your data for anything else. To find out more, click here.