HL-35H Recirculating Coolers (Low Temp. Models)

HL-35H Recirculating Coolers (Low Temp. Models)

SKU#:

8194-10-0724

Powerful recirculating coolers, the HL Models provide a wide range of cooling capacities with temperature reliability. Innovative control system, with a dynamic pumping capacity, and ideal for external temp. applications. Broad range of models to choose from based on working temperature range.

For more information or to place an offline order call 1-800-745-2710

OR

Description

Operating features

Temperature range is from -20 ~40 .
Bright LED water level indicator can be easily identified.
Low fluid level and dry-running protection when triggered shows
an error symbol on the display and an
audible sound.
If the equipment operates at 40 or above due to the external
environment, user can simply identify the
error with a visual or an
audible sound.
Designed to simplify operation with a touch screen type display.
- Easy-to-read VFD display with interactive touch keys.
- Setting and resolution indication 0.1 / .
- Quick keypad lock prevents accidental parameter changes.
- Signal indicator for operation status.

Constructional features

The refrigerating system is designed to reduce energy consumption
and to maximize in refrigerating
efficiency.
Eco-friendly R-404A / R507 refrigerant is used to prevent environmental
pollution. Double service valve is
used to prevent
refrigerant leakage.
Removable gridded vent for reducing condenser contamination
and to facilitate cleanliness.
A wide ( 120mm / 4.72 ) solution inlet is for easy and safe pouring of
solution into the recirculating cooler.
Pressure gauge comes standard for observing the pump pressure
The refrigeration system can be observed easily to diagnostic at a
glance by high / low pressure gauges.
(HL-25H, 35H, 45H, 55H models)
Wheels allow for easy mobility.
The beautifully designed and compact.
Different types of pumps come standard with the recirculating
cooler to provide versatility.
Warning alarm/monitoring system.
- Over temperature.
- Lack of bath solutions.
Variety of accessories is available to meet user s different lab
environment needs. (Refer to accessories)

Specification

Model

HL
-05

HL
-10

HL
-15

HL
-20

HL
-15H

HL
-20H

HL
-25H

HL
-35H

HL
-45H2)

HL
-55H2)

Temperature 1)
(bath fluid :
Ethtylene glycol mix
1:1 Water)

Working Temperature
range
( / )

-20 to 40 / -4 to 104

Temperature stability
at 15 , ( / )

1 / 1.8

Cooling capacity
(bath fluid : Ethtylene
glycol mix 1:1 Water)

at 20 (kw)

0.60

0.07

1.45

1.80

1.45

1.80

2.50

3.30

6.50

7.10

at 10 (kw)

0.50

0.60

1.15

1.50

1.15

1.50

1.80

2.30

4.50

6.00

at 0 (kw)

0.35

0.42

0.86

1.15

0.86

1.15

1.10

1.70

3.00

4.10

at -10 (kw)

0.18

0.32

0.62

0.85

0.62

0.85

0.65

1.00

2.10

2.50

at -20 (kw)

0.04

0.14

0.30

0.40

0.30

0.40

0.40

0.55

1.20

1.50

Pump

Max. flow rate
(L / min, gal / min)

50 / 13.2

40 / 10.57

70 / 18.49

Max. pressure
(bar / psi)

1.0 / 14.5

3.3 / 47.9

6 / 87.0

Dimension

Max,. filling capacity
(L, cu ft)

7.5 / 0.26

13.5 / 0.47

13.5 / 0.47

25 / 0.88

39 / 1.37

For tubing dia
(mm / inch)

20 / 3/4

20 / 3/4

Overall
(WxLxH, mm / inch)

405 620 710 /
15.9 24.4 28.0

515 715 835 /
20.3 28.1 32.9

515 715 835 /
20.3 28.1 32.9

550 900 1140 /
21.7 35.4 44.9

605 1045 1300 /
23.8 41.4 51.2

Net weight (kg / Ibs)

62.85 /
138.6

64.85/
143.0

86.5 /
190.7

87.35 /
192.9

91.8 /
202.4

92.65 /
204.3

141.3 /
310.9

146.3 /
321.9

171 10 /
377 22

176 10 /
388 10

Electric requirement
(230V, 60Hz)

5.0A

5.5A

6.5A

7.5A

6.5A

7.5A

9.5A

13.5A

Cat. No.

AAH
65001K

AAH
65011K

AAH
65021K

AAH
65031K

AAH
65121K

AAH
65131K

AAH
65141K

AAH
65151K

-

-

Electric requirement
(230V, 50Hz)

4.5A

5.0A

6.0A

7.0A

6.0A

7.0A

7.5A

12.5A

Cat. No.

AAH
65002K

AAH
65012K

AAH
65022K

AAH
65032K

AAH
65122K

AAH
65122K

AAH
65142K

AAH
65152K

-

-

Electric requirement
(120V, 60Hz)

10.0A

11.0A

13.0A

15.0A

Cat. No.

AAH
65003K

AAH
65013K

AAH
65023K

AAH
65033K

-

-

-

-

-

-

Electric requirement
(380V, 60Hz)

5.5A

7.0A

Cat. No.

-

-

-

-

-

-

-

-

AAH
65169K

AAH
65179K

Electric requirement
(380V, 50Hz)

5.0A

6.0A

Cat. No.

-

-

-

-

-

-

-

-

AAH
65168K

AAH
65178K

Services

GMI Certified Service means:

Healthy operation of your instrument

Helping to ensure the quality of your results

Lowering cost of ownership through multilple service agreement options

Warranties from 90 days to a year as well as multi-year service agreements are available.

GMI is an ISO 9001:2008 certified company - proof that your instruments pass through a rigorous re-certification process to ensure that it arrives at your laboratory defect-free and in full working order.

We are neither an auction company nor an instrumentation broker. GMI employs our own factory-trained engineers to ensure full control of this process.