Ripple current output devices B5-47/1 in the constant current mode, do not exceed:

0,2% Imax - effective value.

Additional error output stabilized value of ambient temperature for every 10 ° C operating temperature range is less than half the basic error.

Additional error stabilizing output current value when the ambient air temperature for every 10 ° C operating temperature range is less than half the basic error mentioned above.

Output voltage drift over 8 hours of continuous operation and for any 10 minutes, eliminating the time samoprogreva does not exceed the value of basic error.

Offset drift over 8 hours of continuous operation and for any 10 minutes, eliminating the time samoprogreva, does not exceed the basic error.

Devices B5-47/1 should have the manual (front panel) and remote control of the output voltage and output current, and the devices B5-47/1/1 - only manual control.Remote control should be a contact closure 2-13 and 18-29 on the control connector contact 50 of the same connector.Pin control service and the corresponding values ​​of output voltages and currents obtained when the closure are shown in Table.2, 3.The quantity of current flowing through the contacts must be in the range (25-45 mA).

In devices B5-47/1 provides protection against overloads and short circuits at the output of the device by automatically switch from the voltage regulation mode constant current, and vice versa.

The maximum time setting of the output voltage from 0 to 0.9 the maximum value from the date of the management team in voltage does not exceed 100 ms.

The output impedance devices B5-47/1 in voltage in the frequency range from 20 Hz to 200 kHz shall not exceed 3 ohms.

Acceptable modulation factor in voltage does not exceed 5% in the frequency range from 20 Hz to 200 kHz.

Maximum emission output voltage when the load changes from 0.9 to 0 the maximum value in voltage not exceeding 5 V.

Emissions less than 5.The time setting of the output voltage when the load current of 0.9 the maximum value to 0 in voltage does not exceed 100 ms.

DC Power Supply B5-47/1allows serial connection of two devices of the same type.

Electrical isolation output circuits can withstand without breakdown following a test voltage of 500 V DC.

Table 2.

The value to set the output

Voltage, V

Pin Numbers for the REMOTE CONTROL

B5-47/1

2

3

4

5

6

7

8

9

10

11

12

13

0.1 V

1

0

0

0

0

0

0

0

0

0

0

0

0.2 V

0

1

0

0

0

0

0

0

0

0

0

0

0.4 V

0

0

1

0

0

0

0

0

0

0

0

0

0.8 V

0

0

0

1

0

0

0

0

0

0

0

0

1.0 V

0

0

0

0

1

0

0

0

0

0

0

0

2.0 V

0

0

0

0

0

1

0

0

0

0

0

0

4.0 V

0

0

0

0

0

0

1

0

0

0

0

0

8,0 V

0

0

0

0

0

0

0

1

0

0

0

0

10.0 V

0

0

0

0

0

0

0

0

1

0

0

0

20.0 V

0

0

0

0

0

0

0

0

0

1

0

0

Table 3.

The value to set the output

Current, A

Pin Numbers for the REMOTE CONTROL

B5-47/1

18

19

20

21

22

23

24

25

26

27

28

29

0.01 A

1

0

0

0

0

0

0

0

0

0

0

0

0.02 A

0

1

0

0

0

0

0

0

0

0

0

0

0.04 A

0

0

1

0

0

0

0

0

0

0

0

0

0.08 A

0

0

0

1

0

0

0

0

0

0

0

0

0.1 A

0

0

0

0

1

0

0

0

0

0

0

0

0.2 A

0

0

0

0

0

1

0

0

0

0

0

0

0.4 A

0

0

0

0

0

0

1

0

0

0

0

0

0.8 A

0

0

0

0

0

0

0

1

0

0

0

0

1.0 A

0

0

0

0

0

0

0

0

1

0

0

0

2.0 A

0

0

0

0

0

0

0

0

0

1

0

0

-

0

0

0

0

0

0

0

0

0

0

1

0

-

0

0

0

0

0

0

0

0

0

0

0

1

Note.1 - denotes the closure of the specified contact to contact 50 COMMON connector REMOTE CONTROL.

The insulation resistance of these circuits devices B5-47/1 relative to the body - not less than 20 MW, 5 MW, 1 MW, respectively, for normal conditions, higher operating temperatures and humidity. Dielectric strength between any pin of the power cord and the body of the device can withstand without breakdown test voltage of 1500 V AC.

The value of the stress test after test The humidity should be installed by a factor of 0.6.Devices B5-47/1 permit any ground output terminal. DC Power Supply B5-47/1 provides its specifications within the established norms after 30 minutes of switching on.The devices are AC voltage (220 ± 22), frequency (50 ± 0,5) Hz and harmonic content up to 5%.

The power consumed by the device does not exceed 400 VA.Devices allow continuous operation for 8 hours DC Power Supply B5-47/1 retains its specifications within the rules after being in extreme conditions (ambient temperature from 233 to 333 K (-40 to +60 ° C) and the subsequent aging in normal conditions for 2 hours

LLC «Zapadpribor» - a huge selection of measuring equipment for the best value and quality. So you can buy inexpensive devices, we monitor competitors' prices and are always ready to offer a lower price. We only sell quality products at the best prices. On our site you can buy cheaply as the latest innovations and proven equipment from the best manufacturers.

The site includes a special offer «buy at the best price» - if other Internet resources in products presented on our site, the lower price, we will sell it to you even cheaper! Buyers also additional discount for leaving a review or photo of use of our products.

In the price list contains not the whole range of products offered. Prices of goods, not included in the price list can learn to contact the manager. Also, our managers You can get detailed information about how cheap and profitable to buy instrumentation wholesale and retail. Telephone and e-mail for advice on the purchase , delivery or receipt are shown on the description of the goods. We have the most qualified staff, quality equipment and the best price.

LLC «Zapadpribor» - the official manufacturer's dealer test equipment. Our goal - selling high quality goods with best price offer and service to our customers. Our company can not only sell you the necessary instrument, but also offer additional services to its calibration, repair and installation. That you have a pleasant experience when shopping on our site, we have provided special gifts guaranteed to the most popular products.

Plant «META» - is the most reliable manufacturer of equipment for inspection. Brake stand STM produced on this plant.

If there is no site in the descriptive information you need on the device you can always turn to us for help. Our qualified managers will update for you on the technical characteristics of the device from its technical documentation: user manual, certificate, form, operating instructions, diagram. If necessary, we will take a photo of your device, or a device stand.
You can leave feedback on the unit purchased from us, the meter, the device, indicator, or product. Your review for your approval will be published on the site without contact information.Description of the instruments taken from the technical documentation or technical literature. Most pictures of products made directly by our experts prior to shipment of goods. In the description of the device provided the main technical characteristics of the instrument: nominal measuring range, accuracy class, scale, supply voltage, size (size), weight. If the site you see the discrepancy device name (model) specifications, photo or documents attached - let us know - you will get a useful gift with-selling device.
While the need to clarify the overall weight and size or the size of the individual meter you can in our service center. If you need our engineers will help you choose the most complete analog or a suitable replacement for the device you are interested. All counterparts and replacement will be tested in one of our laboratories with the full compliance to your requirements.
Our company carries out repairs and maintenance of measuring equipment in more than 75 different factories of the former Soviet Union and CIS. We also provide such metrological procedure: calibration, tare, graduated, testing of measuring instruments.
If you can make repairs to the device yourself, our engineers can provide you with a complete set of necessary technical documentation: circuit diagram TO, ER, FD, PS. We also have an extensive database of technical and metrological documents: technical specifications (TS), the terms of reference (TOR), GOST (DSTU), the industry standard (OST) test procedure, the method of certification, verification scheme for more than 3,500 types of measuring equipment from the manufacturer this equipment. From the site you can download all the necessary software (the software driver) is required for the product purchased.
We also have a library of legal documents that are related to our field of activity: the law, code, ordinance, decree, temporary position.
Upon request, each measuring device is provided metrological certification or verification. Our staff can represent your interests in such organizations as the metrological Rostest (Rosstandart), State Standard, Derzhspozhiv standard (Derzhspozhivstandart), TSLIT, OGMetr.
Sometimes customers can enter our company name wrong - for example, zapadprybor, zapadprylad, zapadprіbor, zapadprilad, zahіdpribor, zahіdprіbor, zahidpribor, zahidprilad, zahidprіbor, zahidprybor, zahidprylad. That's right - Zapadpribor or zahіdprilad.
LLC "Zapadpribor" is the supplier of ammeters, voltmeters, power meters, frequency meters, phase meters, shunts and other devices such manufacturers of measuring equipment, such as: PO " Elektrotochpribor " (M2044, M2051), Omsk, OAO "Instrument-Making Plant "Vibrator" (M1611, Ts1611), St. Petersburg, JSC "Krasnodar ZIP " (E365, E377, E378), LLC " ZIP Partner" (Ts301 , Ts302 , Ts300) and "ZIP" Yurimov"(M381, C33), Krasnodar, OAO "VZEP" ("Vitebsk factory of electrical appliances") (E8030, E8021), Vitebsk, OAO "Electropribor" (M42300, M42301, M42303, M42304, M42305, M42306), Cheboksary, JSC "Elektroizmeritel" (Ts4342, Ts4352, Ts4353 ) Zhitomir, PJSC "Uman factory "Megger" (F4102, F4103, F4104, M4100), Uman.