Checked and tightened vacuum fittings. Checked and topped off pump oil.Also tightened the drain plug on pump. Verified process program. Was able to do several successful cycles. There is still an intermittant problem with passing the leak check, due to a leak in the process chamber. We are making plans to have the chamber repaired at the annual lab closure in December. If the process aborts, restart the program.