24 DIFFERENT PORTS AVAILABLE
Standard selection of NPT, SAE and BSP ports reduces the amount of adapters required for installation.

LOW COST ACCURACY
± 2.5% of range accuracy in center third of scale; ±4% in upper and lower thirds.

BI-DIRECTIONAL AND REVERSE FLOW OPTION OFFERED
Phosphate ester flow meters are also available in bi-directional and reverse flow versions. Contact Lake Monitors for more information.

Skydrol is a registered trademark of Solutia, Inc.
Pydraul is a registered trademark of Monsanto Co.
Fyrquel is a registered trademark of AKZO Chemicals.
Houghton 900 is a registered trademark of Houghton International.

Engineering Specification

THE PHOSPHATE ESTER IN-LINE FLOW RATE METER SHALL:

Include a direct-reading scale corrected for phosphate ester media.

Use the variable annular orifice technique with compression spring recoil.

1To determine approximate measuring ranges multiply the range listed in the Liquid Flow Rate section of Lake’s Guide to standard meter numbers by 0.93. For example, a P3A6WB10 would have a scale range to 10 GPM *0.93 = 9.3 GPM at full scale.