6/17/2013

iMetry - The example of measurement in lead box - Geiger Counter/Spectrometry (Spectrometer,Spectroscopy) for iPad

Radiation Geigercounter of inside of a lead box(Thickness:50mm)

Spectrometry of inside of a lead box (Thickness:50mm) (log scale)

Radiation Geiger counter of inside of a lead box with copperplate (Thickness: lead block 50mm, copperplate 1mm)

Least descrimination sensitivity of iMetry would be, in this example, relative MDA would be around 0.00024μSv/h@ 1 hour measurement, with 3SD accuracy, under 50mm lead box with 1mm copper shielding, at 0.06 μSv/h background. In this case, cables from imetry should be guided, through V shaped canal.

Sensitivity parameter may vary for each individual difference.

Spectrometry of inside of a lead box with copperplate (Thickness:lead block 50mm, copperplate 1mm) (liner scale)

Spectrometry of inside of a lead box with copperplate (Thickness: lead block 50mm, copperplate 1mm) (log scale)

Radiation Geiger counter of inside of a lead box with copperplate (Thickness: lead block 50mm, copperplate 1mm)

Least descrimination sensitivity of iMetry would be, in this example, relative MDA would be around 0.00024μSv/h@ 1 hour measurement, with 3SD accuracy, under 50mm lead box with 1mm copper shielding, at 0.06 μSv/h background. In this case, cables from imetry should be guided, through V shaped canal.