DL750/DL750P ScopeCorder

Description

This firmware is intended to upgrade the DL750 series ScopeCorder to the version which is said in the above "Versions and Corresponding Functions" page.

Remarks

Please read installation procedures throughout before performing the installation. Each zip file contains firmware program file and an installation procedure.

Firmware Versions and Corresponding Functions

To check the firmware version number, choose MISC -> OVERVIEW on the DL750. The modules and features in the table cannot be used unless supported by your DL's firmware version.
Starting with this version (6.02), the firmware code for the DL750 and DL750P has been unified. So you can use same code for DL750 and DL750P both.

You can enter a setting to specify whether the instrument responds immediately when the START/STOP key is pressed, or whether the instrument only responds if the key is pressed for one second or longer.

4.02 or later

Standard

- Improvement of DualCapture Function

Changed the number of sub waveform screens
/M1 model: Up to 250
/M2 and /M3 model: Up to 500
User can select the display format and the traces to be displayed on the sub waveform window. Displays the list of captured waveforms: Show Map
Mail transmission using the dual capture trigger function: Send Mail
Automated measurement of waveform parameters on the sub waveform (the results are displayed in italics)

- Addition of Parameter Search of the History Memory Function
- Addition of Cursor Functions

In Log mode, the DL750 automatically executes the set operation when measurement stops.

- Added of Function to Change the Size of the Extra Window

4.01 or later

/C10

- Addition of Network Functions

WebDAV server function, SNTP function
Mail attachment of image data

4.01 or later

/G2, /G3

- Displays the Hz Value of the Cut-off Frequency When Applying Filters to DSP or MATH Channels

4.01 or later

Standard

- Numeric monitor display
- Added exponential window to FFT

computation/increased number of points in FFT computation (up to 100 kpoints)

- Normal statistical processing/statistical processing of history memory
- Increased the number of parameters for automated measurement/statistical processing of waveform parameters (up to 24000 to up to 48000)
- Added waveform data save formats for action-on-trigger and GO/NO-GO determination (select from binary, ASCII, and floating point)
- Support for inverted display on the strain module
- Support for the Current Probe (Model 701931)

* Software Agreement

The property rights, proprietary rights, intellectual property rights, and all other rights associated with the software are held by Yokogawa Electric Corporation.

Under no circumstances is any dumping, reverse compiling, reverse assembly, reverse engineering, or any other kind of alteration or revision of this software allowed.

This software is offered free of charge, but no unlimited warranties are made against any defects whatsoever.

Also, Yokogawa may not be able to accept inquiries regarding repair of defects in or questions about this software.

The contents of this software are subject to change without prior notice as a result of continuing improvements to the software\'s performance and functions.

Yokogawa bears no liability for any problems that may occur during download or installation of this software.

Use of the Yokogawa Web site is at the user\'s own risk.

Any parties contributing to the creation or distribution of the contents on the Yokogawa Web site shall bear no responsibility whatsoever for any damages occurring as a result of the use of said contents.

Downloading of the software indicates acceptance of the software agreement.