Cookie Statement

This site uses cookies to store information on your computer. Some are essential to make our site work; others help us improve the user experience. By using the site, you consent to the placement of these cookies. You may at any time change the settings regarding cookies. Read our privacy statement to learn more.

Menu

Philips A04 SPM with OxiMax™ Technology

Overview

Making the Most of the Partnership

We design our technology knowing that the best possible outcomes happen when all parts of the system work together. We address specific needs, but do it seamlessly, without creating upheaval for patients or healthcare organizations.

From the sensors to the front-end electronics and the signal processing, our products are engineered to provide:

If it doesn’t meet our high standards, it doesn’t go to market. We are dedicated to innovation, but will not sacrifice due diligence, rigorous testing or a full understanding of the impact of the product or technology.

Safety

U.S. Federal law restricts this device to sale by or on the order of a physician.

SpO2 Sensors

Wavelength range: 500 to 1,000 nm

Emitted light energy: ≤ 15 m W

Environmental

Module

Operating temperature: 0 to 45°C (32 to 113°F)

Storage temperature: – 40 to 70°C (– 40 to 158°F)

Operating humidity: 95% RH max. at 40°C (104°F)

Storage humidity: 95% RH max. at 65°C (150°F)

Operating altitude: Up to 4,600 m (15,000 ft)

Storage altitude: Up to 15,300 m (50,000 ft)

Philips M1943NL is required to use the Nellcor™ SpO2 module.

Single-patient -use and specialty sensor Accuracy([FOOTNOTE=MAX-N: Clinical functionality has been demonstrated on a population or hospitalized neonate patient. The observed SpO2 accuracy was 2.5% in a study of 42 patients with ages of 1 to 23 days, weighing from 750 to 4100 grams, and 63 observations made spanning a range of 85 to 99% SaO2 while monitored with Nellcor™ OxiMax™ N-595 pulse oximeters. The accuracy specification has been determined between saturations of 80% – 100%],[ANCHOR=],[LINK=]),([FOOTNOTE=MAX-R ± 3.5 digits determined between saturations of 80% – 100%.],[ANCHOR=],[LINK=]) ,1

Saturation (% SpO2 ± 1 SD)

70% to 100% ± 2 digits

60% to 80% ± 3 digits

Low perfusion: 70% to 100% ± 2 digits

Pulse rate: 25 to 250 bpm ± 3 digits

Low perfusion: 25 to 250 bpm ± 3 digits

Compatibility

Compatible with Philips IntelliVue* systems†

MP40

MP50

MP60

MP70

MX80

MX90

MX900

For additional information or to place an order, call Customer Service at 019-232 025 04.

Technology

Get accurate data – quickly

Nellcor™ pulse oximetry with OxiMax™ technology specifies a highly accurate measurement of SpO2, even in some of the most challenging patients.([FOOTNOTE=NCT01720355 at Clinical Trials.gov.],[ANCHOR=],[LINK=])