Fever Screening

Supporting Components

Camera and Device Mounting Package

The camera and device mounting package includes the following items for mounting and
positioning the Infrasight IS640 camera and devices under test:

Mounting Table

Vertical Focusing Stage

XY Positioning Stage

Mounting Table

The mounting table provides a sturdy platform on which to perform
sensitive thermal analysis tests. The table is 45 x 45 x 5.6cm and has a grid
of 17 x 17 M6 threaded holes on 25mm centers for mounting additional
test equipment and fixtures. Includes vibration isolation feet
and table grounding cable with plug.

Vertical Focusing Stage

The vertical stage is 43.8cm tall and provides smooth and precise
manual vertical positioning of the infrared camera. A precision lead
screw moves the camera vertically 2mm for every rotation of the
micrometer dial. Each small increment on the dial moves the
camera vertically 0.010mm (10μm). An aluminum alloy dovetail
slider provides excellent travel straightness, wear resistance, and
requires no lubrication. Using the side ruler and micrometer
dial, the camera can quickly and easily be repositioned to previous
locations. A locking thumbscrew enables users to lock the lead
screw in place to prevent camera movement during testing.

XY Positioning Stage

The XY stage enables small devices to be precisely moved and
positioned within the camera’s field of view. Crossed-roller
bearings offer superior straightness of travel and can bear heavy
loads. The stage platform is 120 x 120mm and has M6 threaded
mounting holes spaced 25mm apart for mounting Micro accessories and test
equipment. Using the x and y micrometer dials, the stage can
be precisely positioned. Each increment on the micrometer
dials moves the stage 0.010mm (10μm). Thumb screws allow the
stage to be locked to in one or both of the x and y directions to
prevent device movement during testing.