Choose Product Options and/or Extra Supplies

Unique in nature, the SoundCare Plus professional ultrasound device comes complete with two different sound heads - 1 cm² and 5 cm² - that can be plugged in simultaneously for more targeted therapy. Simple to operate, the user can switch between wands with the single push of a button.

Product Features:

Comes complete with two sound heads

Used for the treatment of chronic and acute muscular pain

User can switch between dual sound heads with the push of a button

Fully functional 1 MHz and 3 MHz frequencies

20 user-defined presets with pulsed and continuous therapy

Ergonomic handle design

Two Year Manufacturer Warranty

Product Specifications:

Acoustic Frequency: 1 MHz ± 10%; 3 MHz ± 10%

Output Power:

0.5 - 10.0 W ± 20%, when duty factor > 80% for 5 cm²

0.5 - 15.0 W ± 20%, when duty factor < 70% for 5 cm²

0.1 - 2.0 W ± 20%, when duty factor > 80% for 1 cm²

0.1 - 3.0 W ± 20%, when duty factor < 70% for 1 cm²

Pulse Repetition Rate: 100 Hz ± 10%

Duty Factor: 10% - 100%, Stepping 10%

Timer: Adjustable up to 30 minutes

Effective Radiating Area: 5.0 cm² ± 20%; 1.0 cm² ± 20%

Actual Intensity:

3.0 W/cm² ± 20% (1 MHz)

3.0 W/cm² ± 20% (3 MHz)

Raw (Max): 5.0

Beam Type: Collimated

Material of Applicator: Aluminum

Waterproof Grade: IPX7 only for treatment head

tech

good , better than traditional Clinical Ultrasound machine. durable &easy to use .

Did you find this helpful?
4 of 5Found Helpful

Reviewed by: dv from san francisco.
on 12/19/2013

5/5

Mr.

Works perfectly. Would absolutely recommend this unit to anyone.

Did you find this helpful?
2 of 2Found Helpful

Reviewed by: Asbjørn from Norway.
on 8/6/2015

5/5

Great unit!

There are 3 reasons why it is wise to buy this product. Thanks for the assistance in securing the delivery of this product to us.
1. User Friendly
2. Cost Effective
3. High Quality

Did you find this helpful?
1 of 1Found Helpful

Reviewed by: Fernand from Philippines.
on 6/8/2016

5/5

Q:Hello,
I was wondering about the safety of the ultrasound machine. We have ordered one and have tried to use the 3 W/cm^2 option, but for some reason, the probe heats up to unusually high temperatures (~80 C). We've tried on both frequencies, and differing duty cycles, but the high temperature is still there. Is there any reason why this would be happening? Could our probe be defective, or is it that the device heats when used with higher powers?
Thanks!

Asked by: Bruce Wang
- 8/4/2016

A: This would be a question better directed to the manufacturer, as it is possible that there may be a defect somewhere, although this would certainly be a rare and isolated incident if that were the case because this is one of the most popular therapeutic ultrasound devices on the market designed for clinical settings. You may want to contact Roscoe Medical for a potential warranty claim, whose information should also be included in the instruction manual that came with the device.
Answer provided by: TENSPros Management (8/8/2016)

Was this answer helpful?
0 of 0 Customer(s) found this Answer helpful.