The ARCHIE automatically picks up sample from the assigned vial position and delivers it to the introduction module. In between samples, it washes the syringe and needle, utilizing a special dual wash & drain station
tray, mounted beside the sample tray. Next it picks up sample from the assigned sample vial to pre-rinse the syringe with the sample at hand.

Even before you observe which steps it performs, it already starts to inject the sample into the introduction module. Just a few minutes later the first result appears on your screen and the sampler picks up the replicate or next sample. The sampler is set up to work with all standard 2 mL vials on one or two vial trays of 105 positions each. Both vial trays can be equipped as temperature controlled vialt ray (heating as well as cooling for each tray).

An audible signal, LED (RGB) color code and messages in TEIS (software) will indicate the sampler status and performance parameters.