Walk In Test Chamber

THERMOTRON

Thermotron offers Panel, Solid and Stability Walk- In Environmental Chambers for testing large components, assemblies and furnished products. Standard Panel Walk- In chambers comes in various volumes and compressors sizes from 3hp to 25 hp capacity with wide range of features that suits majority of the industry applications. Below is a sample list of standard Panel Walk In chambers. We have several standard configurations available, contact us if the below does not meet your requirement, we will be happy to assist you with suitable equipment.

Model

Temperature Range

Humidity Range

Temperature Change Rate

Panel Walk-In

-68°C to +85°C

20% to 95%RH

1 to 2°C/min

Solid Walk-In

-73°C to +177°C

20% to 95%RH

4 to 5°C/min, or faster

Steady State Walk-In

-20°C to +50°C

20% to 95%RH

N/A

Volume Cu. /ft. (Liters)

Possible Compressor configuration

Refrigeration System

Heater Wattage,KW

Air Flow (CFM)

*Average Heating Rate

*Average
Cooling Rate

286
(8100)

5HP

Single

4.5

1500

2.1

2.2

5-5 HP

Cascade

4.5

1500

2.1

2.2

323
(9147)

5HP

Single

4.5

1500

2.0

2.0

5-5 HP

Cascade

4.5

1,500

2.0

1.6

364
(10308)

5 HP

Single

4.5

1500

1.8

1.8

5-5 HP

Cascade

4.5

1,500

1.8

1.5

410
(11611)

5 HP

Single

4.5

1500

1.7

1.4

5-5 HP

Cascade

4.5

1,500

1.7

1.4

499
(14132)

10 HP

Single

7.5

3000

2.6

3.3

10-10 HP

Cascade

7.5

3,000

2.6

2.8

563
(15944)

10 HP

Single

7.5

3000

2.5

3.1

10-10 HP

Cascade

7.5

3,000

2.5

2.7

605
(17134)

10 HP

Single

12.0

3,000

2.5

2.9

10-10 HP

Cascade

12

3,000

2.5

2.6

683
(19343)

10 HP

Single

12

3,000

2.3

2.9

10-10 HP

Cascade

12

3,000

2.3

2.5

769
(21778)

25HP

Single

15

3,000

2.9

4.3

25-25 HP

Cascade

15

3,000

2.9

3.8

867
(24553)

25HP

Single

15

3,000

2.8

3.9

25-25 HP

Cascade

15

3,000

2.8

3.6

904
(25601)

25HP

Single

15

4,500

2.6

3.9

25-25 HP

Cascade

15

4,500

2.6

3.4

1020
(28886)

25HP

Single

15

4,500

2.5

3.6

25-25 HP

Cascade

15

4,500

2.5

3.3

1097
(31067)

25 HP

Single

18

4,500

2.4

3.6

25-25 HP

Cascade

18

4,500

2.4

3.1

1237
(35032)

25 HP

Single

18

4,500

2.3

3.3

25-25 HP

Cascade

18

4,500

2.3

3.1

1261
(35712)

25 HP

Single

24

6,000

2.2

3.3

25-25 HP

Cascade

24

6,000

2.2

3.0

1422
(40271)

25 HP

Single

24

6,000

2.1

3.1

25-25 HP

Cascade

24

6,000

2.1

2.8

KEY FEATURES

STANDARD

OPTIONAL

4” (10.16 cm) Rigid Urethane Foamed-in-Place Insulation

Air-Cooled Condenser

Cam Action Speed Lock

Shelf

Embedded Steel Straps Around Perimeter

Dry Air Purge

Heavy Duty Inside Floor Surface

Multiple Door Sizes

12″ Color Touch Screen Control Display

Humidity Purification System

USB Port with Flash Drive

LN2 Boost Cooling

40 Gigabyte Hard Drive

Electrical Outlets

Remote Refrigeration Package

Electrical Service 115 / 230/415 V, 50 or 60 Hz with AC 1 or 3 Phase

CE and CSA Compliant

Note: Specifications subject to change without notice. Special order options are available, please check with us.