DasyLAB

Search within DasyLAB

If you are a user of a software or hardware product purchased from Adept Scientific and require technical support, please use our Technical Knowledge Base below. If you can't find a solution on the Knowledge Base then please complete our technical support form or contact us by email or by telephone on +44 (0) 203 695 7810

Using Measurement Computing Hardware with DASYLab 10.0
A Using Measurement Computing Hardware with DasyLAB 10.0 Before installing DasyLAB, it is essential that an up to date version of Instacal is installed. You can download a recent version of Instacal here: ftp://ftp.measurementcomputing.com/downloads/InstaCal/ Run Instacal, and check that your hardwa....

Explaining Blocks and Block Size in DASYLab
6 DASYLab works in a block dependent mode. Each block contains measurement data, called samples, and a block header. The header contains information about the block, including the data type, the start time of the block and the sample interval. The program calculates a time stamp for each sample. Ma....

How can I produce a flat reference curve in DASYLab?
6 The chart recorder and Y/t modules support a reference curve, which can be configured by double clicking the module on the worksheet, and clicking the "Ref. Curve" button. You will see the following options. Show reference curve Specifies that DASYLab displays a reference curve for the selected d....

What is the difference between the chart recorder and the Y/t chart?
A The chart recorder is most suitable for slow data (e.g. up to 1kHz) while the Y/t data is suitable for high speed data. In default mode, the Y/t chart operates like an oscilloscope, and will only update when new data equal to the complete width of the display is available. However, a fast recorder....

DASYLab Error: "Device AI: A module cannot be loaded "
2 The problem results from the fact that although Measurement Computing DASYLab drivers are installed on the computer the actual hardware is not connected to it or configured in Instacal. To resolve this issue close DASYLab, connect the hardware to the computer, run Instacal and configure your hardw....

Using DASYLab to Communicate over RS-232
2 Figure 1: A simple DASYLab worksheet to read data from an instrument via RS-232. Open the Modules menu, and go to the Input/Output section. Paste an RS-232 input module onto the worksheet. (Click OK to the Select New input module at different Interface dialogue.) If an error opening serial interfa....

For the time being we are unable to offer the following product ranges although we are currently working hard to increase the number of products we can offer in the future. Please contact us to talk about alternative products that we may be able to offer you.