Authors

Issue Date

Metadata

Abstract

Objective
The primary aim of this study was to assess the accuracy of an infrared camera and that of a skin thermistor, both commercially available. The study aimed to assess the agreement over a wide range of skin temperatures following cold exposure.
Methods
Fifty-two males placed their right hand in a thin plastic bag and immersed it in 8 °C water for 30 min whilst seated in an air temperature of 30 °C. Following hand immersion, participants removed the bag and rested their hand at heart level for 10 min. Index finger skin temperature (Tsk) was measured with a thermistor, affixed to the finger pad, and an infrared camera measured 1 cm distally to the thermistor. Agreement between the infrared camera and thermistor was assessed by mean difference (infrared camera minus thermistor) and 95% limits of agreement analysis, accounting for the repeated measures over time. The clinically significant threshold for Tsk differences was set at ±0.5 °C and limits of agreement ±1 °C.
Results
As an average across all time points, the infrared camera recorded Tsk 1.80 (SD 1.16) °C warmer than the thermistor, with 95% limits of agreement ranging from −0.46 °C to 4.07 °C.
Conclusion
Collectively, the results show the infrared camera overestimated Tsk at every time point following local cooling. Further, measurement of finger Tsk from the infrared camera consistently fell outside the acceptable level of agreement (i.e. mean difference exceeding ±0.5 °C). Considering these results, infrared cameras may overestimate peripheral Tsk following cold exposure and clinicians and practitioners should, therefore, adjust their risk/withdrawal criteria accordingly.

Journal

URI

DOI

Additional Links

Type

Journal article

Language

en

Description

This is an accepted manuscript of an article published by Elsevier in Journal of Thermal Biology on 03/05/2020, available online: https://doi.org/10.1016/j.jtherbio.2020.102614
The accepted version of the publication may differ from the final published version.

Export search results

The export option will allow you to export the current search results of the entered query to a file. Different
formats are available for download. To export the items, click on the button corresponding with the preferred download format.

By default, clicking on the export buttons will result in a download of the allowed maximum amount of items.

To select a subset of the search results, click "Selective Export" button and make a selection of the items you want to export.
The amount of items that can be exported at once is similarly restricted as the full export.

After making a selection, click one of the export format buttons. The amount of items that will be exported is indicated in the bubble next to export format.