The PerkinElmer NexION 300D ICP-Mass Spectrometer uses 3 quadrupoles with the center being acting as a universal cell to bring you a powerful, state-of-the-art cell based Mass Spectrometry System in a benchtop design. The center universal cell of this mass spectrometer can be used as both a collision cell and as a dynamic reaction cell. The ICP-MS offers users the choice between kinetic energy discrimination as well as DRC by using a scanning quadrupole.

The PerkinElmer NexION 300D ICP-MS uses 3 modes in a single instrument, making it one of the most powerful polyatomic interference removal systems. Depending on the level necessary, users can decide which mode is best for them, standard, collision or reaction. With use of the reaction mode, this ICP-MS system uses highly reactive gas can detect notoriously difficult elements such as iron, arsenic, selenium, magnesium, chromium potassium and more.

This highly stable, flexible ICP-MS instrument utilizes a unique Triple Cone Interface, and Quadrupole Ion deflector offering hour after hour of consistent signal responses. The triple cone interface and quadrupole ion deflector remove most neutrals and photons before even entering the Universal cell, providing exceptional focus and definition. This dual channel mass spectrometer can switch from one mode to another quickly and easily and has no restrictions on the type of gasses you can use. This ICP-MS Instrument removes a tremendous amount of un ionized material, keeping it out of the universal cell, eliminating the need for routine cleaning, minimizing drift and giving you reproducible results. This lack of downtime spent cleaning and recalibrating also translates into higher productivity for your lab.

The PerkinElmer NexION 300D ICP-MS features simultaneous dual mode detector, providing simultaneous measurement of low and high level analytes with nine orders of dynamic range, a large open sample introduction area, low liquid uptake nebulizer, and a free running RF plasma generator. This mass spectrometer save valuable lab space by being able to be placed flush to a wall, the rear is free of connectors. This ICP mass spectrometer also features a four-stage vacuum system, with fast pump down capability, 2-3 times faster than other systems. The NexION 300D mass spectrometer has fully automated X, Y , Z torch positioning, controlled through the NexION’s intuitive and powerful software, from which users can customize the instrument to their needs. The software for this ICP-MS allows users to schedule tasks, notifies of routine maintenance, has an optimization wizard, QC checks, and customizable reporting.

PerkinElmer NexION 300D ICP-Mass Spectrometer Specifications:

Detection limits

Element ng/L (ppt)

9Be < 1

56Fe < 3

59Co < 1

115In < 0.5

238U < 0.5

Sensitivity Element M cps/mg/L

9Be > 3

24Mg > 20

115In > 50

238U > 40

Background signal

Mass 220 < 1 cps

Short-term precision

< 3% RSD

Long-term stability

< 4% RSD over 4 hours

Isotope-ratio precision

< 0.08*% RSD (*or within a factor of two of the counting statistics limit)

Mass calibration stability

< 0.05 amu over 8 hours of continuous operation

Quadrupole peak hop (slew) speed

1.6M amu/sec

Quadrupole scan speed

5000 amu/sec

Abundance sensitivity

Measured at 23Na:

Better than 1.0 x 10-6 at low mass side of peak

Better than 1.0 x 10-7 at high mass side of peak

Transient data acquisition speed

> 3000 temporal data points/sec maximum

Dimensions

57x32x32

PerkinElmer NexION 300D ICP-Mass Spectrometer

AS 93plus Autosampler

Computer Workstation

Perkin Elmer NexION Software Disks

Vacuum Pump

Vacuum Hose

Advanced Filter System

Polyscience Recirculating Chiller

Communication Cables

Power Cables

Two Boxes of Accessories as pictured

Operator's Guide

30-Day Warranty

Table pictured not included

This PerkinElmer NexION 300D ICP-Mass Spectrometer and it's additional components have all been thoroughly inspected and are in excellent condition. All of our lab equipment comes backed with our standard warranty. Should you have any questions, please don't hesitate to contact us.