The high sensitivity MOSFETs are three times as sensitive as the standard MOSFETs and help to maintain reproducibility at low doses. The high sensitivity MOSFETs are ideal for scatter doses to regions at risk, or radiology dose measurements.

• One calibration factor for all photon and electron modalities in the radiotherapy energy range

• Isotropic (± 3% or better for 360°)

• Active region of 0.2 x 0.2 mm

• Permits pinpoint measurement without patient shielding

• Dose-rate and temperature independent

• Unobtrusive in procedures

• Lightweight and flexible

• Multiple dosimeter capability with one reader

Flex Dimensions*: With of Flex: 2.5 mmThickness of Flex (flat part of flex): 0.3 mmLength of Flex (from end of gray cable): 200 mmThickness of Bulb: 1.3 mmLength of Bulb: 8.0 mmDistance to Center of Chip (from end of bulb): 4.0 mmCommon Uses: Dose verification during radiotherapy treatments; beam QA

*Measurements listed above are average values and are meant to serve as a guide for the use of MOSFET dosimeters. For phantom use and tolerances, please contact Best Medical Canada directly for more detailed information.

Note: MOSFET Dosimeters are available with a 10' (3 m) cable upon request.