Related Products

Topcon’s IMAGEnet i-base is a complete solution for image and instrument data capturi...

﻿

Specifications

Angle of coverage

45º

Working range

40,7 mm

Pupil diameter for photography

4,0mm or more / 3,3mm or more when the small pupil diaphragm is used

Type of photography

Color photography and red-free photography

Patient diopter correction range

Without correction lens: -13D to +12D (where split lines are used)

With minus correction lens: -12D to -33D

With plus correction lens: +9D to +40D

Auxiliary function for photography

Auto focus function
(Used only in the split line working range. This can be turned ON/OFF)

Internal Fixation Target

Center/Periphery

Right/left eye automatic detection

Optional position presetting function

Base movement

Back-forth: 46mm, Right-left: 100mm, Up-down: 30mm

Chinrest movement

67mm

Power source

Frequency: 50/60Hz Voltage: AC110, 120, 230, 240V selectable

Weight

24,5kg

Dimensions

272mm(W) x 505mm(D) x 530mm(H)

Power consumptions

400VA(Maximum), 100VA(Normaal)

﻿

Description

The Topcon TRC-NW8 is one of the most advanced cameras from our non-mydriatic camera range. It offers a truly versatile camera capable of non-mydriatic colour & red free angiography.

FundusThe TRC-NW8 non-mydriatic fundus camera comes with a Nikon D-700, 16 megapixel digital camera, enabling the user to zoom in on the finest detail. This particular camera offers a 45 degree angle which covers fovea and macula. Nine fixation targets enable to shoot nine images which can be collated using IMAGEnet i-base to enable a complete overview (IMAGEnet i-base auto-mosaic is needed for this function). Stereo photography and a real red free filter is a standard option on TRC-NW8.

Ease of useTRC-NW8 is joystick operated with a minimum of buttons to push. The auto-shoot, auto-focus and auto-exposure ensure sharp, clear images with a lot of detail. In one movement the practitioner can switch between colour, red free images and fluorescein images.

Patient comfortThe low flash intensity offers the patient increased comfort, which enables the practitioner to shoot more images in less time. A minimum pupil size of 3.3mm is needed.

NHS compliantThe TRC-NW8 has been approved for the national programm of the NHS in the UK for screening of diabetic retinopathy.