The Urology C-Arm Table features motorized actuation of height, X-Y, lateral roll and Trendelenburg motions of the tabletop. Multiple positioning control options, with pre-programmed features, ensure quick and safe tabletop positioning during any phase of a procedure. The rail-mounted main control panel, portable hand-held and foot-operated controllers can be positioned for access from any point around the table.

Main Features:

Accessory Rails: Standard OR accessory rails 1.125" x .375"

(2.86 x .95 cm) on table (rails also on table imaging extension)

Patient Restraints: Two body straps, two ankle straps

Finish: Powder coat

Shipping Weight: 903 lb (409.6 kg)

Patient Capacity:

In urologic position: 500 lb (227 kg)

In use with radiolucent extension: 400 lb (181.4)

Weight tested to four times the patient load rating.

Power: 115 VAC or 230 VAC and Battery. Battery automatically charges when table is plugged in. An additional battery can be charged with optional wall-mounted charger.

The LCD screen of the Main Control Panel displays actual tabletop position (all axes) in real time, and provides menus and prompts for other functions. Up to ten position sets are stored. A "Level" button brings the table to 0° tilt and 0° lateral roll while "Home" both levels and lowers the tabletop to minimum height for patient transfer. The portable hand-held and foot-operated controllers also adjust the table. The Joystick Controller is used for motorized X-Y positioning and features proportional speed control.

More adjustments and features...

The Fowler back adjusts with a simple touch-handle lock/unlock mechanism to an 80° angle with incremental stops at 20°, 30°, 40°, 50°, 60°, 70° and 80°.

The table comes equipped with a lightweight, non-imaging patient transfer extension and a radiolucent extension. The radiolucent extension provides capability for vascular, cardiac, and other general C-Arm applications. Select from knee crutches or the PAL stirrups, arm boards or drain pans, the choice is yours... configure to your exact specifications.

Both AC and battery power are standard. Eliminate power cords with battery operation. An optional battery, with separate charging unit, can provide cordless operation.

The main panel with backlit liquid crystal display mounts on either accessory rail. The LCD screen displays actual tabletop position (all axes) in real time, and provides menus and prompts for other functions. Up to ten position sets may be stored in system memory, and recalled at any time for automatic positioning. A "Level" button brings the table to 0° tilt and 0° lateral roll. "Home" levels the table and lowers the tabletop to minimum height for patient transfer. Control buttons are identified with easily recognizable icons.

The portable hand-held and foot-operated controllers offer the freedom to adjust height, lateral roll, X-Y and Trendelenburg motions from positions away from the main panel, or when hands are busy with other functions. The hand-held controller hangs on the accessory rail when not in use.

The joystick controller for motorized X-Y positioning offers intuitive positioning with proportional speed control. After setting an adjustable icon indicator on the joystick’s base, X-Y motion correlates directionally to actual tabletop motion. To prevent undesired motion by accidental contact with the joystick, a thumb button on top of the joystick must be pressed for activation.

The written material on Claflinequip.com is provided for our customers' information only; it is not intended to be used for diagnosing or treating a disease, medical condition, or prescribing medication. It is strongly recommended that you consult health care professionals for medical advice.