Trace and identify load-side equipment from the circuit breaker while energized.

Components Features and Funcitons:

Receiver:

With two detectors in its tip which are tuned to pick up the electromagnetic signals (“short” mode) or electrostatic signals (“open” mode) gen-erated by the other components: The LED’s and sensitivity settings indicate the relative signal strength to allow easy detection of the signal. Power: 9 volt battery.

Load Signal Generator:

When connected to an energized line, will draw a rapidly fluctuating current pulse from the source, causing the power line to emit an electromagnetic signal which is detected by the receiver in the “short” mode. This allows tracing of line-side conductors without power interruption or inter-ference. 9 - 600V AC or DC. Operating frequency: 32.768KHz.

Transmitter:

Injects a signal into an unenergized conductor. If the conductor is broken (open), no current will flow, so the signal is present as a voltage pulse, which is detected along the conductor’s length by the receivers electrostatic detector (“open” mode). The signal will stop at the open. Power: 9 volt battery. Operating frequency: 32.768 KHz