Wavelength dispersive X-ray fluorescence spectrometer

Rapid elemental analysis for very large and/or heavy samples

AZX 400

Rigaku's unique AZX400 sequential wavelength dispersive X-ray fluorescence (WDXRF) spectrometer was specifically designed to handle very large and/or heavy samples. Accepting samples up to 400 mm diameter, 50 mm thick and 30 kg weight, this system is ideal for analyzing sputtering targets, magnetic disks, or for multilayer film metrology or elemental analysis of large samples.

Customized sample adapter system

Having the versatility to adapt to your specific sample types and analysis needs, the AZX400 is adaptable to various sample sizes and shapes using optional (made to order) adapter inserts. With a variable measurement spot (30 mm to 0.5 mm diameter with 5-step automatic selection) and mapping capability with multi-point measurements to check for sample uniformity, this uniquely flexible instrument can dramatically streamline your quality control processes.

Sample view camera with special lighting

Optional real-time camera allows the analysis point to be viewed on screen. The operator has complete certainty as to what is being measured.

Traditional WDXRF analytical capabilities

All analytical capabilities of a traditional instrument are retained in this "large sample" variant. Analyze beryllium (Be) through uranium (U) with high-resolution, high-precision WDXRF spectroscopy, from solids to liquids and powders to thin films. Analyze wide composition ranges (ppm to tens of percent) and thicknesses (sub Å to mm). Optionally available is diffraction peak interference rejection, for optimal results for single-crystal substrates. The Rigaku AZX400 complies with industry standards SEMI and CE.

Features

Large sample analysis

Up to 400 mm (diameter)

Up to 50 mm (thickness)

Up to 30 kg (mass)

Sample adapter system

Adaptable to various sample sizes

Measurement spot

30 mm to 0.5 mm diameter

5-step automatic selection

Mapping capability

Allows multipoint measurements

Sample view camera (option)

General purpose

Analyze Be - U

Elemental range: ppm to %

Thickness range: sub Å to mm

Diffraction interference rejection (option)

Accurate results for single-crystal substrates

Compliance with industry standards

SEMI, CE marking

Small footprint

50% footprint of the previous model

Product(s):

(Hold down CTRL keyto select multiple items)

Name (first/last):

Company:

E-mail:

Phone:

Ext:

Street address:

City:

State/Province/Region:

(use two-letter abbreviations for US)

Zip/Postal Code:

Country:

Additional comments
(no URLs allowed)

Occasionally we would like to send you news, updates, and special promotions about our products and services. By checking the “I Agree” box and submitting your email address, you are giving us your express consent to send you these types of emails and you are also agreeing to Our Privacy Notice. (click Our Privacy Notice link to review it) and Our Terms & Conditions. (click Our Terms link to review it) I agree