Ultrasonic flow meters FLOMIC FL5034

Solution for water leaks detection

The battery-powered FLOMIC FL5034 ultrasonic flow meterare intended for operational and invoicingmeasurementsof instantaneousflow rate, pressureandconsumptionin water piping in observance of standard EN 14154 in water works, water distribution systems and other industrial applications.

The features of the FLOMICflow metersmake them suitable forwater-leak detectionand tracking and monitoring of the general condition ofwater-supply networks.

Ultrasonic flow meter FL 50X4 design

Theseflow metersare of a compact design with theIP 68protection class. They are characterised by high measurementaccuracyandlong-term stabilityover a wide range of measured values. The meters do not need any external power supply, do not include any moving parts and have significantly lower hydraulic losses.

Themeasurement methodutilised by these meters is a single-beam transit-timepulse method based on evaluation of the time needed for an ultrasonic signal to cross the distance between two measuring transductors. In the standard version, the meter measuresinstantaneous flow rate(in m3/hour) and thetotal volume of waterpassed through the meter (in m3) in the given flow direction.

The measured data are converted intopassive pulse outputsignals. The meter output signals can be connected viadata transfersystems to remote computer control stations. Flowmeter electronics also includesdata storagewith various possibility to connect (GSM module, RS232, optical data probe). This feature can be usefull for distance measuring points without electricity.

Flow meters FLOMIC FL 50X4 can effectively save customers costs everywhere, where needs to measure water and is not possible or is expensive to connect to emergy power grid.

Technical parameters

usage

measurement of total volume of the water passed through and instantaneous flow rate and pressure