Software and Firmware Updates

Software/Firmware revisions are made available at no charge through this web page so that all users can benefit from the latest performance enhancements. The easy upgrade procedure takes only a few minutes to perform.

BioFlo® 510 Updates

Software revision 02.02.0006 incorporates the following solutions/additions to improve the performance of your bioprocess system:• TY-35B Overlay Value was always off (Close) on sterilization mode. This was resolved to allow TY-35B to open when the temperature falls below 121°C

BioFlo® 610 Updates

Software revision 02.03 incorporates the following solutions/additions to improve the performance of your bioprocess controller:• On the sterilize screen, ‘Sparge’ and ‘Overlay’ toggle buttons were changed to be visible to the operator

CelliGen® BLU (1st Generation) Updates

NOTE: This update is specific to 1st generation CelliGen BLU cabinets. Please check the model of your CelliGen BLU cabinet. If your CelliGen BLU model number begins with the prefix "M1363", please follow the steps on this page to complete your update. If your CelliGen BLU model number begins with the prefix "M1374", please see the update CelliGen BLU (2nd Generation)

CelliGen® BLU (2nd Generation) Updates

NOTE: This update is specific to 2nd generation CelliGen BLU cabinets. Please check the model of your CelliGen BLU cabinet. If your CelliGen BLU model number begins with the prefix "M1374", please follow the steps on this page to complete your update. If your CelliGen BLU model number begins with the prefix "M1363", please see the update for CelliGen BLU (1st Generation).

BioCommand® SFI Updates

The updated version of BioCommand SFI allows for the correct interperation of values when used with European operating systems while continuing to report values on US operating systems. This improved version also allows BioCommand SFI to be used with up to 2 each New Brunswick S41i CO2 Incubator Shakers. It trends and archieves critical data including temperature, speed and gas concentration but does not provide batch control.

BioCommand® Track and Trend Updates

NOTE: This update is required for any current customer planning to use BioCommand witha BioFlo® 320. The enhancements made to are specific to the TCP/Modbus protocol used in theBioFlo® 320. There is no change to the functionality of BioCommand for all other New Brunswick™ bioprocess equipment.

NOTE: It is recommended that all customers update their software, however, customers running Revision C BioCommand software packages with all other New Brunswick™ bioprocess equipmentare not required to install the update.

Revision D includes the following enhancements:• Communication over TCP/Modbus protocol• Addition of proportional and integral (P&I) values to programming section• Addition of control mode selection to programming section• Addition of P&I, deadband, and totalizer values to Batch Summary, Datalog, and Reports

BioCommand® Batch Control Updates

NOTE: This update is required for any current customer planning to use BioCommand witha BioFlo® 320. The enhancements made to are specific to the TCP/Modbus protocol used in theBioFlo® 320. There is no change to the functionality of BioCommand for all other New Brunswick™ bioprocess equipment.

NOTE: It is recommended that all customers update their software, however, customers running Revision C BioCommand software packages with all other New Brunswick™ bioprocess equipmentare not required to install the update.

Revision D includes the following enhancements:• Communication over TCP/Modbus protocol• Addition of proportional and integral (P&I) values to programming section• Addition of control mode selection to programming section• Addition of P&I, deadband, and totalizer values to Batch Summary, Datalog, and Reports

BioCommand® Plus Software Updates

NOTE: This update is only for BioCommand Plus 3.30 customers. All other versions of BioCommand Software are not compatible with this software update.

To determine which version software you currently have installed:

Open your BioCommand software program on your PC.

The first screen that comes up identifies both the name (ie. BioCommand Plus) and version number of your software. If you have BioCommand Plus 3.3, proceed with the following update.