Simultaneously measure and display up to 12 different measurement parameters

Capture inrush current, and voltage surge with the peak function

Harmonic measurements to the 50th order

Integration function with automatic range switching

Ability to measure electric energy which is produced or consumed

Pre-compliance testing according to IEC/EN62000-3-2 / 4-7

Standard USB, GPIB, RS232 and LAN interfaces

Line and frequency filter capability for reducing unwanted signal noise

Optional universal breakout box to simplify connection between power meter and DUT

Three-year warranty

Rich Measurement Functions The 5335B is a single-phase AC / DC power meter designed for measuring and analyzing power consumption as well as power quality parameters. The 5335B quickly and accurately measures voltage and current up to 600 Vrms / 20 Arms with 0.1% accuracy and a bandwidth up to of 100 kHz, as well as power factor, frequency, and phase. Additionally, the meter features a powerful integration function, the ability to perform harmonic measurements to the 50th order and an oscilloscope mode.

Universal Breakout BoxThe optional TLBB53 breakout box simplifies AC line connection between the power meter and the DUT, and eliminates the need to cut the power cord and strip wires to connect to the power meter. This breakout box supports easy plug in connection and uses a universal socket to support most plugs used worldwide. A circuit breaker/ switch is also provided for additional protection.

Intuitive User Interface The large 4.3 inch color LCD screen enables easy viewing of configuration and measurements. Use the dedicated function keys to select one of the 4 main measurement modes: meter, harmonic, integral or oscilloscope. The results are displayed in numeric and graphical format. Screenshots can be saved directly to a USB flash drive. The meter is also equipped with standard USB, GPIB, RS232 and LAN interfaces for remote control, data logging and data analysis.

Motor Test

Many industrial products use PWM as a speed control method. The 5335B is able to measure input signals ranging from 0.5 Hz – 100 kHz and input voltages up to 600 V.