Cesar shielded the RF cable and it made a big improvement. Was able to run Pgm 3 for 20 minutes without failures. The foreline pressure varied up to about 270 mT without causing any problems and then decreased to about 145 after a few minutes. Thanks Cesar and Elmer.
An etch rate test on Tox gave 366A/min which is close to normal.