Angle Beam Transducers

ndtXducer® angle beam transducers and wedges allow the use of a complete set of wedges with
different refracted angles for weld inspection or flaw detection, where flaws have angular orientation with
respect to the material surface.

AT-style Angle Beam Transducers and Wedges

AT-style is designed for quick change option for transducers and wedges and for limited access areas or thin
material testing.

Knurled stainless steel case with top Microdot connector

Top Microdot connector

Thread size compatible with many other brands

Custom modifications available

Frequency MHz

Element Diameter

Delay us

Focal Length

Part Number

35

6mm

10

25mm

HM35-25

35

6mm

10

12mm

HM35-12

50

6mm

10

12mm

HM50-12

75

6mm

10

12mm

HM75-12

100

3mm

10

8mm

HM100-8

150

2mm

10

8mm

HM150-8

200

1mm

10

8mm

HM200-8

150

2mm

10

3mm

HM150-3

200

1mm

10

3mm

HM200-3

200

1mm

10

2mm

HM200-2

AS and AW-style Angle Beam Transducers and Wedges

AS-style or standard angle beam transducers and wedges are designed to introduce refracted shear wave from longitudinal wave into a test material by the use of proper wedge.

AW-style transducers and wedges confirm to AWS Code Section D1.1 and they are offered with snail or serrated wedges. Transducers and wedges use industry accepted hole spacing.

Captive screwes included with every transducer.

Top BNC connector is standard.

Custom modifications available.

Frequency MHz

Element Diameter

Delay us

Focal Length

Part Number

35

6mm

10

25mm

HM35-25

35

6mm

10

12mm

HM35-12

50

6mm

10

12mm

HM50-12

75

6mm

10

12mm

HM75-12

100

3mm

10

8mm

HM100-8

150

2mm

10

8mm

HM150-8

200

1mm

10

8mm

HM200-8

150

2mm

10

3mm

HM150-3

200

1mm

10

3mm

HM200-3

200

1mm

10

2mm

HM200-2

AM-style Angle Beam Transducers

AM-style or miniature angle beam transducers are designed to introduce refracted shear wave from longitudinal wave into a test material by the use of proper wedge. Their main application is inspection of welds for flows with suitable orientation. THey work with most commercially available flaw detectors.

Small dimensions make these an ideal choice for limited access applications.