Stability Chamber

"VS511" Series are Stability Chambers designed and manufactured in accordance with ICH guidelines for stability testing. They have a superior accuracy and uniformity of temperature and humidity. Used for Conditioning of samples, Accelerated shelf life studies, Stability testing for drugs and pharma industries, Climate conditioning, etc.

It works on CFC free refrigerant and hermetically sealed compressor. Digitally controlled evaporative humidifier, RH Microcontroller with PID algorithm and advanced RH sensor gives precise humid conditions inside the chamber. Steam injection system has a stainless steel boiler and reservoir. It has a clear view glass inner door. Motorised blower develops air recirculation system which ensures uniform temperature and humidity inside the chamber. Poly-Urethane is used for efficient insulation.

Technical Specifications

Standard Model :

Interior SS 304 with mirror polish. Exterior MS. With epoxy coating.

GMP Model :

Interior SS 316 with mirror polish. Exterior SS 304 with matt finish.

Temp. Range / Accuracy :

20°C to 60°C / ± 0.2°C

Temp. Uniformity :

± 2°C

Temp. Accuracy :

5°C above ambient to 250°C

Humidity Range / Accuracy :

40% RH to 95% RH / ± 2%RH

Humidity Uniformity :

± 3% RH

Temp. Control :

Auto tuned temperature microcontroller with PID algorithm.

Humidity Control :

Advanced RH micro controller with PID algorithm and %RH display.

Heating :

"U" shaped SS tubular air heater.

Refrigerant :

R134a; CFC free refrigerant.

Air Flow :

Forced air re-circulation by means of a motorised blower.

Motor :

1/12 HP, TEFC, F - Class Insulation.

Insulation :

PUF ( Poly Urethane Foam )

Models Standard (s) or GMP (G)

Models

Capacity Litres

Internal dim. H.W.D (cm)

External dim. H.W.D (cm)

Shelves No.s

VS 511 S/G

100

50 x 50 x 40

142 x 66 x 72

2

VS 512 S/G

170

70 x 60 x 40

162 x 76 x 72

3

VS 513 S/G

225

75 x 60 x 50

167 x 76 x 82

3

VS 514 S/G

325

90 x 60 x 60

182 x 76 x 87

4

VS 515 S/G

400

110 x 60 x 60

202 x 76 x 87

5

VS 516 S/G

800

125 x 80 x 80

249 x 96 x 107

6

VS 517 S/G

1000

157 x 80 x 80

249 x 96 x 107

7

resources

Tune your Thermostatic equipment when there is a considerable change in working load or set temperature.It increases efficiencyand saves power.