Real-time polymerase chain reaction detection of parvovirus B19 DNA in blood donations using a commercial and an in-house assay.

MedLine Citation:

PMID:
21175647
Owner:
NLM
Status:
Publisher

Abstract/OtherAbstract:

BACKGROUND: European regulations require testing of manufacturing plasma for parvovirus B19 (B19) DNA to limit the load of this virus to a maximum acceptable level of 10 IU/µL. To meet this requirement, most manufacturers introduced a test algorithm to identify and eliminate high-load donations before making large manufacturing pools of plasma units. Sanquin screens all donations using a commercial assay from Roche and an in-house assay. STUDY DESIGN AND METHODS: Between 2006 and 2009, 6.2 million donations were screened using two different polymerase chain reaction (PCR) assays targeting B19 DNA. Donations with B19 DNA loads of greater than 1 × 10(6) IU/mL showing significant differences in viral load between the two assays were further analyzed by sequencing analysis. RESULTS: A total of 396 donations with B19 DNA loads of greater than 1 × 10(6) IU/mL were identified. Fifteen samples (3.8%) had discordant test results; 10 samples (2.5%) were underquantified by the Roche assay, two samples (0.5%) were underquantified by the in-house assay, and three samples (0.8%) were not detected by the Roche assay. Sequencing analysis revealed mismatches in primer and probe-binding regions. Phylogenetic analysis showed that 12 samples were B19 Genotype 1. The three samples not detected by the Roche assay were B19 Genotype 2. CONCLUSION: This study shows that 3.8% of the viremic B19 DNA-positive donations are not quantified correctly by the Roche or in-house B19 DNA assays. B19 Genotype 1 isolates showing incorrect test results are more common than B19 Genotype 2 or 3 isolates. Newly designed B19 PCR assays for blood screening should preferably have multiplexed formats targeting multiple regions of the B19 genome.