A number of applications like high precision ranging and imaging, LIDAR, micro-materials processing, UV spectroscopy in chemistry and biochemistry will benefit from energetic (300-500uJ) high beam quality sub-nanosecond (~200ps) laser pulses. To address the needs of these applications the new MOPA microlasers from STANDA utilize Master Oscillator Power Amplifier platform that combines company's renowned STA-01 microlaser oscillators together with a highly efficient amplifying stage developed recently. Due to the MOPA configuration the unique spatial and temporal characteristics of the microlasers are preserved while their output power is raised to the multiwatt level. All Master Oscillators as well as Power Amplifiers are produced at Standa facilities.

We value user working space and always try to fit into the tightest environment of a laboratory set up, measuring device or other equipment. The MOPA series continues the traditions of STA-01 microlasers: being the smallest passively Q-switched DPSS laser on the market without compromise to optical performance. Fiber-coupled pump diodes are housed inside the electronic controller greatly reducing heat load and footprint of the laser head. Pump radiation is delivered via 2m long metal-shielded fiber optic cables enabling convenient integration of the laser system into the user equipment. Due to the high pump and output powers the optimized air cooling or water cooling of laser head is required.

Without compromising their compactness and beam quality the STA-01-MOPA-TH STA‑01‑MOPA series lasers can be equipped with second and third harmonic generation crystals for nonlinear frequency conversion. STANDA can offer green and UV lasers with 532 and 355 nm wavelengths accordingly.

Due to the MOPA configuration different modifications of output parameters are possible. Contact us directly or find your local Standa representative if you have laser specification other than given in the table of standard units.

Applications

• Marking

• Electronics (PCB Processing)

• Biomedical

• Instrumentation

• LIDAR and Laser Ranging

• Sensing and Monitoring

• Micro dissection

• Semiconductor Inspection

• Material processing

• MALDI-TOF

• TiSa, OPO pumping

• Laser Induced Fluorescence (LIF)

• Spectroscopy (LIBS, LIPS and PIV)

• Micromachining

• Nonlinear Optics (Harmonics and Supercontinuum Generation)

• Diamond drilling (SiC) and other hard materials

Models

STA-01-MOPA-1

STA-01-MOPA-2

STA-01-MOPA-3

STA-01-MOPA-4

STA-01-MOPA-SH-1

STA-01-MOPA-SH-2

STA-01-MOPA-SH-3

STA-01-MOPA-SH-4

STA-01-MOPA-TH-1

STA-01-MOPA-TH-2

STA-01-MOPA-TH-3

STA-01-MOPA-TH-4

Wavelength, nm

1064

532

355

Average Output Power (max), W

3

2.5

0.2

5

1.5

1.2

0.100

2.5

0.75

0.7

0.050

1.250

Pulse Energy, µJ

300

50

200

100

150

10

100

50

75

5

50

25

Pulse Width (FWHM), ns

0.8

0.4

0.2

0.5

0.7

0.4

0.2

0.5

0.5

0.4

0.2

0.7

Repetition Rate (max), kHz

10

50

1

50

10

50

1

50

10

50

1

50

Peak Power, MW

up to 1.2

Beam Divergence (1/e2, full angle), mrad

< 6

Pulse Spectral Structure

SLM

Spatial Mode

TEM00

Beam Propagation Factor M2

< 1.2

Polarization Ratio (linear)

> 100 : 1

Power Stability RMS, %

<1%

Operating Voltage, V AC

100 – 230

Power Consumption, W

<100

Operating Temperature, °C

20 – 35

Interfaces

USB, External Trigger (TTL rising edge)

Dimensions of Laser Head, mm

79 (W) x 37.5 (H) x 75 (L)

Dimensions of Controller, mm

471 (W) x 157 (H) x 458 (L)

Warranty

12 months

Laser Head Cooling Required:

12WSC1 Cooling system

Thermal Resistance, K/W

≤0.05

Laser Controller Cooling

Provided

Required: 12WSC1 Cooling system

Provided

Required: 12WSC1 Cooling system

Provided

Required: 12WSC1 Cooling system

Thermal Resistance, K/W ≤

0.05

0.05

0.05

Pulse duration of a particular model is factory set to a fixed value from the given range. Contact your Standa representative or Standa directly for more details.Content and specification are subjects to change without notice.