Logic Analyzer Required

The FS224x is designed to run from 0 to 800 MHz with a minimum data valid window of 200 ps. The minimum input voltage required is 400 mv differential.

Signal Loading

When using the FS2240, each target signal is to be isolated from the Robinson Nugent connector by a 266 ohm resistor. The FS2243 and FS2244 include the isolation network in the probing system.

Capture of Transactions

All transactions can be captured except at the time of master clock alignment. During this time 4 bit times worth of data are not acquired in state mode. However, this data can be viewed in Timing Zoom.

Environmental Temperature

Operating: +20 ° to +30 ° C (+68 ° to +86 °F)

Non-operating: -40 ° to +75 ° C (-40 ° to +167 ° F)

Altitude

Operating: 4,600 m (15,000 ft)

Non-operating: 15,300 m (50,000 ft)

Humidity

Up to 90% non-condensing. Avoid sudden, extreme temperature changes that would cause condensation on the Analysis Probe.