TIVA 900/1000

The largest Tiva Washer-Disinfector is designed to meet the increasing processing needs of a CSSD. Fully automatic sliding doors allow for manual or automatic loading and unloading. The two vertically installed washing pumps enable consistent and efficient washer disinfecting performance and a complete drain of the washing circuit. A HEPA H14 air filter is used for injecting clean air into all the chamber parts at different basket levels and into injectors.

15 DIN Baskets : 500 liter chamber

Automatic Single/Double Door

HEPA-filtered forced air drying

LCD display (optional Touch Screen)

40 Programs

3 Chemical dosing pumps

FEATURES

Performance

Washing disinfection temperature fully adjustable up to 93°C

Temperature controlled by means of two independent PT1000 probes

Fresh water intake between each phase of the cycle promotes hygiene and better cleaning

Three chemical dosing pumps with the ability to adjust detergent consumption for each cycle (level control). • Additional dosing pump is optional.

Automatic vertical sliding doors are positioned at a height for convenient loading and unloading using manual or automatic loading and unloading trolleys

Door window made of HST tempered glass

Self-cleaning chamber with rounded edges

Optional

Integrated printer

4th chemical dosing pump

Steam condenser prevents steam from entering the washing area

Turbo drying allows hot air recirculation for shorter cycles

Color touch screen with 65 washing and disinfection programs

CLEANING AND DISINFECTIONThermal DisinfectionThermal disinfection takes place at 93°C with a temperature holding time in compliance with EN ISO 15883. Both temperature and time can be modified to meet the specific customer requirements.

Designed to Stay CleanThe washing chamber is designed with smooth edges and corners to ensure a clean chamber surface and prevent dirty breeding grounds.

LONG LASTING AND SAFEWater QualityBuilt-in large capacity water softeners enhance cleaning and prevents limescale build-up. Connections to cold, warm and DI water are available, with a triple water filter system that captures residues so that they are not re-circulated, thus lengthening pump life