10 to 55 Hz, Double amplitude 1.5 mm 0.06", 2 hours in each of the X, Y, and Z directions

Weight

Approx. 63 g

*1 The above data was obtained using an aluminum target (A5052 t = 1 mm 0.04).When measuring steel or stainless steel targets, refer to the characteristics of linearity for these materials.*2 The above data was obtained using an aluminum target (A5052 t = 1 mm 0.04"). When measuring steel or stainless steel targets, refer to the characteristics of linearity for these materials.*3 The above data was obtained using an aluminum target (A5052 t = 1 mm 0.04). When measuring steel or stainless steel targets, refer to the characteristics of linearity for these materials.*4 NPN output can easily be converted to PNP output by connecting the optional OP-5148 PNP Output Converter.*5 When the distance between the sensor head and the target is 50 % of the measuring range and the operating temperature is 10 to 40 °C (50 to 104 °F) (EX-008) or 0 to 50 °C (32 to 122 °F) (EX-016 andEX-022).