The profilometer of the thicknessCONTROL.TTP series accurately measures the thickness profile of tire tread or side walls during the extrusion process and enable therefore an efficient control during production as well as a fully recorded quality control.

Benefits:

Optimizing the material use and quality

Fast controlling during warm up or product change due to adaptive traversing speed and high-dynamically measurement

No consequential costs due to isotopes or X-rays

Wear-free measurements instead of half time of radiometric tests

Easy integration to already existing lines

Customer-specific adjustable interface

Material parameters:

Material width to 2,000mm

Strip width from 5mm to 550mm

Up to 6 single strips

Material feed up to 60m/min

The systems of the TTP family are equipped with optical sensors and have an application specific operators´ frontend which provides all necessary data regarding the organization of the extrusion process of running surfaces and side walls. Systems of the RTP series can be applied as low-cost solutions.

The TCP 8301.I is based on the light intersection method. On each material side a line is projected crosswise on the measuring object via a laser light source. The profile is detected by a camera. The geometrical data are calculated from the distance of the cameras and the signals from the lower and upper material side. A long-term and temperature stability is achieved by an automated in-situ calibration.

Special features:

Accuracy thickness 60µm

Accuracy width 0.3mm

40 profiles per second

Downloads

The TCP 8301.ET is based on laser triangulation sensors and therefore ensures the measurement being independently from material constants. The system is designed as O-frame, both sensors detect the thickness according to the differential method. In order to ensure highest precision, the laser beams of the sensors are accurately adjusted during the production process using optoelectronic tools which have been specially developed. The system is ideal for monitoring high-dynamically processes due to its high measuring rate. The triangulation sensor applied on the upper side is combined with an eddy current sensor which monitors the distance of the two optical measuring sensors while allowing a software-technical compensation of temperature-induced changes of the measuring gap. Additionally, the compensation sensor is applied according to a special alignment with the measuring task in order to ensure both, a huge measuring gap and highest precision. The TCP 8301.ET is similarly to the TIP8301.ET regarding its functionality. However, they differ concerning the application- or process specific user interface and the corresponding analyzes software.

Special features:

Accuracy thickness ±10µm

Accuracy width 0.2mm

Measuring rate to 20kHz

Downloads

The TCP 8301.CT is based on laser triangulation sensors and therefore ensures the measurement being independently from material constants. The system is designed as O-frame, both sensors detect the thickness according to the differential method. In order to ensure highest precision, the laser beams of the sensors are accurately adjusted during the production process using optoelectronic tools which have been specially developed. The system is ideal for monitoring high-dynamically processes due to its high measuring rate. The O-frame mechanics of the system dispose of a compensation frame which does not vary with temperature and which is used in order to avoid temperature fluctuations of the measuring frame using further sensors. The TCP 8301.CT is similarly to the TIP8301.CT regarding its functionality. However, they differ concerning the application- or process specific user interface and the corresponding analyzes software.

Special features:

Linearity thickness ±5µm

Accuracy width 0.3mm

Measuring rate to 20kHz

Downloads

The RTP 8302.T is based on the laser triangulation sensors and therefore ensures the measurement being independently from material constants. In order to ensure highest precision, the laser beams of the sensors are accurately adjusted during the production process using optoelectronic tools which have been specially developed. The system is ideal for monitoring high-dynamically processes due to its high measuring rate. The C-frame mechanics of the system disposes of an automated in-situ calibration in order to avoid interferences caused by temperature.

Special features:

Linearity from ±5µm

Measuring ranges up to 20mm

Immersion depth from 50° to 1,000mm

Special sizes on request

Measuring rate to 20kHz

Downloads

The RTP 8302.LLT is based on the laser line triangulation sensors and therefore ensures the measurement being independently from surface characteristics of the target material. Combined to point sensors, the system offers more precision in the case of gap width. The C-frame mechanics of the system disposes of an automated in-situ calibration in order to avoid interferences caused by temperature.