Business Sector

Test Types

Specimen

Standard

Traditional testing of stent and stent grafts (to standards such as ASTM F2477) involves subjecting complete devices to pressure pulsation simulating blood flow for a pre-specified number of cycles. Even if the device has not fractured, a simple pass/fail criterion is used for the test to succeed. Industry work is being undertaken to attempt to develop improved in vitro methods for characterizing and evaluating cardiovascular implant devices to provide more information about when and under what conditions fracture will take place.

To enable a representative sample of specimens to be evaluated and to reduce overall test time, multiple samples must be tested on a single system.

The multi-specimen fixture for the ElectroPuls™ E3000 All-Electric Test Instrument was developed to assist cardiovascular implant manufacturers to assess this long-term fatigue characteristics of nickel-titanium (Nitinol) and stainless-steel stent materials and structures.

Each specimen station features a fatigue-rated load cell, precision alignment adjustment and grips that are unique to the material or structure undergoing test. The entire station assembly is submerged into a temperature controlled bath to test the specimens in vitro. The test is controlled by WaveMatrix™ Dynamic Test Software that includes a number of features designed to assist users in running the test. These include integrated temperature control of the fluid bath, a live display of each load cell reading, and trend monitoring of forces to determine each specimen fracture.

The ElectroPulsTM E3000 is a state-of-the-art, all-electric test instrument designed for dynamic
and static testing on a wide range of materials and components. It includes Instron® advanced
digital control electronics, DynacellTM load cell, Console software, and the very latest in testing
technology–hassle-free tuning based on specimen stiffness, electrically operated crosshead lifts,
a T-slot table for flexible test set ups and a host of other user-orientated features. Powered from
a single-phase supply it requires no additional utilities for basic machine operation (for example,
pneumatic air, hydraulics, or water).

ElectroPuls instruments are state-of-the-art electrodynamic test systems for dynamic and static tests on materials & components. Powered entirely from a single-phase mains supply, they feature the latest in testing technology and a host of other user-oriented features.

WaveMatrix™ is intelligent software designed for fatigue and dynamic testing of materials and components. It delivers flexibility - run everything from a simple static ramp, to cyclic waveforms through to complex multi-step, multi-axial tests
The highly visual environment with integrated tabular screens, clear menu structures, time-based matrix test preview and
configurable live test workspace is designed to be intuitive and instill confidence. Packed with intelligent features, such as data
reduction and built-in project organization, designed to simplify your testing.

Please choose a subscription optionOne or more required fields are empty, complete the form to continue.One or more fields contain formatting errors, fix the errors to continue.You must agree to Instron's Privacy Policy in order to continue.

“It is our objective at Instron to provide our customers the best ownership experience by delivering the highest quality products, expert support and world-class service.”