About Memmert

In over 190 countries all over the world, hundreds of thousands of Memmert products have been permanently in use for decades. This is why Memmert is one of the leading suppliers of temperature control appliances worldwide.

It is our hard-lined commitment to quality that has made Memmert a reliable partner in the medical sector and medical research industry ever since the company’s founding in 1947. Of course we are DIN EN ISO 13 485 certified since many years. This medical applications special introduces suitable appliances as well as case studies.

It is our passion for details that determines the outstanding quality of our temperature control appliances. And due our unconditional focus on the requirements of the customers Memmert appliances have been regulars in the labs of the pharmaceutical industry and life sciences for decades now. You’ll find corresponding case studies and appliances on this page.

We are dedicated to good taste. In order to ensure the unrivalled reliability and precision of Memmert incubators, ovens and climate chambers we develop and manufacture all vital components ourselves. Read more on Memmert’s solutions for the food and beverage industry on our special application page.

Case Studies

Measuring the depth of carbonation in concrete

The standard DIN EN 206 assumes concrete structural components to have a service life of at least 50 years. Corrosion of the steel reinforcement in ferroconcrete caused by its exposure to CO2 (carbonation) can shorten the service life. Holcim HüttenZement GmbH is using a Memmert ICH C climate chamber with CO2 control to find out which characteristics concrete needs to have to minimise carbonation.

Climate chamber ICH750

Standard delivery Climate chamber

On this page, you can find all the essential technical data on the Memmert climate chamber ICH. Our customer relations team will be pleased to help if you want further information. If you should require a customised special solution, please contact our technical specialists at myAtmoSAFE@memmert.com.

Temperature

Working temperature range

from -10°C up to +60°C without humidity
(not suitable for long-term storing at sub-zero temperatures. During permanent operation, the glass door may ice over)

Working temperature range

from +10°C up to +60°C with humidity

Setting accuracy temperature

0.1 °C

Temperature

2 Pt100 sensors DIN Class A in 4-wire-circuit for mutual monitoring, taking over functions in case of an error

Humidity

Humidity

humidity supply with distilled water from external tank by self-priming pump

additionally integrated over- and undertemperature protection "ASF", automatically following the setpoint value at a preset tolerance range, alarm in case of over- or undertemperature, heating is switched off in case of overtemperature, compressor in case of undertemperature

Autodiagnostic system

integral fault diagnostics for temperature and humidity control

Alarm

visual and acoustic

Heating concept

Air jacket

air jacket heating system for gentle all-around heating

Heating concept

no drying-up of the load caused by the cooling device due to separation of thermal jacket from interior

Cooling

energy-saving use of CFC-free cooling/heating system (refrigerant R134a)

Defrosting

highly efficient automatic defrosting system

Standard equipment

Entry port

Entry port (silicone), 40 mm clear diameter, moisture tight, can be closed
by a silicone stopper, standard position at the back

Textured stainless steel casing

Electrical data

Voltage
Electrical load

230 V, 50/60 Hz approx. 1350 W

Voltage
Electrical load

115 V, 50/60 Hz approx. 1350 W

Ambient conditions

Set Up

The distance between the wall and the rear of the appliance must be at least 15 cm. The clearance from the ceiling must not be less than 20 cm and the side clearance from walls or nearby appliances must not be less than 5 cm.

Ambient temperature

15 °C to 28 °C (up to 34 °C with limited temperature
range)

Humidity rh

max. 70 %, non-condensing

Altitude of installation

max. 2,000 m above sea level

Overvoltage category

II

Pollution degree

2

Packing/shipping data

Transport information

The appliances must be transported upright

Customs tariff number

8419 8998

Country of origin

Federal Republic of Germany

WEEE-Reg.-No.

DE 66812464

Dimensions approx incl. carton

w x h x d: 1330 x 2150 x 1050 mm

Net weight

approx. 249 kg

Gross weight carton

approx. 319 kg

Standard units are safety-approved and bear the test marks

Temperature-humidity working range ICH

Not all climate chambers are the same. The humidity content of the chamber load, the ambient conditions and the respective temperature-humidity working range are decisive factors in the selection of the right appliance. In the adjacent diagram, you can see the possible temperature/humidity combinations for our climate chamber ICH.

Within the respective temperature-humidity range, condensation-free permanent operation is possible. To which extent condensation may occur in the threshold range depends on the humidity content of the chamber load and the ambient conditions.

This seems not to be the appropriate device for your temperature-humidity-working range?See the other Memmert climate chambers.

About us

Memmert is an innovation leader in the development of heating ovens, incubators, climate chambers as well as waterbaths and oilbaths for a wide range of applications in biological, pharmaceutical and food research as well as industrial quality testing.

Each Memmert appliance is 100% AtmoSAFE

The main objective of the family company has always been: 100% AtmoSAFE. All parameters are controlled with the utmost precision to ensure a controlled atmosphere in our temperature control appliances.