The WaveMaster® IOL 2 with single lens holder for measurement of IOLs in-air

Based on wavefront analysis WaveMaster® IOL 2 measures key parameters of refractive monofocal and toric, spherical or aspherical intraocular lenses using the WaveSensor®, TRIOPTICS' well established Shack-Hartmann sensor. Thus, it is used in research and development as well as in production. Measurement parameters like the refractive power, aberrations and modulation transfer function are calculated by analyzing the wavefront which is obtained in a quick measurement process. Additionally, fully automated measurement of toric lenses (MTF in both meridians, marker recognition) is supported. All aspects of the measurements in-air or in-situ are arranged in a new intutuitively usable interface that guides the user through the process.

Key Features

Multi-purpose measurement system for quality control of all common refractive intraocular lenses: monofocal, toric, spherical and aspherical

Wavefront measurement permits complete power mapping

Power measurement with accuracies as low as one third of the allowed tolerances stated in ISO provide confidence in target achievement.

Measurement of hydrophobic and hydrophilic lenses in air or in the optionally heatable in-situ eye model as per ISO 11979