This machine is applicable to test tensile stress, compressive stress, shearing force, peeling force, and elongation of materials, components, and finished product for metal, plastic, rubber, textile, synthetic chemicals, wire and cable, leather and so on..
1. Use Taiwan TECO servo motor which ensures high efficiency, steady transmission and low noise. The accuracy of speed can be controlled in 2%.
2. The structure is made of the paint-coated aluminum blanking plate. The interior is used the high-accuracy, low-resistance and zero clearance of ball screw and oriented pole which improve loading efficiency and structure rigidity.
3. Using the brand computer as the main control machine, the special testing software can conduct all of testing parameter, work state, collecting data and analysis, result display and printing output.

1. Introduction:
It is suitable for doing testing experiments such as the tensile, compression, bending and shear testing for all kinds of metal materials and nonmetal materials such as plastic,concrete, cement and so on.

Beam impact testers

Model Charpy izod impact JB-300
Brand name: Walter

Charpy izod impact test machine (JB-300)

Introduction:

This Test Instrument is used to determine the impact toughness of metal materials under dynamic load. The pendulum of the machine can be raised or released automatically. It is designed according to national standard GB/T3808 “Charpy Impact Test Method for Metals”
It is also accordance with international standard ISO148, EN10045 and ASTM E23. It has the features of easy operation, high efficiency, safe and reliable. The machine is especially suitable for laboratory, metallurgy industry, machinery production, steel plant and other fields.

Portable tension gauge

Model HF-10
Brand name: Walter

Portable tension gauge (HF-10)

Introduction:
The portable tension gauge has advantages of high precision, easy operation and carry. What's more, it has kinds of testing modes, convenience for conner and it can store and input data to computer for statistical analysis as well as constitutes testers with different functions, combinating with test-beds and clamp. Users can input local acceleration of gravity value to make test more precise.

Specifications:
1. High precision and high resolution.
2. Five testing modes and three display methods for selection
3. N (Newton), kg (Kilogram) and lb (Pound) three measuring units for selection and conversion.
4.Setting function of gravity acceleration
5. Function of peak value maintaining.
6.Automatic peak function.
7.Max and min and comparison value can be set for statistic analysis.
8. Data storage function. 896 testing values can be stored.
9.Data output function. The data can be input into computer through data line for various analyses.
10. Environmental protection.
11. High quality chargeable power supply.
12. 6 digits large screen display.

Torque tester

Model W-1, 2, 5, 10, 20
Brand name: Walter

Digital torque tester (W-1, 2, 5, 10, 20)

Introduction:
Torque Tester is designed to measure the open force and lock force of bottle caps, suction and flexible tube packages. Great stability and precision of Torque Tester makes it a necessity for manufacturing process.

Brinell hardness testers

Model THB-3000
Brand name: Walter

Brinell Hardness Tester (THB-3000)

Inroduction:
The Brinell hardness tester is most commonly used to test materials that have a structure that is too coarse or that have a surface that is too rough to be tested using another test method, e.g., castings and forgings.

Introduction:
Digital Brinell hardness tester is computerized, has high rate of optical measurement and photoelectric sensors system. The operating process and the results are indicated on large LCD screen, the test results can be printed out.

Rockwell hardness tester

Model THR-150A
Brand name: Walter

Manual Rockwell Hardness Tester (THR-150A)

Inroduction:
This tester is a popular type Rockwell hardness testing machine, with its novel appearance and eased operate, can be applied in the workshop of factories and measurement department, for determining the Rockwell hardness of ferrous and non-ferrous metals.

The hardness of rubber and plastic is most commonly measured by the shore Durometer which using either the Shore A or Shore D. Shore A is used for softer rubbers, while the Shore D scale is used for harder ones.

The hardness gauge conforms to the regulations of GB/T531, ISO7619 and ISO 868.

Technical Specifications:

Parameters

Shore A

Shore D

The diameter of pressure needle

1.25mm±0.15 mm

1.25mm±0.15 mm

The diameter/ of the head of pressure needle

0.79mm±0.03 mm

R0.1mm±0.01 mm

Inclination of the taper of pressure needle

35°±0.25°

30°±1°

Stroke of pressure needle

2.5mm±0.04mm

2.5mm±0.04mm

The scope of the dial

0HA-100HA

0HD-100HD

Weight,net(gross)

approx.250g(300g)

approx.250g(300g)

Dimensions (L×W×H)

26×62×115mm

26×62×115mm

Portable hardness tester

Model W-20
Brand name: Walter

Portable Webster Hardness Tester (W-20)

Inroduction:
Webster Hardness Tester is a portable instrument which can perform on-site hardness testing on soft materials. By a quick and easy test, the hardness value can be read out directly with a simple clamp, in accordance with American Standard ASTM B647. A test can be completed by one press. Very fast, simple and portable, small indentation. Suitable for testing the hardness of aluminium profiles, tubings, sheets and other shapes. Used for testing the with long or heavy work pieces or assemblies inconvenient to be sent to the labs.

The Webster hardness value can be converted into the common-used Vickers, Rockwell and Brinell hardness value.
Different models available for testing aluminum alloys, brass, copper alloy and copper.