5017D, 500mW - 500W Avg, 1300W Peak Wideband Power Sensor never requires field calibration, only requires calibration once per year and is traceable to National Institute of Standards and Technology (NIST). The WPS measures True Average Power, Peak Power, and Duty Cycle directly with exceptional accuracy and uses these precise measurements to calculate a wide range of other important factors, such as VSWR, Return Loss, Reflection Coefficient, Crest Factor, Average Burst Power, and CCDF.

500 mW - 500 Watts Avg.1300 Watts Peak (Derate maximum average power rating from 500W at 300MHz to 100W at 1GHZ using a straight line on a log-log scale)

Insertion VSWR

<1.05

Insertion Loss

<0.05 dB

Directivity

28 dB up to 100 MHz, 30 dB from 100 to 1000 MHz

Average Power

Average Forward Power Range: 500 mW - 500 Watts Avg, 1300 Watts Peak (Derate maximum average power rating from 500W at 300MHz to 100W at 1GHZ using a straight line on a log-log scale) *Accuracy, Average Forward Power: ± 4% of reading, + 0.17 WMinimum Forward Power for Reflected Measurement: .5Return Loss: 0.0 to 23 dBVSWR: 1.15 to 99.9Rho: 0.07 to 1.0