Full Stainless Steel Steam Sterilization Model:HA-BD Sterilizing Volume:150L,200L,280L Features: 1. The steam can't go through to chamber when the door haven't close, and the pilot off inform. You can processing dry operate. 2. When the pressure inner over 0, 027MPa. The door be locked and can't be open. 3. If you not choose the sterile time and start the machine, buzzer sound. 4. If you mistake open the door in the course of start, the jacket value auto off and open the chamber value. 5. Lack water the pilot on, heating and sterile pilot off, the buzzer sound and auto release steam. 6. Overload and leakage protection, when leakage or over current occur, it auto off the power. 7. Temperature controlled not work and the inner pressure over 0.165MPa, the safety value auto open.

Cylindrical Sterilizer Machine Model:HA-BC Sterilizing Volume:150L,200L,280L,400L,500L Features: 1.Sterilizing course automatic sterilization controlled,easy to perate. 2.With drying function,suitabledressing drying. 3.With over-temperature,over-pressure auto-protect device. 4.The door opening mechanism can't be perated until the pressure in the chanber is reduced to 0.027Mpa.It can't be start on if the door doesn't close well. 5.The safety valve will be open automatically when the inner pressure over0.24Mpa and the steam be exhausting to the water tank. 6.The power will be cut-off automatically,cut-off the water and alarm when the machine lacing water. 7.The chamber of the sterilizer is made of stainless steel.

Steam Sterilization Model:HA-BA Steam Sterilization Volume:150L,200L,280L Features: 1. The steam can't go through to chamber when the door haven't close, and the pilot off inform. You can processing dry operat. 2. When the pressure inner over 0, 027MPa. The door be locked and can't be open. 3. If you not choose the sterile time and star the machine, buzzer sound. 4. If you mistake open the door in the course of start, the jacker value auto off and open the chamber value. 5. Lack water the pilot on, heating and sterile pilot off, the buzzer sound and auto release steam. 6. Overload and leakage protection, when leakage or over current occur, it auto off the power. 7. Temperature controlled not work and the inner pressure over 0.165MPa, the safety value auto open.

Steam Pressure Sterilizer Model:HA-BB Steam Pressure Sterilizer Volume:150L,200L,280L Features: 1. Sterilizing course automatic sterilization controlled, easy to operate. 2. Autoclave horizontal With drying function, suitable dressing drying. 3. Horizontal Cylindrical Pressure Steam Sterilizer With over-temperature, over-pressure auto-protect device. 4. The door opening mechanism cannot be operated until the pressure in the chamber isreduced to 0.027 Mpa. It can't be start on if the door doesn't close well. 5. The safety valve will be open automatically when the inner pressure over 0.24 MPa andthe steam be exhausting to the water link. 6. The power will be cut-off automatically, cut-off the water and alarm when the machinelacking water. 7. Fully stainless steel structure. 8. With printer.

Table Top Steam Sterilizer Application Range: The sterilizer is a table type in safety and auto-controlled, which is designed for the unit of medical health, pharmacy, science and technology etc. It is suitable for sterilizing of surgical, dental and eye instruments, glassware medicine, culture medium and biological dressing, food and other goods. Technical Features: 1.Indicator light indicates working state; 2.Maximum temperature 134°C, suitable for 4-6 minutes flash sterilizing; 3.With over temperature & over pressure auto-protection; 4.Safe protection of water lacking; 5.After sterilization it will automatically cut the power and then alarm; 6.Fully stainless steel structure;

Type:TS-A20, TS-A24 Use: The sterilizer is a table type in safety and auto-controlled, which is designed for the unit of medical health, pharmacy, science and techmology etc. And mostly suitable for sterilizing of surgical, dental and eye instruments, glassware medicine, culturemedium and biological dressing, food and other goods. Characteristics: 1. Standard Class B with three times vacuum and drying, the remaining temperature of sterilized instrument is less than 0.2%. 2. LCD display working status, touch keys. 3. The vacuum data can reach -0.8bar, it fits various instrument, including package, unpackaged, solid, hollow,multi-apertures and inset pipes. It makes sure the instrument can be sterilized completely. 4. Fully computer controlled LCD display bright show. The interface is modularized panel for easy operation. 5. It is installed with BOWIE&DICK which measures the penetration of water steam. 6. It has vacuum text which can make sure the ability of texting vacuuming. 7. Fast and independent steam generator. 8. Built-in mini printer which can record the sterilizer information. 9. Safety valve,Safety lock door system,Pressure or temperature overload protection, Alarm system.

Application Range: The sterilizer is a table type in safety and auto-controlled, which is designed for the unit of medical health, pharmacy, science and technology etc. It is suitable for sterilizing of surgical, dental and eye instruments, glassware medicine, culture medium and biological dressing, food and other goods. Technical Features: 1.Indicator light indicates working state; 2.For 4-6 minutes rapidly sterilizing; 3.Sterilizing temperature and time can be preset; 4.Steam-water inner circulation system, no steam discharge and the environment for sterilizing will be clean and dry; 5.Over temperature & over pressure auto-protection; 6.Safe protection of water lacking; 7.Exhaust the cool air automatically; 8.Door safety lock system; 9.With three stainless steel sterilizing plates; 10.The chamber of the sterilizer is made of stainless steel; 11.Automatically shut off with beep reminding after sterilization;

Application Range: The sterilizer is a table type in safety and auto-controlled, which is designed for the unit of medical health, pharmacy, science and technology etc. It is suitable for sterilizing of surgical, dental and eye instruments, glassware medicine, culture medium and biological dressing, food and other goods. Technical Features: 1.Indicator light indicates working state; 2.For 4-6 minutes rapidly sterilizing; 3.Sterilizing temperature and time can be preset; 4.Steam-water inner circulation system, no steam discharge and the environment for sterilizing will be clean and dry; 5.Over temperature & over pressure auto-protection; 6.Safe protection of water lacking; 7.Exhaust the cool air automatically; 8.Door safety lock system; 9.With four stainless steel sterilizing plates; 10.The chamber of the sterilizer is made of stainless steel; 11.Automatically shut off with beep reminding after sterilization;

Table Top Steam Sterilizer Application Range: The sterilizer is a table type in safety and auto-controlled, which is designed for the unit of medical health, pharmacy, science and technology etc. It is suitable for sterilizing of surgical, dental and eye instruments, glassware medicine, culture medium and biological dressing, food and other goods. Technical Features: 1.Indicator light indicates working state; 2.For 4-6 minutes rapidly sterilizing; 3.Sterilizing temperature and time can be preset; 4.Steam-water inner circulation system, no steam discharge and the environment for sterilizing will be clean and dry; 5.Over temperature & over pressure auto-protection; 6.Safe protection of water lacking; 7.Exhaust the cool air automatically; 8.Door safety lock system; 9.With three stainless steel sterilizing plates; 10.Fully stainless steel structure; 11.Automatically shut off with beep reminding after sterilization;

Application Range: The sterilizer is a table type in safety and auto-controlled, which is designed for the unit of medical health, pharmacy, science and technology etc. It is suitable for sterilizing of surgical, dental and eye instruments, glassware medicine, culture medium and biological dressing, food and other goods. Technical Features: 1.Standard Class B with three vacuum and drying, the remaining temperature of sterilized instrument is less than 0.2%; 2.LCD display working status, touch keys; 3.The vacuum data can reach -0.8bar, it fits various instrument, including package, unpackaged, solid, hollow, multi-apertures and inset pipes. It makes sure the instrument can be sterilized completely; 4.Fully computer controlled LCD display bright show. The interface is modularized panel for easy operation; 5.It is installed with BOWIE&DICK which measures the penetration of water steam; 6.It has vacuum test which can make sure the ability of vacuum testing; 7.Fast and independent steam generator; 8.Built-in mini printer which can record the sterilization information; 9.Equipped with safety valve, safety lock door system, pressure lock system, pressure or temperature overload protection and alarm system; 10.It inactivates the HIV, HBV, BSE and bacillus efficiently; 11.With drying function;

Table Top Steam Sterilizer Application Range: The sterilizer is a table type in safety and auto-controlled, which is designed for the unit of medical health, pharmacy, science and technology etc. It is suitable for sterilizing of surgical, dental and eye instruments, glassware medicine, culture medium and biological dressing, food and other goods. Technical Features: 1.Indicator light indicates working state; 2.For 4-6 minutes rapidly sterilizing; 3.Sterilizing temperature and time can be preset; 4.Steam-water inner circulation system, no steam discharge and the environment for sterilizing will be clean and dry; 5.Over temperature & over pressure auto-protection; 6.Safe protection of water lacking; 7.Exhaust the cool air automatically; 8.Door safety lock system; 9.With three stainless steel sterilizing plates; 10.Fully stainless steel structure; 11.Automatically shut off with beep reminding after sterilization;

Portable Pressure Steam Sterilizer Application Range: The product uses saturated steam to rapidly and efficiently sterilize articles. It can be used in hospitals, public health centers, health stations, clinics in factories and mines, scientific research institutes, etc to sterilize medical equipment, surgical dressings, glass utensils, solutions, substrata. It can also be used as cooker in plateau areas and be used to make high quality drinking water in industrial and mining enterprises. Technical Features: 1.Fully stainless steel structure; 2.Double scale indication pressure gauge; 3.Electric heated; 4.Easy to operate, safe and reliable;

Table Top Steam Sterilizer Application Range: The sterilizer is a table type in safety and auto-controlled, which is designed for the unit of medical health, pharmacy, science and technology etc. It is suitable for sterilizing of surgical, dental and eye instruments, glassware medicine, culture medium and biological dressing, food and other goods. Technical Features: 1.Indicator light indicates working state; 2.Maximum temperature 134°C, suitable for 4-6 minutes flash sterilizing; 3.With over temperature & over pressure auto-protection; 4.Safe protection of water lacking; 5.After sterilization it will automatically cut the power and then alarm; 6.Fully stainless steel structure;

Application Range: The sterilizer is a table type in safety and auto-controlled, which is designed for the unit of medical health, pharmacy, science and technology etc. It is suitable for sterilizing of surgical, dental and eye instruments, glassware medicine, culture medium and biological dressing, food and other goods. Technical Features: 1.Indicator light indicates working state; 2.For 4-6 minutes rapidly sterilizing; 3.Sterilizing temperature and time can be preset; 4.Steam-water inner circulation system, no steam discharge and the environment for sterilizing will be clean and dry; 5.Over temperature & over pressure auto-protection; 6.Safe protection of water lacking; 7.Exhaust the cool air automatically; 8.Door safety lock system; 9.With three stainless steel sterilizing plates; 10.The chamber of the sterilizer is made of stainless steel; 11.Automatically shut off with beep reminding after sterilization;