•Pc controller measurement can be achieved for more accurate and flexible requirement

Application

* Medicine/Pharmaceutical Industry

* Environmental Monitoring

* Commodity Inspection

* Food Inspection

* Agricultural Chemistry

* Teaching in Colleges & Universities

* Metallurgy

* Geology

* Machine Manufacturing

* Petrochemical Industries

* Water and Waste Water Labs

* Food and Beverage Labs

Standard Configuration

* Glass Cell :4 nos.

* Quartz cells : 2 nos.

* Instruments Cover : 1 no.

* Software CD : 1 no.

* USB Cable : 1 no.

* Operational Manual : 1no.

* Software Manual :1 no.

* Software key : 1no.

DOUBLE BEAM UV-VIS Spectrophotometer with more accuracy and flexible requirements. The two detectors are used to measure sample and reference respectively and simultaneously for optimizing measurement accuracy. It has wide wavelength range satisfying requirement of various fields, such as biochemical research and industry, pharmaceuticals analysis and production, education, environment protection, food industry etc.

Basic Mode

To measure the Absorbance and transmittance

Quantitative

1. Coefficient Method

2. Standard Curve Up to 10 Standard sample may be used to establish a curve. Four methods for fitting a curve through the calibration points : Linear fit. Linear fit through zero, Square fit and cubic fit.

DNA/Protein Test

Concentration and DNA purity are quickly and easily calculated: Absorbance

2. Hi, Medium and low scan speed are available. They vary from 100 to 3600 nm/min

3. Wavelength are scanned from high to low so that the instrument waits at high WL. And it minimizes the degradation of UV sensitive samples.

Kinetics

This mode may be used for time course scanning or reaction rate calculations.

Abs v/s time graphs is displayed on the screen in real time, Wait time and measurement time up to 12 hours may be entered with time interval of 0.5,1,2,5,10,30 seconds and one min. Post-run manipulation includes re-scaling, curve tracking and selection of the part of the curve required for rate calculation. Rate is calculated using a linear regression algorithm before multiplying be the entered factor.