Contains six cuffs of different sizesName: Automatic Tourniquet System

Place of Origin: China

Brand: Y&Z.OEM.ODM

Size: 330*310*405mm

Weight: 5.6kg

The maximum noise: ≤80dB

Color: White

Model No.: YTQ-F

Cuff:with six cuffs,12”*2,18”*2, 34”*2

Application: mainly used for auxiliary oppression hemostasis in
hospital for surgery

Package: Master export carton

MOQ: 1 piece

Payment terms: 30% deposit by TT,the balance paid upon the copy of
B/L.

Description:

The ATS Tourniquet system is an automatic microprocessor-based
pneumatic and output stability pressure medical appliance.
Self-testing at boot start, for certain types of equipment
malfunction, the unit will also display error messages that
identify the cause of error, thus reducing fault isolation time.
Multiple Audible and visual alarms will alert user of abnormal
conditions. The machine also has ability to independently inflate
the gas or also can at the same time. The “IVRA” function suitable
for orthopaedic operation. Built-in recharge the lion-battery,
which can assures machine still on work even if a sudden power
outage (lasting more than 4H with full load). Large, bright four
sets LED displays, which respectively displaying the pressure and
time of master and second gas channel. The machine will alarm when
there just have 10mins,5min, 2min left, besides the alarm time can
also be setup. The apparatus can alarm in time after setup, the
mostly we can set 4 group alarm time. The alarm sound has 8 levels,
which also can adjust . The set data can be stored more than three
months, the data including the pressure, time, sound level of power
off. The memory lasting time will be a little variation depends on
inside battery or other conditions.

Applications:

The automatic Tourniquet system is an automatic
microprocessor-based pneumatic and output stability pressure
medical appliance. The cuff inflated, and the compression acting on
the patient's limb and blockade bleeding of limbs, to reduce
bleeding.

The automatic Tourniquet system has function of pressure setting
and pressure stabilization, instead of manual

Specifications:

Maximum Pressure

90KPa (650mmHg)

Time Set

0—99 minutes

Power Supply

110v, 50HZ /220,60HZ

Power of Whole Machine

No more than 25 W

Range of Pressure

1KPa—90KPa

Pressure Accuracy

±5 mmHg pressure set

Time Alarm set Ranges

2-240 min

Time Accuracy

0.5±1 sec

Li-ion battery

14.8V,≥1000mAh

Normal working environment temperature range

5℃~35℃

The normal work of relative humidity

≤80%

Transportation and storage environment temperature

5℃~45℃

Transportation and storage of relative humidity

≤80%

Competitive Advantage:

1. Accurate monitoring of tourniquet time.

2. High convenience for user and patient.

3. Visual and audible alarms when an unsafe condition is detected.

4. Various materials, colors, sizes are available.

5. Low noise and long lifespan.

6. Adopting imported transducer to calculate and equalize the
pressure.