$300

$700

GW Instek is launching to the market a new series of safety tester with high durability and multiple user-protection design. The GPT-9800 Series look to meet a variety of safety standards such as IEC, EN, UL, CSA, GB, JIS and other safety-related requirements for the safety tests of electronic products and components.

A total of 4 safety testers in the GPT-9800 Series, namely GPT-9804, GPT-9803, GPT-9802 and GPT-9801, are available for various applications. The GPT-9804 is a 4-in-1 model capable of performing AC withstanding, DC withstanding, insulation resistance and ground bond tests. The GPT-9803 is a 3-in-1 model capable of performing AC withstanding, DC withstanding as well as insulation resistance tests. The GPT-9802 is capable of performing both AC and DC withstanding tests, whereas the GPT-9801 is able to perform AC withstanding test. The series of safety tester is built upon a high-efficiency PWM amplifier platform with AC 200VA maximum output capacity to impede the influence from the voltage fluctuation of the input AC source. This ensures a stable voltage supply for all the tests of the GPT-9800 Series.

200VA AC Test Capacity

240x64 Ice Blue Dot matrix LCD

Manual/Auto Mode

Function Key for quick selecting

High intensity flash for Caution & Status indication

Safety INTERLOCK function

Zero Crossing Turn-on Operation

Controllable Ramp-up Time

True RMS Current Measurement

High resolution: 1uA for measuring current, 2V for setting voltage

PWM amplifier to enhance the power efficiency and reliable testing

Max. 100 memory block for test condition (step) setting. And each step can be named individually.

Remote terminal on the front panel for “start” and “stop” control

Interface : RS-232C, USB device, Signal I/O and GPIB (optional)

AC WITHSTANDING

Output-Voltage Range

0.100kV~ 5.000kV ac

Output-Voltage Resolution

2V/step

Output-Voltage Accuracy

± (1% of setting + 5V) [no load]

Maximum Rated Load

200 VA (5kV/40mA)

Maximum Rated Current

40mA (0.5kV< V ≦5kV)

10mA (0.1kV≦ V ≦0.5kV)

Output-Voltage Waveform

Sine wave

Output-Voltage Frequency

50 Hz / 60 Hz selectable

Voltage Regulation

± (1% of reading + 5V) [full load → no load]

Voltmeter Accuracy

± (1% of reading + 5V)

Current Measurement Range

0.001mA~40.0mA

Current Best Resolution

0.001mA / 0.01mA /0.1mA

Current Measurement Accuracy

±(1.5% of reading + 30uA)

Window Comparator Method

Yes

ARC Detect

Yes

RAMP (Ramp-Up Time)

0.1s~999.9s

TIMER (Test Time)

OFF, 0.5s~999.9s

GND

ON/OFF selectable

DC WITHSTANDING (GPT-9804/9803/9802)

Output-Voltage Range

0.100kV~ 6.000kV dc

Output-Voltage Resolution

2V/step

Output-Voltage Accuracy

± (1% of setting + 5V) [no load]

Maximum Rated Load

50W (5kV/10mA)

Maximum Rated Current

10mA (0.5kV< V ≦6kV)

2mA (0.1kV≦ V ≦0.5kV)

Voltage Regulation

± (1% of reading + 5V) [full load → no load]

Voltmeter Accuracy

± (1% of reading + 5V)

Current Measurement Range

0.001mA~ 10.0mA

Current Best Resolution

0.001mA /0.01mA /0.1mA

Current Measurement Accuracy

±(1.5% of reading + 30uA)

Window Comparator Method

Yes

ARC Detect

Yes

RAMP (Ramp-Up Time)

0.1s~999.9s

TIMER (Test Time)

OFF, 0.5s~999.9s

GND

ON/OFF selectable

INSULATION RESISTANCE (GPT-9804/9803)

Output Voltage

50V~1000V dc

Output-Voltage Resolution

50V/step

Output-Voltage Accuracy

± (1% of setting + 5V)

Resistance Measurement Range

1MW~ 9500MW

Test Voltage

50V≦V <500V

1~50MΩ：±(5% of reading + 1 MΩ)

51~2000MΩ：±(10% of reading + 1 MΩ)

500V≦ V ≦1000V

1~500MΩ：±(5% of reading + 1 MΩ)

501~9500MΩ：±(10% of reading + 1 MΩ)

Window Comparator Method

Yes

RAMP (Ramp-Up Time)

0.1s~999.9s

TIMER (Test Time)

1s~999.9s

GND

OFF (fixed)

Ground Bond (GPT-9804)

Output-Current

03.00A~30.00A ac

Output-Current Resolution

0.01A

Output-Current Accuracy

3A≦I≦8A：±(1% of reading + 0.2A)

8A30A：±(1% of reading + 0.05A)

Test-Voltage

6Vac max (open circuit)

Test-Voltage Frequency

50Hz/60Hz selectable

Resistance Measurement Range

0.1mW~ 650.0mW

Resistance Measurement Resolution

0.1mW

Do you have a AC 200VA AC/DC Withstanding Voltage/Insulation Resistance/Ground Bond Tester that needs to be calibrated?