The Chatillon® K-DFE Series is ideal for functional capacity evaluations. This compact, easy-to-use force gauge is features a load averaging function ideal for job task analysis and functional capacity evaluations. Averaging can be based on a load setpoint with hold time more ...

Features

The CHATILLON® K-DFE Series is ideal for functional capacity evaluations and job task analysis. This compact, easy-to-use force gauge has a measurement accuracy of better than 0.25% full scale. A large, easy-to-read, high resolution dot matrix LCD display supports a variety of measurements including functional capacity (Load Averaging) based on a force or a time duration. The gauge also displays normal and peak readings, pass/fail results and statistics based on up to 10 stored results. Measurements are displayed in ozf, gf, lbf, kgf and N units. The display can be inverted and displayed results may be “hidden” during testing. The K-DFE gauge comes standard with carrying case, battery adapter/charger and NIST Certificate of Calibration with data. A variety of testing fixtures are supplied standard. The K-DFE gauge has a standard RS232 digital output for exporting data to a personal computer using our TCD WEDGE application software.

Features

Advanced Operating Modes

- Functional Capacity (Load Averaging)

- Force-Based

- Time Duration-Based

- Normal

- Peak Strength

- Average Strength

- Strength Limits

- Pass-Fail Limits

Statistical Calculations

- Mean with Maximum and Minimum Values

- Coefficient of Variation with Mean and Standard Deviation

- Standard Deviation with Variance and Mean

- % Differentiation

Integral Loadcells

- Accuracy +0.25% Full Scale

- Mechanical Overload Protection to 150% Full Scale

Simple Operator Interface

- High Resolution Dot-Matrix

- Menus with Prompts for Easy Use

- Dedicated and Function Keys with Navigation Pod

NIST Calibration with Data

- Available IEC/ISO17025 Cert with Uncertainty

2 Year Warranty

Notice
The K-DFE digital force gauge is a general purpose force measurement instrument. It is not intended for applications requiring Class 2 designation by the United States Food & Drug Administration or by Health Canada