Orthopedic Surgical Table

Model: DST-700 Name: Orthopedic Surgical Table Introduction The superb YUDA DST-700 series operating table system is the result of systematic improvements to offer the best support for the patient and surgical team in OR room. YUDA offers unique X-ray capabilities and extreme articulations for...

Introduction
The superb YUDA DST-700 series operating table system is the result of systematic improvements to offer the best support for the patient and surgical team in OR room. YUDA offers unique X-ray capabilities and extreme articulations for perfect and ergonomic access.
With its versatile table tops, the DST-700 series can be use across all surgical disciplines

Features:
1.Extreme Load Capacity
Maximum capacity 430 kg / 950 lbs patient weight.
2.Longitudinal Shift for X-ray Compatibility
Motorized longitudinal shift of up to 350 mm (14 in) and the use of carbon-fiber modules provide excellent C-arm access.
3.Ergonomic Comfort
The height adjustment from 660 to 1000 mm (25.9 to 39.3 in) permits ergonomic working positions, both seated and standing.
4.Easy Click
The Easy Click mounting point allows the quick use of a broad range of socket-mounted modules .
5.Control Units
The operating table can be controlled in different ways: corded hand control, remote control, For emergency situations, there is an override panel on the table column. The color display can be used to store individual operating table positions that can be activated at any time. Status reports, instructional notes and fault messages are shown on the display.
6.One reset-function as CPR . After surgery finished, just touch CPR function, it gives a time saving to use.
7.Gynecology and Urology
With the addition of a variety of knee crutches, leg holders or leg plates at the mounting points as required, The seat plate extension is employed for urology interventions.
8.Neurosurgery
In addition to the prone and supine positions, special accessories and devices put the patient in a seated position for complete access during cranial surgery.

Technical data

Bed Panel Size(full carbon fiber material)

(2020 x 535) ± 10mm

Bed height from the ground(Without pads)

Min height: (695 ± 20) mmMax height: (995 ± 20) mm

Lifting up and down (electric)

(300 ± 20) mm

Horizontal move route(electric)

(330 ± 20) mm

Head and foot tilt (electric)

Head tilt: 30° ± 2 °

Foot tilt: 30° ± 2 °

Lateral tilt(electric)

Left: 25° ± 2 °

Right: 25° ± 2 °

Back plate fold(electric)

Upward: 65 ° ± 2 °

Downward: 20 ° ± 2 °

Leg plate fold(manual)

Upward: 35 ° ± 5 °

Downward: 90 ° ± 5 °

Head section fold(manual)

Upward: 45 ° ± 5 °

Downward: 90 ° ± 5 °

One key reset function

Electric

battery

One week (80-100 times)

Table panel X-ray absorption coefficient

≤ 0.44mmAL

Bed lock

Electric

Power supply

AC: 220V 50Hz 300VA

DC: 24V (Panasonic lead-acid batteries)

Net weight /capacity weight

300kg(661lbs) /430kg( 950lbs)

About my tables
Germany Engineer unique design
Core parts from Germany
International leading technology
Suitable for human body design Services:Complete customer orders as required on time. In the service of the product, we have a special after-sales service team to relieve the worries of the customers.