Open Data supplied by Natural Environment Research Council (NERC)

Sea Bird Electronics SBE13 Dissolved Oxygen Sensor

The SBE 13 was designed as an auxiliary sensor for Sea Bird SBE 9plus, but can fitted in custom instrumentation applications. When used with the SBE 9 Underwater Unit, a flow-through plenum improves the data quality, as the pumping water over the sensor membrane reduces the errors caused by oxygen depletion during the periods of slow or intermittent flushing and also reduces exposure to biofouling.

The output voltage is proportional to membrane current (oxygen current) and to the sensor element's membrane temperature (oxygen temperature), which is used for internal temperature compensation.

Two versions of the SBE 13 are available: the SBE 13Y uses a YSI polarographic element with replaceable membranes to provide in situ measurements up to 2000 m depth and the SBE 13B uses a Beckman polarographic element to provide in situ measurements up to 10500 m depth, depending on the sensor casing. This sensor includes a replaceable sealed electrolyte membrane cartridge.

The SBE 13 instrument has been out of production since 2001 and has been superseded by the SBE 43.

Sea-Bird Electronics SBE 911 and SBE 917 series CTD profilers

The SBE 911 and SBE 917 series of conductivity-temperature-depth (CTD) units are used to collect hydrographic profiles, including temperature, conductivity and pressure as standard. Each profiler consists of an underwater unit and deck unit or SEARAM. Auxiliary sensors, such as fluorometers, dissolved oxygen sensors and transmissometers, and carousel water samplers are commonly added to the underwater unit.

Underwater unit

The CTD underwater unit (SBE 9 or SBE 9 plus ) comprises a protective cage (usually with a carousel water sampler), including a main pressure housing containing power supplies, acquisition electronics, telemetry circuitry, and a suite of modular sensors. The original SBE 9 incorporated Sea-Bird's standard modular SBE 3 temperature sensor and SBE 4 conductivity sensor, and a Paroscientific Digiquartz pressure sensor. The conductivity cell was connected to a pump-fed plastic tubing circuit that could include auxiliary sensors. Each SBE 9 unit was custom built to individual specification. The SBE 9 was replaced in 1997 by an off-the-shelf version, termed the SBE 9 plus , that incorporated the SBE 3 plus (or SBE 3P) temperature sensor, SBE 4C conductivity sensor and a Paroscientific Digiquartz pressure sensor. Sensors could be connected to a pump-fed plastic tubing circuit or stand-alone.

Temperature, conductivity and pressure sensors

The conductivity, temperature, and pressure sensors supplied with Sea-Bird CTD systems have outputs in the form of variable frequencies, which are measured using high-speed parallel counters. The resulting count totals are converted to numeric representations of the original frequencies, which bear a direct relationship to temperature, conductivity or pressure. Sampling frequencies for these sensors are typically set at 24 Hz.

The temperature sensing element is a glass-coated thermistor bead, pressure-protected inside a stainless steel tube, while the conductivity sensing element is a cylindrical, flow-through, borosilicate glass cell with three internal platinum electrodes. Thermistor resistance or conductivity cell resistance, respectively, is the controlling element in an optimized Wien Bridge oscillator circuit, which produces a frequency output that can be converted to a temperature or conductivity reading. These sensors are available with depth ratings of 6800 m (aluminium housing) or 10500 m (titanium housing). The Paroscientific Digiquartz pressure sensor comprises a quartz crystal resonator that responds to pressure-induced stress, and temperature is measured for thermal compensation of the calculated pressure.

Additional sensors

Optional sensors for dissolved oxygen, pH, light transmission, fluorescence and others do not require the very high levels of resolution needed in the primary CTD channels, nor do these sensors generally offer variable frequency outputs. Accordingly, signals from the auxiliary sensors are acquired using a conventional voltage-input multiplexed A/D converter (optional). Some Sea-Bird CTDs use a strain gauge pressure sensor (Senso-Metrics) in which case their pressure output data is in the same form as that from the auxiliary sensors as described above.

Deck unit or SEARAM

Each underwater unit is connected to a power supply and data logging system: the SBE 11 (or SBE 11 plus ) deck unit allows real-time interfacing between the deck and the underwater unit via a conductive wire, while the submersible SBE 17 (or SBE 17 plus ) SEARAM plugs directly into the underwater unit and data are downloaded on recovery of the CTD. The combination of SBE 9 and SBE 17 or SBE 11 are termed SBE 917 or SBE 911, respectively, while the combinations of SBE 9 plus and SBE 17 plus or SBE 11 plus are termed SBE 917 plus or SBE 911 plus .

Fixed Station Information

Station Name

Ellett Line/Extended Ellett Line Station E

Category

Offshore location

Latitude

57° 31.98' N

Longitude

12° 37.98' W

Water depth below MSL

1658.0 m

Ellett Line/Extended Ellett Line: Station E

Station E is one of the fixed CTD stations, which together form The Extended Ellett Line. The line lies between Iceland and the Sound of Mull (Scotland) crossing the Iceland Basin and Rockall Trough via the outcrop of Rockall. As part of this initiative, CTD dips, together with associated discrete sampling of the water column, have typically been carried out annually at this station since September 1996.

Prior to September 1996, Station E was part of a shorter repeated survey section, consisting of 35 fixed stations, known as The Ellett Line (originally termed the Anton Dohrn Seamount Section). This line incorporated those stations across the Rockall Trough and Scottish shelf between Rockall and the Sound of Mull and was visited at regular intervals (usually at least once a year) between 1975 and January 1996.

Fixed Station Information

Station Name

Ellett Line

Category

Offshore route/traverse

Ellett Line

The Ellett Line is a hydrographic transect consisting of 35 individual fixed stations which were occupied, usually at least once a year, between 1975 and 1996. The time series is named after the scientist David Ellett, who coordinated the survey work at Dunstaffnage Marine Laboratory (DML), near Oban. The transect ran between the north west coast of Scotland to the small outcrop of Rockall, via the Anton Dohrn Seamount - a prominent bathymetric feature in the Rockall Trough (see map). STD/CTD dips and associated water sampling for the analysis of nutrients were routinely performed during each station occupation.

In 1996 the transect was lengthened to incorporate new additional fixed stations crossing the Iceland Basin from Rockall to Iceland. This transect, which is still routinely occupied annually, is now known as the Extended Ellett Line and is a collaborative effort between scientists at Dunstaffnage Marine Laboratory and the Southampton site of the National Oceanography Centre (NOC).

Map of standard stations (1975-1996)

Map produced using the GEBCO Digital Atlas

The white triangles indicate the nominal positions of the Ellett Line stations (1975- 1996). Measurements made along the Ellett Line lie within a box bounded by co-ordinates 56° 40.02' N, 13° 42.0' W at the south west corner and 57° 37.2' N, 6° 7.98' W at the north east corner.

Nominal Ellett Line stations (1975-1996)

Listed below are nominal details of the standard hydrographic stations that formed the Ellett Line between 1975 and January 1996.