Electronet series ULT-200 is a Two Wire Ultrasonic Level Transmitters specially designed for non-contact type level measurement.

Ultrasonic level measurement is based on the time required by ultrasonic pulses to travel from the sensor & surface of liquid & back to the sensor. Ultrasonic level transmitters are the best for applications such as slurries, corrosive liquids & wastewater. The transmitter provides 4-20 mA DC continuous output with local indication through LCD. The measuring range is available from 0.6 to 8 mtr.

Additional Information:

Item Code: ULT200

Yes! I am Interested

View PDF

Get CALLBACK

ULT 200B : Battery Operated Ultrasonic Level Transmitter

Ask For Price

Product Price: Get Latest Price

Product Details:

Brand

ELECTRONET

Response Time

Less than 500 mSec

Model

ULT 200B

Measuring Range

0.6 to 6 mtr

Process Connection

1) 2 inch BSP (M) Threaded 2) Other on request

Accuracy

+/- 0.25% of F. S.

lectronet series ULT-200B is a Battery Operated Ultrasonic Level Transmitters specially designed for non-contact type level measurement. Ultrasonic level measurement is based on the time required by ultrasonic pulses to travel from the sensor and surface of liquid and back to the sensor. Ultrasonic level transmitters are the best for applications such as slurries, corrosive liquids, and wastewater without obstacle throughout the beam. The transmitter provides a local indication through LCD.

The measuring range is available from 0.6 to 12 meters.

Additional Information:

Item Code: ULT200B

Yes! I am Interested

View PDF

Get CALLBACK

ULTTX22 : Two Wire Ultrasonic Level Transmitter

Ask For Price

Product Price: Get Latest Price

Product Details:

Measuring Range

0.6 to 6 Meter

Power Supply

24 V DC

Power Consumption

600 mW

Measuring Frequency

25 Khz

Protection

IP 66

Temperature Coefficient

+- 0.01 % Per Degree C

Electronet series ULT-TX-22 is a Two Wire Ultrasonic Level Transmitters specially designed for non-contact type level measurement.

Ultrasonic level measurement is based on the time required by ultrasonic pulses to travel from the sensor & surface of liquid & back to the sensor. Ultrasonic level transmitters are the best for applications such as slurries, corrosive liquids & wastewater. The transmitter provides 4-20 mA DC continuous output. The measuring range is available from 0.6 to 6 mtr.

Electronet series ELGWR-40 4-wire TDR-Sensor with a single rod or coaxial probe for continuous level measurement and point level detection in liquids, with analog and switching output. The innovative TDR technology enables direct, precise and highly reliable continuous level measurement as well as point level detection in almost every liquid – independent of changing process conditions (such as density, conductivity, temperature, pressure, vapour and turbulence). ELGWR-40 has almost no installation restrictions it can be mounted in small tanks, tall and narrow nozzles and it measures precisely even with difficult tank geometries or close to interfering structures. ELGWR-40 is also especially suitable in bypass chambers and stilling wells. It is suitable for all types of processing and storage applications and has an exceptional performance in liquids with low dielectric constant (i.e. low reflectivity) such as oils and hydrocarbons

Additional Information:

Item Code: ELGWR40

Yes! I am Interested

View PDF

Get CALLBACK

ELMLG100 : Magnetic Liquid Level Gauge

Ask For Price

Product Price: Get Latest Price

Product Details:

Material

SS316

Model

ELMLG-100

Brand

ELECTRONET

Power Supply

24 V DC

Probe MOC

SS316

Process connection

Half inch NPT (F) / Flange

Output

4 to 20 mA / 1 to 5 V

Electronet series Magnetic Liquid Level GaugeELMLG-100 is an instrument for indicating the fluid level.

It has a magnetic float inside of the non-magnetic chamber and an indicator outside of it. The indicator indicates the fluid level by means of the magnetic force of the magnet installed inside of the float.

The indicator is completely separated from the chamber. The level of liquid and gaseous phases can be distinguished clearly by the Bi-color. Magnetic level indicators are low-maintenance alternatives to sight glasses and other level indicators. They provide non-invasive level indication while reducing leak points and fugitive emissions.

Additional Information:

Item Code: ELMLG100

Yes! I am Interested

View PDF

Get CALLBACK

LCT01 : Capacitance Level Transmitter

Ask For Price

Product Price: Get Latest Price

Product Details:

Model

LCT-01

Color

Blue

Isolated Power Supply

24 V DC, External

Process Connection

1) 1 inch BSP Threaded 2) 1 inch ASA 150, Flanged

Accuracy

+/- 1% F. S.

Output

1) 4 to 20 mA DC, 2 Wire 2) 0 - 5 V DC

Electronet series LCT-01 is Capacitance Type Level Transmitters. Vessel wall and probe form two electrodes. If the vessel is of nonconductive material, the probe will be the double-rod type.

The distance between electrode and surface area of electrodes remain unchanged. The variable is the depth of the material being measured which represents the dielectric constant between two electrodes. Air and vacuum have relative dielectric constant as 1 and that of liquids, it is greater than 1.

The capacitance of the capacitor, therefore, depends on how much material lies between the probe and vessel wall i.e. whether the probe is covered with or free from the material. The capacitor changes with a change in the level of the material and provides corresponding 4-20 mA DC continuous output.

Additional Information:

Item Code: LCT01

Yes! I am Interested

View PDF

Get CALLBACK

Eltar20 : Through Air Radar Level Transmitter

Ask For Price

Product Price: Get Latest Price

Product Details:

Accuracy

+/- 0.1% of max. range in lab using 4-20mA current output & +/-0.25%of max. range(typically in field

Model

ELTAR-20

Maximum Pressure

3 bar

Probe MOC

S S316 (For other requirement please consult factory)

Conduit Entry

Half inch NPT X 2

Operation

Pulse Radar

Electronet series ELTAR-20 Radar Level Transmitter is high-frequency level measurement instrument with the maximum measuring distance up to 30 meters. The antenna is further optimized and the new-type microprocessor can perform a higher rate of signal analysis and processing, making the instrument available for complex measurement conditions such as reactors and solid silos.