Yokogawa GS820 Multi Channel Source Measure Unit

The GS820 is a highly accurate and highly functional 2-channel programmable DC voltage/current source that incorporates voltage/current generation and measurement functions. The GS820 is equipped with two analog channels with each channel consisting of a constant voltage source VS, a constant current source IS, a voltmeter VM, and an ammeter IM. The two source measure channels are isolated.

Features:

The output and measurement resolutions are 5.5 digits

Four-quadrant operation consisting of source operation (current source) and sink operation (current sink) is available with ranges up to 7 V and 3.2 A or 18 V and 1.2 A

Isolated 2-channel source and measurement function

Source and measurement ranges: 7 V and 3.2 A or 18 V and 1.2 A

Minute current ranges with 200-nA or 1-pA resolution

Generate arbitrary waveforms consisting of up to 100,000 points at 100-μs intervals