Syringes

Syringe Selection Guide

If a syringe is being used by hand, a manual syringe should be selected. If a syringe is installed in an AOC autosampler, then choose the appropriate syringe and volume to suit your instrument and application.

Syringe Style

Fixed Needle

Recommended for use with an autosampler because the probability of needle bending is minimal

Minimal sample carryover

Removable Needle

Better option for inexperienced users

Reduce cost if used in precipitation of salts or in situations in which it may bend easily (such as thick septum)

Available for many methods by changing needles

Gas Tight

For Luer Lock needles and fittings Specifically designed to install and secure

Luer Lock needles easily

Needle Tip Styles

Cone: Autosampler

The cone-shaped needle tip is specially developed to withstand multi injection demands and improve septum lifetime when used with an autosampler. The cone design effectively "parts" the septum during piercing instead of cutting it, as would a bevel needle.

Bevel: Manual Injection

The standard general-purpose needle tip style supplied with many syringes is a 20 ° bevel tip. It is the preferred option for manual injection when piercing the septum in exactly the same place is difficult. The bevel tip is designed for optimum septum penetration and prevention of septum coring.

Side Hole Dome

Samples are filled and dispensed through the side hole, eliminating septum plugging of the needle. Ideal for large-volume gas injections. The solid domed tip minimizes septum damage.

Plunger Style

PTFE Tipped Plunger

PTFE Tip enables gastight analysis

Suitable for both liquid and gas samples

Ideal for analysis of highly viscous liquid because it reduces the possibility of the plunger sticking due to micro particles (samples should have no effects with PTFE)

Plunger is replaceable

Syringe for AOC

Syringe for AOC-20i/20s

Syringe
Volume

PTFE Tipped
Plunger

Needle Length
(mm)

Needle Gauge

Needle OD
(mm)

Needle Tip

P/N

Replacement Plunger
P/N

Replacement Needle
P/N

5 μL

-

42

23

0.63

Cone

221-75173

-

-

10 μL

✓

42

23

0.63

Cone

221-74469

-

-

10 μL

-

42

23

0.63

Cone

221-34618

-

-

10 μL

✓

42

23

0.63

Cone

221-75174

221-75174-02

221-75174-01

50 μL

-

42

23

0.63

Cone

221-45243

-

-

250 μL

✓

42

23

0.63

Cone

221-45244

221-45244-01

-

Description

P/N

Syringe Volume

Syringe for OCI

221-37282-02

10 μL

Elastic Syringe for AOC *

221-49548

10 μL

*Plunger is made of titanium. It has less wear debris compared to metal, making it less hard and ideal for hydrous sample.

Syringe for AOC-5000 Plus

Syringe
Volume

PTFE Tipped
Plunger

Needle Length
(mm)

Needle Gauge

Needle OD
(mm)

Needle Tip

P/N

10 μL

-

50

26

0.47

Cone

221-75175

10 μL

-

50

23

0.63

Cone

221-75175-01

10 μL

-

80

26

0.47

Cone

221-75176

10 μL

-

80

23

0.63

Cone

221-75176-01

10 μL

✓

50

26

0.47

Cone

221-75175-02

10 μL

✓

50

23

0.63

Cone

221-75175-03

25 μL

✓

50

23

0.63

Cone

221-75177

1 mL

✓

50

26

0.47

Side Hole Dome

221-75178

1 mL

✓

50

23

0.63

Side Hole Dome

221-75178-01

2.5 mL

✓

50

26

0.47

Side Hole Dome

221-75179

2.5 mL

✓

50

23

0.63

Side Hole Dome

221-75179-01

Syringe for AOC-6000

Syringe
Volume

PTFE Tipped
Plunger

Needle Length
(mm)

Needle Gauge

Needle OD
(mm)

Needle Tip

P/N

1 μL

-

57

23

0.63

Cone

225-19744-01

5 μL

-

57

26

0.47

Cone

225-19744-02

10 μL

-

57

26

0.47

Cone

225-19744-03

10 μL

✓

57

26

0.47

Cone

225-19744-04

25 μL

✓

57

26

0.47

Cone

225-19744-05

50 μL

✓

57

26

0.47

Cone

225-19744-06

100 μL

✓

57

26

0.47

Cone

225-19744-07

250 μL

✓

57

26

0.47

Cone

225-19744-08

500 μ

✓

57

26

0.47

Cone

225-19744-09

1 mL

✓

57

23

0.63

Side Hole Dome

225-19744-10

2.5 mL

✓

65

23

0.63

Side Hole Dome

225-19744-11

Manual Syringe

Shimadzu manual syringes are available from 5 μL to 500 μL, and can be widely used in your laboratory work.

Fixed Needle Style

Syringe Volume

PTFE Tipped Plunger

Needle Length(mm)

Needle Gauge

Needle OD
(mm)

Needle Tip

P/N

Replacement Plunger

P/N

Qty

5 μL

-

50

26

0.47

Bevel

221-75170

-

-

10 μL

-

50

26

0.47

Bevel

670-12552-01

-

-

10 μL

✓

50

26

0.47

Bevel

221-75170-01

221-75170-02

2

25 μL

-

50

25

0.5

Bevel

670-12510-31

-

-

25 μL

✓

50

25

0.5

Bevel

221-75171

221-75171-01

1

50 μL

-

50

25

0.5

Bevel

670-12510-36

-

-

50 μL

✓

50

25

0.5

Bevel

221-75172

221-75172-03

1

100 μL

-

50

25

0.5

Bevel

670-12510-18

-

-

100 μL

✓

50

25

0.5

Bevel

221-75172-01

221-75172-04

1

250 μL

-

50

25

0.5

Bevel

670-12510-19

-

-

250 μL

✓

50

25

0.5

Bevel

221-75172-02

221-75172-05

1

500 μL

-

50

25

0.5

Bevel

670-12510-20

-

-

Removable Needle Style

Syringe Volume

PTFE Tipped Plunger

Needle Length(mm)

Needle Gauge

Needle OD
(mm)

Needle Tip

P/N

Replacement Plunger
P/N

Replacement Needle
P/N

10 μL

✓

50

26

0.47

Bevel

670-12553-21

670-12553-33

670-12510-95

25 μL

-

50

25

0.5

Bevel

670-12510-74

-

670-12510-96

50 μL

-

50

25

0.5

Bevel

670-12510-75

-

100 μL

-

50

25

0.5

Bevel

670-12510-76

-

250 μL

-

50

25

0.5

Bevel

670-12510-77

-

500 μL

-

50

25

0.5

Bevel

670-12510-78

-

Syringe for Gas Analysis

Luer Lock Needle Style

Syringe Volume

PTFE Tipped
Plunger

P/N

Replacement Plunger
P/N

1 mL

✓

221-54778-01

221-54778-11

5 mL

✓

221-54778-02

221-54778-12

10 mL

✓

221-54778-03

221-54778-13

25 mL

✓

221-54778-04

221-54778-14

50 mL

✓

221-54778-05

221-54778-15

100 mL

✓

221-54778-06

221-54778-16

Luer Lock Needle

Needle Length(mm)

Needle Gauge

Needle OD(mm)

Needle Tip

Qty

P/N

50

23

0.63

Bevel

5

221-54778-51

50

19

1.07

Bevel

5

221-54778-52

50

14

2.1

Bevel

5

221-54778-54

Syringe Valve

Diag. #

Syringe Volume

Description

Qty

P/N

①

5 mL to 2 L

Push Button Valve for Luer Lock Needle

1

221-54778-49

②

50 μL to 2 L

Push Button Valve for Luer Lock Needle and Tip Syringe

1

221-54778-50

Removable Needle Style

Gastight syringes with a capacity of milliliter quantity level. The needle is 30mm long and is of a screw-in type, which ensures an easy exchange.

Further dimensions and specifications on request. For Research Use Only. Not for use in diagnostic procedures. This page may contain references to products that are not available in your country. Please contact us to check the availability of these products in your country.