-HPT700 Series DP transducer/transmitter is a high reliability,
stability, and accuracy DP transmitters. It is---widely used in the
field of liquid, gas pressure test, water, oil, and mildly
corrosive liquids measurement.The product material is 1Cr18Ni9Ti stainless steel, imported
oil-filled pressure sensor, dedicated amplifier and V / I
conversion circuit transmitter, the specific anti-high voltage,
high current, high magnetic field, polar lead and other
characteristics.---- HPT700 Series DP transducer/transmitter are piezoresistive
differential pressure sensor as sensing element. Silicon oil is
filled in between die and two diaphragms, when measured
differential pressure is added on two diaphragms the pressure could
be transferred onto die through silicon oil. Sensor die connects
with amplifier circuit through wires, using semi-conductor’s
piezoresistive effect, transforming differential pressure signal
into electric signal. The whole product is used for differential
pressure measurement of petroleum, chemi-industry, power station
and hydrology, etc.-

-

-

Model: Parameter:

HPT700-H Series

Pressure Range:

0-5kPa...... 3500kPa Optional

Overload:

150%F.S.

Static Pressure

21,25 Bar (Optional)

Accuracy:

(Linearity Hysteresis

Repeatability)

≤±0.1%F.S-

≤±0.25%F.S

≤ ±0.5%F.S

Optional

Stability:

Standard: 0.1%F.S,--- Max:0.2%F.S

Working Temp:

-40°C~85°C

Storage Temp:

-40°C~125°C

Temperature Compensation:

-20°C~70°C

Medium compatible:

Compatible with 316L Stainless Steel

Electronic Wire:

2 Wire

3 Wire

Output:

4~20mA

0~5V

0~10 V

Power Supply:

9~36Vdc

5~36Vdc/ac

12~36Vdc

Load resistance:

(U-10)/0.02(Ω )

>100kΩ -

Insulate resistance:

>100M Ω -@100V

-

Electronic connection:

Fixed cable and water proof IP68

Terminal Box DIN43650 IP65.

Pressure

connect port:

1/4’‘-18NPT Female hose connection

RG=Ø8gas mouth optional. (by order)

Response time:

≤10ms

Pressure Type:

Differential Pressure (Gauge pressure)

Certificate approving:

ExiaIICT5 and CE Certificate.

-

EMC Standard:

electromagnetic radiation:EN50081-1/-2

electromagnetic susceptibility:EN50082-2

Lighting protection:

withstand voltage of Air conduction:8000V,

withstand voltage of Shell and Cable: 4000V

Water Proof:

IP65

Weight:

Net weight is about 0.5KG, Full Packing weight is about 0.65KG
(Included 3 Meters Cable)