MT Pro Single/Multi – Channel Integrated Interferometer

Dimension Technology releases MT Pro, a new Single/Multi Channel Integrated Interferometer. MT Pro is the upgraded MPO interferometer with image resolution up to 1.5um. Equipped with functions like auto focus, PASS/FAIL analysis and auto calibration on reference mirror, MT Pro has optimized repeat-ability. The testing time is 0.5 second for single fibre measurement and 5 second for MT12 measurement. With field of view as 4.3mm*3.3mm, MT Pro is capable for MT16 and up to MT72 measurement. In the special designed fixture platform, 0° and 8° fixtures can be quickly switched without extra tools or fixtures needed. MT Pro is calibrated with 0.1nm laser interferometer and the high accuracy of measurement is committed. The ferrule to be tested is mounted with frame. Modularized software interface makes the measurement intuitive and user-friendly.

Product Parameters

Up to 1.5um image resolution

MT Pro uses an upgraded optical system to accurately restore the details on the MPO facets and profile. The accurate raw data ensures the trust-able 3D testing results.

High repeat-ability

10 continuous testing results without plug-in and plug-out.

Repeat-ability of Fibre Height, 12 core MPO

Repeat-ability of Fibre Core Dip, 12 core MPO

Accurate testing result

Each MT Pro system is calibrated with 0.1nm laser interferometer. The accuracy and consistency are committed for the key parameters for MT/MPO measurement, including ROC, Fiber height and Core dip.

Auto focus

MT Pro can be configured to start a measurement from Auto Focus function. The Auto Focus function can eliminate the difference on each Fiber channels. Further more, with Auto Focus function, the measurement procedure is significantly simplified, especially for APC connectors.

Auto calibration with reference mirror

MT Pro applies reference mirror to calibrate the instrument automatically and maintains the peak performance at all conditions. Compared with manual tuning stage and data compensation, the auto calibration is quick and accurate.

Unique frame mounting fixture for MT ferrule

Dimension uses its patented frame mounting fixture to hold the MT ferrules to be tested.

1. The frame mounting fixture for MT ferrule is compatible with MT4, MT8, MT12, MT16,MT24, MT32, MT48 and MT72. Only one fixture is needed to test all MT ferrules.2. Committed accuracy and repeat-ability for Ferrule angle measurement .3. Endurable and long life time fixture.4. Safe to PIN hole of DUT.5. Helpful to analyze the accuracy of polishing jig.

Quick switch between 0° and 8° fixtures

With the special designed fixture platform, 0° and 8° fixtures can be switched quickly. No extra calibration is needed. The high repeat-ability and reproducibility are guaranteed. The fixture platform is compatible for all types of MT/MPO PC and APC products.

One click operation

Only one clicking is needed to complete the whole measurement procedures, including auto focus, scanning, analysis and all calculations. The testing report will be ready in seconds.

Intuitive software interface and excellent 3D rebuilding

MT Pro applies intuitive and modularized software interface. The real time image, 3D profile, cross section, roughness and testing results are displayed in the user interface. The operation and configuration are convenient and user-friendly.

The details of the 3D profile visualize the polishing result and helps to analyze and improve the polishing process.

Quick measurement

The MT Pro hardware and software design improves the speed for MPO measurement. It takes only 0.5 second for single fibre measurement or 5 second for 12 core MPO measurement.