Open Data supplied by Natural Environment Research Council (NERC)

Bissett-Bermann 9040 Conductivity Temperature and Depth

The basic configuration of the B-B 9040 CTD incorporates pressure, temperature and conductivity sensors which could be logged digitally. This system also made it possible to derive other parameters, such as salinity, depth and sound velocity.

The instrument was versatile and it was possible to attach a dissolved oxygen sensor or to change the CTD housing, allowing it to obtain data from deeper layers in the water column. The accuracy for salinity is ±0.02 ppt , and ±0.02°C for temperature.

Data collection

The instrument used was a Bissett Berman 9040 CTD system and the data were logged on a Hewlett Packard 9820 and stored in an integer format. Instrument lowering and raising speeds were between 0.5 m/s and 1 m/s. An acoustic pinger was placed above the CTD to give an accurate depth measurement, this could then be used to check the CTD pressure calibration. An NIO bottle with reversing thermometers was placed above the pinger, within 2m of the CTD system. A bottle sample was taken at the bottom of the cast providing the temperature and salinity are uniform at that point. If large temperature or salinity gradients were present then the bottle sample was triggered at a suitable site on the upcast. A surface salinity sample was also taken at the start of the dip.

Calibration

The CTD was not calibrated in the laboratory. The manufacturer's calibration was used and water samples taken to check the calibration and apply corrections where necessary.

Temperature

The manufacturer's calibration was used to convert the raw data to to physical units using the equation below:

Temperature (°C) = (10 6 /Pt -2238.68/55.84)

where Pt is the temperature period in microseconds

These values were then plotted against the water bottle (i.e. reversing thermometer) temperatures and a regression line fitted to the data such that:

Temperature(WB) = m x Temperature(CTD) + c

Then the regression coefficients (m and c) were applied to correct the CTD temperature data - these are given in the table below.

Conductivity

The manufacturer's calibration was used to convert the raw data to to physical units using the equation below:

Conductivity (mmho/cm) = (10 6 /Pc - 4995)/58.12 + 10

where Pc is the conductivity period in microseconds

The water bottle salinities and corrected CTD temperatures were used to calculate the water bottle conductivity values. These values were then plotted against the CTD conductivities and a regression line fitted to the data such that:

Conductivity(WB) = m x Conductivity(CTD) + c

Then the regression coefficients were applied to correct the CTD conductivity data - these are given in the table below.

Pressure

The depths from the acoustic pinger were noted where the bottle samples were taken and then used to check the calibration of the pressure sensor - unless calibration values were available from the reversing thermometers. The equation below was used to convert the pressure period to physical units.

Pressure = (10 6 /Pd - 9712)/0.26267

where Pd is the pressure period in microseconds

A regression fit was carried out using the calibration values and the slope and intercept determined. The pressure values could then be corrected using:

Pressure (CORR) = m x Pressure(CTD) + c

The fit of the CTD data to the water bottle calibration data is given in the table below:

Variable

Slope (m)

Intercept (c)

Standard Deviation

Temperature (°C)

1.0023

0.0422

0.006

Conductivity (mmho/cm)

1.0026

-0.0234

0.000

Pressure (dbar)

0.9898

-1.2486

2.706

Data Processing

Obvious wild points were edited out of the calibration file and the calibration programs run to obtain values for the slopes and intercepts for temperature, pressure and conductivity. These were then applied to the uncalibrated data. Conductivities were converted to conductivity ratios and then converted to salinities using UNESCO recommended routines and sigma-t was calculated. The data values were then sieved to ensure a minimum separation between pressure values of 1 dbar. The data were then visually inspected and major spikes flagged.

Fixed Station Information

Station Name

Ellett Line/Extended Ellett Line Station G

Category

Offshore location

Latitude

57° 29.52' N

Longitude

11° 51.00' W

Water depth below MSL

1812.0 m

Ellett Line/Extended Ellett Line: Station G

Station G is one of the fixed CTD stations, which together form The Extended Ellett Line. The line lies between Iceland and the Sound of Mull (Scotland) crossing the Iceland Basin and Rockall Trough via the outcrop of Rockall. As part of this initiative, CTD dips, together with associated discrete sampling of the water column, have typically been carried out annually at this station since September 1996.

Prior to September 1996, Station G was part of a shorter repeated survey section, consisting of 35 fixed stations, known as The Ellett Line (originally termed the Anton Dohrn Seamount Section). This line incorporated those stations across the Rockall Trough and Scottish shelf between Rockall and the Sound of Mull and was visited at regular intervals (usually at least once a year) between 1975 and January 1996.

Fixed Station Information

Station Name

Ellett Line

Category

Offshore route/traverse

Ellett Line

The Ellett Line is a hydrographic transect consisting of 35 individual fixed stations which were occupied, usually at least once a year, between 1975 and 1996. The time series is named after the scientist David Ellett, who coordinated the survey work at Dunstaffnage Marine Laboratory (DML), near Oban. The transect ran between the north west coast of Scotland to the small outcrop of Rockall, via the Anton Dohrn Seamount - a prominent bathymetric feature in the Rockall Trough (see map). STD/CTD dips and associated water sampling for the analysis of nutrients were routinely performed during each station occupation.

In 1996 the transect was lengthened to incorporate new additional fixed stations crossing the Iceland Basin from Rockall to Iceland. This transect, which is still routinely occupied annually, is now known as the Extended Ellett Line and is a collaborative effort between scientists at Dunstaffnage Marine Laboratory and the Southampton site of the National Oceanography Centre (NOC).

Map of standard stations (1975-1996)

Map produced using the GEBCO Digital Atlas

The white triangles indicate the nominal positions of the Ellett Line stations (1975- 1996). Measurements made along the Ellett Line lie within a box bounded by co-ordinates 56° 40.02' N, 13° 42.0' W at the south west corner and 57° 37.2' N, 6° 7.98' W at the north east corner.

Nominal Ellett Line stations (1975-1996)

Listed below are nominal details of the standard hydrographic stations that formed the Ellett Line between 1975 and January 1996.