RESULTS:

Vascular seals created with the Small Jaw handset failed at a mean (95% CI) of 17.2 cycles (9.6-24.8) and a minimum of 10 cycles. Vascular seals created with the Impact failed at a mean of 20 cycles (18.4-21.6) and a minimum of 17 cycles. The majority of seal failures (73%; 95% CI 39%-94%) immediate leaked during vessel filling. The rate of vascular seal failure increased after the initial failure. Failure was associated with histologic disparities in tissue apposition.

CONCLUSION:

Repeated use and resterilization resulted in failure of the vascular seal due to inadequate tissue apposition after a minimum of 10 cycles.

CLINICAL SIGNIFICANCE:

Surgeons reusing and resterilizing LigaSure handsets (ForceTriad platform) should consider discarding handsets after 9 cycles for the Small Jaw and after 16 cycles for the Impact. Handsets should be immediately discarded after any intraoperative identification of vascular seal failure.