Designed and equipped specifically for seed implantation procedures the Brachytherapy C-Arm Table features a radiolucent extension that provides capability for brachytherapy treatment involving the upper body and general C-Arm applications. Motorized actuation of height, X-Y, lateral roll and Trendelenburg motions of the tabletop, with pre-programmed features, ensure quick and safe tabletop positioning during any phase of a procedure.

Control made easyThe LCD screen of the Main Control Panel displays actual tabletop position (all axes) in real time, and provides menus and prompts for other functions. Up to ten position sets are stored. A "Level" button brings the table to 0° tilt and 0° lateral roll while "Home" both levels and lowers the tabletop to minimum height for patient transfer. The portable hand-held and foot-operated controllers also adjust the table. The Joystick Controller is used for motorized X-Y positioning and features proportional speed control.

More adjustments and features...The Fowler back adjusts with a touch-handle lock/unlock mechanism to an 80° angle with incremental stops at 20°, 30°, 40°, 50°, 60°, 70° and 80°.

The table comes with a lightweight non-imaging patient transfer extension and a radiolucent extension. The radiolucent extension is made for vascular, cardiac, and other generalC-Arm applications. A standard OR accessory rail makes stepper positioning quick and easy. Select from knee crutches or the PAL stirrups, standard or Dura Board arm boards—configure the table to your exact specifications.Both AC and battery power are standard. Eliminate power cords with battery operation. An optional battery, with separate charging unit, can provide cordless operation.

Both AC and battery power are standard. When operating on battery, there are no power cords to catch on feet or block table wheels. Battery charging can be accomplished during off hours by plugging in the AC power cord. An optional battery with separate charging unit can provide continuous cordless operation.

Ordering Information:

Product Videos

A review of assembly and operation of the Biodex Urology and Brachytherapy C-Arm tables. For more detailed information, please refer to the user manual (http://www.biodex.com/sites/default/files/058800man_06011revd.pdf)