Specifications

A range of standard and custom CTC autosampler syringes manufactured to achieve the highest level of accuracy and precision possible. These C-Line syringes also feature a new sealing system based on a novel metal spring for working with enhanced tightness, excellent performance over a large temperature range and temperature gradients, increased lifetime compared to traditional headspace syringes and increased accuracy and reproducibility of headspace GC analysis. Syringes in this category are either 700 and 7000 series Microliter or 1700 and 1000 series Gastight syringes. Microliter syringes have a stainless-steel plunger that is hand fitted to the glass barrel and are designed for liquid sampling. Gastight syringes have a PTFE plunger tip that creates a tight connection between the syringe barrel and plunger, allowing for liquid or gas sampling.

Made of PTFE/polymer for better temperature stability

Features flange for easy installation

Adjustable plunger button for plunger tip clearance

Hub and needle design for zero carryover

C-Line Syringe Terminations

Fixed Needle (FN) features a direct attachment of the needle to the barrel and eliminates contact between the sample solvent and the adhesive reducing carryover

Special Cemented Needle (SN) shares the same characteristics as a standard cemented needle syringe but comes in a special length

Knurled Hub (KH) with removable plunger/needle assembly, knurled hub is used exclusively on the 7000 series syringes

Headspace Syringe Terminations

Luer Tip Cemented Needle (LTN) for mid volume syringes

Needles cemented into the glass syringe barrel at a point corresponding to the zero graduation mark, hence dead volume is limited to the internal volume of the needle

Not autoclavable

Needle gauge determined by the syringe volume and are not user-selectable