Rigaku NEX QC EDXRF for Analysis of Sulfur in Gypsum

Applied Rigaku Technologies, Inc. has announced a new empirical method for the analysis of sulfur in gypsum. Gypsum is a soft sulfate mineral added to Portland cement to prevent flash setting of concrete.

Application Note #1255 demonstrates the performance of the Rigaku NEX QC energy dispersive X-ray fluorescence (EDXRF) spectrometer in the monitoring of gypsum for quality control during the cement production process.

X-ray fluorescence (XRF) spectroscopy is integral to control of the production process in modern cement plants, enabling a cement maker to produce a consistent product.

Energy dispersive X-ray fluorescence spectrometry is a routinely employed for screening and quality control, providing rapid compositional data at various stages of production.

Whether at the quarry, cement plant or industrial manufacturing site, the sulfur content of the gypsum as it is processed needs to be closely monitored to ensure the quality and proper formulations of the final products.

EDXRF analyzers are typically implemented to monitor the addition of gypsum throughout the manufacturing process.

For this method, each gypsum sample is prepared by grinding the material to dry, homogeneous powder <200 mesh (<75um particle size).

Approximately 8 grams of powder is then placed in a 32mm sample cup and packed to an even and consistent sample. Empirical calibration was developed using a set of six assayed standards with sulfur content ranging between 12-17%.

Analytical results and precision data are shown from the measurement of the low and high Sulfur content calibration standards. Detection limits were in part determined by the calibration range and analysis time.

The data confirm that the NEX QC analyzer yields excellent results for the measurement of sulfur in gypsum, without the need for helium purge. Other elements in the gypsum, including calcium, can also be measured.

The method shows how the Rigaku NEX QC analyzer, with its simple, modern touch screen interface, enables reliable and efficient measurement to meet analytical needs at the quarry or cement plant when testing gypsum and finished cement, as well as during the manufacturing of other gypsum products.

Join For Free

Access to this exclusive content is for Technology Networks Premium members only.

Join Technology Networks Premium for free access to:

Exclusive articles

Presentations from international conferences

Over 4,000+ scientific posters on ePosters

More Than 5,300+ scientific videos on LabTube

35 community eNewsletters

Sign In

Forgotten your details? Click Here
If you are not a member you can join here

*Please note: By logging into TechnologyNetworks.com you agree to accept the use of cookies. To find out more about the cookies we use and how to delete them, see our privacy policy.