Welcome to HVAC-Talk.com, a non-DIY site and the ultimate Source for HVAC Information & Knowledge Sharing for the industry professional! Here you can join over 150,000 HVAC Professionals & enthusiasts from around the world discussing all things related to HVAC/R. You are currently viewing as a NON-REGISTERED guest which gives you limited access to view discussions

To gain full access to our forums you must register; for a free account. As a registered Guest you will be able to:

Participate in over 40 different forums and search/browse from nearly 3 million posts.

On another note, I thought of a way for Digi-Cool DRSA-1100 and 1600 owners to mod their units to do SH and SC with 2 probes.
Those models only support one temperature prob, so normally the probe needs to be moved from one line to the other when switching between SH and SC.

Do a mod to plug 2 probes in, like I did, but wire them to a DPDT switch mounted where the existing prob connection port is.
To switch between measuring SH and SC, you just hit the mode button to change to calculating SH or SC, and flip the switch to use the appropriate probe.

Thanks for posting the pictures.
It looks like they are using the same sensor in those as they do in the pipe wrap probes, which is good.
The actual thermistor is on the back side of the metal that looks like it is wrapped around the foam like material. The metal protects the thermistor, and aids heat transfer to it.

I guess I'll go ahead and order a couple of them, and keep the pipe wrap probes as spares, and for when I'm working on larger equipment.

From seeing all the mountings it looks like the 4011 pipe wrap allows for the least ambient air around the sensor next would be the wireless clamp then the wired clamp I know from testing the wireless one vs the pipe strap there is a difference. From the look of it the clamp version will be even further off than the wireless version in the field. The pipe strap looks like the best one as far as far as precise measurements and at the least cost. Last one I bought was 46 bucks!!! (@J-Mart chuckcrj)

On another note, I thought of a way for Digi-Cool DRSA-1100 and 1600 owners to mod their units to do SH and SC with 2 probes.
Those models only support one temperature prob, so normally the probe needs to be moved from one line to the other when switching between SH and SC.

Do a mod to plug 2 probes in, like I did, but wire them to a DPDT switch mounted where the existing prob connection port is.
To switch between measuring SH and SC, you just hit the mode button to change to calculating SH or SC, and flip the switch to use the appropriate probe.

Like this.............
These are my 1600's with the DPDT switch modification. I did this over the past summer. I got the idea from someone on a youtube video. The inside is not as neat as your modification. But they are still water proof and work great. They are 1 degree apart between the 2 probe. I did have them measured against other thermometers and are all within 2 degrees of each other like your previous post. I'm happy with the results and I don't care what clover thinks. Yes, I have the AK900's but these are still my go to gauges.

IMO I don't get why the Digi-cool haters out there still say what they say. Digi-cools were built for techs to alter if needed and still be amazingly accurate.