The Levelogger Edge temperature and water level datalogger features a new Hastelloy pressure sensor and a titanium based PVD coating adding increased corrosion resistance and even greater stability over a wide range of temperature and pressure conditions. The Hastelloy sensor features the same excellent 0.05 % of full scale accuracy of previous leveloggers while being more robust, able to withstand 2 times over the rated depth without sustaining permanent damage. The temperature compensated range has been improved to 32–122 °F, and the barologger can now make temperature corrected pressure readings down to 14 °F. Thermal response time has increased to 1 minute for every 18 °F change, allowing temperature readings and compensation to stabilize quickly.

NEW The DataGrabber accessory allows you to directly download data from a levelogger to a USB thumbdrive for later analysis (direct read cable or DataGrabber optical adapter required)

Solinst now offers the Aquavent, a vented water level logger with all the features and ease of use you have come to expect from Solinst loggers.

The Levelogger Edge water level datalogger can store up to 40,000 readings in various logging modes including linear, event and user schedules with future start, future stop, repeat, pause, and real time views. Using a special compression mode, the levelogger edge can store up to 120,000 readings. Logging intervals can be set as low as 0.125/second, and can run for up to 10 years (based on 1 reading per second intervals) from its lithium battery. The Levelogger Edge inherits SDI-12 compatibility, faraday cage design protecting against electrical surges and lighting, a durable maintenance free enclosure, and small dimensions (7/8″ x 6.25″). Weight has also been reduced to only 4.6 oz.

Solinst manufacturers two other water level dataloggers, the cost effective Levelogger Junior Edge which provides an economical solution for many applications, and the LTC Levelogger Junior for adding conductivity measurements in addition to water level and temperature.

Along with the improvements to the Levelogger water level datalogger, Solinst has improved the PC software. It now allows barometric compensation on multiple data files simultaneously, increased levelogger programming options, and many user interface design improvements. The USB optical reader's communication speed has also been increased to 38,400 bps.

In addition to the PC software included with every Levelogger communication kit, a handheld instrument, the Solinst Leveloader, is also available to view data.

The Barologger Edge features all the improvements to the Levelogger Edge and increased accuracy over previous Barologger versions. Using a Barologger is the easiest and most accurate way to compensate for atmospheric pressure fluctuations in your data. One Barologger can be used to compensate the data for every levelogger in a 20 mile radius (for every 1000 ft change in elevation).

A Solinst Rainlogger allows you to add precipitation data to levelogger deployments for the correlation of rainfall events and water level data.

Solinst has also redesigned the Levelogger well caps for use with either suspension wire or direct read installations. Every well cap now has two direct read ports (one for a levelogger, and one for an optional barologger), a eyelet for suspension wires, a vent to allow the equalization of barometric pressure within the well, and can be secured using a lock.

Levelogger Edge Features

Memory for up to 120,000 level and temperature readings

Self-Test Diagnostic Utility

Stainless steel construction with Titanium based PVD coating

Able to withstand up to 2x rated depth without damage

Fast thermal response

Direct read or suspension wire deployment options

Increased thermal compensation temperature range

Durable, water-tight and maintenance free design

Cellular or radio telemetry options available, SDI-12 compatibility

10 year battery life at 1 reading per minute

Solinst Levelogger Edge Ordering Instructions

A complete Levelogger Edge water level datalogger system consists of a levelogger, a deployment solution, and a communications kit.

For each Levelogger you wish to deploy you must choose either a direct read cable or a suspension wire of appropriate length as a deployment solution. Direct read cables allow you to download data from the Levelogger without removing it, suspension wires require you to bring it to the surface.

Atmospheric pressure compensation can be provided by a Barologger. You only need 1 Barologger for every levelogger in a 20 mile radius, or 1000 ft difference in elevation.

Communication kits are available in both USB and RS232 serial versions and include everything you need to program and download data from any number of leveloggers. A basic communication kit is required for programming and downloading data from the levelogger, direct read communication kits contain everything in the standard kit and a cable for connecting to direct read deployments. In addition to the communication options, the DataGrabber and App Interface are also available for quick and easy downloads in the field.

Locking well caps for 2" wells and adapters for 4" wells that support either direct read or suspension wire deployment solutions are available. Levelogger well caps have direct read ports for both a levelogger and a barologger.

Field Supplies

Great Prices for Project Supplies

Did you know that Equipco also sells environmental field supplies?

Whether you need calibration standards, tubing, or other supplies like gloves and labels, we stock everything you need to finish the job. Our high quality field supplies, including NIST traceable calibration solutions and gases, come at significant savings over manufacturer brands.

Used Equipment

Great Discounts for Used Equipment

Did you know that Equipco also sells used equipment?

Whenever we have used equipment to sell, whether like-new equipment used only for demonstration or a former rental, well maintained and available for a great price, find what you’re looking for here and check back for updates.

RKI Eagle 2

Multiple Gas Monitor

RKI’s Eagle 2 is an advanced portable gas detector capable of monitoring up to six sensors simultaneously including a PID. 16 different sensors are available providing for hundreds of configuration possibilities.

Landtec GEM5000

Landfill Gas Analyzer

The GEM5000 is the latest in the line of Landtec's GEM landfill gas analyzers. Now lighter, smaller, more accurate, and with an improved user interface, the only instrument you need to take into the field just got even better.

Levelogger Edge

Water Level Datalogger

Advanced and durable, the Levelogger Edge is the latest water level datalogger from Solinst. An all new sensor design manages to achieve excellent accuracy while being more robust then previous models and remaining economical.

MultiRAE Series

Multiple Threat Monitor

The MultiRAE Series of handheld gas detectors includes three models for detection of up to six threats simultaneously. Choose from 25 toxic sensors, 3 PID sensors with a 9.8 eV lamp option, and even a gamma radiation sensor.