FPU400 Series peristaltic pumps offer simplicity at a
reduced cost. The motor turns the
rollers, which squeeze fluid through the precision-bore
high-tolerance tubing in a wavelike motion, acting like
a positive displacement pump. The fluid doesn’t touch
the pump; it only comes into contact with the tubing.
This makes the pump suitable for fluids carrying small
particles. FPU400 pumps continuously overocclude
the tubing, which makes the pump self-priming and
non-siphoning. They are easy to maintain, with few
moving parts and no seals or valves to clog, clean,
or replace. As one section of tube fatigues, just move
the tubing to a section that hasn’t been under the
rollers and continue pumping. Or simply replace the
supplied tubing. The replacement 230 mm (9") lengths
of precision-bore tubing ensure repeatable pumping
rates time after time. Union connectors for
hooking up system tubing are sold seperately. The replacement 230 mm
(9") lengths of tubing come without connectors. The
pumps are supplied standard with a mounting panel.
The AC-powered pumps are also available in a rugged
polycarbonate enclosure for bench mounting. Maximum
backpressure is 15 psi without derating (up to 25 psi
@ 75% of maximum flow). Maximum suction head is
8.5 m (28'). Flow rate repeatability from one pump to
the next or when replacing tubing is within 8% of
maximum flow rate listed. Shipping weight is 1.6 kg
(3.5 lb) for all sizes. These pumps are ideal for
metering/dosing, repetitive dispensing, and general
liquid/gas transfer, and they have a 100% duty cycle
for continuous operation.

† All prices on this site are shown in Malaysian Ringgit.Note: FPU400 Series comes complete with four 230 mm (9") lengths of Norprene tubing (1/16, 1/8, 3/16, and 1/4"), 2 rollers, mounting plate and operator’s manual.