This small hand held pulse oximeter is full featured and is FDA and CE registered. The CMS-60D oximeter has alarms and is suitable for a variety of monitoring needs including hospital, clinic, home care and physical care for individuals that are in training. Pilots has also found it useful for monitoring their condition while in flight. It is suitable for both continuous monitoring and spot check.

Featuring a large OLED display which allows you to read in the dark and during severe weather conditions, the CMS-60D Handheld Pulse Oximeter provides fast, real-time and accurate readings including SpO2, Pulse Rate, with a bar graph indicating pulse intensity and pulse waveform. This item is small, lightweight and easy-to-carry. This unit also has included SPO2 data management software; several manufacturers charge $300-$400 for this capability.

Can use for adult or for children with the optional pediatric probe.

The CMS-60D auditory and visual alarms notify you of the following conditions : if the finger is out of position, weak battery, if it measures a reading beyond set limits and even when the probe is disconnected. Several display modes and adjustable screen brightness levels allow patients and doctors to see the readings more comfortably and conveniently.

Powered by 2 AA dry batteries or rechargeable batteries (not included), the CMS-60D has the ability to store 24 hours worth of data including both measured pulse rate value and SpO2 value. Readings can be uploaded to your computer easily using the free software and usb cable that is included with the device.

Warranty:

1 year limited manufacturer warranty

Specifications

SPO2 - Digital

Range: 0%-99%

Resolution:±1%

Accuracy:±2% (70-99%)

0--69%, unspecified

Pulse Rate

Range: 30bpm-250bpm

Resolution: 1bpm

Measurement Performance in low perfusion

Oxyhemoglobin and Pulse value can be shown accurately when pulse-filling rate is 0.4%

Power supply

Requirement:2.6V DC-3.6V DC

Battery

Type: Interior battery, BF

Requirement:2 AA dry batteries

Interference Resistance Capacity Against Ambient Light

Deviation is smaller than ±1% between value of Oxyhemoglobin measured in natural lighting indoor condition and present lighting sources and that meandered in dark room.

Environmental

Operating Temperature: 10°C-40°C

Storage Temperature: -40°C- +60°C

Operating humidity: 30%RH-75%RH

Storage Humidity:5%RH-95%RH

Operating Pressure:700hPa-1060hPa

Storage Pressure: 500hPa – 1060hPa

Dimensions

Size:110*60*23 mm

Weight:6.3oz((180g with batteries)

Display

Type: blue OLED

Parameter: SpO2 (2-digit), PR (3-digit),bar graph,Pleth waveform

Adjustable brightness leve

Operating system Microsoft Windows CP/Vista/7

Processor: Pentium IV 1.8G or higher

Memory: 256MB or higher

Disk Drive: 40 GB or higher

Adjustable brightness level; data refresh rate 1 sec

1 year warranty

Features

Free software to upload and analyze up to 24 hours of data at a time (WinXP/2000/Win7 and Vista).