Features1.Completely controlled by PC, can flexibly select the flame and graphite furnace atomizer (optional).2.Integrated floated optical platform design can obviously improve the optical system shock resistance andkeep stable though use the optical signal for a long term.3.Eight light stands can be changed automatically and preheat the eight lights meantime as well as optimizethe working condition of the hollow cathode lamp.4..Position adjusting: the best height of the flame burner and can automatically set the front and rear positions.5.Fully automated wavelength scanning and peak searching.6..Complete safety chains protection equipment: the function of warning and automatic safety protectiontowards the wrong burner, leakage of the gas, under voltage of air and the abnormal flameout.7.Deuterium lamp and self-absorption background regulation.8.Data processing: super strong database, possesses more than 500 data self-storage and cut-off storagefunction, can store the analyzed result with the form of EXCEL and the testing method and the result can berandomly called.9.Measuring method: flame absorption method and emission method.10.Result printing: parameter printing, data result printing and diagram printing.

COMPLETE AUTOMATIC CONTROLLING SYSTEMWith the help of the software, the following can be easily achieved1. Selection of the element lamp2. Up-down-front-rear adjustment of the lifter3. Adjustment of the optical energy4. Selection of the slit5. Determination of wavelength scanning and peak searching6. Selection of the atomizer7. Setting of the background deduction method8. Controlling of the gas flow9. Automatic flaming and flaming out10.Setting of the graphite furnace testing method

ADVANCED GRAPHITE FURNACE TEMPERATURECONTROLLING TECHNOLOGY1.The bringing in of PID technology can effectively overcome the influence on the temperature rising process caused by the voltage fluctuation and the resistance change to make more accurate controlling process. The combination of the 3ms/time fast sampling technique can make more accurate andreliable testing data. 2.The fast heating capacity can improve the flexibility of the elements further3. Use the ordinary power source of 220V without need of dynamic power of 380V4.The maximum procedure heating capacity setting of 20 levels can make a more convenient and easiertest of different samples5.Three grades adjustable gas flow can accustom to more application needs6.Can timely alarm when the gas and water is stopped and insufficient gas and water, can avoid theequipment damage and measuring error.

SAFE, RELIABLE AND CONVENIENT FLAMESYSTEM

1.EPC can control the flow of Acetylene (C2H2) more accurately and it is a kind of system which also canoperate easily2. Efficient atomization system enables a higher sensitivity

3.The whole operation system has a high security as the fire safety system can alarm whenever the electricity is cut off, abnormal flame occurs, a lack of pressure happens or the burner does not match well. And it will automatically turn off the gas, prohibit temper ing. Thus it keeps the operation people and equipment from harm and damage.

MULTI-FUNCTIONAL SOFTWARE WORKSTATION1.A workstation that is supported WINDOWS XP/2000/NT2.The rich menu brings great convenience to the use of customer3.Convenient conversion between different menus makes the operation more easily4.Various analytical correction methods provide the users more choices5.Basic default parameter settings enable even the beginners can do the normal operation6. Flexible storage, editing and printing methods give the user largest support