ค้นหา

Frontier™ Floor Mounted™

Esco Frontier Floor Mounted Fume Hood is designed to provide comfortable
space when users have to deal with large apparatus and hazardous
containers. This fume hood is built with horizontal or vertical sliding
sashes for ease of access when transporting apparatus into the hood.
Although this type of fume hood is sometimes denoted as a walk-in fume
hood, it is a misnomer. The user must not enter the hood while an
activity generating hazardous fumes exists or when suspected
concentration of fumes exists inside the chamber.

Key Benefits:

ASHRAE 110 certified / EN14175 certified

UL classified

Excellent aesthetics

5 degree sloped front

Superior containment at 0.3m/s

High energy savings

Disclaimer: Esco FrontierTM is UL classified. Due to the custom nature of the range, some options do not qualify for the UL label.

Sash is clearly labeled with operating instructions and illustrations.

General Specifications, Frontier® Floor Mounted™ Fume hood

Model

220-240 VAC, 50/60 Hz, 1ø

EFF-4UBRVW-8

2090025

EFF-4UBUVW-8

2090342

EFF-5UBRVW-8

2090012

EFF-5UBUVW-8

2090631

EFF-6UBRVW-8

2090026

EFF-6UBUVW-8

2090632

EFF-6UBRHW-8

2090028

EFF-6UBUHW-8

EFF-8UBRVW-8

2090027

EFF-8UBUVW-8

2090633

EFF-8UBRHW-8

2090029

EFF-8UBUHW-8

110-120 VAC, 50/60 Hz, 1ø

EFF-4UBRVW-9

2090272

EFF-4UBUVW-9

2090273

EFF-5UBRVW-9

2090557

EFF-5UBUVW-9

2090634

EFF-6UBRVW-9

2090558

EFF-6UBUVW-9

2090283

EFF-6UBRHW-9

EFF-6UBUHW-9

EFF-8UBRVW-9

2090559

EFF-8UBUVW-9

2090635

EFF-8UBRHW-9

EFF-8UBUHW-9

Nominal Size

1.2 meters (4’)

1.5 meters (5’)

1.8 meters (6’)

2.4 meters (8’)

External Dimensions

(W x D x H)

1220 x 965 x 2688 mm

(48.0” x 38.0”x 106.0”)

1525 x 965 x 2688 mm

(60.0” x 38.0”x 106.0”)

1830 x 965 x 2688 mm

(72.0” x 38.0”x 106.0”)

2440 x 965 x 2688 mm

(96.0” x 38.0”x 106.0”)

Internal Dimensions *

(W x D x H)

996 x 710 x 1586 mm

(39.2” x 28.0”x 65.0”)

1301 x 710 x 1586 mm

(51.2” x 28.0”x 65.0”)

1606 x 710 x 1586 mm

(63.2” x 28.0”x 65.0”)

2216 x 710 x 1586 mm

(87.2” x 28.0”x 65.0”)

Exhaust Volume/ Static Pressure Required

Face Velocity

Sash Opening

0.4 m/s (80 fpm)

Design Opening:

457 mm (18.0”)

1160 cmh at 18 Pa

(682 cfm at 0.08" WG)

1507 cmh at 23 Pa

(887 cfm at 0.10" WG)

1855 cmh at 28 Pa

(1092 cfm at 0.12" WG)

2551 cmh at 20 Pa

(1501 cfm at 0.09" WG)

0.5 m/s (100 fpm)

Design Opening:

457 mm (18.0”)

1449 cmh at 28 Pa

(853 cfm at 0.12" WG)

1884 cmh at 36 Pa

(1109 cfm at 0.15" WG)

2319 cmh at 44 Pa

(1365 cfm at 0.19" WG)

3189 cmh at 31 Pa

(1877 cfm at 0.13" WG)

0.4 m/s (80 fpm)

Full Open:

736 mm (29.0”)

1805 cmh at 20 Pa

(1062 cfm at 0.09" WG)

2346 cmh at 34 Pa

(1381 cfm at 0.15" WG)

2888 cmh at 51 Pa

(1700 cfm at 0.22" WG)

3971 cmh at 24 Pa

(2337 cfm at 0.10" WG)

0.5 m/s (100 fpm)

Full Open:

736 mm (29.0”)

2256 cmh at 31 Pa

(1328 cfm at 0.13" WG)

2933 cmh at 52 Pa

(1726 cfm at 0.22" WG)

3610 cmh at 80 Pa

(2124 cfm at 0.34" WG)

4964 cmh at 38 Pa

(2921 cfm at 0.16" WG)

Exhaust Outlet Diameter

305 mm (12.0”)

Number of Exhaust Collars

1

1

1

2

Fluorescent Lamp Intensity

975 lux (90 Ft-candles)

948 lux (88 Ft-candles)

919 lux (85-Ft candles)

971 lux (90 Ft-candles)

Controller

Rocker Switches (Option to upgrade to Sentinel XL™

Construction

Main Body

Electrogalvanized steel with Epoxy-polyester hybrid Isocide™

Internal Liner (default)

Esco Resinate™

Worktop (default)

No worktop (Option to purchase low height support stand with worktop)

Sash Specifications

Sash material

temperedlaminatedandframedsashglass

Sash configuration

Vertical

Vertical/ Horizontal

Maximum Sash Opening (two vertical Sash up)

1647 mm (64.8”)

Electrical

Cabinet Full Load Amps (FLA)

32 A

Cabinet Nominal Power

100 W (lighting only)

Esco offers a variety of options and accessories to meet application requirements:

Additional service fixtures and electrical outlets.

Fully configurable optional Esco Sentinel XL™ microprocessor controller for hood performance monitoring and continuous display of face velocity. The alarm system will trigger should the face velocity within the work zone falls below the recommended setting.

Esco offers a variety of options and accessories to meet application requirements:

Additional service fixtures and electrical outlets.

Fully configurable optional Esco Sentinel XL™ microprocessor controller for hood performance monitoring and continuous display of face velocity. The alarm system will trigger should the face velocity within the work zone falls below the recommended setting.