Tell a Colleague

If you think this article may interest another healthcare professional then enter their email address below and click send to email them details.

POLYAIR® 100 & POLYAIR® 60

The POLYAIR® 100 and POLYAIR® 60 are named after their cell height (in millimetres).

POLYAIR® 100 = 100mm / 10cm / 4 inches high

POLYAIR® 60 = 60mm / 6cm / 2.4 inches high

Maximum user weight for POLYAIR® cushions is 28.3 stone (180kg).

3 Year warranty on all POLYAIR® cushions, and 2 years on covers.

Both cushions are avalable in either single or two compartments. (Limited sizes available for single compartment cushions)

The two compartment cushions are suitable to be used any way round (depending on access to the inflation valves), to provide a solution to either a left/right or a front/back configuration. The separate compartments can be inflated to the same level, or at different levels to manage situations such as pelvic imbalance.

Inflation and Checking on 2 Compartment Cushions.

Inflating the 2 compartment cushions is the same as inflating the single compartment cushions, except that it has to be done twice. The hand pump supplied with the cushion inflates the cushion quickly, and easily.

With the use of the pressure gauge, attached to the pump, setting the inflation for users is simple:

Switch the pressure gauge on - the lights will flash.

Attach the pump hose to the cushion. Inflate the cushion until the MIN light is on steadily. (Repeat on second compartment, if the cushion has two compartments)

Seat the user on the cushion correctly.

Gently press the orange air-release button to allow the user to become immersed into the cushion. Keep checking the setting of the cushion, until appropriate setting has been reached. (See full set up guide here)

Two Compartment Cushion Codes

POLYAIR® 100 Code

Size

POLYAIR® 60 Code

N/A

14x14 in (36x36cm)

P301C36361HW

P302C40401HW

16x16 in (40x40cm)

P301C40401HW

P302C40431HW

16x17 in (40x43cm)

P301C40431HW

P302C40461HW

16x18 in (40x46cm)

P301C40461HW

P302C43431HW

17x17 in (43x43cm)

P301C43431HW

P302C46401HW

18x16 in (46x40cm)

P301C46401HW

P302C46461HW

18x18 in (46x46cm)

P301C46461HW

P302C51461HW

20x18 in (51x46cm)

P301C51461HW

P302C51501HW

20x20 in (51x50cm)

P301C51501HW

All cushions are supplied with one water-resistant, breathable cover, with an anti-slip base and two eyelets for access to the valve(s).

If you would like to purchase a spare cover at the time of purchasing your cushion, simply replace "1HW" at the end of the cushion code with "2HW".

Single Compartment Cushion Codes

POLYAIR® 100 Code

Size

POLYAIR® 60 Code

P302CM40401HW

16X16 in (40X40cm)

P301CM40401HW

P302CM43431HW

17x17 in (43x43cm)

P301CM43431HW

P302CM46461HW

18x18 in (46x46cm)

P301CM46461HW

All cushions are supplied with one water-resistant, breathable cover, with an anti-slip base and two eyelets for access to the valve(s).

If you would like to purchase a spare cover at the time of purchasing your cushion, simply replace "1HW" at the end of the cushion code with "2HW".

Unfortunately, we are unable to offer special sizes in these cushions.