1.3.1 Communication Wellhead and Connector Cables

A Levelogger USB PC Interface Cable can also be used to connect from an App/DataGrabber Connector Cable to the USB Port on a PC. See the Levelogger User Guide for USB Driver installation Instructions for the PC Interface Cable.

The AquaVent Wellhead fits conveniently onto a 2" (50 mm) well casing using the well cap base (a 4" adaptor is also available). There are two options: the SP Wellhead has a 10-pin connection (Solinst Protocol) for cables communicating with Solinst software and accessories; the SPX Wellhead has an additional 12-pin connection for cables communicating with third party dataloggers or telemetry systems using MODBUS or SDI-12 protocols (see separate User Guides).

There are two Connector Cable options for the SP Wellhead:

The USB Connector Cable connects to a PC for communication with Levelogger Software.

App/DataGrabber Connector Cable for connection to a Levelogger App Interface or DataGrabber. See separate User Guide/Instructions.

Note:

The small internal AquaVent logger battery is only used to maintain the internal clock of the AquaVent logger when it is not connected to the Wellhead, and during any firmware upgrades.

Each Wellhead comes with four (4) user-replaceable 1.5V AA lithium batteries that power the AquaVent (see Section 8.1 for battery installation instructions and battery life estimates). They can last 8 years based on 1 reading per minute.

The Wellheads contain multiple desiccants and have a hydrophobic filter where the Vented Cable terminates at surface. There is no need to replace desiccants or the filters, they are designed to provide protection from moisture over the lifetime of the instrument. The filter protects the cable from moisture, while still allowing air to flow through the vent tube, resulting in the cancellation effect of barometric pressure on the AquaVent logger.

Note:

Solinst recommends using 1.5V AA lithium batteries in the Wellhead.

A 2" Well Cap Assembly is supplied with each Wellhead, along with a Support Hanger Bracket to secure the Vented Cable. See section 8.2 for installation instructions.

Note:

The Wellhead Connector Cables to be used with the SP Wellhead are identified with a orange band at the 10-pin connection. The 10-pin (Solinst Protocol) connection on the Wellhead is also labeled orange.