Texto completo

Resumen

After construction of the LYSS (Light cYcling Stressing Source) in early 2014, several CPV receivers, with and without secondary optical element (SOE) have been aged under fast transient illumination cycling,. The test plan for Madrid consisted of 50000 cycles. Receivers with poor heat spreaders showed low reliability but those with thicker metal layers passed the test well. The operation of LYSS along 8 months, after more than 250000 cycles, did not show any significant failure, except lamp reposition every 120 hours, in average. The equipment seems valid for unveiling weak receiver designs with respect to intensive illumination, in steady and transient modes.