DIN Rail Temperature Transmitter

TX2DR (Pt100)

TX2DR is a 35 mm DIN rail mounted “all-in-one” 2-wire temperature transmitter. Its high reliability industrial design offers some rare functions, e.g. 4-20 mA or 20-4 mA selection. The transmitter converts low level input signals from field sensors to a proportional 4 - 20 mA current signal that is ideal for driving panel meters, recorders, controllers, computers and other instrumentation. TX2DR receives input from Pt100 RTD sensors through a 3-wire connection and can be designed on request for other inputs such as Pt50, Pt200, Pt500 and Pt1000. Relieved of a display, this transmitter is a cost-effective option. 1 LED in the front of the unit indicates it is operating. It is accurate to 0.1% of span. TX2DR is adjustable for 8 overlapping ranges in °C or °F and gives a temperature linear output. All selections are made by solder jumpers and it is calibrated via ‘Fine’ ZERO/SPAN potentiometers. The transmitter is powered by a 10 - 30 V DC supply. TX2DR - TC mV is another version of this transmitter designed for input from J, L, T, K & N thermocouples or millivolts linear input.

Multi-range

Multi-input : E, J, K, L, N, R, S and T thermocouples or millivolts

Solder jumpers for

8 SPAN ranges, 25 to 600 C° / 45 to 1080 F°

4 ZERO ranges, -100 to +70 °C / -148 to +158 °F

Input-Output Isolation

Upscale/downscale selectable sensor break detection

Pt50, Pt200, Pt500, Pt1000 also available

Applications

Temperature

Industries

Customization

Not Available

Please choose from our customization options and enter under comments while inquiring about this product`