* It is recommended that surgeons personalize their A. Constant based on their surgical techniques and equipment, experience and pos- operative results. For more information please visit Hanita Lenses web.

Attributes

Advanced Optical Design

The BunnyLens MF was designed using the most advanced tools by a professional R&D team of optical and mechanical engineers. The optical diffractive apodized aspheric profile was developed using proprietary algorithms and the Zemax™ software. Calculations were optimized in order to provide optimal energy distribution in different light conditions, minimize spherical aberrations and generate +3.0 D addition power for near vision, equivalent to +2.4D at the spectacle plane.

Apodized Diffractive Optics

The diffractive steps are located in the 4mm central zone, suiting pupil sizes in various lighting conditions. The steps follow an apodization function, meaning they reduce in height according to an algorithm designed so the patient can function comfortably all day long. The aspheric surface integrated with the diffractive profile is the BunnyLens AF aberration free optic, which has been clinically proven to provide excellent vision in all conditions.