Fluke i30 AC/DC Current Clamp

Product overview: Fluke i30 AC/DC Current Clamp

The i30 current clamp is based on Hall Effect technology for use in measurement of both DC and AC current. The i30 may be used in conjunction with multimeters, recorders and other suitable recording instruments for accurate non-intrusive current measurement.

Specifications: Fluke i30 AC/DC Current Clamp

Electrical Specifications

Specified current range

30 mA to 30 A DC, 30 mA to 20 A AC rms

Usable current range

5 mA to 30 A DC, 30 mA to 20 A AC rms

Crest Factor

1.4 in

Output sensitivity

100 mV/A

Accuracy (at +25°C)

±1% of reading ±2 mA

Resolution

±1 mA

Load impedance

> 10 k Ohms and ≤ 100pF

Conductor position sensitivity

±1% relative to center reading

Frequency range

DC to 20 kHz (-0.5 dB)

Temperature coefficient

±0.01% of reading/°C

Power supply

9 V Alkaline, MN1604/PP3, 30 hours, low battery indicator

Working voltage (see Safety Standards section)

300 V AC rms or DC

General Specifications

Maximum conductor size

19 mm diameter

Output connection

4 mm safety connector

Output zero

Manual adjust via thumbwheel

Cable length

1.5 meters

Operating temperature range

0°C to +50°C

Storage temperature range (with battery removed)

-20°C to +85°C

Operating humidity

15% to 85% (non-condensing)

Weight

250 g

Safety

IEC 61010-1, Pollution Degree 2

IEC 61010-2- 032: CAT III 300V

Use of the probe on uninsulated conductors is limited to 300 V ac rms or dc andfrequencies below 1 kHz.