My GMC-320+ died last week and I want to put my CDV-700 into service in the interim. Does wapp4netc monitor one of the rs-232 signals for pulses only (converted to USB) or does it only read standardized data format on the RX data line of the rs-232 port? In other words, can I customize an RS-232/USB board I have to read the count pulses from the CDV-700 such that wapp4netc can use it?Thanks,Del

As of this evening I have a CDV-700 sending data to the network. I used a USB/Serial interface and fed the negative pulses from the earphone jack to the RX data input on the Serial side of the interface. The wapp4netc application immediately began tallying the counts. Success.

Who is online

Users browsing this forum: No registered users and 3 guests

You cannot post new topics in this forumYou cannot reply to topics in this forumYou cannot edit your posts in this forumYou cannot delete your posts in this forumYou cannot post attachments in this forum