Product Information

Don't carry and spend money on a loop calibrator, thermocouple calibrator, RTD calibrator, frequency calibrator, pressure calibrator (with optional modules), when you can handle all of these in a Fluke 725 Multifunction Calibrator. Measure and source mA, volts, temperature (RTDs and thermocouples), frequency, ohms, and pressure, using optional pressure modules.

The 725 has a split screen display which permits viewing input and output values simultaneously. The upper display allows you to measure volts, current, and pressure only. The lower display allows you to measure and source volts, current, pressure, resistance temperature detectors, thermocouples, frequency, and ohms. For valve and I/P tests, you can source mA while measuring pressure. The 725 has auto-stepping and auto-ramping for remote testing, plus 25% stepping for fast linearity tests.

From the moment you pick it up, the Fluke 725 is ready to perform. Its simple controls, without menus, make operation easy. Its memory functions make set up fast. And its rugged design lets it work as hard as you do.

What's In The Box

One set of AC72 alligator clips

One set of stackable test leads

One set of TL75 test leads

User manuals

NIST- traceable calibration report and data

Specifications

Measurement Accuracy

Voltage DC

30.000 V

0.02%+ 2 counts (upper display)

30.000 V

0.02%+ 2 counts (lower display)

100.00 mV

0.02%+ 2 counts

-10.00 mV to 75.00 mV

0.025 % + 1 count (via TC connector)

Current DC

24.000 mA:

0.02%+ 2 counts

Resistance

0.0 to 400.0 Ω

0.1Ω (4-wire), 0.15Ω (2- and 3-wire)

401 to 1500 Ω

0.5Ω (4-wire), 1Ω (2- and 3-wire)

1500 to 3200 Ω

1Ω (4-wire), 1.5Ω (2- and 3-wire)

Frequency

2.0 to 1000.0 CPM

0.05 % + 1 count

1.0 to 1100.0 Hz

0.05 % + 1 count

1.00 to 10.00 kHz

0.05 % + 1 count

Sensitivity

1 V peak-to-peak-minimum

Pressure

Accuracy

from 0.025% of range using any of 8 intrinsic safe pressure modules. (for detailed specifications refer to pressure modules in options and accessories)
Modules available for differential, gauge, vacuum, absolute, dual and high pressure.