Information Conveyed to Urologists About Laser Fiber Diameter is Incorrect

A new study investigating lasers used for urological surgery revealed that neither the total nor the core diameters of laser fibers correspond to the advertised laser fiber diameter.

Information Conveyed to Urologists About Laser Fiber Diameter is Incorrect

Furthermore, there are serious differences between manufacturers of fibers with a supposedly equal diameter.

The new study, conducted by Dr. Peter Kronenberg of Hospital Fernando Fonseca in Amadora, Portugal, and Prof. Olivier Traxer of the Hôpital Tenon in Paris, France, aimed to objectively confirm the diameter between laser fibers of supposed equal thickness by different brands.

According to Dr. P. Kronenberg, who recently presented the findings at the 2nd Meeting of the EAU Section of Urolithiasis (EULIS) in Copenhagen, Denmark, urologists need to know the exact technical specifications of the material they use.

"If the information conveyed to them, whether written on a product label or transmitted by an industry representative, is incorrect, their judgments and the decisions they make based on this knowledge may have surgical repercussions," he explained.

"And yet our measurements and survey suggest that most information conveyed to urologists regarding laser fiber diameter is incorrect and that the large majority of industry representatives are not aware of the real diameter of the laser fibers they represent."

In the course of the study 14 different brand-new laser fibers of 6 leading brands (with advertised diameters of 200, 270, 272, 273, 365, and 400 µm, including single use and re-usable fibers) were assessed through light microscopy. Multiple measurements of both the total diameter (including fiber coating) and the fiber core diameter were performed and compared to their respective advertised diameter. A 10% diameter deviation was considered as the maximum tolerance margin to be acceptable due to manufacture procedures. In addition to that, representatives of those brands at the top two major urology congresses were questioned on what the real diameter of their fibers were and if said diameter was the total diameter or only its core.

The total and the core diameters measured were both significantly different from the advertised diameter in all fibers (p<0.00001). If one considers the advertised diameter as the total diameter, none of the fibers had a measured diameter within the 10% tolerance range. The median increase between total diameter and advertised diameter was 85.3% (range 50.7-116.7%), with four of the fibers being more than double as thick as promoted.

If the advertised diameter corresponds to the fiber core diameter, only one fiber respected the 10% margin, the rest of the fibers having a median increase in core diameter of 30.9% (range 16.7-80.1%). The measurement of core diameters of fibers with a supposed equal diameter revealed differences of up to 100 µm in core diameter between manufacturers.

"Due to the ever increasing miniaturization of the instruments today, small differences in laser fiber diameter have repercussions on flexibility, irrigation flow and indirectly on visibility and operating time of ureterorenoscopy," commented Kronenberg on the findings. "And we detected differences in all fibers tested, sometimes well over 100%."

Additionally, in the enquiry, 4 out of 12 representatives said the advertised diameter to be the core diameter, acknowledging that the total fiber diameter was thicker than advertised. Only 1 of those 4 representatives admitted that the core diameter of his fibers was in fact bigger than advertised. The majority of representatives promoted the advertised diameter as the total diameter.

Referring to the implications of the study, Kronenberg explained that such large discrepancies found in the course of the study might warrant the researchers to look into other technical specifications of these and other materials used in endourology.

"It is highly necessary to see if the materials we use are in conformity with the provided information and if there are significant inter-manufacturer differences between materials with supposedly equal characteristics."

Your comments are automatically posted once they are submitted. All comments are however constantly reviewed for spam and irrelevant material (such as product or personal advertisements, email addresses, telephone numbers and website address). Such insertions do not conform to our policy and 'Terms of Use' and are either deleted or edited and republished.Please keep your comments brief and relevant.This section may also have questions seeking help. If you have the information you are welcome to respond, but please ensure that the information so provided is genuine and not misleading.

Disclaimer - All information and content on this site are for information and educational purposes only. The information should not be used for either diagnosis or treatment or both for any health related problem or disease. Always seek the advice of a qualified physician for medical diagnosis and treatment.Full Disclaimer