Six Panel InstaCube Saliva Drug Test

In the event this product is out of stock, would you like a similar product of equal or greater value?:

No - Please call if there are any changes

Yes - Please upgrade

Product sold in cases of 25 tests

1-3 cases

4-11 cases

12-19 cases

20-39 cases

40+ cases

$167.25

$154.75

$148.75

$141.25

$133.75

($6.69 each)

($6.19 each)

($5.95 each)

($5.65 each)

($5.35 each)

* Discounts may vary based on options above

Six Panel InstaCube Saliva Drug Test

With a sleek cube style design, the InstaCube 6 panel saliva drug test has been providing high quality results to clients. With the highest level of drug testing science, we have manufactured a device that is destined to be a leader in the drug screening industry. Our InstaCube saliva drug tests have been put through rigorous quality control and testing to ensure its accuracy and ease of use. Oral drug tests are quickly becoming the preferred method of drug testing due to their less intrusive process and affordability.

The InstaCube 6 panel Oral Drug Test is intended to screen for the presence of drugs and drug metabolites in saliva. This test is for professional in vitro diagnostic use only.

The Oral Fluid Test is a state of the art lateral flow chromatographic immunoassay for the qualitative detection of drugs and drug metabolites in oral fluid at the following cut-off concentrations:

Bring saliva drug test to room temperature (1530?C) prior to testing. Instruct the donor to not place anything in the mouth including food, drink, gum, and tobacco products for at least 10 minutes prior to collection.

Step 1:

Bring the pouch to room temperature before opening it. Remove the test device from the pouch and use it as soon as quickly as possible.

Step 2:

Remove the collector from the sealed pouch. Take out of the collector cap; Insert the sponge into the mouth. Close mouth and move the sponge around for oral fluid collection. Soak sponge in oral fluid and swab the inside of the mouth and tongue. Collect oral fluid for 3 minutes until sponge is soft and fully saturated, No hard spots should be felt on the sponge when saturated.

Step 3:

Place the test device on a clean and level surface. Remove the collection sponge from the mouth and insert it sponge first into the screening device, push down until the collector cap sealed with the device tightly. (See illustration 2?

Step 4:

Leave test device upright on flat surface and keep upright while test is running. Wait for the colored lines to appear in test results area. Read the results at 10 minutes.

Note: Once the collection sponge locks in place, the device is airtight, tamper evident, and ready to be disposed or sent to lab for confirmation (on presumptive positive results)

INTERPRETATION OF RESULTS

Negative Result

NEGATIVE: Two colored bands appear on the membrane. One band appears in the control region (C) and another band appears in the test region (T). Certain lines may appear lighter or thinner than other lines. A line is to be considered a line whether it is faint, light or dark.

*NOTE: The shade of red in the test line region (Drug/T) will vary, but it should be considered negative whenever there is even a faint pink line.

Preliminary Positive Result:

POSITIVE: Only one colored band appears, in the control region (C). No apparent colored band appears in the test region (T).

Invalid Result:

INVALID: Control band fails to appear. Results from any test which has not produced a control band at the specific read time must be discarded. Please review the procedure and repeat with a new test. If the problem persists, discontinue using the kit immediately and contact your local distributor.

The InstaCube Oral Drug Test provides only a qualitative, preliminary analytical result. A secondary analytical method must be used to obtain a confirmed result. Gas chromatography/mass spectrometry (GC/MS) or gas chromatography/tandem mass spectrometry (GC/MS/MS) is the preferred confirmatory method.

There is a possibility that technical or procedural errors, as well as other interfering substances in the oral fluid specimen may cause erroneous results.

A positive test result does not indicate the concentration of drug in the specimen or the route of administration.

STORAGE AND STABILITY

Store as packaged in the sealed pouch either at room temperature or refrigerated (230°C). The test device is stable through the expiration date printed on the sealed pouch.
The test device must remain in the sealed pouch until use. DO NOT FREEZE. Do not use beyond the expiration date.