Easy to learn. Fast to use.The DS-C features the most powerful and intuitive software in its class. Pre-installed DeNovix EasyApps® and a breakthrough Android™ touchscreen interface ensure error-free operation and sample results under 2 seconds. Every instrument is ready right out of the box and backed by an industry-leading three-year warranty.

Autosave millions of samples.

Common quantification assays pre-configured

Easily create custom assays

Fluorescence Freedom!Add a DeNovix USB FX Module to enable powerful fluorescence quantification using the DS-C. DeNovix Fluorescence EasyApps are preloaded on the instrument and instantly become enabled when the USB model is connected. Use fluorescent assays for dsDNA, RNA and protein quantification from DeNovix or several other manufacturers. Four fluorescent channels encompass a wide fluorophore excitation (361-662 nm) and emission (435-740 nm) range. Learn more about the FX Fluorometer Module.

Unrivaled Connectivity and Storage.Built-in Wi-Fi, Ethernet and USB allow easy export of the data you want in the format you need. 32 GB of memory allows storage for millions of results and over 1 million samples.

Email a csv or color jpg of your results right from EasyApps

Automatically save data to network drives or LIMS

Flexible export to easily rename and save the fields you desire

Print results or graphs to network and label printers

Save to USB

DS-C Spectrophotometers are available in your choice of four DeNovix colors.

DS-C Cuvette Spectrophotometer

Attributes

Light Source

Pulsed Xenon flash lamp

Wavelength Range

190
to
840
nm

Wavelength Resolution

.5
nm

Depth

27
cm

Width

20
cm

Power Requirements

10 W; max 30 W

Weight

1.5
kg

Additional Specifications

Beam height:8.5 mm

Heating:37-45 °C +/- 0.5 °C

Pathlength:Up to 10 mm

Absorbance Range:0.0008-1.5 AU (1 cm)

Maximum Concentration:75 ng/µl dsDNA (1 cm)

Detection Limit:0.002 mg/ml BSA; 0.04 ng/µl dsDNA (1 cm)

Detector Type:2048 element CCD

Spectral Resolution:1.5 nm (FWHM at Hg 253.7 nm)

Absorbance Precision:0.015 AU (1 cm), or 1%, whichever is greater

Absorbance Accuracy:1.5% at 0.75 AU at 260 nm

Absorbance Range:0.015 – 1.5 (1 cm equivalent)

Colors:Arctic White, Brazilian Blue, Fire Red, Tungsten Silver

Measurement Time for Spectrophotometry:2 seconds

Measurement Time for Fluorescence:2 seconds

Operating Voltage:12 VDC

Approvals:CE, UL/CSA, FCC and Japan CAB

Manufacture Location:USA

Features

Intuitive EasyApps Software

Wi-Fi, Ethernet, e-mail, Networks and Print

32GB for Millions of Results and Methods

Optional Fluorescence Upgrade

Photos

Manuals and Resources

Note: We appreciate the support of individuals and companies making their documents and resources available to our community. If you find a document or resource that does not belong please let us know by emailing info@labwrench.com