Inductance box R5000 (R 5000, R-5000, p5000, p 5000, p-5000)It is intended for use as a measure of the mutual inductance in DC and AC frequency up to 10 kHz.Rated frequency range: up to 5000Hz.Extended frequency range: up to 10000 Hz.The highest value of the mutual inductance mH store 11,111Accuracy class 0.2

To control the differential transformer devices powered by the AC power frequency 60 Hz, during their production and use.Maximal change of mutual inductance M is ± 13,99 mH.The limit of variation of the angle of dielectric losses 4-11 °.The limit changes residual mutual inductance M is ±0150mkGn.Dimensions: 405x315x240 mm, weight 12 kg.

It is intended to be used as a model measures inductance in AC circuits with a maximum operating frequency of 1500 10000 Hz, depending on the nominal value of exemplary action.
Nominal values: 1Gn, 0.1 GN, 0.001, GN, GN 0.0001
Error: 0.1%

It is intended for use as a measure inductance measurement circuits AC frequency from 20 Hz to 10000 Hz by connecting all decades except the decade "x10", designed for operation at frequencies up to 2500 Hz.