Specialised Monitors

Tiger Select - Benzene

Expanding on the Phocheck Tiger family, the Tiger Select is unique in offering a Total Aromatic Compounds mode in addition to the Benzene specific mode. Tiger Select utilizes the Ion Science high output 10.0 eV detection system displaying a reading for Total Aromatic Compounds (TACs) in the PPB range immediately on start-up. Should aromatics be detected, a Draeger benzene pre-filter tube can be easily attached to ensure rapid detection and selective measurement of benzene. Replacing the 10eV lamp / sensor with the 10.6 allows the instrument to be used as a conventional VOC monitor. In this mode the range is in the PPM but optional PPB and data logging can also be added.

With 10eV lamp, screening for possible benzene is simple and rapid. When aromatic levels remain below 1ppm, benzene levels can be assumed to be below 1ppm and no further action is required.

If levels of aromatic above 1ppm are detected, a Dreager tube is added and a second test to confirm that benzene is present and to quantify the levels.

FEATURES

Resolution 10 PPB

Max range 40 PPM (in Benzene Select mode)

Max range 4000 PPM (in TAC mode)

Run time up to 24hrs

Data Logging 120,000pts

Calibration 2 or 3 pt.

Certified Intrinsically Safe (USETLC)

Tiger Select Kit also includes the standard 10.6eV lamp / sensor allowing the instrument to be used for conventional VOC sampling in the same way as the Phocheck Tiger. In this mode PPM VOC capability is standard. Optional PPB as well as data logging capability is optional.

Part Numbers

ISPTXSLBME0512

TIGER SELECT, STANDARD

ID8103511

BENZENE PREFILTER TUBE 10/PACK

IS861300

HEALTH & SAFETY MODE (ADD ALARM SETTINGS FOR TWA & STEL)

IS861301

PPB MODE (AUTORANGES RESOLUTION FROM 0.1 PPM TO 1 PPB)

IS861303

FULL DATA LOGGING (AT PROGRAMMABLE TIME INTERVALS OR SINGLE POINT KEY ACTIVATED)