The Signal Fire AI-8C optical fiber fusion splicer is the successor to the AI-8 model. The manufacturer slightly changed the design and housing of the device, and redesigned the power and charging system, further increasing its reliability. The number of splices that can be made per battery charge cycle has increased to 240. The AI-8C optical fiber fusion splicer uses the high-speed image process technology and special exact orientation technology, so that the whole process of fiber's fusion can be finished in 10 seconds automatically.

Attention! By default, the device is configured in auto mode. In most cases, any adjustments of the default settings are not required. No calibration is required - it must be done only in case of problems with obtaining the correct splice. Available in the Google Play and AppStore application is required, for example, to change the default splicing program or calibration due to extremely changing environmental conditions.

Changing the parameters and configuration of the splicer is possible only through the Bluetooth interface with dedicated applications on Android / IPhone.

SPECIFICATION

Search for similar

Fiber optic type:

Multi-Mode (MM) - G.651,

Single-Mode (SM) - G.652 & G.657,

Single-Mode (DS / NZDS) Non-Zero Dispersion Shifted Fiber (G.655)

Cladding diameter:

80 ... 150 µm

Outer coating diameter:

0.1 ... 1 mm

Fiber cleaved length:

16 mm

Orientation technology:

core alignment, cladding alignment

Display:

5.1 " LCD, 800 x 480 pxImage: display X and Y axis simultaneously

Average fusion loss:

0.01 dB (MM)0.025 dB (SM)0.04 dB DS / NZDS

Average fusion time:

6 s

Average heat time:

15 s

Fusion programs:

(factory applications) + (user applications)

Internal light:

Internal heater:

Set fusions parameters:

Surface angle test

fiber forward

prefuse power

prefuse time

arc power

arc time, etc

Fusion storage:

Mechanical strength test:

2 N

Computer interface:

USB 2.0 - Possibility to charge smartphones

Power supply:

13.5 V DC / 4 A (power adapter included)

Battery power supply:

11 V DC Li-Ion battery, 7800 mAh

Electrodes lifetime:

~ 3000 fusions

Operation temp:

-15 °C ... 50 °C

Permissible relative humidity:

< 95 % (non-condensing)

Weight:

2.1 kg

Dimensions:

130 x 122 x 148 mm - splicer

270 x 237 x 293 mm - toolbox for tools and splicing machine

Guarantee:

3 years

PRESENTATION

Top view:

Operation elements of the splicer:

Removable battery allows for continuous operation, and replacement is very easy and instant:

Side view:

Fiber cleaver tool:

In the kit:

The kit includes a practical box housing a fusion splicer, the necessary tools and a small table and stool: