eddyMax 4U II ®

Number of Frequencies: 4 adjustable frequency channels.Adaption of the transmitter outputvoltage to the probe, range -40up to 8 dB adjustable in 0.1 dB increments

Probe Matching:Adjustable preamplifier for optimalmatching to the sensitivity of the probe, range 0 up to 78 dB in 6 dB increments with signal level indicator.

Amplifier: Total gain range from -48 up to 126 dB main gain range from -48 bis 48 dB,adjustable in 0,1dB increments,preamplifier range from 0 up to 78 dB,X/Y axis spread from -20 dB up to 20 dB adjustable in 0.1 dB increments

Phase: 360° in 0.1°- increments

Filter: Adjustable low pass and high pass filter range from 0.1 Hz up to 5000 Hz

Flexible Channels:

Depending on the inspection task and the application used for the inspection several channels for signal processing including signal mixing channels areavailable.

Signal Display:

Coloured signal display in impedance and chart mode, switchable to display in impedance mode with several signal windows.The dot can be displayed in store ornonstore mode with highlighted signal trace.

Modes of Operation / Probe Types:The instrument can be used for static testing with hand probes or dynamic operation with rotation probes. All types of probes like SR-probe, bridgeprobes and reflection probes can be connected.

Protocol Functions:- Hardcopy of the screen display- Text editor for labeling the screen display prior to the output- Protocol output with text editor and documentation of the instrument parameters of all used channels