2 or 3 Year Extended Warranties are available for this product. Please refer to the Attachments tab for further information on the program.

Description

The Amron External Hyperbaric Chamber Conditioning System (Model 9000-ECS) is the supply source for chilled or heated water which is circulated through the Amron Internal Hyperbaric Chamber Conditioning System (Amron Model 9100-ICS2) or any heating/cooling module using water as the media.

The Amron External Hyperbaric Chamber Conditioning System (Model 9000-ECS) is manufactured specifically for heating and cooling double lock decompression chambers. When used in conjunction with one or more internal units, the Amron External Hyperbaric Chamber Conditioning System provides temperature and humidity control by cooling and dehumidifying, or heating the hyperbaric chamber environment.

Optional Amron Remote Control Switch 9000-RCS with 25 ft. of cable permits the remote selection of Heat, Cool, or Power-Off. Single switch for easy installation in control console.

2 or 3 Year Extended Warranties are available for this product. Please refer to the Attachments tab for further information on the program.

Videos

Additional Information

Search Weight

75

Brand

Amron

Managed

FEATURED PRODUCT

GSA

Yes

DOD

Yes

Clearance Item

No

Video

No

FAQ

Product FAQ's

Ask a Question

*Name

*Email

*Visibility

*Question

*Security Code Message:

If you have difficulty in reading the image above then refresh your browser a few times until you see an image that is clear enough to copy.

Does the 9000-ECS External Hyperbaric Chamber Conditioning System come with the glycol and water cooling media?

No, prior to operating the system you will have to fill the unit with 50% ethylene or propylene glycol and 50% water. Please note: Do not operate the system without fluid, as the circulating pump requires fluid for lubrication and cooling.

What size chamber is the 9000-ECS External Hyperbaric Chamber Conditioning System unit able to cool/heat?

The 9000-ECS Environmental Control System was designed to provide chilled or heated fluid for temperature control of a standard 54” deck decompression chamber but in some situations it can cool/heat chambers as large as 72”. The 9000-ECS is used in conjunction with an internal conditioning unit (heat exchanger) located inside the deck decompression chamber. On systems larger than 54” please use two internal conditioning units or more as needed.

Any hand operated siphon pump will be able to remove the fluid/antifreeze from the 9000-ECS reservoir. If you want to drain the entire system, you can simply remove the hose connection at the base of the reservoir and allow the fluid to drain out.

There is no problem installing two 9000-ECS units on a chamber. This has been done many times on larger chambers where they require a faster response time for cooling and heating. Please note that each 9000-ECS can support up to three (3) 9100-ICS2’s.