ergoselect 8

The classic examination couch that converts to a full-fledged reclining ergometer with a single hand movement. The ergometry unit is retracted below the couch surface and raised up as needed. Adjustable, removable shoulder supports provide a comfortable counter-surface during exercise tests. The ergoselect 8 design is a perfect fit for both requirements and stands out for its modern and reliable technology.

Details

The ergoselect 8 examination couch stands out for its surface area of 200 cm x 75 cm, with a height of 68 cm, and a pad thickness of 7.5 cm.

The large headrest can be lifted by up to 45° with gas springs. Laterally attached DIN rails enable the attachment of standard accessories (e.g. IV pole).

Max. patient weight up to 225 kg

The robust mechanical design of powder-coated tubular steel guarantees the highest level of safety and trouble-free operation, even with continuous use.

ergoselect 8 is also available in a heavy-duty version for morbidly obese patients up to patient weights of 300 kg.

Functional reclining ergometer

The ergometry unit can be removed from the couch surface and locked in place with a single hand movement. A module for control via PC-ECG or ECG recorder is integrated by default.

As an alternative, further remote control terminals are available on a separate stand. These enable the simple performance of individual ergometric test protocols and manual load adjustment.

Headrest and shoulder supports

To support the patient during a supine ergometric test, headrest and bilaterally adjustable shoulder supports can be attached and simply adjusted to the patient’s height.

Speed display (cadence)

During the ergometric test, the patient display shows the current speed for control, while additional icons warn against overly low or high speed.

The display can be folded away laterally when not in use.

Standard-compliant precision

Thanks to the use of reliable components and high-quality measuring electronics, our devices don’t just meet the standard specifications for medical ergometers, but far exceed them - over the entire load range from 6 to 1000 watts. Moreover, the special two-stage belt drive ensures whisper-quiet operation during an ergometric test.

ECG connection/universal interfaces

All leading ECG manufacturers support a control option for ergoline ergometers in their exercise ECG or ergospirometry systems.

ergoline ergometers offer a wide range of connection options. In addition to the standard RS-232 and USB interface, wireless connections via Bluetooth or WLAN are available as well.

Color variants

The pad cushions are available in many colors to match any practice or hospital interior design.