Preloaded with all necessary software and drivers, this datalogger plugs directly into a USB Port to easily program, analyze, and printout ambient air data.
Temperature, humidity, and atmospheric (barometric) pressure is seamlessly generated into a GMP Compliant full color PDF report with both upper and lower control
limits visible. Not only can a detailed PDF be easily transported on the logger, but the included Excel feature allows the data to be further manipulated by using the
CSV file. The unit is self-contained leaving no need for cradles, cables, software or drivers. Logging up to 16,000 readings each for Temperature, RH and Barometric
Pressure at user defined intervals with date/time stamps is performed in either metric or imperial units of measure. A color-coded threshold alarm light is visible at a distance whether mounted by the included wall mount, freestanding, or using the built-in easel kickstand. The Large LCD display cycles through real-time readings during the recording period and the user also has the option to mark times of interest, adding them to the data report even if it is outside of the programmed interval window. The Min/Max of any time period can be seen for all three measurements even if the device is actively recording. Simply press the button to view this result. Compliant to EN12830:2018, GMP, GLP and is available with an NIST traceable certificate of compliance.

Dimensions: 4 ¼" × 1 ¾" × ¾" (110 x 45 x 17 mm)
Weight: 1.7 oz (50 g)

Specifications

Range

Resolution

Accuracy

Temperature:

-22 to 158°F-30 to 70°C

0.1°F or C

±1.0°F±0.5°C

RH:

0.1 to 80%

0.1%

±3% (@25°C & 10 to 80% RH)otherwise ±5%

Barometer:

300 to 1100 hPa8.9 to 32.5 inHg

0.1

±3 hPa (0 to 60°C)±5 hPa (-20 to 0&degC)

Standards

Certificate of Compliance Traceable to N.I.S.T. (optional)
Specifies that the standards against which the product has been certified or calibrated were themselves calibrated by the National Institute of Standards and Technology (N.I.S.T.),
or with N.I.S.T. calibrated equipment under conditions specified by N.I.S.T.

Sper Scientific warrants this product against defects in materials and workmanship for a the stated period from the date of purchase, and agrees to repair or replace any defective unit without charge. If your model has since been discontinued, an equivalent Sper Scientific product will be substituted if available. This warranty does not cover probes, batteries, or damage resulting from accident, misuse, or abuse.
Opening the meter to expose its electronics will break the waterproof seal and void the warranty.