Monnit Wireless Humidity Sensor - Commercial AA Battery Powered

Also Available As:

Information

Price

$104.00

Format

Frequency

900 MHz

Part#

MNS-9-W2-HU-RH

Lead Length

Certification

Power

Switch

Monnit wireless humidity sensors allow you to monitor the relative humidity of the air within a room or enclosure. Ideal for monitoring humidity within greenhouses, industrial spaces, museums, saunas and humidors. They can also be used for residential applications such as controlling mold, mildew or dust mites.

NIST certified product is available, and can be selected after choosing 3 foot lead length.

Product Description

Monnit wireless humidity sensors allow you to monitor the relative humidity of the air within a room or enclosure. Ideal for monitoring humidity within greenhouses, industrial spaces, museums, saunas and humidors. They can also be used for residential applications such as controlling mold, mildew or dust mites.

NIST certified product is available, and can be selected after choosing 3 foot lead length.

The standard version of this sensor is powered by two replaceable 1.5 V AA sized batteries (included with purchase). This sensor is also available with a line power option. The line powered version of this sensor has a barrel power connector allowing it to be powered by a standard 3.0 - 3.6 V power supply (Included with Line Power option upgrade). The line powered version also uses two standard 1.5 V AA batteries as backup for uninterrupted operation in the event of line power outage.

Power options must be selected at time of purchase, as the internal hardware of the sensor must be changed to support the selected power requirements.

Notes:&nbsp

Product Features

Scientific grade sensor.

+/- 3% accuracy (between 10% - 90% RH).

* Actual range may vary depending on environment.** Battery life is determined by sensor reporting frequency and other variables.

What does this wireless sensor's data look like in software?

What's in the box?

Wireless Sensor

Mounting Hardware

2 x AA Batteries

Humidity Sensor Specifications

RH Operating Range

0 – 100% RH

Accuracy

+/- 3% under normal conditions (10% - 90% RH)

Energy Consumption

80uW (at 12bit, 3V, 1 measurement / s)

RH Response Time

8 sec (tau63%)

AA Specifications

Battery Type

2 Replaceable AA 1.5 V Batteries

Dimension

4.375 in x 2.470 in x 1.12 in

Sensing Element

2.0 - 3.6 V (2 x AA 1.5V Batteries or Optional Power Supply)

Device Range

250 - 300 ft. non-line-of-sight device range*

Battery Life

At 1 hour heartbeat setting, included AA batteries will typically last 3-4 Years. **