Measuring transducer E842

Description E842

Technical characteristics:

№

Type

Iin., mA

Iout., mA

Iconsump. for measured circuit, VA

Load resistance, Ohm

Accuracy class

1

Е842А

0..0,50..10..2,50..5

0..5

not more than 1

0 – 2500

1

2

Е842С

0..20

not more than 1,5

0 – 500

0,5

Dimentions - 70х80х77 mm.Weight- not more than 0,3 kg.Power supply not needed. Powered from measured circuit.Time of setteng working condition (worm-up time):– 5 min for Е842А;– 10 min for Е842С.Time of setting output signal in case of input signal hopping to any value in measured range – not more than 0,1 s.Worktime between failures – not less than 50 000 h.Average lifetime – not less than 10 years.

Е842 indexes:

Е842А - output current (0..5) mAЕ842С - output current (0..20) mA

Catalog:

ABLogger is the device intended for condition monitoring of battery charging devices and electrical systems equipped with a battery (vehicle electrical on-board systems, alternative power sources).