SV 104IS Intrinsically Safe Noise Dosimeter

bestseller

Type approved

The SV 104IS is the intrinsically safe version of the revolutionary personal noise dosimeter, the SV 104. Both instruments offer a new approach to occupational health and safety noise monitoring presenting features such as 1/1 octave band real-time analysis and audio events recording functions which are a new quality in an instrument of this size. All results are clearly displayed on the amazing OLED screen which offers excellent visibility even in full daylight or darkness.

This personal dosimeter has a robust 1/2" MEMS microphone enabling easy calibration using most commonly available acoustic calibrators. The new microphone has a large dynamic range of the 90 dB which allows to measure noise from 60 dBA to 140 dBA. The long list of microphone advantages includes also the autocalibration feature and TEDS memory that stores the calibration info in the microphone itself. The auto-calibration means performing the acoustic calibration automatically once the microphone is inserted into the calibrator.

The SV 104IS is a cable-free dosimeter and is typically attached to the user’s shoulder, close to the ear using the mounting clips supplied. The instrument works with Svantek’s health and safety software package, “Supervisor”, that provides various tools for data analysis and reporting. The docking station supports data transfer to the PC through the infrared interface as well as handles the battery charging. Rechargeable batteries of SV 104IS are able to power the instrument for 50 hours.

The standard kit includes SV104IS noise dosimeter with a SV 27IS shock resistant MEMS and a windscreen with a stainless steel mounting thread. The dosimeter has an inbuilt 64 MB memory and an Infrared interface for communication with PC software (license for PC software is included). Each SV 104IS has its factory calibration certificate and 36-MONTHS WARRANTY CARD for noise dosimeter and LIFE-TIME WARRANTY card for a microphone

Optional Functions

The option for 1/1 OCTAVE real-time analysis allows accurate and correct selection of hearing protectors. When presented as a spectrogram, the 1/1 octave can be used for quick verification of noise sources in the time history. It can be activated at any time, by entering an activation code.

The AUDIO EVENTS RECORDING option works during measurement and is logged in parallel to time history so it can be played back in the PC software. The settings, like triggers or recording time, are adjustable. It can be activated at any time, by entering an activation code.

Optional Accessories

SA 104-1 Docking Station for Single Dosimeter

SA 104-5 Docking Station for 5 Dosimeters

SA 144 Carrying Case for 5 Dosimeters and Docking Station for 5 Units

SA 147 Waterproof Carrying Case for Noise Dosimeter and Single Docking Station