Introduction: Three protection: operator, sample and environment. Airflow system: 70% air recirculation, 30% air exhaustA2 Cabinet is suitable for working with microbiological research in the absence of volatile or toxic chemicals and radionuclide.

Advantage: 1. Motorized front window. 2. Time reserve function.3. HEPA filter life and UV life indicator. 4. Automatic air speed adjustable with filter block.5. Work area surrounded by negative pressure, it can ensure maximum safety in work area.6. Audio and visual alarm (Filter replacement, window over height, abnormal Air flow velocity etc).7. Most accessories are standard. No need to pay more, save your money.8. Remote control: all functions can be realized with it, making the operation much easier and more convenient.9. Foot switch: adjust front window height by foot during experiment, to avoid airflow turbulence caused byarm movement.10. Interlock function: UV lamp and front window; UV lamp and blower, fluorescent lamp; blower and front window.

Model

BSC-1100IIA2-X

BSC-1300IIA2-X

BSC-1500IIA2-X

BSC-1800IIA2-X

BSC-2000IIA2-X

Internal Size(W*D*H)

940* 600*660 mm

1150*600*660 mm

1350*600*660mm

1710*600*660 mm

1800*600*660mm

External Size(W*D*H)

1100*750*2250mm

1300*750*2250mm

1500*750*2250mm

1870*750*2290mm

1950*800*2305mm

Tested Opening

Safety Height = 200 mm

Max Opening

420mm

420mm

400mm

400mm

400mm

Inflow Velocity

0.53±0.025 m/s

Downflow Velocity

0.33±0.025 m/s

HEPA Filter

Two, 99.999% efficient at 0.3 um. Filter life indicator.

Front Window

Motorized. Two-layer laminated toughened glass ≥ 5mm. Anti UV.

Noise

EN12469 ≤ 58 dB / NSF49 ≤ 61 dB

UV Lamp

30W*1

30W* 1

40W* 1

40W*2

40W* 1

UV timer, UV life indicator, emission of 253.7 nanometers for most efficient decontamination.

Lamp

LED Lamp

Fluorescent Lamp

12W*2

14W*2

16W*2

40W*2

40W*2

Illumination

≥1000Lux

Consumption

760W

800W

900W

900W

1500W

Waterproof Sockets

Two, total load of two sockets: 500W

Display

LCD display: exhaust filter and downflow filter pressure, filter and UV lamp working time, inflow and downflow velocity, filter life, humidity and temperature, system working time etc.