John M. Pauly

Reid Weaver Dennis Professor

Electrical Engineering

Bio

Bio

Interests include medical imaging generally, and magnetic resonance imaging (MRI) in particular. Current efforts are focused on medical applications of MRI where real-time interactive imaging is important. Two examples are cardiac imaging, and the interactive guidance of interventional procedures. Specific interests include rapid methods for the excitation and acquisition of the MR signal, and the reconstruction of images from the data acquired using these approaches.

Abstract

A calibrationless parallel imaging technique developed previously for (1)H MRI was modified and tested for hyperpolarized (13)C MRI for applications requiring large FOV and high spatial resolution. The technique was demonstrated with both retrospective and prospective under-sampled data acquired in phantom and in vivo rat studies. A 2-fold acceleration was achieved using a 2D symmetric EPI readout equipped with random blips on the phase encode dimension. Reconstructed images showed excellent qualitative agreement with fully sampled data. Further acceleration can be achieved using acquisition schemes that incorporate multi-dimensional under-sampling.

Abstract

Selective RF pulses are commonly designed with the desired profile as a low pass filter frequency response. However, for many MRI and NMR applications, the spectrum is sparse with signals existing at a few discrete resonant frequencies. By specifying a multiband profile and releasing the constraint on "don't-care" regions, the RF pulse performance can be improved to enable a shorter duration, sharper transition, or lower peak B1 amplitude. In this project, a framework for designing multiband RF pulses with improved performance was developed based on the Shinnar-Le Roux (SLR) algorithm and convex optimization. It can create several types of RF pulses with multiband magnitude profiles, arbitrary phase profiles and generalized flip angles. The advantage of this framework with a convex optimization approach is the flexible trade-off of different pulse characteristics. Designs for specialized selective RF pulses for balanced SSFP hyperpolarized (HP) (13)C MRI, a dualband saturation RF pulse for (1)H MR spectroscopy, and a pre-saturation pulse for HP (13)C study were developed and tested.

Abstract

To develop and assess motion correction techniques for high-resolution pediatric abdominal volumetric magnetic resonance images acquired free-breathing with high scan efficiency.First, variable-density sampling and radial-like phase-encode ordering were incorporated into the 3D Cartesian acquisition. Second, intrinsic multichannel butterfly navigators were used to measure respiratory motion. Lastly, these estimates are applied for both motion-weighted data-consistency in a compressed sensing and parallel imaging reconstruction, and for nonrigid motion correction using a localized autofocusing framework. With Institutional Review Board approval and informed consent/assent, studies were performed on 22 consecutive pediatric patients. Two radiologists independently scored the images for overall image quality, degree of motion artifacts, and sharpness of hepatic vessels and the diaphragm. The results were assessed using paired Wilcoxon test and weighted kappa coefficient for interobserver agreements.The complete procedure yielded significantly better overall image quality (mean score of 4.7 out of 5) when compared to using no correction (mean score of 3.4, P

Abstract

This study assessed T1ρ relaxation dispersion, measured by magnetic resonance imaging (MRI), as a tool to noninvasively evaluate cartilage material and biochemical properties. The specific objective was to answer two questions: (1) does cartilage initial elastic modulus (E 0) correlate with T1ρ dispersion effects and (2) does collagen or proteoglycan content correlate with T1ρ dispersion effects?Cadaveric patellae with and without visible cartilage damage on conventional MR were included. T2 and T1ρ relaxation times at 500 and 1000 Hz spin-lock field amplitudes were measured. We estimated T1ρ dispersion effects by measuring T1ρ relaxation time at 500 and 1000 Hz and T2 relaxation time and using a new tool, the ratio T1ρ/T2. Cartilage initial elastic modulus, E 0, was measured from initial response of mechanical indentation creep tests. Collagen and proteoglycan contents were measured at the indentation test sites; proteoglycan content was measured by their covalently linked sulfated glycosaminoglycans (sGAG). Pearson correlation coefficients were determined, taking into account the clustering of multiple samples within a single patella specimen.Cartilage initial elastic modulus, E 0, increased with decreasing values of T1ρ/T2 measurements at both 500 Hz (P = 0.034) and 1000 Hz (P = 0.022). 1/T1ρ relaxation time (500 Hz) increased with increasing sGAG content (P = 0.041).T1ρ/T2 ratio, a new tool, and cartilage initial elastic modulus are both measures of water-protein interactions, are dependent on the cartilage structure, and were correlated in this study.

Abstract

The development of catheters and guidewires that are safe from radiofrequency (RF) -induced heating and clearly visible against background tissue is a major challenge in interventional MRI. An interventional imaging approach using a toroidal transmit-receive (transceive) coil is presented. This toroidal transceiver allows controlled, low levels of RF current to flow in the catheter/guidewire for visualization, and can be used with conductive interventional devices that have a localized low-impedance tip contact.Toroidal transceivers were built, and phantom experiments were performed to quantify transmit power levels required for device visibility and to detect heating hazards. Imaging experiments in a pig cadaver tested the extendibility to higher field strength and nonphantom settings. A photonically powered optically coupled toroidal current sensor for monitoring induced RF currents was built, calibrated, and tested using an independent image-based current estimation method.Results indicate that high signal-to-noise ratio visualization is achievable using milliwatts of transmit power-power levels orders of magnitude lower than levels that induce measurable heating in phantom tests. Agreement between image-based current estimates and RF current sensor measurements validates sensor accuracy.The toroidal transceiver, integrated with power and current sensing, could offer a promising platform for safe and effective interventional device visualization.

Abstract

The concept of a "radiofrequency safety prescreen" is investigated, wherein dangerous interactions between radiofrequency fields used in MRI, and conductive implants in patients are detected through impedance changes in the radiofrequency coil.The behavior of coupled oscillators is reviewed, and the resulting, observable impedance changes are discussed.A birdcage coil is loaded with a static head phantom and a wire phantom with a wire close to its resonant length, the shape, position, and orientation of which can be changed. Interactions are probed with a current sensor and network analyzer.Impedance spectra show dramatic, unmistakable splitting in cases of strong coupling, and strong correlation is observed between induced current and scattering parameters.The feasibility of a new, low-power prescreening technique has been demonstrated in a simple phantom experiment, which can unambiguously detect resonant interactions between an implanted wire and an imaging coil. A new technique has also been presented which can detect parallel transmit null modes for the wire.

Abstract

A calibrationless parallel imaging reconstruction method, termed simultaneous autocalibrating and k-space estimation (SAKE), is presented. It is a data-driven, coil-by-coil reconstruction method that does not require a separate calibration step for estimating coil sensitivity information.In SAKE, an undersampled, multichannel dataset is structured into a single data matrix. The reconstruction is then formulated as a structured low-rank matrix completion problem. An iterative solution that implements a projection-onto-sets algorithm with singular value thresholding is described.Reconstruction results are demonstrated for retrospectively and prospectively undersampled, multichannel Cartesian data having no calibration signals. Additionally, non-Cartesian data reconstruction is presented. Finally, improved image quality is demonstrated by combining SAKE with wavelet-based compressed sensing.Because estimation of coil sensitivity information is not needed, the proposed method could potentially benefit MR applications where acquiring accurate calibration data is limiting or not possible at all.

Abstract

Magnetic resonance spectroscopy of hyperpolarized substrates allows for the observation of label exchange catalyzed by enzymes providing a powerful tool to investigate tissue metabolism and potentially kinetics in vivo. However, the accuracy of current methods to calculate kinetic parameters has been limited by T1 relaxation effects, extracellular signal contributions, and reduced precision at lower signal-to-noise ratio.To address these challenges, we investigated a new modeling technique using metabolic activity decomposition-stimulated echo acquisition mode. The metabolic activity decomposition-stimulated echo acquisition mode technique separates exchanging from nonexchanging metabolites providing twice the information as conventional techniques.This allowed for accurate measurements of rates of conversion and of multiple T1 values simultaneously using a single acquisition.The additional measurement of T1 values for the reaction metabolites provides further biological information about the cellular environment of the metabolites. The new technique was investigated through simulations and in vivo studies of transgenic mouse models of cancer demonstrating improved assessments of kinetic rate constants and new T1 relaxation value measurements for hyperpolarized (13) C-pyruvate, (13) C-lactate, and (13) C-alanine.

Abstract

In metabolic MRI with hyperpolarized contrast agents, the signal levels vary over time due to T1 decay, T2 decay following RF excitations, and metabolic conversion. Efficient usage of the nonrenewable hyperpolarized magnetization requires specialized RF pulse schemes. In this work, we introduce two novel variable flip angle schemes for dynamic hyperpolarized MRI in which the flip angle is varied between excitations and between metabolites. These were optimized to distribute the magnetization relatively evenly throughout the acquisition by accounting for T1 decay, prior RF excitations, and metabolic conversion. Simulation results are presented to confirm the flip angle designs and evaluate the variability of signal dynamics across typical ranges of T1 and metabolic conversion. They were implemented using multiband spectral-spatial RF pulses to independently modulate the flip angle at various chemical shift frequencies. With these schemes we observed increased SNR of [1-(13)C]lactate generated from [1-(13)C]pyruvate, particularly at later time points. This will allow for improved characterization of tissue perfusion and metabolic profiles in dynamic hyperpolarized MRI.

Abstract

Metabolic imaging with hyperpolarized [1-(13)C]-pyruvate can rapidly probe tissue metabolic profiles in vivo and has been shown to provide cancer imaging biomarkers for tumor detection, progression, and response to therapy. This technique uses a bolus injection followed by imaging within 1-2 minutes. The observed metabolites include vascular components and their generation is also influenced by cellular transport. These factors complicate image interpretation, especially since [1-(13)C]lactate, a metabolic product that is a biomarker of cancer, is also produced by red blood cells. It would be valuable to understand the distribution of metabolites between the vasculature, interstitial space, and intracellular compartments. The purpose of this study was to better understand this compartmentalization by using a perfusion and diffusion-sensitive stimulated-echo acquisition mode (STEAM) MRSI acquisition method tailored to hyperpolarized substrates. Our results in mouse models showed that among metabolites, the injected substrate (13)C-pyruvate had the largest vascular fraction overall while (13)C-alanine had the smallest vascular fraction. We observed a larger vascular fraction of pyruvate and lactate in the kidneys and liver when compared to back muscle and prostate tumor tissue. Our data suggests that (13)C-lactate in prostate tumor tissue voxels was the most abundant labeled metabolite intracellularly. This was shown in STEAM images that highlighted abnormal cancer cell metabolism and suppressed vascular (13)C metabolite signals.

Abstract

MRI using receiver arrays with many coil elements can provide high signal-to-noise ratio and increase parallel imaging acceleration. At the same time, the growing number of elements results in larger datasets and more computation in the reconstruction. This is of particular concern in 3D acquisitions and in iterative reconstructions. Coil compression algorithms are effective in mitigating this problem by compressing data from many channels into fewer virtual coils. In Cartesian sampling there often are fully sampled k-space dimensions. In this work, a new coil compression technique for Cartesian sampling is presented that exploits the spatially varying coil sensitivities in these nonsubsampled dimensions for better compression and computation reduction. Instead of directly compressing in k-space, coil compression is performed separately for each spatial location along the fully sampled directions, followed by an additional alignment process that guarantees the smoothness of the virtual coil sensitivities. This important step provides compatibility with autocalibrating parallel imaging techniques. Its performance is not susceptible to artifacts caused by a tight imaging field-of-view. High quality compression of in vivo 3D data from a 32 channel pediatric coil into six virtual coils is demonstrated.

Abstract

In this work, we present a new MR spectroscopy approach for directly observing nuclear spins that undergo exchange, metabolic conversion, or, generally, any frequency shift during a mixing time. Unlike conventional approaches to observe these processes, such as exchange spectroscopy (EXSY), this rapid approach requires only a single encoding step and thus is readily applicable to hyperpolarized MR in which the magnetization is not replenished after T(1) decay and RF excitations. This method is based on stimulated-echoes and uses phase-sensitive detection in conjunction with precisely chosen echo times in order to separate spins generated during the mixing time from those present prior to mixing. We are calling the method Metabolic Activity Decomposition Stimulated-echo Acquisition Mode or MAD-STEAM. We have validated this approach as well as applied it in vivo to normal mice and a transgenic prostate cancer mouse model for observing pyruvate-lactate conversion, which has been shown to be elevated in numerous tumor types. In this application, it provides an improved measure of cellular metabolism by separating [1-(13)C]-lactate produced in tissue by metabolic conversion from [1-(13)C]-lactate that has flowed into the tissue or is in the blood. Generally, MAD-STEAM can be applied to any system in which spins undergo a frequency shift.

Abstract

MR scans are sensitive to motion effects due to the scan duration. To properly suppress artifacts from nonrigid body motion, complex models with elements such as translation, rotation, shear, and scaling have been incorporated into the reconstruction pipeline. However, these techniques are computationally intensive and difficult to implement for online reconstruction. On a sufficiently small spatial scale, the different types of motion can be well approximated as simple linear translations. This formulation allows for a practical autofocusing algorithm that locally minimizes a given motion metric--more specifically, the proposed localized gradient-entropy metric. To reduce the vast search space for an optimal solution, possible motion paths are limited to the motion measured from multichannel navigator data. The novel navigation strategy is based on the so-called "Butterfly" navigators, which are modifications of the spin-warp sequence that provides intrinsic translational motion information with negligible overhead. With a 32-channel abdominal coil, sufficient number of motion measurements were found to approximate possible linear motion paths for every image voxel. The correction scheme was applied to free-breathing abdominal patient studies. In these scans, a reduction in artifacts from complex, nonrigid motion was observed.

Abstract

When evaluating the severity of valvular stenosis, the peak velocity of the blood flow is routinely used to estimate the transvalvular pressure gradient. One-dimensional Fourier velocity encoding effectively detects the peak velocity with an ungated time series of spatially resolved velocity spectra in real time. However, measurement accuracy can be degraded by the pulsatile and turbulent nature of stenotic flow and the existence of spatially varying off-resonance. In this work, we investigate the feasibility of improving the peak velocity detection capability of one-dimensional Fourier velocity encoding for stenotic flow using a novel echo-shifted interleaved readout combined with a variable-density circular k-space trajectory. The shorter echo and readout times of the echo-shifted interleaved acquisitions are designed to reduce sensitivity to off-resonance. Preliminary results from limited phantom and in vivo results also indicate that some artifacts from pulsatile flow appear to be suppressed when using this trajectory compared to conventional single-shot readouts, suggesting that peak velocity detection may be improved. The efficiency of the new trajectory improves the temporal and spatial resolutions. To realize the proposed readout, a novel multipoint-traversing algorithm is introduced for flexible and automated gradient-waveform design.

Abstract

Spin echo pulse sequences are used to produce clinically important T(2) contrast. However, conventional 180° radiofrequency pulses required to generate a spin echo are highly susceptible to the B(1) inhomogeneity at high magnetic fields such as 7 Tesla (7 T), resulting in varying signal and contrast over the region of interest. Adiabatic 180° pulses may be used to replace conventional 180° pulses in spin echo sequences to provide greater immunity to the inhomogeneous B(1) field at 7 T. However, because the spectral profile of an adiabatic 180° pulse has nonlinear phase, pairs of these pulses are needed for proper refocusing, resulting in increased radiofrequency power deposition and long minimum echo times. We used the adiabatic Shinnar Le-Roux method to generate a matched-phase adiabatic 90°-180° pulse pair to obviate the need for a second adiabatic 180° pulse for phase refocusing. The pulse pair was then reformulated into a single self-refocused pulse to minimize the echo time, and phantom and in vivo experiments were performed to validate pulse performance. The self-refocused adiabatic pulse produced transmit profiles that were substantially more uniform than those achieved using a conventional spin echo sequence.

Abstract

Radio-frequency ablation (RFA) is an effective minimally invasive treatment for tumors. One primary source of difficulty is monitoring and controlling the ablation region. Currently, RFA is performed at 460 kHz, for which magnetic resonance imaging (MRI) could play a role given its capability for temperature monitoring and tumor visualization. If instead the ablation were to be performed at the MRI Larmor frequency, then the MR capability for B(1) field mapping could be used to directly visualize the radio-frequency (RF) fields created by the ablation currents. Visualizing the RF fields may enable better control of the ablation currents, enabling better control of lesion shape and size and improving repeatability. We demonstrate the feasibility of performing RFAs at 64 MHz and show preliminary results from imaging the RF fields from the ablation. The post-ablation RF fields show an increase in current density in the ablated region, consistent with an increase in conductivity of the ablated tissue.

Abstract

A rapid echo planar imaging sequence for dynamic imaging of [1-(13)C] lactate and [1-(13)C] pyruvate simultaneously was developed. Frequency-based separation of these metabolites was achieved by spatial shifting in the phase-encoded direction with the appropriate choice of echo spacing. Suppression of the pyruvate-hydrate and alanine resonances is achieved through an optimized spectral-spatial RF waveform. Signal sampling efficiency as a function of pyruvate and lactate excitation angle was simulated using two site exchange models. Dynamic imaging is demonstrated in a transgenic mouse model, and phantom validations of the RF pulse frequency selectivity were performed.

Abstract

Magnetic resonance imaging (MRI) pulse sequence consoles typically employ closed proprietary hardware, software, and interfaces, making difficult any adaptation for innovative experimental technology. Yet MRI systems research is trending to higher channel count receivers, transmitters, gradient/shims, and unique interfaces for interventional applications. Customized console designs are now feasible for researchers with modern electronic components, but high data rates, synchronization, scalability, and cost present important challenges. Implementing large multichannel MR systems with efficiency and flexibility requires a scalable modular architecture. With Medusa, we propose an open system architecture using the universal serial bus (USB) for scalability, combined with distributed processing and buffering to address the high data rates and strict synchronization required by multichannel MRI. Medusa uses a modular design concept based on digital synthesizer, receiver, and gradient blocks, in conjunction with fast programmable logic for sampling and synchronization. Medusa is a form of synthetic instrument, being reconfigurable for a variety of medical/scientific instrumentation needs. The Medusa distributed architecture, scalability, and data bandwidth limits are presented, and its flexibility is demonstrated in a variety of novel MRI applications.

Abstract

Stimulated-echoes in MR can be used to provide high sensitivity to motion and flow, creating diffusion and perfusion weighting as well as T(1) contrast, but conventional approaches inherently suffer from a 50% signal loss. The super stimulated-echo, which uses a specialized radio-frequency (RF) pulse train, has been proposed in order to improve the signal while preserving motion and T(1) sensitivity. This paper presents a novel and straightforward method for designing the super stimulated-echo pulse train using inversion pulse design techniques. This method can also create adiabatic designs with an improved response to RF transmit field variations. The scheme was validated in phantom experiments and shown in vivo to improve signal-to-noise ratio (SNR). We have applied a super stimulated-echo to metabolic MRI with hyperpolarized (13)C-labeled molecules. For spectroscopic imaging of hyperpolarized agents, several repetition times are required but only a single stimulated-echo encoding is feasible, which can lead to unwanted motion blurring. To address this, a super stimulated-echo preparation scheme was used in which the diffusion weighting is terminated prior to the acquisition, and we observed a SNR increases of 60% in phantoms and 49% in vivo over a conventional stimulated-echo. Experiments following injection of hyperpolarized [1-(13)C] -pyruvate in murine transgenic cancer models have shown improved delineation for tumors since signals from metabolites within tumor tissues are retained while those from the vasculature are suppressed by the diffusion preparation scheme.

Abstract

The article "A k-space analysis of small-tip-angle excitation" introduced a spatial frequency interpretation of the effect of RF excitation pulses. This introduction describes where the initial ideas for this paper came from, and traces out some of the applications that have been developed using this perspective.

Abstract

In this article, cross-relaxation imaging is applied to human ex vivo knee cartilage, and correlations of the cross-relaxation imaging parameters with macromolecular content in articular cartilage are reported. We show that, unlike the more commonly used magnetization transfer ratio, the bound pool fraction, the cross-relaxation rate (k) and the longitudinal relaxation time (T(1)) vary with depth and can therefore provide insight into the differences between the top and bottom layers of articular cartilage. Our cross-relaxation imaging model is more sensitive to macromolecular content in the top layers of cartilage, with bound pool fraction showing moderate correlations with proteoglycan content, and k and T(1) exhibiting moderate correlations with collagen.

Abstract

The purpose of this article is to review some of the basic principles of imaging and how metal-induced susceptibility artifacts originate in MR images. We will describe common ways to reduce or modify artifacts using readily available imaging techniques, and we will discuss some advanced methods to correct readout-direction and slice-direction artifacts.The presence of metallic implants in MRI can cause substantial image artifacts, including signal loss, failure of fat suppression, geometric distortion, and bright pile-up artifacts. These cause large resonant frequency changes and failure of many MRI mechanisms. Careful parameter and pulse sequence selections can avoid or reduce artifacts, although more advanced imaging methods offer further imaging improvements.

Abstract

Non-Cartesian imaging provides many advantages in terms of flexibility, functionality, and speed. However, a major drawback to these imaging methods is off-resonance distortion artifacts. These artifacts manifest as blurring in spiral imaging. Common techniques that remove the off-resonance field inhomogeneity distortion effects are not sufficient, because the high order concomitant gradient fields are nontrivial for common imaging conditions, such as imaging 5 cm off isocenter in an 1.5 T scanner. Previous correction algorithms are either slow or do not take into account the known effects of concomitant gradient fields along with the field inhomogeneities. To ease the correction, the distortion effects are modeled as a non-stationary convolution problem. In this work, two fast and accurate postgridding algorithms are presented and analyzed. These methods account for both the concomitant field effects and the field inhomogeneities. One algorithm operates in the frequency domain and the other in the spatial domain. To take advantage of their speed and accuracy, the algorithms are applied to a real-time cardiac study and a high-resolution cardiac study. Both of the presented algorithms provide for a practical solution to the off-resonance problem in spiral imaging.

Abstract

Magnetic resonance imaging (MRI) near metallic implants is often hampered by severe metal artifacts. To obtain distortion-free MR images near metallic implants, SEMAC (Slice Encoding for Metal Artifact Correction) corrects metal artifacts via robust encoding of excited slices against metal-induced field inhomogeneities, followed by combining the data resolved from multiple SEMAC-encoded slices. However, as many of the resolved data elements only contain noise, SEMAC-corrected images can suffer from relatively low signal-to-noise ratio. Improving the signal-to-noise ratio of SEMAC-corrected images is essential to enable SEMAC in routine clinical studies. In this work, a new reconstruction procedure is proposed to reduce noise in SEMAC-corrected images. A singular value decomposition denoising step is first applied to suppress quadrature noise in multi-coil SEMAC-encoded slices. Subsequently, the singular value decomposition-denoised data are selectively included in the correction of through-plane distortions. The experimental results demonstrate that the proposed reconstruction procedure significantly improves the SNR without compromising the correction of metal artifacts.

Abstract

Hyperpolarized 13C MR spectroscopic imaging can detect not only the uptake of the pre-polarized molecule but also its metabolic products in vivo, thus providing a powerful new method to study cellular metabolism. Imaging the dynamic perfusion and conversion of these metabolites provides additional tissue information but requires methods for efficient hyperpolarization usage and rapid acquisitions. In this work, we have developed a time-resolved 3D MR spectroscopic imaging method for acquiring hyperpolarized 13C data by combining compressed sensing methods for acceleration and multiband excitation pulses to efficiently use the magnetization. This method achieved a 2 sec temporal resolution with full volumetric coverage of a mouse, and metabolites were observed for up to 60 sec following injection of hyperpolarized [1-(13)C]-pyruvate. The compressed sensing acquisition used random phase encode gradient blips to create a novel random undersampling pattern tailored to dynamic MR spectroscopic imaging with sampling incoherency in four (time, frequency, and two spatial) dimensions. The reconstruction was also tailored to dynamic MR spectroscopic imaging by applying a temporal wavelet sparsifying transform to exploit the inherent temporal sparsity. Customized multiband excitation pulses were designed with a lower flip angle for the [1-(13)C]-pyruvate substrate given its higher concentration than its metabolic products ([1-(13)C]-lactate and [1-(13)C]-alanine), thus using less hyperpolarization per excitation. This approach has enabled the monitoring of perfusion and uptake of the pyruvate, and the conversion dynamics to lactate and alanine throughout a volume with high spatial and temporal resolution.

Abstract

A relationship between T1? relaxation time and glycosaminoglycan (GAG) content has been demonstrated in chemically degraded bovine cartilage, but has not been demonstrated with quantitative biochemistry in human cartilage. A relationship has also been established between T2 relaxation time in cartilage and osteoarthritis (OA) severity. We hypothesized that T1? relaxation time would be associated with GAG content in human cartilage with normal T2 relaxation times.T2 relaxation time, T1? relaxation time, and glycosaminoglycan as a percentage of wet weight (sGAG) were measured for top and bottom regions at 7 anatomical locations in 21 human cadaver patellae. For our analysis, T2 relaxation time was classified as normal or elevated based on a threshold defined by the mean plus one standard deviation of the T2 relaxation time for all samples.In the normal T2 relaxation time subset, T1? relaxation time correlated with sGAG content in the full-thickness and bottom regions, but only marginally in the top region alone. sGAG content decreased significantly with age in all regions.In the subset of cartilage specimens with normal T2 relaxation time, T1? relaxation time was inversely associated with sGAG content, as hypothesized. A predictive model, which accounts for T2 relaxation time and the effects of age, might be able to determine longitudinal trends in GAG content in the same person based on T1? relaxation time maps.

Abstract

High-quality magnetic resonance imaging (MRI) requires precise control of the transmit radio-frequency (RF) field. In parallel excitation applications such as transmit SENSE, high RF power linearity is essential to cancel aliased excitations. In widely-employed class AB power amplifiers, gain compression, cross-over distortion, memory effects, and thermal drift all distort the RF field modulation and can degrade image quality. Cartesian feedback (CF) linearization can mitigate these effects in MRI, if the quadrature mismatch and dc offset imperfections inherent in the architecture can be minimized. In this paper, we present a modified Cartesian feedback technique called "frequency-offset Cartesian feedback" (FOCF) that significantly reduces these problems. In the FOCF architecture, the feedback control is performed at a low intermediate frequency rather than dc, so that quadrature ghosts and dc errors are shifted outside the control bandwidth. FOCF linearization is demonstrated with a variety of typical MRI pulses. Simulation of the magnetization obtained with the Bloch equation demonstrates that high-fidelity RF reproduction can be obtained even with inexpensive class AB amplifiers. Finally, the enhanced RF fidelity of FOCF over CF is demonstrated with actual images obtained in a 1.5 T MRI system.

Abstract

Diffusion imaging and bound pool fraction (BPF) mapping are two quantitative magnetic resonance imaging techniques that measure microstructural features of the white matter of the brain. Diffusion imaging provides a quantitative measure of the diffusivity of water in tissue. BPF mapping is a quantitative magnetization transfer (qMT) technique that estimates the proportion of exchanging protons bound to macromolecules, such as those found in myelin, and is thus a more direct measure of myelin content than diffusion. In this work, we combined BPF estimates of macromolecular content with measurements of diffusivity within human white matter tracts. Within the white matter, the correlation between BPFs and diffusivity measures such as fractional anisotropy and radial diffusivity was modest, suggesting that diffusion tensor imaging and bound pool fractions are complementary techniques. We found that several major tracts have high BPF, suggesting a higher density of myelin in these tracts. We interpret these results in the context of a quantitative tissue model.

Abstract

Multi echo chemical shift-based water-fat separation methods allow for uniform fat suppression in the presence of main field inhomogeneities. However, these methods require additional scan time for chemical shift encoding. This work presents a method for water-fat separation from undersampled data (CS-WF), which combines compressed sensing and chemical shift-based water-fat separation. Undersampling was applied in the k-space and in the chemical shift encoding dimension to reduce the total scanning time. The method can reconstruct high quality water and fat images in 2D and 3D applications from undersampled data. As an extension, multipeak fat spectral models were incorporated into the CS-WF reconstruction to improve the water-fat separation quality. In 3D MRI, reduction factors of above three can be achieved, thus fully compensating the additional time needed in three-echo water-fat imaging. The method is demonstrated on knee and abdominal in vivo data.

Abstract

Parallel excitation uses multiple transmit channels and coils, each driven by independent waveforms, to afford the pulse designer an additional spatial encoding mechanism that complements gradient encoding. In contrast to parallel reception, parallel excitation requires individual power amplifiers for each transmit channel, which can be cost prohibitive. Several groups have explored the use of low-cost power amplifiers for parallel excitation; however, such amplifiers commonly exhibit nonlinear memory effects that distort radio frequency pulses. This is especially true for pulses with rapidly varying envelopes, which are common in parallel excitation. To overcome this problem, we introduce a technique for parallel excitation pulse design that yields pulses with smoother envelopes. We demonstrate experimentally that pulses designed with the new technique suffer less amplifier distortion than unregularized pulses and pulses designed with conventional regularization.

Abstract

Temperature estimation in proton resonance frequency (PRF) shift MR thermometry requires a reference, or pretreatment, phase image that is subtracted from image phase during thermal treatment to yield a phase difference image proportional to temperature change. Referenceless thermometry methods derive a reference phase image from the treatment image itself by assuming that in the absence of a hot spot, the image phase can be accurately represented in a smooth (usually low order polynomial) basis. By masking the hot spot out of a least squares (?(2)) regression, the reference phase image's coefficients on the polynomial basis are estimated and a reference image is derived by evaluating the polynomial inside the hot spot area. Referenceless methods are therefore insensitive to motion and bulk main field shifts, however, currently these methods require user interaction or sophisticated tracking to ensure that the hot spot is masked out of the polynomial regression. This article introduces an approach to reference PRF shift thermometry that uses reweighted ?(1) regression, a form of robust regression, to obtain background phase coefficients without hot spot tracking and masking. The method is compared to conventional referenceless thermometry, and demonstrated experimentally in monitoring HIFU heating in a phantom and canine prostate, as well as in a healthy human liver.

Abstract

Adiabatic pulses are a special class of radio frequency (RF) pulses that may be used to achieve uniform flip angles in the presence of a nonuniform B(1) field. In this work, we present a new, systematic method for designing high-bandwidth (BW), low-peak-amplitude adiabatic RF pulses that utilizes the Shinnar-Le Roux (SLR) algorithm for pulse design. Currently, the SLR algorithm is extensively employed to design nonadiabatic pulses for use in magnetic resonance imaging and spectroscopy. We have adapted the SLR algorithm to create RF pulses that also satisfy the adiabatic condition. By overlaying sufficient quadratic phase across the spectral profile before the inverse SLR transform, we generate RF pulses that exhibit the required spectral characteristics and adiabatic behavior. Application of quadratic phase also distributes the RF energy more uniformly, making it possible to obtain the same spectral BW with lower RF peak amplitude. The method enables the pulse designer to specify spectral profile parameters and the degree of quadratic phase before pulse generation. Simulations and phantom experiments demonstrate that RF pulses designed using this new method behave adiabatically.

Abstract

Patients with long-wire medical implants are currently prevented from undergoing magnetic resonance imaging (MRI) scans due to the risk of radio frequency (RF) heating. We have developed a simple technique for determining the heating potential for these implants using reversed radio frequency (RF) polarization. This technique could be used on a patient-to-patient basis as a part of the standard prescan procedure to ensure that the subject's device does not pose a heating risk. By using reversed quadrature polarization, the MR scan can be sensitized exclusively to the potentially dangerous currents in the device. Here, we derive the physical principles governing the technique and explore the primary sources of inaccuracy. These principles are verified through finite-difference simulations and through phantom scans of implant leads. These studies demonstrate the potential of the technique for sensitively detecting potentially dangerous coupling conditions before they can do any harm.

Abstract

To develop a method that combines parallel imaging and compressed sensing to enable faster and/or higher spatial resolution magnetic resonance (MR) imaging and show its feasibility in a pediatric clinical setting.Institutional review board approval was obtained for this HIPAA-compliant study, and informed consent or assent was given by subjects. A pseudorandom k-space undersampling pattern was incorporated into a three-dimensional (3D) gradient-echo sequence; aliasing then has an incoherent noiselike pattern rather than the usual coherent fold-over wrapping pattern. This k-space-sampling pattern was combined with a compressed sensing nonlinear reconstruction method that exploits the assumption of sparsity of medical images to permit reconstruction from undersampled k-space data and remove the noiselike aliasing. Thirty-four patients (15 female and 19 male patients; mean age, 8.1 years; range, 0-17 years) referred for cardiovascular, abdominal, and knee MR imaging were scanned with this 3D gradient-echo sequence at high acceleration factors. Obtained k-space data were reconstructed with both a traditional parallel imaging algorithm and the nonlinear method. Both sets of images were rated for image quality, radiologist preference, and delineation of specific structures by two radiologists. Wilcoxon and symmetry tests were performed to test the hypothesis that there was no significant difference in ratings for image quality, preference, and delineation of specific structures.Compressed sensing images were preferred more often, had significantly higher image quality ratings, and greater delineation of anatomic structures (P < .001) than did images obtained with the traditional parallel reconstruction method.A combination of parallel imaging and compressed sensing is feasible in a clinical setting and may provide higher resolution and/or faster imaging, addressing the challenge of delineating anatomic structures in pediatric MR imaging.

Abstract

A new approach to autocalibrating, coil-by-coil parallel imaging reconstruction, is presented. It is a generalized reconstruction framework based on self-consistency. The reconstruction problem is formulated as an optimization that yields the most consistent solution with the calibration and acquisition data. The approach is general and can accurately reconstruct images from arbitrary k-space sampling patterns. The formulation can flexibly incorporate additional image priors such as off-resonance correction and regularization terms that appear in compressed sensing. Several iterative strategies to solve the posed reconstruction problem in both image and k-space domain are presented. These are based on a projection over convex sets and conjugate gradient algorithms. Phantom and in vivo studies demonstrate efficient reconstructions from undersampled Cartesian and spiral trajectories. Reconstructions that include off-resonance correction and nonlinear l(1)-wavelet regularization are also demonstrated.

Abstract

A rapid technique for mapping of T(2) relaxation times is presented. The method is based on the conventional single-echo spin echo approach but uses a much shorter pulse repetition time to accelerate data acquisition. The premise of the new method is the use of a constant difference between the echo time and pulse repetition time, which removes the conventional and restrictive requirement of pulse repetition time > T(1). Theoretical and simulation investigations were performed to evaluate the criteria for accurate T(2) measurements. Measured T(2)s were shown to be within 1% error as long as the key criterion of pulse repetition time/T(2) > or =3 is met. Strictly, a second condition of echo time/T(1) < 1 is also required. However, violations of this condition were found to have minimal impact in most clinical scenarios. Validation was conducted in phantoms and in vivo T(2) mapping of healthy cartilage and brain. The proposed method offers all the advantages of single-echo spin echo imaging (e.g., immunity to stimulated echo effects, robustness to static field inhomogeneity, flexibility in the number and choice of echo times) in a considerably reduced amount of time and is readily implemented on any clinical scanner.

Abstract

This study evaluates how characterization of tissue heterogeneity of myocardial infarction by cardiovascular magnetic resonance (CMR) is associated with cardiovascular events (CVE) in patients with ischemic cardiomyopathy (ICM).Prior studies demonstrated that the quantification of myocardial scar volume by CMR is superior to left ventricular end-diastolic volume, left ventricular end-systolic volume, and left ventricular ejection fraction (LVEF) in predicting future CVE in ICM patients. Evaluation of infarct heterogeneity by measuring infarct core and border zones through CMR might have a higher association with CVE.Seventy patients (mean LVEF: 25 +/- 11%) considered for revascularization or medical management +/- implantable cardiac defibrillator were enrolled. A 1.5-T GE MRI (Signa, GE Healthcare, Milwaukee, Wisconsin) was used to acquire cine and delayed enhancement images. The patients' core and border zones of infarcted myocardium were analyzed and followed for CVE.Larger infarct border zone and its percentage of myocardium were found in the 29 patients (41%) who had CVE (median 13.3 g [interquartile range (IQR) 8.4 to 25.1 g] vs. 8.0 g [IQR 3.0 to 14.5 g], p = 0.02 and 7.8% [IQR 4.9% to 17.0%] vs. 4.1% [IQR 1.9% to 9.3%], p = 0.02, respectively). The core infarct zone and its percentage of myocardium, left ventricular end-diastolic volume, left ventricular end-systolic volume, and LVEF were not statistically significant. Sub-analysis of the medical management and revascularization patients with CVE demonstrated that the medically managed patients had a larger border zone, whereas there was no difference between border and core zones in the revascularization group (p < 0.05).Quantification of core and border zones and their percentages of myocardium through CMR is associated with future CVE and might assist in the management of patients with ICM.

Abstract

Partially parallel imaging with localized sensitivities is a fast parallel image reconstruction method for both Cartesian and non-Cartesian trajectories, but suffers from aliasing artifacts when there are deviations from the assumption of perfect localization. Such reconstructions would normally crop the individual coil images to remove the artifacts prior to combination. However, the sampling densities in variable-density k-space trajectories support different field-of-views for separate regions in k -space. In fact, the higher sampling density of low frequencies can be used to reconstruct a bigger field-of-view without introducing aliasing artifacts and the resulting image signal-to-noise ratio (SNR) can be improved. A novel, fast variable-density parallel imaging method is presented, which reconstructs different field-of-views from separate frequencies according to the local sampling density in k-space. Aliasing-suppressed images can be produced with high SNR-efficiency without the need for accurate estimation of coil sensitivities and complex or iterative computations.

Abstract

A modified Cartesian feedback method called "frequency-offset Cartesian feedback" and based on polyphase difference amplifiers is described that significantly reduces the problems associated with quadrature errors and DC-offsets in classic Cartesian feedback power amplifier control systems.In this method, the reference input and feedback signals are down-converted and compared at a low intermediate frequency (IF) instead of at DC. The polyphase difference amplifiers create a complex control bandwidth centered at this low IF, which is typically offset from DC by 200-1500 kHz. Consequently, the loop gain peak does not overlap DC where voltage offsets, drift, and local oscillator leakage create errors. Moreover, quadrature mismatch errors are significantly attenuated in the control bandwidth. Since the polyphase amplifiers selectively amplify the complex signals characterized by a +90° phase relationship representing positive frequency signals, the control system operates somewhat like single sideband (SSB) modulation. However, the approach still allows the same modulation bandwidth control as classic Cartesian feedback.In this paper, the behavior of the polyphase difference amplifier is described through both the results of simulations, based on a theoretical analysis of their architecture, and experiments. We then describe our first printed circuit board prototype of a frequency-offset Cartesian feedback transmitter and its performance in open and closed loop configuration. This approach should be especially useful in magnetic resonance imaging transmit array systems.

Abstract

To demonstrate accelerated imaging with both artifact reduction and different contrast mechanisms near metallic implants.Slice-encoding for metal artifact correction (SEMAC) is a modified spin echo sequence that uses view-angle tilting and slice-direction phase encoding to correct both in-plane and through-plane artifacts. Standard spin echo trains and short-TI inversion recovery (STIR) allow efficient PD-weighted imaging with optional fat suppression. A completely linear reconstruction allows incorporation of parallel imaging and partial Fourier imaging. The signal-to-noise ratio (SNR) effects of all reconstructions were quantified in one subject. Ten subjects with different metallic implants were scanned using SEMAC protocols, all with scan times below 11 minutes, as well as with standard spin echo methods.The SNR using standard acceleration techniques is unaffected by the linear SEMAC reconstruction. In all cases with implants, accelerated SEMAC significantly reduced artifacts compared with standard imaging techniques, with no additional artifacts from acceleration techniques. The use of different contrast mechanisms allowed differentiation of fluid from other structures in several subjects.SEMAC imaging can be combined with standard echo-train imaging, parallel imaging, partial-Fourier imaging, and inversion recovery techniques to offer flexible image contrast with a dramatic reduction of metal-induced artifacts in scan times under 11 minutes.

Abstract

Hyperpolarized [1-(13)C]-pyruvate is an exciting new agent for the in vivo study of cellular metabolism and a potential cancer biomarker. The nature of the hyperpolarized signal poses unique challenges because of its short duration and the loss of magnetization with every excitation. In this study, we applied a novel and efficient time-resolved MR spectroscopic imaging (MRSI) method to investigate in a prostate cancer model the localized temporal dynamics of the uptake of [1-(13)C]-pyruvate and its conversion to metabolic products, specifically [1-(13)C]-lactate. This hyperpolarized (13)C method used multiband excitation pulses for efficient use of the magnetization. This study demonstrated that regions of tumor were differentially characterized from normal tissue by the lactate dynamics, where tumors showed later lactate detection and longer lactate duration that was statistically significant (P < 0.001). Compared to late-pathologic-stage tumors, early- to intermediate-stage tumors demonstrated significantly (P < 0.01) lower lactate total signal-to-noise ratio (SNR), with similar temporal dynamic parameters. Hyperpolarized pyruvate dynamics provided uptake, perfusion, and vascularization information on tumors and normal tissue. Large, heterogeneous tumors demonstrated spatially variable uptake of pyruvate and metabolic conversion that was consistent with cellularity and necrosis identified by histology. The results of this study demonstrated the potential of this new hyperpolarized MR dynamic method for improved cancer detection and characterization.

Abstract

High polarization of nuclear spins in liquid state through hyperpolarized technology utilizing dynamic nuclear polarization has enabled the direct monitoring of (13)C metabolites in vivo at a high signal-to-noise ratio. Acquisition time limitations due to T(1) decay of the hyperpolarized signal require accelerated imaging methods, such as compressed sensing, for optimal speed and spatial coverage. In this paper, the design and testing of a new echo-planar (13)C three-dimensional magnetic resonance spectroscopic imaging (MRSI) compressed sensing sequence is presented. The sequence provides up to a factor of 7.53 in acceleration with minimal reconstruction artifacts. The key to the design is employing x and y gradient blips during a fly-back readout to pseudorandomly undersample k(f)-k(x)-k(y) space. The design was validated in simulations and phantom experiments where the limits of undersampling and the effects of noise on the compressed sensing nonlinear reconstruction were tested. Finally, this new pulse sequence was applied in vivo in preclinical studies involving transgenic prostate cancer and transgenic liver cancer murine models to obtain much higher spatial and temporal resolution than possible with conventional echo-planar spectroscopic imaging methods.

Abstract

The currents induced in long conductors such as guidewires by the radio-frequency (RF) field in magnetic resonance imaging (MRI) are responsible for potentially dangerous heating of surrounding media, such as tissue. This paper presents an optically coupled system with the potential to quantitatively measure the RF currents induced on these conductors. The system uses a self shielded toroid transducer and active circuitry to modulate a high speed light-emitting-diode transmitter. Plastic fiber guides the light to a photodiode receiver and transimpedance amplifier. System validation included a series of experiments with bare wires that compared wire tip heating by fluoroptic thermometers with the RF current sensor response. Validations were performed on a custom whole body 64 MHz birdcage test platform and on a 1.5 T MRI scanner. With this system, a variety of phenomena were demonstrated including cable trap current attenuation, lossy dielectric Q-spoiling and even transverse electromagnetic wave node patterns. This system should find applications in studies of MRI RF safety for interventional devices such as pacemaker leads, and guidewires. In particular, variations of this device could potentially act as a realtime safety monitor during MRI guided interventions.

Abstract

Conventional spectral-spatial pulses used for water-selective excitation in proton resonance frequency-shift MR thermometry require increased sequence length compared to shorter wideband pulses. This is because spectral-spatial pulses are longer than wideband pulses, and the echo time period starts midway through them. Therefore, for a fixed echo time, one must increase sequence length to accommodate conventional spectral-spatial pulses in proton resonance frequency-shift thermometry. We introduce improved water-selective spectral-spatial pulses for which the echo time period starts near the beginning of excitation. Instead of requiring increased sequence length, these pulses extend into the long echo time periods common to PRF sequences. The new pulses therefore alleviate the traditional tradeoff between sequence length and fat suppression. We experimentally demonstrate an 11% improvement in frame rate in a proton resonance frequency imaging sequence compared to conventional spectral-spatial excitation. We also introduce a novel spectral-spatial pulse design technique that is a hybrid of previous model- and filter-based techniques and that inherits advantages from both. We experimentally validate the pulses' performance in suppressing lipid signal and in reducing sequence length compared to conventional spectral-spatial pulses.

Abstract

MRI has been used extensively to noninvasively track the location of cells labeled with superparamagnetic iron-oxide nanoparticles (SPIOs) in vivo. Typically, SPIOs are employed as a negative contrast agent which makes it difficult to differentiate labeled cells from extraneous sources of inhomogeneity and actual voids in the image. As a result, several novel approaches have been put forth to obtain positive contrast from SPIOs. One technique proposed by Cunningham et al. utilizes spectrally selective pulses to excite and refocus spins in the vicinity of the SPIOs. Although the frequency selectivity of this technique provides effective positive contrast, the lack of slice selectivity results in interfering signal from sources of off-resonance outside the slice of interest. We have developed a self-refocused spatial-spectral (SR-SPSP) pulse to achieve slice-selective spin-echo imaging of off-resonant spins. Using a self-refocused pulse affords flexibility in echo-time selection since the spin echo may be placed at any time after the end of the pulse. The spatial selectivity achieved by the SR-SPSP RF pulse eliminates background signal from out-of-slice regions and reduces the on-resonant water suppression requirements. Phantom and in vivo data demonstrate that positive contrast and slice-selectivity are achieved using this novel RF pulse.

Abstract

Time series of in vivo magnetic resonance images exhibit high levels of temporal correlation. Higher temporal resolution reconstructions are obtained by acquiring data at a fraction of the Nyquist rate and resolving the resulting aliasing using the correlation information. The dynamic imaging experiment is modeled as a linear dynamical system. A Kalman filter based unaliasing reconstruction is described for accelerated dynamic magnetic resonance imaging (MRI). The algorithm handles arbitrary readout trajectories naturally. The reconstruction is causal and very fast, making it applicable to real-time imaging. In vivo results are presented for cardiac MRI of healthy volunteers.

Abstract

Magnetic resonance imaging (MRI) near metallic implants remains an unmet need because of severe artifacts, which mainly stem from large metal-induced field inhomogeneities. This work addresses MRI near metallic implants with an innovative imaging technique called "Slice Encoding for Metal Artifact Correction" (SEMAC). The SEMAC technique corrects metal artifacts via robust encoding of each excited slice against metal-induced field inhomogeneities. The robust slice encoding is achieved by extending a view-angle-tilting (VAT) spin-echo sequence with additional z-phase encoding. Although the VAT compensation gradient suppresses most in-plane distortions, the z-phase encoding fully resolves distorted excitation profiles that cause through-plane distortions. By positioning all spins in a region-of-interest to their actual spatial locations, the through-plane distortions can be corrected by summing up the resolved spins in each voxel. The SEMAC technique does not require additional hardware and can be deployed to the large installed base of whole-body MRI systems. The efficacy of the SEMAC technique in eliminating metal-induced distortions with feasible scan times is validated in phantom and in vivo spine and knee studies.

Abstract

Variable-rate selective excitation (VERSE) is a radio frequency (RF) pulse reshaping technique. It is most commonly used to reduce the peak magnitude and specific absorption rate (SAR) of RF pulses by reshaping pulses and gradient waveforms to reduce RF magnitude while preserving excitation profiles. In this work, a general time-optimal VERSE algorithm for multidimensional and parallel transmit pulses is presented. Time optimality is achieved by translating peak RF limits to gradient upper bounds in excitation k-space. The limits are fed into a time-optimal gradient waveform design technique. Effective SAR reduction is achieved by reducing peak RF subject to a fixed pulse length. The presented method is different from other VERSE techniques in that it provides a noniterative time-optimal multidimensional solution, which drastically simplifies VERSE designs. Examples are given for 1D and 2D single channel and 2D parallel transmit pulses.

Abstract

In this work, a shim method that minimizes the maximum off-resonance frequency (min-max shim) in balanced steady-state free precession (bSSFP) is tested for brain imaging at 3T with constant and linear shim terms. The method demonstrates improvement of spatial coverage and banding artifact reduction over standard least-squares shimming. In addition, a new method (modified min-max shim) is introduced. This method reduces boundary band regions where the artifact is inevitable due to the excessive off-resonance frequency distribution. In comparison to standard least-squares shimming, the min-max based shim method either eliminates or reduces the size of banding artifacts. The method can be used to increase the signal-to-noise ratio (SNR) in bSSFP imaging or to increase the functional contrast in bSSFP functional MRI (fMRI) by allowing a longer usable repetition time (TR).

Abstract

Ultrashort echo time imaging with half RF pulse excitation is challenging as eddy currents induced by the slice-select gradient distort the half pulse slice profile. This work presents two pulses with T(2)-dependent slice profiles that are less sensitive to eddy currents. The double half pulse improves the slice selectivity for long T(2) components, while the inverted double half pulse suppresses the unwanted long T(2) signal. Thus, both approaches prevent imperfect cancellation of out-of-slice signal from contaminating the desired slice. Experimental results demonstrate substantially improved slice selectivity and R(2)* quantitation accuracy with these pulses. These pulses are effective in making short T(2) imaging and quantitation less sensitive to eddy currents and provide an alternative to time-consuming gradient characterization.

Abstract

Ultrashort echo time imaging with half RF pulse excitation is sensitive to eddy currents induced by the slice-select gradient that distorts the half pulse slice profile. This work demonstrates improvements in the half pulse profile by using spatial saturation on both sides of the imaged slice to suppress the out-of-slice magnetization. This effectively improves the selectivity of the half pulse excitation profile. A quadratic phase RF pulse with high bandwidth and selectivity was used to achieve a wide saturation band with sharp edges. Experimental results demonstrate substantially improved slice selectivity and R(2)* quantitation accuracy obtained with the out-of-slice saturation. This approach is effective in making short T(2) imaging and quantitation with half pulses less sensitive to eddy currents.

Abstract

The advantages of using a 3 Tesla (T) scanner for MR spectroscopic imaging (MRSI) of brain tissue include improved spectral resolution and increased sensitivity. Very selective saturation (VSS) pulses are important for maximizing selectivity for PRESS MRSI and minimizing chemical shift misregistration by saturating signals from outside the selected region. Although three-dimensional (3D) PRESS MRSI is able to provide excellent quality metabolic data for patients with brain tumors and has been shown to be important for defining tumor burden, the method is currently limited by how much of the anatomic lesion can be covered within a single examination. In this study we designed and implemented cosine modulated VSS pulses that were optimized for 3T MRSI acquisitions. This provided improved coverage and suppression of unwanted lipid signals with a smaller number of pulses. The use of the improved pulse sequence was validated in volunteer studies, and in clinical 3D MRSI exams of brain tumors.

Abstract

Hyperpolarized 13C offers high signal-to-noise ratios for imaging metabolic activity in vivo, but care must be taken when designing pulse sequences because the magnetization cannot be recovered once it has decayed. It has a short lifetime, on the order of minutes, and gets used up by each RF excitation. In this paper, we present a new dynamic chemical-shift imaging method that uses specialized RF pulses designed to maintain most of the hyperpolarized substrate while providing adequate SNR for the metabolic products. These are multiband, variable flip angle, spectral-spatial RF pulses that use spectral selectivity to minimally excite the injected prepolarized 13C-pyruvate substrate. The metabolic products of lactate and alanine are excited with a larger flip angle to increase SNR. This excitation was followed by an RF amplitude insensitive double spin-echo and an echo-planar flyback spectral-spatial readout gradient. In vivo results in rats and mice are presented showing improvements over constant flip angle RF pulses. The metabolic products are observable for a longer window because the low pyruvate flip angle preserves magnetization, allowing for improved observation of spatially varying metabolic reactions.

Abstract

Dynamic nuclear polarization and dissolution of a (13)C-labeled substrate enables the dynamic imaging of cellular metabolism. Spectroscopic information is typically acquired, making the acquisition of dynamic volumetric data a challenge. To enable rapid volumetric imaging, a spectral-spatial excitation pulse was designed to excite a single line of the carbon spectrum. With only a single resonance present in the signal, an echo-planar readout trajectory could be used to resolve spatial information, giving full volume coverage of 32 x 32 x 16 voxels every 3.5s. This high frame rate was used to measure the different lactate dynamics in different tissues in a normal rat model and a mouse model of prostate cancer.

Abstract

To improve the slice profile and image quality of R2* mapping in the iceball during cryoablation with ultrashort echo time (UTE) imaging by compensating for eddy currents induced by the selective gradient when half-pulse radiofrequency (RF) excitation is employed to achieve UTEs.A method to measure both B0 and linear eddy currents simultaneously is first presented. This is done with a least-square fitting process on calibration data collected on a phantom. Eddy currents during excitation are compensated by redesigning the RF pulse and the selective gradient accordingly, while that resultant from the readout gradient are compensated for during image reconstruction. In vivo data were obtained continuously during the cryoablation experiments to calculate the R2* values in the iceball and to correlate them with the freezing process.Image quality degradation due to eddy currents is significantly reduced with the proposed approaches. R2* maps of iceball throughout the cryoablation experiments were achieved with improved quality.The proposed approaches are effective for compensating eddy currents during half-pulse RF excitation as well as readout. TEs as short as 100 microsec were obtained, allowing R2* maps to be obtained from frozen tissues with improved quality.

Abstract

A fast and simple algorithm for designing time-optimal waveforms is presented. The algorithm accepts a given arbitrary multidimensional k-space trajectory as the input and outputs the time-optimal gradient waveform that traverses k-space along that path in minimum time. The algorithm is noniterative, and its run time is independent of the complexity of the curve, i.e., the number of switches between slew-rate limited acceleration, slew-rate limited deceleration, and gradient amplitude limited regions. The key in the method is that the gradient amplitude is designed as a function of arc length along the k-space trajectory, rather than as a function of time. Several trajectory design examples are presented.

Abstract

High polarization of nuclear spins in liquid state through dynamic nuclear polarization has enabled the direct monitoring of 13C metabolites in vivo at very high signal-to-noise, allowing for rapid assessment of tissue metabolism. The abundant SNR afforded by this hyperpolarization technique makes high-resolution 13C 3D-MRSI feasible. However, the number of phase encodes that can be fit into the short acquisition time for hyperpolarized imaging limits spatial coverage and resolution. To take advantage of the high SNR available from hyperpolarization, we have applied compressed sensing to achieve a factor of 2 enhancement in spatial resolution without increasing acquisition time or decreasing coverage. In this paper, the design and testing of compressed sensing suited for a flyback 13C 3D-MRSI sequence are presented. The key to this design was the undersampling of spectral k-space using a novel blipped scheme, thus taking advantage of the considerable sparsity in typical hyperpolarized 13C spectra. Phantom tests validated the accuracy of the compressed sensing approach and initial mouse experiments demonstrated in vivo feasibility.

Abstract

Passband balanced-steady-state free precession (b-SSFP) fMRI is a recently developed method that utilizes the passband (flat portion) of the b-SSFP off-resonance response to measure MR signal changes elicited by changes in tissue oxygenation following increases in neuronal activity. Rapid refocusing and short readout durations of b-SSFP, combined with the relatively large flat portion of the b-SSFP off-resonance spectrum allows distortion-free full-brain coverage with only two acquisitions. This allows for high-resolution functional imaging, without the spatial distortion frequently encountered in conventional high-resolution functional images. Finally, the 3D imaging compatibility of the b-SSFP acquisitions permits isotropic-voxel-size high-resolution acquisitions. In this study we address some of the major technical issues involved in obtaining passband b-SSFP-based functional brain images with practical imaging parameters and demonstrate the advantages through breath-holding and visual field mapping experiments.

Abstract

Adiabatic pulses are useful in achieving uniform excitation profiles in the presence of B(1)-inhomogeneity. At higher fields, this inhomogeneity becomes more severe, further amplifying the need for B(1)-insensitive excitation. Although gradient modulation techniques for slice-selective adiabatic excitation have been introduced, a pulse that falls within the gradient and RF amplifier limits for most commercial human scanners is currently unavailable. In this work, we present an alternative gradient modulated approach for pulse design that achieves adiabatic slice selection with significantly lower RF peak power requirements. The resulting Slice-selective Tunable-flip AdiaBatic Low peak-power Excitation (STABLE) pulse consists of an oscillating gradient in conjunction with a BIR-4-like RF envelope that is sampled by many short spatial subpulses to achieve spatial selectivity. Simulations show that the expected spatial profile as well as the off-resonance behavior of the pulse remain invariant for a range of B(1) values. Phantom and in vivo results demonstrate the adiabaticity and slice selectivity of the STABLE pulse.

Abstract

Proton magnetic resonance spectroscopic imaging ((1)H MRSI) is a useful technique for measuring metabolite levels in vivo, with Choline (Cho), Creatine (Cre), and N-Acetyl-Aspartate (NAA) being the most prominent MRS-detectable brain biochemicals. (1)H MRSI at very high fields, such as 7T, offers the advantages of higher SNR and improved spectral resolution. However, major technical challenges associated with high-field systems, such as increased B(1) and B(0) inhomogeneity as well as chemical shift localization (CSL) error, degrade the performance of conventional (1)H MRSI sequences. To address these problems, we have developed a Position Resolved Spectroscopy (PRESS) sequence with adiabatic spatial-spectral (SPSP) refocusing pulses, to acquire multiple narrow spectral bands in an interleaved fashion. The adiabatic SPSP pulses provide magnetization profiles that are largely invariant over the 40% B(1) variation measured across the brain at 7T. Additionally, there is negligible CSL error since the transmit frequency is separately adjusted for each spectral band. in vivo (1)H MRSI data were obtained from the brain of a normal volunteer using a standard PRESS sequence and the interleaved narrow-band PRESS sequence with adiabatic refocusing pulses. In comparison with conventional PRESS, this new approach generated high-quality spectra from an appreciably larger region of interest and achieved higher overall SNR.

Abstract

To reduce Gibbs ringing artifact in three-dimensional (3D) articular knee cartilage imaging with linear prediction (LP).A reconstruction method using LP in 3D was applied to truncated data sets of six healthy knees. The technique first linearizes the data before applying the prediction algorithm. Three radiologists blindly reviewed and ranked images of the full, truncated, and predicted data sets. Statistical analysis of the radiologists' reviews was performed for image quality, clinical acceptability of the images, and equivalence with the gold standard.LP applied to 3D knee cartilage imaging allows for 40% decreased scan time while providing image quality with statistical equivalence to a full data set.3D spoiled gradient echo imaging (SPGR) knee cartilage imaging requires significant scan time. This 40% reduction in scan time will allow such scans to be more feasible without sacrificing clinical acceptability.

Abstract

Nonuniform, non-Cartesian k-space trajectories enable fast scanning with reduced motion and flow artifacts. In such cases, the data are usually convolved with a kernel and resampled onto a Cartesian grid before reconstruction. For trajectories such as undersampled variable-density spirals, the mainlobe width of the kernel for undersampled high spatial frequencies has to be larger to limit the amount of aliasing energy. Continuously varying the kernel extent is time consuming. By dividing k-space into several annuli and using appropriate mainlobe widths for each, the aliasing energy and noise can be reduced at the expense of lower resolution towards the edge of the field of view (FOV). Resolution can instead be preserved at the center of the FOV, which is expected to be free of artifacts, without any artifact reduction. The image reconstructed from each annulus can be deapodized separately. The method can be applied to most k-space trajectories used in MRI.

Abstract

Prepolarized magnetic resonance imaging (PMRI) uses two pulsed electromagnets to achieve high-field image quality with the benefits of low-field data acquisition. The principal challenge with all resistive MRI systems is the implementation of a highly precise magnet current supply. The noise current through the magnet is fundamentally limited by the current transducer used to provide feedback and the voltage reference used to generate the demand signal. Field instability in the main field magnet can both corrupt the received data and degrade the robustness of Carr¿Purcell¿Meiboom¿Gill (CPMG) echo trains, which are paramount to efficient imaging in PMRI. In this work, we present the magnet control system that achieved sufficient field stability for PMRI at $0.5/0.13$ T, identify the dominant sources of noise in the control system, examine the imaging artifacts that can occur if the field stability is insufficient, and identify how the design can be improved for better field stability, should it be required for future implementations of PMRI.

Abstract

The sparsity which is implicit in MR images is exploited to significantly undersample k-space. Some MR images such as angiograms are already sparse in the pixel representation; other, more complicated images have a sparse representation in some transform domain-for example, in terms of spatial finite-differences or their wavelet coefficients. According to the recently developed mathematical theory of compressed-sensing, images with a sparse representation can be recovered from randomly undersampled k-space data, provided an appropriate nonlinear recovery scheme is used. Intuitively, artifacts due to random undersampling add as noise-like interference. In the sparse transform domain the significant coefficients stand out above the interference. A nonlinear thresholding scheme can recover the sparse coefficients, effectively recovering the image itself. In this article, practical incoherent undersampling schemes are developed and analyzed by means of their aliasing interference. Incoherence is introduced by pseudo-random variable-density undersampling of phase-encodes. The reconstruction is performed by minimizing the l(1) norm of a transformed image, subject to data fidelity constraints. Examples demonstrate improved spatial resolution and accelerated acquisition for multislice fast spin-echo brain imaging and 3D contrast enhanced angiography.

Abstract

The transgenic adenocarcinoma of mouse prostate (TRAMP) mouse is a well-studied murine model of prostate cancer with histopathology and disease progression that mimic the human disease. To investigate differences in cellular bioenergetics between normal prostate epithelial cells and prostate tumor cells, in vivo MR spectroscopic (MRS) studies with non-proton nuclei, such as (13)C, in the TRAMP model would be extremely useful. The recent development of a method for retaining dynamic nuclear polarization (DNP) in solution permits high signal-to-noise ratio (SNR) (13)C MRI or MRSI data to be obtained following injection of a hyperpolarized (13)C agent. In this transgenic mouse study, this method was applied using a double spin-echo (DSE) pulse sequence with a small-tip-angle excitation RF pulse, hyperbolic-secant refocusing pulses, and a flyback echo-planar readout trajectory for fast (10-14 s) MRSI of (13)C pyruvate (pyr) and its metabolic products at 0.135 cm(3) nominal spatial resolution. Elevated (13)C lactate (lac) was observed in both primary and metastatic tumors, demonstrating the feasibility of studying cellular bioenergetics in vivo with DNP hyperpolarized (13)C MRSI.

Abstract

Ultrashort echo time (UTE) imaging is a technique that can visualize tissues with sub-millisecond T(2) values that have little or no signal in conventional MRI techniques. The short-T(2) tissues, which include tendons, menisci, calcifications, and cortical bone, are often obscured by long-T(2) tissues. This paper introduces a new method of long-T(2) component suppression based on adiabatic inversion pulses that significantly improves the contrast of short-T(2) tissues. Narrow bandwidth inversion pulses are used to selectively invert only long-T(2) components. These components are then suppressed by combining images prepared with and without inversion pulses. Fat suppression can be incorporated by combining images with the pulses applied on the fat and water resonances. Scaling factors must be used in the combination to compensate for relaxation during the preparation pulses. The suppression is insensitive to RF inhomogeneities because it uses adiabatic inversion pulses. Simulations and phantom experiments demonstrate the adiabatic pulse contrast and how the scaling factors are chosen. In vivo 2D UTE images in the ankle and lower leg show excellent, robust long-T(2) suppression for visualization of cortical bone and tendons.

Abstract

Dynamic nuclear polarization of metabolically active compounds labeled with (13)C has been introduced as a means for imaging metabolic processes in vivo. To differentiate between the injected compound and the various metabolic products, an imaging technique capable of separating the different chemical-shift species must be used. In this paper, the design and testing of a pulse sequence for rapid magnetic resonance spectroscopic imaging (MRSI) of hyperpolarized (13)C is presented. The pulse sequence consists of a small-tip excitation followed by a double spin echo using adiabatic refocusing pulses and a "flyback" echo-planar readout gradient. Key elements of the sequence are insensitivity to calibration of the transmit gain, the formation of a spin echo giving high-quality spectral information, and a small effective tip angle that preserves the magnetization for a sufficient duration. Experiments in vivo showed three-dimensional coverage with excellent spectral quality and SNR.

Abstract

The purpose of this work was to present and evaluate a new method for directly designing T2-selective preparation pulses. Using a modified Shinnar-Le-Roux (SLR) transform, the design of T2-selective pulses becomes equivalent to designing a pair of polynomials one of which represents the longitudinal magnetization and the other the transverse magnetization. The polynomials enable one to directly analyze the various tradeoffs involved in the design. To evaluate the new method, a short-T2-selective magnetization preparation pulse was designed. Following the preparation pulse, a 2D Fourier transform (2DFT) multislice gradient echo sequence was used for imaging. For verification Bloch equation simulations were performed along with both in vivo and phantom scans. Phantom scans showed good signal suppression of long-T2 species. This is supported by good long-T2 signal suppression seen on the in vivo images. Simulations indicate that the pulse is robust to +/-150 Hz B0 inhomogeneities and +/-10% B1 inhomogeneities.

Abstract

Prostate MR spectroscopic imaging (MRSI) at 3T may provide two-fold higher spatial resolution over 1.5T, but this can result in longer acquisition times to cover the entire gland using conventional phase-encoding. In this study, flyback echo-planar readout trajectories were incorporated into a Malcolm Levitt's composite-pulse decoupling sequence (MLEV)-point-resolved spectroscopy sequence (PRESS) to accelerate the acquisition of large array (16 x 16 x 8), high spatial (0.154 cm(3)) resolution MRSI data by eight-fold to just 8.5 minutes. Artifact free, high-quality MRSI data was obtained in nine prostate cancer patients. Easy data reconstruction and the robustness of the flyback echo-planar encoding make this technique particularly suitable for the clinical setting. The short acquisition time provided by this method reduces the 3T prostate MRI/MRSI exam time, allows longer repetition times, and/or allows the acquisition of additional MR acquisitions within the same exam.

Abstract

In transition-band steady-state free precession (SSFP) functional MRI (fMRI), functional contrast originates from a bulk frequency shift induced by a deoxygenated hemoglobin concentration change in the activated brain regions. This frequency shift causes a magnitude and/or phase-signal change depending on the off-resonance distribution of a voxel in the balanced-SSFP (bSSFP) profile. However, in early low-resolution studies, only the magnitude signal activations were shown. In this paper the task-correlated phase-signal change is presented in a high-resolution (1 x 1 x 1 mm3) study. To include this phase activation in a functional analysis, a new complex domain data analysis method is proposed. The results show statistically significant phase-signal changes in a large number of voxels comparable to that of the magnitude-activated voxels. The complex-data analysis method successfully includes these phase activations in the activation map and thus provides wider coverage compared to magnitude-data analysis results.

Abstract

Certain applications of MRI, such as bilateral breast imaging, require simultaneous imaging of multiple volumes. Although image data can be acquired sequentially, the SNR is often improved if both slabs are excited and imaged together, typically with phase encoding across a volume including both slabs and the space between them. The use of independent phase modulation of multiple slabs eliminates the need to encode empty space between slabs, which can result in a significant time reduction. Each slab is excited with a phase proportional to phase-encode number such that the slab positions in the acquired data are shifted to reduce empty space. With careful consideration this technique is compatible with different pulse sequences (e.g., spin-echo, gradient-echo, RF spoiling, and balanced SSFP (bSSFP)) and acceleration strategies (e.g., partial k-space and parallel imaging). This technique was demonstrated in phantoms and applied to bilateral breast imaging, where scan times were reduced by 20-30%.

Abstract

A 3 T MLEV-point-resolved spectroscopy (PRESS) sequence employing optimized spectral-spatial and very selective outer-voxel suppression pulses was tested in 25 prostate cancer patients. At an echo time of 85 ms, the MLEV-PRESS sequence resulted in maximally upright inner resonances and minimal outer resonances of the citrate doublet of doublets. Magnetic resonance spectroscopic imaging (MRSI) exams performed at both 3 and 1.5 T for 10 patients demonstrated a 2.08+/-0.36-fold increase in signal-to-noise ratio (SNR) at 3 T as compared with 1.5 T for the center citrate resonances. This permitted the acquisition of MRSI data with a nominal spatial resolution of 0.16 cm3 at 3 T with similar SNR as the 0.34-cm3 data acquired at 1.5 T. Due to the twofold increase in spectral resolution at 3 T and the improved magnetic field homogeneity provided by susceptibility-matched endorectal coils, the choline resonance was better resolved from polyamine and creatine resonances as compared with 1.5 T spectra. In prostate cancer patients, the elevation of choline and the reduction of polyamines were more clearly observed at 3 T, as compared with 1.5 T MRSI. The increased SNR and corresponding spatial resolution obtainable at 3 T reduced partial volume effects and allowed improved detection of the presence and extent of abnormal metabolite levels in prostate cancer patients, as compared with 1.5 T MRSI.

Abstract

Ultrashort echo time (UTE) imaging has shown promise as a technique for imaging tissues with T2 values of a few milliseconds or less. These tissues, such as tendons, menisci, and cortical bone, are normally invisible in conventional magnetic resonance imaging techniques but have signal in UTE imaging. They are difficult to visualize because they are often obscured by tissues with longer T2 values. In this article, new long-T2 suppression RF pulses that improve the contrast of short-T2 species are introduced. These pulses are improvements over previous long-T2 suppression pulses that suffered from poor off-resonance characteristics or T1 sensitivity. Short-T2 tissue contrast can also be improved by suppressing fat in some applications. Dual-band long-T2 suppression pulses that additionally suppress fat are also introduced. Simulations, along with phantom and in vivo experiments using 2D and 3D UTE imaging, demonstrate the feasibility, improved contrast, and improved sensitivity of these new long-T2 suppression pulses. The resulting images show predominantly short-T2 species, while most long-T2 species are suppressed.

Abstract

For in vivo magnetic resonance imaging at high field (> or =3 T) it is essential to consider the homogeneity of the active B(1) field (B(1)+), particularly if surface coils are used for RF transmission. A new method is presented for highly rapid B(1)+ magnitude mapping. It combines the double angle method with a B(1)-insensitive magnetization-reset sequence such that the choice of repetition time (TR) is independent of T(1) and with a multislice segmented (spiral) acquisition to achieve volumetric coverage with adequate spatial resolution in a few seconds. Phantom experiments confirmed the accuracy of this technique even when TR < T(1), with the side effect being lowered SNR. The speed of this method enabled B(1)+ mapping in the chest and abdomen within a single breath-hold. In human cardiac imaging, the method enabled whole-heart coverage within a single 16-s breath-hold. Results from phantoms and healthy volunteers at 1.5 T and 3 T are presented.

Abstract

Recently a novel T2 selective imaging method based on linear combination (LC) filtering was developed. By linearly combining images acquired with different echo times LC filtering is able to generate images showing only tissues with a preselected range of T2 relaxation times. In this study the use of LC filtering in knee imaging was investigated. Three LC filters were designed: a short LC filter for imaging the knee meniscus, a medium LC filter for articular cartilage, and a long LC filter for synovial fluid. To verify the filter designs, eight phantoms with different T2 relaxation times were imaged. In addition, in vivo images were acquired from four asymptomatic volunteers and a subject with cartilage damage. T2 maps were also generated using the same source images. Signal-to-noise ratio (SNR) measurements were made of the meniscus, cartilage, and fluid regions on the three LC filtered images. The highest SNR was seen in the target tissue on each of the LC filtered images. LC filtering is a new method that can selectively image knee tissues based on their T2.

Abstract

In functional MRI (fMRI) the resonance frequency shift induced from respiration is a major source of physiological noise. In transition-band SSFP fMRI the respiration-induced resonance offset not only increases noise interference, it also shifts the activation band. This leads to a reduction in the contrast-to-noise ratio (CNR) and the potential for varying contrast levels during the experiment. A novel real-time method that compensates for the respiration-induced resonance offset frequency is presented. This method utilizes free induction decay (FID) phase information to measure the resonance offset. For compensation, one can update the resonant frequency in real time by changing the transmit RF pulse and receiver phases to track the measured offset. The results show decreased signal power in the respiration frequency band and increased numbers of activated voxels with higher Z-scores compared to uncompensated experiments.

Abstract

Multislice breath-held coronary imaging techniques conventionally lack the coverage of free-breathing 3D acquisitions but use a considerably shorter acquisition window during the cardiac cycle. This produces images with significantly less motion artifact but a lower signal-to-noise ratio (SNR). By using the extra SNR available at 3 T and undersampling k-space without introducing significant aliasing artifacts, we were able to acquire high-resolution fat-suppressed images of the whole heart in 17 heartbeats (a single breath-hold). The basic pulse sequence consists of a spectral-spatial excitation followed by a variable-density spiral readout. This is combined with real-time localization and a real-time prospective shim correction. Images are reconstructed with the use of gridding, and advanced techniques are used to reduce aliasing artifacts.

Abstract

The resolution in conventional BOLD FMRI is considerably lower than can be achieved with other MRI methods, and is insufficient for many important applications. One major difficulty in robustly improving spatial resolution is the poor image quality in BOLD FMRI, which suffers from distortions, blurring, and signal dropout. This work considers the potential for increased resolution with a new FMRI method based on balanced SSFP. This method establishes a blood oxygenation sensitive steady-state (BOSS) signal, in which the frequency sensitivity of balanced SSFP is used to detect the frequency shift of deoxyhemoglobin. BOSS FMRI is highly SNR efficient and does not suffer from image distortions or signal dropout, making this method an excellent candidate for high-resolution FMRI. This study presents the first demonstration of high-resolution BOSS FMRI, using an efficient 3D stack-of-segmented EPI readout and combined acquisition at multiple center frequencies. BOSS FMRI is shown to enable high-resolution FMRI data (1 x 1 x 2 mm(3)) in both visual and motor systems using standard hardware at 1.5 T. Currently, the major limitation of BOSS FMRI is its sensitivity to temporal and spatial field drift.

Abstract

If an individual can learn to directly control activation of localized regions within the brain, this approach might provide control over the neurophysiological mechanisms that mediate behavior and cognition and could potentially provide a different route for treating disease. Control over the endogenous pain modulatory system is a particularly important target because it could enable a unique mechanism for clinical control over pain. Here, we found that by using real-time functional MRI (rtfMRI) to guide training, subjects were able to learn to control activation in the rostral anterior cingulate cortex (rACC), a region putatively involved in pain perception and regulation. When subjects deliberately induced increases or decreases in rACC fMRI activation, there was a corresponding change in the perception of pain caused by an applied noxious thermal stimulus. Control experiments demonstrated that this effect was not observed after similar training conducted without rtfMRI information, or using rtfMRI information derived from a different brain region, or sham rtfMRI information derived previously from a different subject. Chronic pain patients were also trained to control activation in rACC and reported decreases in the ongoing level of chronic pain after training. These findings show that individuals can gain voluntary control over activation in a specific brain region given appropriate training, that voluntary control over activation in rACC leads to control over pain perception, and that these effects were powerful enough to impact severe, chronic clinical pain.

Abstract

To study the feasibility of a combined high spatial and temporal resolution real-time spiral MRI sequence for guiding coronary-sized vascular interventions.Eight New Zealand White rabbits (four normal and four with a surgically-created stenosis in the abdominal aorta) were studied. A real-time interactive spiral MRI sequence combining 1.1 x 1.1 mm(2) in-plane resolution and 189-msec total image acquisition time was used to image all phases of an interventional procedure (i.e., guidewire placement, balloon angioplasty, and stenting) in the rabbit aorta using coronary-sized devices on a 1.5 T MRI system.Real-time spiral MRI identified all rabbit aortic stenoses and provided high-temporal-resolution visualization of guide-wires crossing the stenoses in all animals. Angioplasty balloon dilatation and deployment of coronary-sized copper stents in the rabbit aorta were also successfully imaged by real-time spiral MRI.Combining high spatial and temporal resolution with spiral MRI allows real-time MR-guided vascular intervention using coronary-sized devices in a rabbit model. This is a promising approach for guiding coronary interventions.

Abstract

The spatial resolution of conventional magnetic resonance spectroscopic imaging-(MRSI) is typically coarse, mainly due to SNR limitations. The increased signal available with higher field scanners and new array coils now permits higher spatial resolution, but conventional chemical shift imaging (phase encoding) limits the spatial coverage possible in a patient-acceptable acquisition time. The "flyback" echo-planar trajectory is particularly insensitive to errors and provides data that are simple to process. In this study, high-efficiency gradient waveforms for flyback echo-planar MRSI were designed and implemented. Normal volunteer studies at 3 T showed the feasibility of acquiring high spatial resolution with large coverage in a short scan time (2048 voxels in 2.3 min and 4096 voxels in 8.5 min). The trajectories were insensitive to errors in timing and require only a modest (10 to 30%) penalty in SNR relative to conventional phase encoding using the same acquisition time.

Abstract

Multiple receiver-coil data collection is an effective approach to reduce scan time. There are many parallel imaging techniques that reduce scan time using multiple receiver coils. One of these methods, partially parallel imaging with localized sensitivities (PILS), utilizes the localized sensitivity of each coil. The advantages of PILS over other parallel imaging methods include the simplicity of the algorithm, good signal-to-noise ratio (SNR) properties, and the fact that there is no additional complexity involved in applying the algorithm to arbitrary k-space trajectories. This PILS method can be further improved to provide truly parallel broadband imaging with the use of multiple-demodulation hardware. By customizing the demodulation based on each coil's location, the k-space sampling rate can be chosen based on each coil's localized sensitivity region along the readout direction. A simulated demodulation of data from 2D Fourier transform (FT) and spiral trajectories is shown to demonstrate the method's feasibility.

Abstract

In areas of highly pulsatile and turbulent flow, real-time imaging with high temporal, spatial, and velocity resolution is essential. The use of 1D Fourier velocity encoding (FVE) was previously demonstrated for velocity measurement in real time, with fewer effects resulting from off-resonance. The application of variable-density sampling is proposed to improve velocity measurement without a significant increase in readout time or the addition of aliasing artifacts. Two sequence comparisons are presented to improve velocity resolution or increase the velocity field of view (FOV) to unambiguously measure velocities up to 5 m/s without aliasing. The results from a tube flow phantom, a stenosis phantom, and healthy volunteers are presented, along with a comparison of measurements using Doppler ultrasound (US). The studies confirm that variable-density acquisition of kz-kv space improves the velocity resolution and FOV of such data, with the greatest impact on the improvement of FOV to include velocities in stenotic ranges.

Abstract

The range of RF coils that can be used in combined X-ray/MR (XMR) systems is limited because many conventional coils contain highly X-ray attenuating materials that are visible in the X-ray images and potentially obscure patient anatomy. In this study, an X-ray compatible coil design that has minimal X-ray attenuation in the field of view (FOV) of the X-ray image is presented. In this design, aluminum is used for the loop conductor and discrete elements of the coil are eliminated from the X-ray FOV. A surface coil and an abdominal phased array coil were built using the X-ray compatible design. X-ray attenuation and MR imaging properties of the coils were evaluated and compared to conventional coils. The X-ray compatible phased array coil was used to image patients during two interventional procedures in the XMR system. The X-ray compatible coils allowed for fluoroscopic X-ray image acquisition, without degradation by the coil, while maintaining excellent MR imaging qualities.

Abstract

Reconstruction of magnetic resonance images from data not falling on a Cartesian grid is a Fourier inversion problem typically solved using convolution interpolation, also known as gridding. Gridding is simple and robust and has parameters, the grid oversampling ratio and the kernel width, that can be used to trade accuracy for computational memory and time reductions. We have found that significant reductions in computation memory and time can be obtained while maintaining high accuracy by using a minimal oversampling ratio, from 1.125 to 1.375, instead of the typically employed grid oversampling ratio of two. When using a minimal oversampling ratio, appropriate design of the convolution kernel is important for maintaining high accuracy. We derive a simple equation for choosing the optimal Kaiser-Bessel convolution kernel for a given oversampling ratio and kernel width. As well, we evaluate the effect of presampling the kernel, a common technique used to reduce the computation time, and find that using linear interpolation between samples adds negligible error with far less samples than is necessary with nearest-neighbor interpolation. We also develop a new method for choosing the optimal presampled kernel. Using a minimal oversampling ratio and presampled kernel, we are able to perform a three-dimensional (3-D) reconstruction in one-eighth the time and requiring one-third the computer memory versus using an oversampling ratio of two and a Kaiser-Bessel convolution kernel, while maintaining the same level of accuracy.

Abstract

Magnetic resonance spectroscopic imaging (MRSI) has proven to be a powerful tool for the metabolic characterization of prostate cancer in patients before and following therapy. The metabolites that are of particular interest are citrate and choline because an increased choline-to-citrate ratio can be used as a marker for cancer. High-field systems offer the advantage of improved spectral resolution as well as increased magnetization. Initial attempts at extending MRSI methods to 3 T have been confounded by the J-modulation of the citrate resonances. A new pulse sequence is presented that controls the J-modulation of citrate at 3 T such that citrate is upright, with high amplitude, at a practical echo time. The design of short (14 ms) spectral-spatial refocusing pulses and trains of nonselective refocusing pulses are described. Phantom studies and simulations showed that upright citrate with negligible sidebands is observed at an echo time of 85 ms. Studies in a human subject verified that this behavior is reproduced in vivo and demonstrated that the water and lipid suppression of the new pulse sequence are sufficient for application in prostate cancer patients.

Abstract

Contrast agents incorporating superparamagnetic iron-oxide nanoparticles have shown promise as a means to visualize labeled cells using MRI. Labeled cells cause significant signal dephasing due to the magnetic field inhomogeneity induced in water molecules near the cell. With the resulting signal void as the means for detection, the particles behave as a negative contrast agent, which can suffer from partial-volume effects. In this paper, a new method is described for imaging labeled cells with positive contrast. Spectrally selective RF pulses are used to excite and refocus the off-resonance water surrounding the labeled cells so that only the fluid and tissue immediately adjacent to the labeled cells are visible in the image. Phantom, in vitro, and in vivo experiments show the feasibility of the new method. A significant linear correlation (r = 0.87, P < 0.005) between the estimated number of cells and the signal was observed.

Abstract

The aim of this work was to show the potential utility of a novel rapid 3D fat-suppressed MRI method for joint imaging.Phase-sensitive steady-state free precession provides rapid 3D joint imaging with robust fat suppression and excellent cartilage delineation.

Abstract

A 3-echo linear combination myelin imaging method is presented. The echo times and weights are chosen such that the signal-to-noise ratio (SNR) of myelin-water is maximized, and signals from other white matter components are sufficiently suppressed. Interfering tissue water and cerebrospinal fluid (CSF) signals are much stronger than myelin due to their longer T2 and abundance. By carefully optimizing the echo times a 50-fold tissue water suppression is achieved along with a 10-fold CSF suppression. For comparison 4, 5, and 32 echo filters are also designed using the same method. The SNR efficiency of these filters is very similar. The 3-echo filter design was validated by phantom scans. In addition, multislice in vivo myelin images were acquired from both a healthy volunteer and a multiple sclerosis patient. Total scan time was 5 min. A uniform T2 filter is also designed to pass all white matter species with uniform gain. The myelin-water fraction of the in vivo 3-echo data set is then measured by dividing the myelin image by the uniformly filtered image. Obtained myelin-water fractions compare well with previous work.

Abstract

Magnetic resonance imaging (MRI) in the presence of metallic objects suffers from slice-selection errors and in-plane distortions. View angle tilting (VAT) corrects for in-plane distortions by adding a gradient on the slice-select axis during readout, but can suffer from image blurring. This work demonstrates that the major source of blurring is a slice profile modulation of the data, and proposes several solutions to prevent such blurring. Multiple high-bandwidth readouts are demonstrated to reduce the blurring while improving the signal-to-noise ratio (SNR) over a single high-bandwidth readout.

Abstract

Spectral-spatial RF (SSRF) pulses allow simultaneous selection in both frequency and spatial domains. These pulses are particularly important for clinical and research MR spectroscopy (MRS) applications for suppression of large water and lipid resonances. Also, the high bandwidth of the subpulses (5-10 kHz) greatly reduces the spatial-shift errors associated with different chemical shifts. However, the use of high-bandwidth subpulses along with enough spectral bandwidth to measure a typical range of metabolite frequencies (e.g., 300 Hz at 3 T) can require RF amplitudes beyond the limits of the RF amplifier of a typical scanner. In this article, a new method is described for designing nonlinear-phase 180 degrees SSRF pulses that can be used for spectral editing. The novel feature of the pulses is that the spectral profile develops as a symmetric sweep, from the outside edges of the spectral window towards the middle, so that coupled components are tipped simultaneously and over a short interval. Pulses were designed for lactate editing at 1.5 T and 3 T. The spectral and spatial spin-echo profiles of the new pulses were measured experimentally. Spectra acquired in phantom experiments showed a well-resolved, edited lactate doublet, with 91% to 93% editing efficiency.

Abstract

The proton resonance frequency (PRF) shift provides a means of measuring temperature changes during minimally invasive thermotherapy. However, conventional PRF thermometry relies on the subtraction of baseline images, which makes it sensitive to tissue motion and frequency drift during the course of treatment. In this study, a new method is presented that eliminates these problems by estimating the background phase from each acquired image phase. In this referenceless method, a polynomial is fit to the background phase outside the heated region in a weighted least-squares fit. Extrapolation of the polynomial to the heated region serves as the background phase estimate, which is then subtracted from the actual phase. The referenceless method is demonstrated on a phantom during laser heating, 0 degrees temperature rise images of in vivo human liver, interstitial laser ablation of porcine liver, and transurethral ultrasound ablation of canine prostate. A good correlation between temperature maps reconstructed with the referenceless and subtraction methods was found.

Abstract

Refocused steady-state free precession (SSFP) is limited by its high sensitivity to local field variation, particularly at high field strengths or the long repetition times (TRs) necessary for high resolution. Several methods have been proposed to reduce SSFP banding artifact by combining multiple phase-cycled SSFP acquisitions, each differing in how individual signal magnitudes and phases are combined. These include maximum-intensity SSFP (MI-SSFP) and complex-sum SSFP (CS-SSFP). The reduction in SSFP banding is accompanied by a loss in signal-to-noise ratio (SNR) efficiency. In this work a general framework for analyzing banding artifact reduction, contrast, and SNR of any multiple-acquisition SSFP combination method is presented. A new sum-of-squares method is proposed, and a comparison is performed between each of the combination schemes. The sum-of-squares SSFP technique (SOS-SSFP) delivers both robust banding artifact reduction and higher SNR efficiency than other multiple-acquisition techniques, while preserving SSFP contrast.

Abstract

Real-time cardiac and coronary MRI at 1.5T is relatively "signal starved" and the 3T platform is attractive for its immediate factor of two increase in magnetization. Cardiac imaging at 3T, however, is both subtly and significantly different from imaging at 1.5T because of increased susceptibility artifacts, differences in tissue relaxation, and RF homogeneity issues. New RF excitation and pulse sequence designs are presented which deal with the fat-suppression requirements and off-resonance issues at 3T. Real-time cardiac imaging at 3T is demonstrated with high blood SNR, blood-myocardium CNR, resolution, and image quality, using new spectral-spatial RF pulses and fast spiral gradient echo pulse sequences. The proposed sequence achieves 1.5 mm in-plane resolution over a 20 cm FOV, with a 5.52 mm measured slice thickness and 32 dB of lipid suppression. Complete images are acquired every 120 ms and are reconstructed and displayed at 24 frames/sec using a sliding window. Results from healthy volunteers show improved image quality, a 53% improvement in blood SNR efficiency, and a 232% improvement in blood-myocardium CNR efficiency compared to 1.5T.

Abstract

Diffusion-weighted imaging (DWI) has strong potential as a diagnostic for early cartilage damage, with clinical impact for diseases such as osteoarthritis. However, in vivo DWI of cartilage has proven difficult with conventional methods due to the short T2. This work presents a 3D steady-state DWI sequence that is able to image short-T2 species with high SNR. When combined with 2D navigator correction of motion-induced phase artifacts, this method enables high resolution in vivo DWI of cartilage. In vivo knee images in healthy subjects are presented with high SNR (SNR = 110) and submillimeter in-plane resolution (0.5 x 0.7 x 3.0 mm(3)). A method for fitting the diffusion coefficient is presented which produces fits within 10% of literature values. This method should be applicable to other short-T2 tissues, such as muscle, which are difficult to image using traditional DWI methods.

Abstract

A dynamic real-time (dRT) architecture has been developed to address limitations in magnetic resonance coronary angiography (MRCA). A prospective clinical trial of 45 patients suspected of coronary artery disease was conducted to determine clinical utility of this integrated real-time system.Clinical implementation of MRCA is not performed routinely today. However, improved anatomic coverage, image quality, and scan flexibility may enhance its clinical utility. A novel real-time architecture addresses these challenges through instantaneous reconfiguration between real-time (RT) and high-resolution (HR) imaging sequences with dynamic selection of the desired element on a custom-designed receiver coil.A total of 45 subjects were recruited consecutively to evaluate scan time, anatomic coverage, image quality, and detection of coronary lesions. Using a modern PC, the dRT switches from RT to gated HR imaging sequence in one repetition time (39 ms). Magnetic resonance imaging (MRI) scanning was performed using a custom-designed coronary coil consisting of two four-inch phase-array circular elements enabled with real-time selection of the desired coil element.All studies were completed in less than 45 minutes and required a mean of 12 breath holds (16 heartbeats). Of the total number of coronary segments, 91% (357/394) were visualized. Excellent or good image quality was achieved in 86% of the segments. Blinded analysis of the coronary arteries revealed sensitivity of 93% and specificity of 88% in the detection of coronary stenoses.The integrated environment of dRT provides a rapid and flexible scan protocol for MRCA while achieving wide anatomical coverage, high image quality, and reliable detection of coronary stenosis in short scan time.

Abstract

The standard method for FMRI, using the blood oxygenation level dependent (BOLD) effect, has significant limitations that result from the coupling of functional contrast to sources of image artifact. We have developed an alternative method for FMRI based on balanced-SSFP imaging. This method uses the balanced-SSFP phase profile to invert the signal in deoxygenated blood relative to oxygenated blood. The resulting blood oxygenation sensitive steady-state (BOSS) signal decouples functional contrast from imaging, enabling significantly better image quality than BOLD FMRI. BOSS FMRI is very SNR-efficient, achieves strong functional contrast and is relatively immune to susceptibility gradients. In this paper, we present results validating the ability to detect functional activity using BOSS FMRI. One of the potential advantages of BOSS FMRI is the ability to acquire high-resolution data due to the SNR efficiency of balanced-SSFP. Preliminary high resolution results (1 x 1 x 2 mm/sup 3/) at 1.5 T are presented.

Abstract

It is not currently known whether subjects can learn to voluntarily control activation in localized regions of their own brain using neuroimaging. Here, we show that subjects were able to learn enhanced voluntary control over task-specific activation in a chosen target region, the somatomotor cortex. During an imagined manual action task, subjects were provided with continuous direction regarding their cognitive processes. Subjects received feedback information about their current level of activation in a target region of interest (ROI) derived using real-time functional magnetic resonance imaging (rtfMRI), and they received automatically-adjusted instructions for the level of activation to achieve. Information was provided both as continously upated graphs and using a simple virtual reality interface that provided an image analog of the level of activation. Through training, subjects achieved an enhancement in their control over brain activation that was anatomically specific to the target ROI, the somatomotor cortex. The enhancement took place when rtfMRI-based training was provided, but not in a control group that received similar training without rtfMRI information, showing that the effect was not due to conventional, practice-based neural plasticity alone. Following training, using cognitive processes alone subjects could volitionally induce fMRI activation in the somatomotor cortex that was comparable in magnitude to the activation observed during actual movement. The trained subjects increased fMRI activation without muscle tensing, and were able to continue to control brain activation even when real-time fMRI information was no longer provided. These results show that rtfMRI information can be used to direct cognitive processes, and that subjects are able to learn volitionally regulate activation in an anatomically-targeted brain region, surpassing the task-driven activation present before training.

Abstract

The extension of MR imaging to new applications has demonstrated the limitations of the architecture of current real-time systems. Traditional real-time implementations provide continuous acquisition of data and modification of basic sequence parameters on the fly. We have extended the concept of real-time MRI by designing a system that drives the examinations from a real-time localizer and then gets reconfigured for different imaging modes. Upon operator request or automatic feedback the system can immediately generate a new pulse sequence or change fundamental aspects of the acquisition such as gradient waveforms excitation pulses and scan planes. This framework has been implemented by connecting a data processing and control workstation to a conventional clinical scanner. Key components on the design of this framework are the data communication and control mechanisms, reconstruction algorithms optimized for real-time and adaptability, flexible user interface and extensible user interaction. In this paper we describe the various components that comprise this system. Some of the applications implemented in this framework include real-time catheter tracking embedded in high frame rate real-time imaging and immediate switching between real-time localizer and high-resolution volume imaging for coronary angiography applications.

Abstract

MR coronary angiography (MRCA) has been demonstrated successfully at 3 Tesla (T). However, the advantages remain unclear. No systematic comparison of MRCA between 1.5 T and 3 T has been performed. Therefore, anatomic coverage, image quality, signal-to-noise ratio (SNR), contrast-to-noise ration (CNR), and susceptibility artifacts were compared in 23 subjects.Identical real-time (RT) and high-resolution (HR) sequences were implemented on the GE 1.5 T (Signa Twinspeed) and 3.0 T (Signa VH/i) whole body systems (GE, Milwaukee, WI). Both scanners were equipped with high-performance gradient systems capable of 40 mT/m peak amplitude and 150 mT/m/ms slew rate. Real-time localization of the coronary arteries was followed by a cardiac-gated, breath-hold HR sequence. Twenty-three subjects were recruited consecutively and underwent both 3 T and 1.5 T MRCA within one week. Coronary coverage based on the number of coronary segments visualized, image quality using a grading scale, SNR, CNR, and presence of susceptibility artifacts were analyzed. A significant improvement in SNR (47%), CNR (30%), and image quality were seen in 3 T. However, a significant increase in susceptibility artifacts was also noted.MRCA at 3 T significantly improves SNR, CNR, and image quality at the expense of susceptibility artifacts. Further optimization of the imaging parameters at 3 T may facilitate clinical implementation of MRCA.

Abstract

To test the feasibility of using real-time phase contrast (PC) magnetic resonance imaging (MRI) to track velocities (1-20 cm/second) of skeletal muscle motion.To do this we modified a fast real-time spiral PC pulse sequence to accommodate through-plane velocity encoding in the range of -20 to +20 cm/second. We successfully imaged motion of the biceps brachii and triceps brachii muscles during elbow flexion and extension in seven unimpaired adult subjects using real-time PC MRI.The velocity data demonstrate that the biceps brachii and the triceps brachii, antagonistic muscles, move in opposite directions during elbow flexion and extension with velocity values in the muscle tissue ranging from -10 to +10 cm/second.With further development, real-time PC MRI may provide a means to analyze muscle function in individuals with neurologic or movement disorders who cannot actively complete the repeated motions required for dynamic MRI techniques, such as cine PC MRI, that are more commonly used in musculoskeletal biomechanics applications.

Abstract

A technique is presented for the acquisition of temperature maps in the presence of variable respiratory motion using the proton resonance frequency (PRF) shift. The technique uses respiratory triggering, diaphragm position determination with a navigator echo, and the collection of multiple baseline images to generate temperature maps. Laser ablations were performed in an ex vivo liver phantom undergoing variable simulated respiratory motion and in vivo in four porcine livers, demonstrating a reduction of artifacts in the computed temperature maps compared with conventional single baseline techniques, both uncorrected and corrected for motion.

Abstract

Blood oxygenation level dependent (BOLD) functional MRI (fMRI) is an important method for functional neuroimaging that is sensitive to changes in blood oxygenation related to brain activation. While BOLD imaging has good spatial coverage and resolution relative to other neuroimaging methods (such as positron emission tomography (PET)), it has significant limitations relative to other MRI techniques, including poor spatial resolution, low signal levels, limited contrast, and image artifacts. These limitations derive from the coupling of BOLD functional contrast to sources of image degradation. This work presents an alternative method for fMRI that may over-come these limitations by establishing a blood oxygenation sensitive steady-state (BOSS) that inverts the signal from deoxygenated blood relative to the water signal. BOSS fMRI allows the imaging parameters to be optimized independently of the functional contrast, resulting in fewer image artifacts and higher signal-to-noise ratio (SNR). In addition, BOSS fMRI has greater functional contrast than BOLD. BOSS fMRI requires careful shimming and multiple acquisitions to obtain a precise alignment of the magnetization to the SSFP frequency response.

Abstract

Motion during diffusion-weighted imaging (DWI) introduces phase errors that can cause significant artifacts in brain images. One method of correcting these errors uses additional navigator data to measure the phase corruptions. Standard navigator methods correct for rigid-body motion but cannot correct for nonrigid deformations of the brain related to the cardiac cycle. This work derives a generalized reconstruction that corrects for nonrigid motion based on a least-squares formulation. Since this reconstruction has the disadvantage of being computationally expensive, an approximation is presented, called a refocusing reconstruction. The refocusing reconstruction is both efficient and straightforward. Each readout is multiplied in image space by the phase conjugate of the navigator image, and these rephased readouts are then summed. The conditions under which the refocusing reconstruction is sufficient are considered and methods to improve the quality of refocused images are discussed. In particular, synchronization of the acquisition to the cardiac cycle can provide data that is well-conditioned to the refocusing reconstruction without incurring the large time penalty traditionally associated with cardiac gating. These methods are applied to steady-state DWI, a promising pulse sequence that is particularly sensitive to motion-induced phase artifacts. The refocusing reconstruction is shown to significantly improve SS-DWI over standard rigid-body corrections.

Abstract

A spiral high-resolution coronary artery imaging sequence (SH) interfaced with real-time localization system (RT) has been developed. A clinical study of 40 patients suspected of coronary artery disease (CAD) was conducted. Segmented k-space acquisition techniques have dominated magnetic resonance coronary angiography (MRCA) over the last decade. Although a recent multicenter trial using this technique demonstrated encouraging results, the technique was hampered by low specificity. Spiral k-space acquisition had demonstrated several advantages for MRCA. Therefore, a first clinical trial implementing spiral high-resolution coronary imaging sequence with real-time localization (SH-RT) was performed.A clinical study of 40 patients suspected of CAD undergoing X-ray angiography was conducted to analyze the clinical reliability of this novel imaging system. The SH-RT had been designed to exploit the unique capability of two imaging sequences. The RT allowed a rapid localization of the coronary arteries. Then SH achieved multislice acquisition during a short breath-hold with submillimeter resolution. The MRCA data were analyzed for scan time, anatomic coverage, image quality, and accuracy in detecting CAD. In 40 subjects, SH achieved 0.7 to 0.9 mm resolution with 14-heartbeat breath-holds. Excellent or good image quality was achieved in 78% (263/337) of the coronary segments. Blinded consensus reading among three observers generated sensitivity of 76% and specificity of 91% in the detection of CAD compared with X-ray angiography. The MRCA imaging sequence implementing a novel spiral k-space acquisition technique enabled rapid and reliable imaging of the CAD in submillimeter resolution with short breath-holds.

Abstract

The high prevalence of osteoarthritis continues to demand improved accuracy in detecting cartilage injury and monitoring its response to different treatments. MRI is the most accurate noninvasive method of diagnosing cartilage lesions. However, MR imaging of cartilage is limited by scan time, signal-to-noise ratio (SNR), and image contrast. Recently, there has been renewed interest in SNR-efficient imaging sequences for imaging cartilage, including various forms of steady-state free-precession as well as driven-equilibrium imaging. This work compares several of these sequences with existing methods, both theoretically and in normal volunteers. Results show that the new steady-state methods increase SNR-efficiency by as much as 30% and improve cartilage-synovial fluid contrast by a factor of three. Additionally, these methods markedly decrease minimum scan times, while providing 3D coverage without the characteristic blurring seen in fast spin-echo images.

Abstract

To perform short echo time MR spectroscopic imaging of the lung parenchyma on normal volunteers.A short echo time projection-reconstruction spectroscopic imaging sequence was implemented on a commercial 1.5T whole body MRI scanner. Images and spectra of the lung parenchyma were obtained from five normal volunteers. Breath-held spectroscopic imaging was also performed.Spectroscopic imaging of short-T2* species allows visualization of different anatomic structures based upon their frequency shifts. A characteristic peak from the parenchyma was seen at three ppm from water frequency.Short echo time MR spectroscopic imaging of the lung parenchyma was demonstrated in normal volunteers. This method may improve proton imaging of the lungs and add specificity to the diagnosis of pulmonary disease.

Abstract

Although MR spectroscopic imaging (MRSI) of the prostate has demonstrated clinical utility for the staging and monitoring of cancer extent, current acquisition methods are often inadequate in several aspects. Conventional 180 degrees pulses can suffer from chemical shift misregistration, and have high peak-power requirements that can exceed hardware limits in many prostate MRSI studies. Optimal water and lipid suppression are also critical to obtain interpretable spectra. While complete suppression of the periprostatic lipid resonance is desired, controlled partial suppression of water can provide a valuable phase and frequency reference for data analysis and an assessment of experimental success in cases in which all other resonances are undetectable following treatment. In this study, new spectral-spatial RF pulses were developed to negate chemical shift misregistration errors and to provide dualband excitation with partial excitation of the water resonance and full excitation of the metabolites of interest. Optimal phase modulation was also included in the pulse design to provide 40% reduction in peak RF power. Patient studies using the new pulses demonstrated both feasibility and clear benefits in the reliability and applicability of prostate cancer MRSI.

Abstract

Cryosurgery has been shown to be an effective therapy for prostate cancer. Temperature monitoring throughout the cryosurgical iceball could dramatically improve efficacy, since end temperatures of at least -40 degrees C are required. The results of this study indicate that MR thermometry based on tissue R(*)(2) has the potential to provide this information. Frozen tissue appears as a complete signal void on conventional MRI. Ultrashort echo times (TEs), achievable with half pulse excitation and a short spiral readout, allow frozen tissue to be imaged and MR characteristics to be measured. However, half pulse excitation is highly sensitive to eddy current distortions of the slice-select gradient. In this work, the effects of eddy currents on the half pulse technique are characterized and methods to overcome these effects are developed. The methods include: 1) eddy current compensated slice-select gradients, and 2) a correction for the phase shift between the first and second half excitations at the center of the slice. The effectiveness of these methods is demonstrated in R(*)(2) maps calculated within the frozen region during cryoablation.

Abstract

An interactive real-time imaging system capable of rapid coronary artery imaging is described. High-resolution spiral and circular echo planar trajectories were used to achieve 0.8 x 1.6 mm2 resolution in 135 ms (CEPI) or 1.13 x 1.13 mm2 resolution in 189 ms (spirals), over a 20-cm FOV. Using a sliding window reconstruction, display rates of up to 37 images/sec were achieved. Initial results indicate this technique can perform as a high-quality 2D coronary localizer and with SNR improvement may enable rapid screening of the coronary tree.

Abstract

Refocused steady-state free precession (SSFP) imaging sequences have recently regained popularity as faster gradient hardware has allowed shorter repetition times, thereby reducing SSFP's sensitivity to off-resonance effects. Although these sequences offer fast scanning with good signal-to-noise efficiency, the "transient response," or time taken to reach a steady-state, can be long compared with the total imaging time, particularly when using 2D sequences. This results in lost imaging time and has made SSFP difficult to use for real-time and cardiac-gated applications. A linear-systems analysis of the steady-state and transient response for general periodic sequences is shown. The analysis is applied to refocused-SSFP sequences to generate a two-stage method of "catalyzing," or speeding up the progression to steady-state by first scaling, then directing the magnetization. This catalyzing method is compared with previous methods in simulations and experimentally. Although the second stage of the method exhibits some sensitivity to B(1) variations, our results show that the transient time can be significantly reduced, allowing imaging in a shorter total scan time. Magn Reson Med 46:149-158, 2001.

Abstract

A multislice real-time imaging technique is described which can provide continuous visualization of the entire left ventricle under resting and stress conditions. Three dynamically adjustable slices containing apical, mid, and base short axis views are imaged 16 times/sec (48 images/sec), with each image providing 3.12 mm resolution over a 20 cm field of view. Initial studies indicate that this technique is useful for the assessment of LV function by providing simultaneous real-time visualization of all 16 wall segments. This technique may also be used for stress LV function and, when used in conjunction with contrast agents, myocardial perfusion imaging. Magn Reson Med 45:371-375, 2001.

Abstract

A method was developed for quantitating the temperature within frozen tissue with the magnetic resonance (MR) parameter R2*. The pulse sequence uses half-pulse excitation and a short spiral readout to achieve echo times as short as 0.2 msec. Fiber-optic temperature sensors were inserted into bovine liver tissue. The tissue was frozen at one end while being held warm at the other end. Once steady state was reached, the parameter R2* was measured. A linear dependence of R2* on temperature was demonstrated. R2* is independent of freeze number and of the orientation of the temperature gradient with respect to the main magnetic field. Feasibility in a canine prostate during cryosurgery is demonstrated. J. Magn. Reson. Imaging 2001;13:99-104.

Abstract

In many applications of dynamic MR imaging, only a portion of the field-of-view (FOV) exhibits considerable variations in time. In such cases, a prior knowledge of the static part of the image allows a partial-FOV reconstruction of the dynamic section using only a fraction of the raw data. This method of reconstruction generally results in higher temporal resolution, because the scan time for partial-FOV data is shorter. The fidelity of this reconstruction technique depends, among other factors, on the accuracy of the prior information of the static section. This information is usually derived from the reconstructed images at previous time frames. This data, however, is normally corrupted by the motion artifact Because the temporal frequency contents of the motion artifact is very similar to that of the dynamic section, a temporal low-pass filter can efficiently remove this artifact from the static data. The bandwidth of the filter can be obtained from the rate of variations inside and outside the dynamic area. In general, when the temporal bandwidth is not spatially uniform, a bank of low-pass filters can provide a proper suppression of the motion artifact outside the dynamic section. This reconstruction technique is adapted for spiral acquisition and is successfully applied to cardiac fluoroscopy, doubling the temporal resolution.

Abstract

A real-time interactive color flow MRI system capable of rapidly visualizing cardiac and vascular flow is described. Interleaved spiral phase contrast datasets are acquired continuously, while real-time gridding and phase differencing is used to compute density and velocity maps. These maps are then displayed using a color overlay similar to what is used by ultrasound. For cardiac applications, 6 independent images/sec are acquired with in-plane resolution of 2.4 mm over a 20 cm field of view (FOV). Sliding window reconstruction achieves display rates up to 18 images/sec. Appropriate tradeoffs are made for other applications. Flow phantom studies indicate this technique accurately measures velocities up to 2 m/sec, and accurately captures real-time velocity waveforms (comparable to continuous wave ultrasound). In vivo studies indicate this technique is useful for imaging cardiac and vascular flow, particularly valvular regurgitation. Arbitrary scan planes can be quickly localized, and flow measured in any direction.

Abstract

A graphical user interface (GUI) has been developed which enables interactive feedback and control to the real-time diminishing variance algorithm (DVA). This interactivity allows the user to set scan parameters, view scan statistics, and view image updates during the course of the scan. In addition, the DVA has been extended to simultaneously reduce motion artifacts in three dimensions using three orthogonal navigators. Preliminary in vivo studies indicate that these improvements to the standard DVA allow for significantly improved consistency and robustness in eliminating respiratory motion artifacts from MR images, particularly when imaging the coronary arteries.

Abstract

A new, fast, spectrally selective steady-state free precession (SSFP) imaging method is presented. Combining k-space data from SSFP sequences with certain phase schedules of radiofrequency excitation pulses permits manipulation of the spectral selectivity of the image. For example, lipid and water can be resolved. The contrast of each image depends on both T1 and T2, and the relative contribution of the two relaxation mechanisms to image contrast can be controlled by adjusting the flip angle. Several potential applications of the technique, referred to as linear combination steady-state free precession (LCSSFP), are demonstrated: fast musculoskeletal, abdominal, angiographic, and brain imaging.

Abstract

A new fast, spectrally selective imaging method called fluctuating equilibrium magnetic resonance is presented. With all gradients refocused over a repetition interval, certain phase schedules of radiofrequency excitation pulses produce an equilibrium magnetization that fluctuates from excitation to excitation, thus permitting simultaneous acquisition of several images with different contrast features. For example, lipid and water images can be rapidly acquired. The effective echo time can be adjusted using the flip angle, thus providing control over the T(2) contribution to the contrast. Several applications of the technique are presented, including fast musculoskeletal, abdominal, breast, and brain imaging, in addition to MR angiography. A technique for combining lipid and water images generated with this sequence for angiography is described and other potential applications are suggested. Magn Reson Med 42:876-883, 1999.

Abstract

The high incidence of osteoarthritis and the recent advent of several new surgical and non-surgical treatment approaches have motivated the development of quantitative techniques to assess cartilage loss. Although magnetic resonance (MR) imaging is the most accurate non-invasive diagnostic modality for evaluating articular cartilage, improvements in spatial resolution, signal-to-noise ratio (SNR), and contrast-to-noise ratio (CNR) would be valuable. Cartilage presents an imaging challenge due to its short T(2) relaxation time and its low water content compared with surrounding materials. Current methods sacrifice cartilage signal brightness for contrast between cartilage and surrounding tissue such as bone, bone marrow, and joint fluid. A new technique for imaging articular cartilage uses driven equilibrium Fourier transform (DEFT), a method of enhancing signal strength without waiting for full T(1) recovery. Compared with other methods, DEFT imaging provides a good combination of bright cartilage and high contrast between cartilage and surrounding tissue. Both theoretical predictions and images show that DEFT is a valuable method for imaging articular cartilage when compared with spoiled gradient-recalled acquisition in the steady state (SPGR) or fast spin echo (FSE). The cartilage SNR for DEFT is as high as that of either FSE or SPGR, while the cartilage-synovial fluid CNR of DEFT is as much as four times greater than that of FSE or SPGR. Implemented as a three-dimensional sequence, DEFT can achieve coverage comparable to that of other sequences in a similar scan time. Magn Reson Med 42:695-703, 1999.

Abstract

Phase contrast velocity imaging is a standard method for accurate in vivo flow measurement. One drawback, however, is that it lengthens the scan time (or reduces the achievable temporal resolution) because one has to acquire two or more images with different flow sensitivities and subtract their phases to produce the final velocity image. Without this step, non-flow-related phase variations will give rise to an erroneous, spatially varying background velocity. In this paper, we introduce a novel phase contrast velocity imaging technique that requires the acquisition of only a single image. The idea is to estimate the background phase variation from the flow-encoded image itself and then have it removed, leaving only the flow-related phase to generate a corrected flow image. This technique is sensitive to flow in one direction and requires 50% less scan time than conventional phase contrast velocity imaging. Phantom and in vivo results were obtained and compared with those of the conventional method, demonstrating the new method's effectiveness in measuring flow in various vessels of the body. Magn Reson Med 42:704-713, 1999.

Abstract

To evaluate the performance of portal venous puncture with use of magnetic resonance (MR) guidance, and to place a transjugular intrahepatic portosystemic shunt (TIPS) in a swine model.A study of 12 swine was performed to evaluate the ability of interventional MR imaging to guide portal vein puncture and TIPS placement. Six swine had catheters placed in the right hepatic vein under C-arm fluoroscopy. A nitinol guide wire was left in the vein and the animals were then moved into an open configuration MR imaging unit. A TIPS needle set was used to puncture the portal vein using MR fluoroscopy. The animals were transferred to the C-arm, and venography confirmed portal vein puncture. A follow-up study was performed in six additional swine to place a TIPS using only MR imaging guidance. MR tracking was used to advance a catheter from the right atrium into the inferior vena cava. Puncture of the portal vein was performed and a nitinol stent was placed, bridging the hepatic parenchyma. MR venogram confirmed placement.Successful portal vein puncture was achieved in all animals. The number of punctures required decreased from 12 in the first animal to a single puncture in the last eight swine. A stent was successfully placed across the hepatic tract in all six swine.Real-time MR imaging proved to be a feasible method to guide portal vein puncture and TIPS placement in pigs.

Abstract

Several methods were investigated to improve the depiction of biopsy needles in radiofrequency (RF)-refocused magnetic resonance imaging. Distortion correction is performed by the use of view angle tilting (VAT): a gradient is employed on the slice-select axis during readout. Needle conspicuity is increased by offsetting the gradient echo from the spin echo and by inverting the 90 degrees RF pulse slice-select gradient. VAT effectively re-registers in-plane shifts. Since this method changes the projection angle through the slice, some structures appear blurred, while other structures appear sharper. VAT does not correct errors in slice selection. Offsetting the spin echo from the gradient echo increases needle conspicuity but can result in a shift in the apparent location of the needle. Inverting the 90 degrees slice-select gradient effectively increases the needle conspicuity with no shift in the needle location. These methods provide an easy and interactive means to manipulate needle artifacts but should be used cautiously.

Abstract

We conducted an initial clinical trial of a newly developed cardiac magnetic resonance imaging (CMRI) system. We evaluated left ventricular (LV) function in 85 patients to compare the clinical utility of the CMRI system with echocardiography, the current noninvasive gold standard.Conventional CMRI systems require cardiac-gating and respiratory compensation to synthesize a single image from data acquired over multiple cardiac cycles. In contrast, the new CMRI system allows continuous real-time dynamic acquisition and display of any scan plane at 16 images/s without the need for cardiac gating or breath-holding.A conventional 1.5T Signa MRI Scanner (GE, Milwaukee, Wisconsin) was modified by the addition of an interactive workstation and a bus adapter. The new CMRI system underwent clinical trial by testing its ability to evaluate global and regional LV function. The first group (A) consisted of 31 patients with acceptable echocardiography image quality. The second group (B) consisted of 31 patients with suboptimal echocardiography image quality. The third group (C) consisted of 29 patients with severe lung disease or congenital cardiac malformation who frequently have suboptimal echo study. Two independent observers scored wall motion and image quality using the standard 16-segment model and rank-order analysis.CMRI evaluation was complete in less than 15 min. In group A, no significant difference was found between ECHO and CMRI studies (p = NS). In group B, adequate visualization of wall segments was obtained 38% of the time using ECHO and 97% of the time using CMRI (p < 0.0001). When grouped into coronary segments, adequate visualization of at least one segment occurred in 18 of 30 patients (60%) with ECHO and in all 30 patients (100%) with CMRI (p < 0.0001). In group C, adequate visualization of the wall segments was obtained in 58% (CI 0.53-0.62) of the time using echocardiography and 99.7% (CI 0.99-1.0) of the time using CMRI (p < 0.0001).The new CMRI system provides clinically reliable evaluation of LV function and complements suboptimal echocardiography. In comparison with the conventional CMRI, the new CMRI system significantly reduces scan time, patient discomfort and associated cost.

Abstract

Traditional T2-based imaging techniques are geared toward imaging long-T2 species. Traditional techniques are, therefore, not optimal in clinical situations where the information of interest lies in the short-T2 species. T2-selective RF excitation (TELEX) is a technique for obtaining a T2-based contrast that highlights short-T2 values while suppressing long-T2 values-opposite to traditional T2 contrast. Previously, TELEX has been demonstrated qualitatively to highlight only very short-T2 values (T2 approximately 0.001 s). When applied to longer T2 values (T2 > or = 0.01 s), TELEX becomes sensitive to deltaB0 non-uniformities. This restricts its application to problems in which the T2 of interest is very short. In this study, TELEX is characterized quantitatively. Furthermore, a bandwidth broadening scheme is developed that reduces the deltaB0 sensitivity of TELEX. This permits the technique to be applied to longer T2 values. The capabilities and limitations of a practical implementation of TELEX are discussed.

Abstract

Cartilage injury resulting in osteoarthritis is a frequent cause of disability in young people. Osteoarthritis, based on either cartilage injury or degeneration, is a leading cause of disability in the United States. Over the last several decades, much progress has been made in understanding cartilage injury and repair. Magnetic resonance (MR) imaging, with its unique ability to noninvasively image and characterize soft tissue, has shown promise in assessment of cartilage integrity. In addition to standard MR imaging methods, MR imaging contrast mechanisms under development may reveal detailed information regarding the physiology and morphology of cartilage. MR imaging will play a crucial role in assessing the success or failure of therapies for cartilage injury and degeneration.

Abstract

A proton spectroscopic imaging sequence was used to investigate the feasibility of lactate imaging in head and neck tumors. The sequence employs a two-shot lactate editing method with inversion recovery for additional lipid suppression, and a restricted field of view to suppress motion artifacts. Variations in acquisition parameters and two different receive coils were investigated on twelve patients. Elevated lactate was detected in three patients, no lactate was observed in seven patients, and two studies were inconclusive because of severe motion or inhomogeneity artifacts. Best results were obtained with an anterior/posterior neck coil at a 288 ms echo time (TE).

Abstract

For quantitative velocity measurement, we have developed a technique that acquires full velocity spectra without cardiac gating. After a cylindrical excitation restricts imaging to one spatial dimension, data are acquired while an oscillating gradient is played out. After each excitation, an image of velocity versus spatial location is obtained. For a given spatial location, a series of these images can be used to form an image of velocity versus time. Acquisition times are much shorter than for phase-contrast imaging or Fourier-encoded velocity imaging, obviating the need for cardiac gating. Although a two-shot version of this technique has been presented previously, we have developed a one-shot version that offers higher temporal resolution for a given velocity resolution and superior off-resonance properties.

Abstract

The acquisition of complete three-dimensional (3D), segmented gradient-echo data sets to visualize the coronary arteries can be both time consuming and sensitive to motion, even with use of multiple breath-holding or respiratory gating. An alternate hybrid approach is demonstrated here, in which real-time interactive imaging is first used to locate an optimal oblique coronary scan plane. Then, a limited number of contiguous slices are acquired around that plane within a breath-hold with use of two-dimensional (2D) segmented gradient-echo imaging. Dual inversion nulling is used to suppress fat and myocardium. Finally, if needed, a limited reformat of the data is performed to produce images from relatively long sections of the coronaries. This approach yields relatively rapid visualization of portions of the coronary tree. Several different methods are compared for interactively moving the scan plane.

Abstract

An interleaved echo-planar imaging (EPI) technique is presented for the rapid acquisition of isotropic diffusion-weighted images of stroke patients. Sixteen isotropic diffusion-weighted images at three b values are acquired in less than 3 min. A spiral navigator echo is used to measure the constant and linear phase shifts across the head in both the x and y directions which result from motion during the isotropic diffusion- sensitizing gradients. The measured k-space errors are corrected during a gridding reconstruction. The gridding kernel has a constant width in kx and a variable width in ky which eliminates variable data-density ghosts. The resulting isotropic diffusion-weighted images have excellent lesion-to-normal brain contrast, very good spatial resolution, and little sensitivity to susceptibility effects in the base of the brain. Examples of diffusion-weighted images and ADC maps from several stroke patients are shown.

Abstract

A real-time interactive MRI system capable of localizing coronary arteries and imaging arrhythmic hearts in real-time is described. Non-2DFT acquisition strategies such as spiral-interleaf, spiral-ring, and circular echo-planar imaging provide short scan times on a conventional scanner. Real-time gridding reconstruction at 8-20 images/s is achieved by distributing the reconstruction on general-purpose UNIX workstations. An X-windows application provides interactive control. A six-interleaf spiral sequence is used for cardiac imaging and can acquire six images/s. A sliding window reconstruction achieves display rates of 16-20 images/s. This allows cardiac images to be acquired in real-time, with minimal motion and flow artifacts, and without breath holding or cardiac gating. Abdominal images are acquired at over 2.5 images/s with spiral-ring or circular echo-planar sequences. Reconstruction rates are 8-10 images/s. Rapid localization in the abdomen is demonstrated with the spiral-ring acquisition, whereas peristaltic motion in the small bowel is well visualized using the circular echo-planar sequence.

Abstract

Reliable fat suppression is especially important with fast imaging techniques such as echo-planar (EPI), spiral, and fast spin-echo (FSE) T2-weighted imaging. Spectral-spatial excitation has a number of advantages over spectrally selective presaturation techniques, including better resilience to B0 and B1 inhomogeneity. In this paper, a FSE sequence using a spectral-spatial excitation pulse for superior fat suppression is presented. Previous problems maintaining the CPMG condition are solved using simple methods to accurately program radio-frequency (RF) phase. Next an analysis shows how B0 eddy currents can reduce fat suppression effectiveness for spectral-spatial pulses designed for conventional gradient systems. Three methods to compensate for the degradation are provided. Both the causes of the degradation and the compensation techniques apply equally to gradient-recalled applications using these pulses. These problems do not apply to pulses designed for high-speed gradient systems. The spectral-spatial FSE sequence delivers clinically lower fat signal with better uniformity than spectrally selective pre-saturation techniques.

Abstract

A new generation of high power gradient systems which allow much faster MR imaging as well as shorter echo times has recently become available. Some of these high-speed gradient systems impose limits on the percentage of time during which the gradient can change in amplitude (slewing duty cycle). While this limitation may be immaterial to many 2DFT and echo planar imaging methods, a traditional circular spiral trajectory is difficult to use on these systems because its gradient waveforms change during the entire course of the trajectory so that the slewing duty cycle during the readout period is 100%. We describe a piecewise-linear spiral trajectory which is composed of linear segments and rounded corners. This trajectory reduces the slewing duty cycle while maintaining the desirable imaging properties of circular spirals including interleaving by simple gradient rotation. For one representative example, the slewing duty cycle is reduced to 46%. A conventional gridding method was used for image reconstruction, but a new numerical algorithm to calculate the density compensation factor was required. Use of piecewise-linear spiral trajectories reduces the impact imposed by limited gradient slewing duty cycle.

Abstract

Dual phase-compensating spectral/spatial echo-planar (EP) spin-echo (SE) pulses were incorporated into the point resolved spectroscopy (PRESS) excitation sequence to improve water and lipid suppression for 1H chemical shift imaging (CSI) and to decrease the dependence of the PRESS box location upon chemical shift. The asymmetric EPSE pulses (either minimum or maximum phase in the chemical shift domain) were substituted for the two PRESS SE pulses to yield zero phase spectra. Three different pulses were designed and tested at 1.5 T. Pulse 1, targeted for brain CSI (TE > 85 msec), passed choline to lipid resonances, suppressed water, and rephased the methyl lactate doublet independently of TE. Pulse 2, targeted for general purpose shorter TE PRESS, possessed both high chemical shift and spatial domain bandwidths. Pulse 3, designed for prostate CSI, passed choline to citrate resonances while suppressing lipids and water. The three pulses possessed spatial bandwidths ranging between 3.3 and 5.0 kHz, more than three times higher than that offered by one-dimensional SE pulses of equivalent maximum B1 amplitude. Phantom and in vivo experimental results demonstrated that, for EPSE pulses 1 and 2, suppression factors higher than 10(4) were achieved. The increased spatial bandwidths resulted in less contamination by signals from outside the designated PRESS excited region and a significant improvement in the uniformity of metabolite intensities for voxels located near edges of the PRESS box.

Abstract

We have developed a technique to accurately null the longitudinal magnetization (Mz) of background material. This suppression involves first saturating the longitudinal magnetization (Mz) of a region, and then applying several nonselective inversions. The inversions are timed relative to the saturation such that Mz is nulled across a broad range of T1 at a predetermined time after the initial saturation. B1 and B0 inhomogeneity, which could lead to inaccurate suppression, are dealt with by the combination of a multiple tip saturation sequence and four adiabatic inversion pulses. The suppression sequence can be used to form projective angiograms by selectively tagging the imaging region with the saturation pulse. After the inversions are played out, a projection taken through the tag region when Mz is nulled will only contain signal from blood that has flown into the region after the saturation. Since only two dimensions are acquired, the technique can acquire gated projection angiograms in reasonable scan times. Representative inflow MIR angiograms of the carotid arteries and renal arteries show excellent background suppression.

Abstract

Non-2DFT k-space readout strategies are useful in fast imaging but prone to blurring when reconstructed off resonance. Field inhomogeneities or susceptibility variations, coupled with a long readout time, are the major sources of this artifact. Correction methods based on a priori off-resonance information such as an acquired field map have been proposed in the literature. An alternative approach estimates the spatially varying off-resonance frequency from the data itself before applying a correction. In this latter approach there is a trade-off between the extent of correction and the chance of increased artifact due to estimation error. This paper introduces an improved algorithm for field map estimation which is both faster and more robust than the existing method. It uses a multi-stage estimation of the field map, starting from a coarse estimate both in frequency and space and proceeds towards higher resolution. The new algorithm is applied to phantom and in vivo images acquired with radial and spiral sequences to give sharper images.

Abstract

Field inhomogeneities or susceptibility variations produce blurring in images acquired using non-2DFT k-space readout trajectories. This problem is more pronounced for sequences with long readout times such as spiral imaging. Theoretical and practical correction methods based on an acquired field map have been reported in the past. This paper introduces a new correction method based on the existing concept of frequency segmented correction but which is faster and theoretically more accurate. It consists of reconstructing the data at several frequencies to form a set of base images that are then added together with spatially varying linear coefficients derived from the field map. The new algorithm is applied to phantom and in vivo images acquired with projection reconstruction and spiral sequences, yielding sharply focused images.

Abstract

To evaluate short echo time (TE) projection reconstruction magnetic resonance (MR) imaging in the detection of cartilage lesions.Twenty-seven cartilage regions of 10 human patellar specimens were examined with the following MR sequences: short TE projection reconstruction (repetition time msec/TE msec, 400/0.15), fat-suppressed three-dimensional spoiled gradient-recalled acquisition in the steady state (Spoiled GRASS) (50/10, 60 degrees flip angle), and magnetization transfer contrast (MTC) subtraction (400/6). MR findings were correlated with histopathologic grading of the cartilage.For detection of cartilage lesions, sensitivity of projection reconstruction imaging (100%) was significantly greater (P = .03) than that of MTC (62%) but not significantly greater (P > .05) than that of Spoiled GRASS (81%) imaging. Accuracy of projection reconstruction was significantly greater than that of MTC (P = .004) and Spoiled GRASS (P = .03) imaging. Unmasking of collagen fibers was most predictive of abnormal signal intensity of the cartilage with all sequences.In vitro, short TE projection reconstruction MR imaging provides superior delineation of cartilage lesions when compared with two other sequences. On Spoiled GRASS and MTC images, signal intensity of the superficial layer of cartilage is not a reliable sign for surface integrity.

Abstract

Spiral imaging has a number of advantages for fast imaging, including an efficient use of gradient hardware. However, inhomogeneity -induced blurring is proportional to the data acquisition duration. In this paper, we combine spiral data acquisition with a RARE echo train. This allows a long data acquisition interval per excitation, while limiting the effects of inhomogeneity. Long spiral k-space trajectories are partitioned into smaller, annular ring trajectories. Each of these annular rings is acquired during echoes of a RARE echo train. The RARE refocusing RF pulses periodically refocus off-resonant spins while building a long data acquisition. We describe both T2-weighted single excitation and interleaved RARE spiral sequences. A typical sequence acquires a complete data set in three excitations (32 cm FOV, 192 x 192 matrix). At a TR = 2000 ms, we can average two acquisitions in an easy breath-hold interval. A multifrequency reconstruction algorithm minimizes the effects of any off-resonant spins. Though this algorithm needs a field map, we demonstrate how signal averaging can provide the necessary phase data while increasing SNR. The field map creation causes no scan time penalty and essentially no loss in SNR efficiency. Multiple slice, 14-s breath-hold scans acquired on a conventional gradient system demonstrate the performance.

Abstract

Dynamic cardiac imaging in MRI is a very challenging task. To obtain high spatial resolution, temporal resolution, and signal-to-noise ratio (SNR), single-shot imaging is not sufficient. Use of multishot techniques resolves this problem but can cause motion artifacts because of data inconsistencies between views. Motion artifacts can be reduced by signal averaging at some cost in increased scan time. However, for the same increase in scan time, other techniques can be more effective than simple averaging in reducing the artifacts. If most of the energy of the inconsistencies is limited to a certain region of kappa-space, increased sampling density (oversampling) in this region can be especially effective in reducing motion artifacts. In this work, several variable-density spiral trajectories are designed and tested. Their efficiencies for artifact reduction are evaluated in computer simulations and in scans of normal volunteers. The SNR compromise of these trajectories is also investigated. The authors conclude that variable-density spiral trajectories can effectively reduce motion artifacts with a small loss in SNR as compared with a uniform density counterpart.

Abstract

Echo-planar imaging (EPI) is very susceptible to flow artifacts. Two ways to improve its flow properties are presented. First, "partial flyback" is proposed to reduce artifacts arising from flow in the readout direction. Near the center of k-space, only the even echoes of the EPI echo-train are used. Partial flyback is shown to improve the readout-flow properties at the expense of a slight worsening of the phase-encode flow and off-resonance properties. We recommend that the flyback region acquire 95% of the energy in k-space. Second, "inside-out" EPI is used to reduce artifacts arising from flow in the phase-encode direction. Data collection begins at the center of k-space, with separate interleaves to acquire the top and bottom, halves of k-space. Partial flyback is combined with partial-Fourier EPI and inside-out EPI. Partial-flyback inside-out EPI has worse off-resonance properties than partial-flyback partial-Fourier EPI but demonstrates better flow properties and does not require partial k-space reconstruction.

Abstract

Magnetic resonance angiography (MRA) was performed by using RF pulses designed to excite a limited spatial extent in two orthogonal directions. The restriction in the second spatial dimension can be used to increase inflow enhancement and to improve small field-of-view imaging. A rectangular excitation was produced with an "echo-planar" k-space trajectory and a sinc-modulated RF waveform. In vivo images have demonstrated that vessels are more clearly delineated with the two-dimensional excitation. Aliasing artifacts in small field-of-view imaging are significantly reduced, although in some cases complete elimination is not possible due to the nature of the gradient trajectory.

Abstract

This work describes a diffusion-weighted (DW) interleaved echo-planar imaging (IEPI) method for use on either conventional whole-body scanners or scanners equipped with high-speed gradient and receiver hardware. In combination with cardiac gating and motion correction with a pair of orthogonal navigator echoes, the presented method is time-efficient, compensates for patient motions, and is less sensitive to image distortions than single-shot methods. The motion-correction scheme consists of correction for constant and linear phase terms found from the orthogonal navigator echoes. The correction for the linear phase term in the phase-encode direction includes gridding the data to the Cartesian grid. The DW IEPI was used to image a phantom rotating about the slice-select direction, and motion correction was performed to eliminate ghost artifacts arising from motion in either the readout- or phase-encoding directions. DW IEPI with motion correction was performed on a normal volunteer and on a patient with a 26-day-old region of ischemia over much of the right hemisphere.

Abstract

Image contrast with gradient-recalled echo sequences (GRE) used for fMRI can have both blood oxygen level-dependent (BOLD) and inflow components, and the latter is often undesirable. A dual-echo technique can be used to differentiate these mechanisms, because modulation of signal from inflow is common to both echoes, whereas susceptibility and diffusion-related signal losses are larger in the second echo. An efficient dual-echo interleaved spiral sequence was developed for use with a conventional scanner. It uses a k-space trajectory that spirals out from the origin while the first echo is collected, then spirals back in while collecting the second echo. Decomposition of the data provides separate images of the inflow and T2-weighted components. Results demonstrate the decomposition with phantom experiments and with photic stimulation in normal volunteers.

Abstract

Water molecules associated with collagen have short transverse (T2) relaxation times. Projection-reconstruction techniques are able to achieve an echo time (TE) much shorter than conventional techniques, allowing imaging of tissues with T2 < 5 ms. Using these techniques, a conventional 1.5-T MRI human imaging system can directly image collagen-associated water from knee menisci and tendons in normal volunteers and patients. Long-T2 suppression improves the contrast between these structures and the surrounding tissue with long-T2 relaxation times. Spectroscopic imaging provides improved lipid/water registration and information about chemical composition and relaxation times. Direct imaging of tendons and menisci may provide more information about these structures and provide a new way to assess both injury and repair.

Abstract

Spiral imaging has a number of advantages for ultrafast data acquisition. However, implementation on high-field small-bore systems requires carefully addressing the issues of inhomogeneity-induced blurring and gradient hardware constraints. In this paper, spiral imaging on a 40-cm-bore 4.7T CSI Omega System (Bruker Instruments) is discussed. A constant-voltage gradient waveform design algorithm is developed to reduce readout times as well as minimize waveform distortions due to gradient amplifier nonlinearities. Residual errors are then measured and taken into account in the image reconstruction procedure. Multiple spiral interleaves as well as a multifrequency reconstruction algorithm are used to decrease blurring of off-resonance spins. Both phantom and in vivo images demonstrate the performance of the resulting pulse sequences.

Abstract

The imaging of dynamic processes in the body is of considerable interest in interventional examinations as well as kinematic studies, and spiral imaging is a fast magnetic resonance imaging technique ideally suited for such fluoroscopic applications. In this manuscript, magnetic resonance fluoroscopy pulse sequences in which interleaved spirals are used to continuously acquire data and reconstruct one movie frame for each repetition time interval are implemented. For many applications, not all of k-space needs to be updated each frame, and nonuniform k-space sampling can be used to exploit this rapid imaging strategy by allowing variable update rates for different spatial frequencies. Using the appropriate reconstruction algorithm, the temporal updating rate for each spatial frequency is effectively proportional to the corresponding k-space sampling density. Results from a motion phantom as well as in in vivo gadolinium diethylenetriaminopentaacetic acid (Gd-DTPA) bolus tracking studies in a rat model demonstrate the high temporal resolution achievable using these techniques as well as the tradeoffs available with nonuniform sampling densities. This paper focuses on the acquisition of real-time dynamic information, and all images presented are reconstructed retrospectively. The issues of real-time data reconstruction and display are not addressed.

Abstract

In vivo phosphorus spectroscopy requires very short acquisition delays in order to capture the signal from components with short transverse relaxation times (T2). The echo time typical of standard slice selective spin-echo pulses are too long for this application, so hard pulse, free induction decay (FID) acquisitions have frequently been used instead. With FID, however, there is an interval between the time of coherence and data acquisition (acquisition delay), with resulting baseline distortions. In this paper we describe the design of a new short TE, slice-selective, composite spin-echo pulse with echo times as short as 2.5 ms. With a long TR, fully relaxed, multislice spectra can be collected. This technique will be useful for assessing in vivo, changes in brain phospholipid activity associated with psychiatric and neurological diseases.

Abstract

The accurate measurement of flow velocity has long been a subject of NMR research. In the field of medical imaging, a variety of techniques primarily based on the principle of Fourier encoding have been described. Due to time constraints, necessary trade-offs exist between spatial versus velocity spectral resolution. In general, either the average velocity of individual pixels is displayed or velocity spectral determinations are made at the cost of spatial localization. The recent development of multidimensional excitation pulses makes spatial localization possible during the excitation phase of the pulse sequences. This approach, coupled with time varying gradient readout, can be used to obtain single-shot localized velocity spectra. Using these concepts, we have obtained in vivo real-time measurements of localized velocity spectra on our clinical imager.

Abstract

A technique is presented for collecting the spin velocity distribution as a function of position and time. It uses a multidimensional excitation pulse to select a cylinder, giving localization in two dimensions. Resolution in the third spatial dimension is achieved in the readout. During readout, an oscillating gradient encodes the acquired data in both one spatial dimension (x) and one velocity dimension (v). Two acquisitions (42 ms each) are needed to get a complete coverage of kx--kv space, which makes this technique real-time. The data is interpolated from the nonuniformly sampled kx--kv space to a Cartesian frame with a gridding scheme to take advantage of the Fast Fourier Transform. The technique was successfully applied to phantoms and normal volunteers, giving reasonable real-time measurements of velocity.

Abstract

A spectroscopic imaging sequence incorporating a two-shot lactate editing method was used in two human brain studies to image lactate and NAA. The subtractive editing method allows separate images of lactate, NAA, and lipids to be collected during a single study with no SNR penalty. The sequence uses a spectral-spatial excitation for slice selection and water suppression, and employs inversion recovery and an echo time of 136 ms for additional lipid suppression.

Abstract

The echo-planar k-space trajectory can be used as the basis for any two-dimensional selective pulse. The main application is spectral-spatial pulses, which must be based on the echo-planar trajectory. In this paper we show how echo-planar spin-echo (EPSE) pulses may be designed.

Abstract

It is shown that a conventional 1.5-T magnetic resonance (MR) imaging system can help characterize some of the key components of atherosclerotic plaque ex vivo. Fresh human aorta with atheromata was suspended in solutions of agarose and manganese chloride and heated to body temperature. The specimens were imaged with modified Dixon and projection-reconstruction imaging sequences. The specimens were then examined histologically to obtain direct correlation between images, spectra, and histologic characteristics. The results show that vessel wall and plaque components can be identified by means of their MR characteristics and correlated with their histologic appearance. The authors were able to identify normal vessel wall components, such as adventitial lipids and smooth muscle. They were also able to identify and localize plaque components such as fibrous tissue, calcification, lipids, and possible areas of hemorrhage and hemosiderin deposition.

Abstract

Imaging of the lungs presents a unique challenge for MR technology. Signal from lung parenchyma is limited by low proton density, the effects of motion and susceptibility, and a multiexponential T2 relaxation time. We have illustrated the use of spin echo and projection reconstruction MR sequences in imaging lung parenchyma, and the use of a GRE sequence in imaging the pulmonary vascular tree.

Abstract

In theory, multidimensional pulses can be designed to be selective in any number of dimensions. In practice, available gradient power has enforced a limit to two dimensions. We show here that three-dimensional pi pulses are feasible on commercial imaging machines provided that the range of off-resonance frequencies are limited.

Abstract

Projection reconstruction (PR) techniques are shown to have intrinsic advantages over spin-warp (2DFT) methods with respect to diminished artifacts from respiratory motion. The benefits result from (1) portrayal of artifacts as radial streaks, with the amplitude smallest near the moving elements; (2) streak deployment perpendicular to the direction of motion of moving elements and often residing outside the anatomic boundaries of the subject; (3) inherent signal averaging of low spatial frequencies from oversampling of central k-space data. In addition, respiratory-ordered view angle (ROVA) acquisition is found to diminish residual streaking significantly by reducing interview inconsistencies. Comparisons of 2DFT and PR acquisitions are made with and without ROVA. Reconstructions from magnitude-only projections are found to have increased streaks from motion-induced phase shifts.

Abstract

For several non-2D Fourier transform imaging methods, off-resonant reconstruction does not just cause geometric distortion, but changes the shape of the point spread function and causes blurring. This effect is well known for projection reconstruction and spiral k-space scanning sequences. We introduce here a method that automatically removes blur introduced by magnetic field inhomogeneity and susceptibility without using a resonant frequency map, making these imaging methods more useful. In this method, the raw data are modulated to several different frequencies and reconstructed to create a series of base images. Determination of degree of blur is done by calculating a focusing measure for each point in each base image and a composite image is then constructed using only the unblurred regions from each base image. This method has been successfully applied to phantom and in vivo images using projection-reconstruction and spiral-scan sequences.

Abstract

The authors have developed a pulse sequence for imaging lung parenchyma with projection reconstruction magnetic resonance (MR) imaging that reduces the effects of motion and susceptibility. In this study, the projection reconstruction technique was further modified by optimizing MR signal frequencies for reconstructing the images. This was done by means of one of two methods. With the first method, a susceptibility map was derived from the raw image data and this map was used to indicate the optimal frequencies for reconstructing the images. The second method of susceptibility correction was a postprocessing technique in which the optimal reconstruction frequencies were selected with use of specific focusing criteria to generate the least blurred image. The effect of using susceptibility map correction on a phantom was demonstrated, and both of these methods were used to improve the visibility of pulmonary structures on images of subjects with normal and abnormal lungs.

Abstract

Spectroscopic images of the brain have great potential in disease diagnosis and treatment monitoring. Unfortunately, interfering lipid signals from subcutaneous fat and poor water suppression due to magnetic field inhomogeneities can make such images difficult to obtain. A pulse sequence that uses inversion recovery for lipid suppression and a spectral-spatial refocusing pulse for water suppression is introduced. In contrast to methods that eliminate fat signal by restricting the excited volume to lie completely within the brain, inversion-recovery techniques allow imaging of an entire section without such restrictions. In addition, the spectral-spatial pulse was designed to provide water suppression insensitive to a reasonable range of B0 and B1 inhomogeneities. Several data processing algorithms have also been developed and used in conjunction with the new pulse sequence to produce metabolite maps covering large volumes of the human brain. Images from single- and multisection studies demonstrate the performance of these techniques.

Abstract

Using the technique of separable k-space excitation, we have designed a two-dimensional selective adiabatic pulse that inverts magnetization from a square region in the xy plane with insensitivity to RF variations. We also have designed a two-dimensional adiabatic pulse that inverts selectively in frequency and in one spatial dimension. The pulses should be useful for both MR imaging and spectroscopy. We present experimental results to demonstrate that the two-dimensional adiabatic pulses are feasible on commercial MR imaging systems.

Abstract

In a magnetic resonance imaging system, an RF power amplifier is employed to boost an RF pulse to sufficient strength to excite the nuclear spins in a subject. The nonideal behavior of this amplifier distorts a selective-excitation pulse, and this distortion in turn degrades the slice profile. We have found two types of nonideal behavior particularly troublesome: nonlinearity and incidental phase modulation. One of their effects is the introduction of an unwanted "skirt" in the out-of-slice region of a slice profile. We present an effective method of correction in which a selective-excitation pulse is prewarped to compensate for the distortion.

Abstract

A three-dimensional projection reconstruction technique is described for imaging boron-11 distributions, with potential application to boron neutron capture therapy. The method samples a spherical volume of k space uniformly to obtain a 32 x 32 x 32 matrix with voxel size of 0.42 cm3. A signal-to-noise ratio (S/N) of 3 was obtained in 8.5 minutes in a phantom containing 75 micrograms/mL of boron in borocaptate sodium (BSH). Images were obtained in a dog after cessation of an intravenous infusion of BSH and again 30 minutes later, with a maximum boron S/N of about 12. Boron levels in the brain dropped about 6%-8% and were more diffusely distributed on the images obtained 30 minutes after BSH infusion.

Abstract

When time-varying gradients are used for imaging, the off-resonance behavior does not just cause geometric distortion as is the case with spin-warp imaging, but changes the shape of the impulse response and causes blurring. This effect is well known for projection reconstruction and spiral k-space scanning sequences. The authors introduce a reconstruction and homogeneity correction method to correct for the zeroth order effects of inhomogeneity using prior knowledge of the inhomogeneity. In this method, the data are segmented according to collection time, reconstructed using some fast, linear algorithm, correlated for inhomogeneity, and then superimposed to yield a homogeneity corrected image. This segmented method is compared to a conjugate phase reconstruction in terms of degree of correction and execution time. The authors apply this method to in vivo images using projection-reconstruction and spiral-scan sequences.

Abstract

In flow-imaging experiments with 2-D Fourier transform sequences, the time difference between phase encoding and readout leads to a potentially misleading displacement artifact. This artifact arises in regions of rapid flow and high shear, and manifests as an intensity distortion in addition to a bulk shift. We have studied methods of mitigating the artifact, including offset-echo acquisition, backward-evolving phase encoding, moment-compensated phase encoding, and projection-reconstruction imaging. Experiments on flow phantoms verified the nature and reduction of this displacement artifact. Of the four methods studied, the projection-reconstruction sequence proved to be the most effective, completely eliminating the artifact.

Abstract

Magnetic susceptibility effects in magnetic resonance (MR) imaging of normal lung parenchyma occur because of magnetic-field inhomogeneities induced by the microscopic heterogeneity of the lung. The effects on MR imaging of the lung are loss of signal from intravoxel phase dispersion (measured with T2') and a shift in the macroscopic resonant frequency from that of water toward that of air (delta v). These effects of MR imaging at 1.5 T were quantitated by measuring T2' decay and delta v at different locations in the lungs of two adult volunteers and one excised inflated human lung. The average T2' was 7 msec in the excised inflated specimen and 6.3 msec in normal in vivo lungs. There was a gravitational increase in T2' from nondependent to dependent lung. T2' increased to 35 msec in atelectatic lung tissue and to more than 140 msec in tumor. The macroscopic resonant lung frequency increased to 3.6 ppm more than that of mediastinal muscle. These values are important for developing MR pulse sequences appropriate for imaging lung parenchyma.

Abstract

Magnetic resonance (MR) imaging of lung parenchyma is limited by the low proton density and short T2 in the lung as well as the effects of susceptibility and motion. The MR imaging appearance of lung parenchyma was investigated with a pulse sequence that offers some solutions to these problems. This sequence employs projection reconstruction (PR) acquisition gradients and a section-selective excitation pulse designed to eliminate the need to refocus and to allow low-frequency k-space data to be collected with minimal delay. Echo times as short as 50 microseconds can be achieved, producing a proton-density-weighted image. An excised inflated lung specimen and specimens from human subjects with normal lungs (n = 3), pulmonary arteriovenous malformations (n = 1), bronchogenic carcinoma (n = 1), and bullous lung disease with lung metastases (n = 1) were examined. Signal intensity from lung parenchyma and visibility of pulmonary structures were superior on images obtained with the PR MR imaging technique compared with spin-echo images.

Abstract

Proton spectroscopy and spectroscopic imaging in the human brain require the elimination of both water and lipid signals. Strong lipid signals from subcutaneous fat are usually eliminated by confining the excited volume to lie wholly within the skull. Water suppression, however, can be difficult due to both B0 and RF inhomogeneities, which are particularly troublesome in imaging experiments where a relatively large region-of-interest (ROI) is typical. In this paper, we discuss the use of multidimensional selective-excitation pulses (e.g., pulses that are simultaneously selective along two axes) to both define the ROI and provide the necessary water suppression. Pulse sequences providing three-dimensional localization along with water suppression that is insensitive to a range of B0 and RF inhomogeneities are described. Spectra and spectroscopic images (voxel volume = 3.4 cc. acquisition time = 38 min) of various 1H metabolites from a patient with an astrocytoma show clear differences between normal and cancerous tissues and demonstrate the ability of these techniques to be used in vivo.

Abstract

An overview of the Shinnar-Le Roux (SLR) algorithm is presented. It is shown how the performance of SLR pulses can be very accurately specified analytically. This reveals how to design a pulse that produces a specified slice profile and allows the pulse designer to trade off analytically the parameters describing the pulse performance. Several examples are presented to illustrate the more important tradeoffs. These include linear-phase and minimum- and maximum-phase pulses. Linear-phase pulses can be refocused with a gradient reversal and can be used as spin-echo pulses. Minimum- and maximum-phase pulses have better slice profiles, but cannot be completely refocused.

Abstract

Using a k-space interpretation of small-tip excitation, a single excitation pulse has been designed that is simultaneously selective in space and resonant frequency. An analytic expression for the response of this pulse has been derived. The pulse has been implemented on a 1.5-T imaging system. The pulse has been applied to a rapid gradient-echo imaging sequence that forms both water and fat images within a breath-holding interval. These rapid images are free of the chemical shift artifacts at organ boundaries that typically afflict conventional rapid images. The pulse can be applied to a variety of other sequences, such as multislice water/fat sequences and rapid k-space scanning sequences.

Abstract

In the selective inversion recovery method for projection angiography, upstream blood is tagged by an inversion excitation and then allowed to flow into the imaged region. The subtraction of this first image from a second image acquired without the tagging leaves a signal from only the selectively tagged blood. Pulse sequence design involves consideration of the duration of the blood transit interval, excitation timing and cardiac gating, static material suppression, inversion excitation pulses, and flow compensation. Each of these considerations must be viewed with respect to the particular application. The method has demonstrated potential application to areas such as the carotid arteries, aortic arch, and peripheral vessels.

Abstract

This paper describes several methods for magnetic resonance angiography that create projection images based solely on flowing blood. To both remove static tissue from the image and generate signals from blood, two classes of methods considered are temporal subtraction and cancelling excitation. Temporal subtraction, analogous to digital subtraction angiography with live and mask images, is performed via phase or magnitude differences in blood signals, while cancelling excitation is characterized by its removal of static structures by selectively exciting only flowing material. Means of projection imaging which incorporate these flow-sensitive methods include variations of thick-slice 2-D spin-wrap imaging, line-scan imaging, and volumetric imaging with time-varying gradients.