Total

Product

The most popular choice by far. Most people suffer from hyperhidrosis on three or more zones. The Dermadry Total was build for hands, feet and underarms treatments. It is therefore the complete cost effective solution to excessive sweating.

$349

Technical Data

Treatment Area: Underarms, Hands, Feet

Current Type: Pulsed Current

Type of device: Tap-Water Iontophoresis Machine

Intended Use: Treating Excessive Sweating (Hyperhidrosis)

Device Classification: Medical Device Class II (Canada)

Hands

Feet

Underarms

Current strength range

1-15 mA

1-25 mA

1-8 mA

Treatment duration (predefined)

20 min

20 min

15 min

Polarity alternating frequency (predefined)

5 min

5 min

2min 30s

Maximum tension

48 V

55 V

30 V

Contraindications

Do not use this device if you have any of the following conditions:

Cardiac pacemaker

ICD (implantable cardioverter defibrillator)

Suspected or diagnosed heart problems (e.g. cardiac arrhythmia)

Seizure disorders (epilepsy)

Pregnant or suspected pregnancy

Metal-containing intrauterine device (IUD)

Metallic implants

Large skin lesions and lesions that cannot be covered with petroleum jelly

Tingling and stinging sensations. Muscle numbness (paresthesia) may occur. Slight pain could be felt at the beginning of the treatment or after the polarity alternating sequence. If you feel any of the above side effects, reduce intensity during next treatment.

Increased sweating: After the first few treatments, you may experience an increase in sweating. This symptom will subside after a few treatments.

Small electric shocks during treatment: In very rare cases, a harmless electric shock may occur if treatment is interrupted suddenly. To avoid this risk, remove your hands, feet or underarm electrodes slowly. Hands, feet and underarms electrodes can be safely removed at any time during treatment.

Skin dryness: Skin may become dry, small lesions or scaling may occur. To reduce these symptoms, use a moisturizing cream after treatments.

Aluminum may cause minor allergic reactions for some users. They may develop red, bumpy, scaly, itchy or swollen skin at the point of contact. If you notice any allergic reaction developing on the treated area of skin, stop using the device and contact Dermadry’s Customer Service.