• Check that EZ137 S1 is in NORM position.
• Check zero adjustment EZ137 R2 (measure at R11) R1 ADZA.
• Check input scanner on EZ137.
• Program error code to ERBUS (error buffer system, service
mode 7.4 and 7.3) and check if this error will be followed by another one (this error code + 1). If so, replace the board.
• Check that filament current limits on EZ126 are properly adjusted.

i have all manuals for this system of you need it contact me,
magmesme@gmail.com
Error 3249 mean : kV set point > Â±4% tolerance from EZ133 X22 , adjustment R2 1 V - 20 kV measured after every desk
data change

Ad Statistics

15,000 service professionals from over 50 different countries have trained in over 35 courses in radiology, mammography, CT, ultrasound, networking, PACS, & DICOM. Classroom & lab facilities to accommodate up to 60 students.