PDR2000A Two-Channel Digital Power Supply and Readout

PDR2000A Two-Channel Digital Power Supply and Readout

Overview

The PDR2000A is a versatile, low cost solution for powering and displaying up to two (2) Baratron® transducers. Both instruments are simultaneously powered, so users may alternate between readings with the press of a button. Readings are also available on demand via the RS-232 interface.

Low cost solution for powering and displaying two Baratron® transducers

Gauge Select, Full Scale Adjust, Calibrate, Zero, Set Point 1 High, Set Point 1 Low, Set Point 2 High, Set Point 2 Low, Torr/mbar Select

Features

Two Configurable Process Trip Points

Two process relay trip points are available and may be assigned to either of the two manometer channels. Upper and lower settings allow the user complete control over the hysteresis of the control trip point. The relays have isolated outputs and are rated for 2 Amps at 240 VAC.

Displays Transducer Input in Torr or mbar Units

The PDR2000 provides ±15 VDC, 750 mA to power a capacitance manometer and displays the 0 to 10 VDC pressure transducer output in Torr or mbar Units. A 9-pin type "D" connector is used to interface the Baratron® manometers.

Two Configurable Process Trip Points

Two process relay trip points are available and may be assigned to either of the two manometer channels. Upper and lower settings allow the user complete control over the hysteresis of the control trip point. The relays have isolated outputs and are rated for 2 Amps at 240 VAC.

Displays Transducer Input in Torr or mbar Units

The PDR2000 provides ±15 VDC, 750 mA to power a capacitance manometer and displays the 0 to 10 VDC pressure transducer output in Torr or mbar Units. A 9-pin type "D" connector is used to interface the Baratron® manometers.