Overview

All in One Imaging Solution: Observation, Image Capture, Measurement, and Sharing

The DSX510i inverted digital microscope is designed for observations and analysis of metal structures. The easy-to-use, all-in-one imaging system provides detailed analysis and reports. An intuitive touch screen-based user interface and up to 9,000x magnification enable users of various experience levels to obtain reliable measurement and inspection results.

Observation

Technology that Reveals What Couldn’t Be Seen Before

High NA — Low-Aberration Field Lenses

With lenses that have higher NA and lower aberration than current digital microscopes, plus improved evenness of light intensity, the DSX510 series of digital microscopes offer high resolution equal to the very best light microscopes.

Zoom Optics Cover a Wide Range of Magnification

Optical Zoom: Get Closer to the Sample

The DSX510i offers optical zoom of up to 13X and digital zoom of up to 30X. A single optical lens can cover the typical magnification range of conventional light microscopes.

Zoom in on the sample

Two Lenses Can Be Mounted at Once: Nosepiece

Two lenses can be mounted at once for an even greater magnification range (70X to 9000X maximum range).

Macro Map: Always Know the Location

As zoom magnification is adjusted to a higher level, the area seen at one time is reduced — Macro map automatically records a full field-of-view image in a separate macro window.

A Variety of Observation Methods, Freely Usable

Quickly and Easily Acquire the Images Required for Observation

The DSX510i offers a variety of observation modes that deliver the highresolution images users expect from high-end optics.

Choose the Observation Method with One Easy Click

A click of a button delivers the required image on screen

With virtually every industrial observation method just a click away, the DSX510i makes it easy to choose the right one for the task at hand. No complicated adjustments needed — simply choose between modes (brightfield, darkfield, MIX [BF + DF], Differential Interference Contrast, polarized light) and start creating high-end images that meet precise demands.

Advanced Image Processing

HDR: High-Definition Visuals Beyond the Human Eye

Sample appearance can vary depending on quality of material, surface conditions, or illumination methods. One of a variety of observation methods made possible by the advanced digital technology of DSX510i, the HDR (High Dynamic Range) function, combines several images taken at different exposures to accurately correct brightness differences on the sample surface, delivering a more consistently accurate sample rendition.

WiDER: Easy Inspection of Samples with High Reflectance Difference

The DSX510i eliminates this issue with WiDER, a proprietary image processing system that takes care of high-contrast problems with one click. No blackouts. No glare.

Color Enhancement: See Only What’s Required

The color enhancement feature of DSX510i allows color to be placed where it is needed while leaving the rest of the image in black-and-white, making it much easier to locate defects. Ideal for focusing on one particular defect.

Anyone Can Make Observations in Optimum Conditions

Best Image Function Ensures Optimal Performance from Any Operator

The DSX510i will set all the necessary parameters to achieve that image. The Best image function ensures optimal images, whether looking for defects, uneven surfaces, or foreign objects. With Best image, anyone can operate the system — beginner to expert — and it can be customized for each operator.

Repeatability: Easily Recall Any Inspection (Observation) Setting

The mechanisms of DSX510i fully digitalized so that every image taken or saved has the conditions it was taken under recorded with the image. If an image is captured with the DSX510i, the operator can retrieve the conditions of capture from the image data with one click, enabling additional observations under the same conditions and settings.

Capturing

Capture Full-Focus Images or 3D Images

EFI: View Uneven Samples in Focus Across the Entire Image

With its EFI (Extended Focal Image) capability, the DSX510i can obtain a clear, in-focus image of an entire sample with one click — no matter how uneven the surface. During EFI, several images are taken while the point of focus is moved up and down. From these images, the areas where the sample was in focus are combined into one image where the whole sample is in focus, allowing precise inspection of uneven surfaces. Olympus’s EFI capturing speed is now faster than ever.

3D Image: One Click Shows the Sample in 3D

With one click, the DSX510i can capture image samples in three dimensions, allowing examination from any angle and a view of the sample as it actually is. With detailed 3D images, sample features or unevenness can be viewed and measured. Height differences and volume can also be measured, making it easier to accurately analyze the sample.

Live Panorama Capturing Covers Large Areas Fast

There is no longer such a problem as “outside the field of view.” With Live panorama, simply move the observation position on the screen, and the motorized stage will move the sample to that position. As the stage moves, the system automatically stitches images into a large single field of view, in real time.

Auto Panorama: One Click Captures a Large Field of View

Simply put the sample on the stage, and start the process with one click. The stage moves in a spiral and the feature captures the required area automatically.

Manual Panorama: Prioritize a Required Area

Obtains real-time image stitching of the desired area in a very short time, by moving the stage vertically, laterally and obliquely.

Automatic Stitching Gives High Quality, High Value-Added Image

High Quality Panorama

Set the number of images, the length, and the starting point, then start the image stitching process. This executes pattern matching and corrects shading, resulting in a high-quality and high value-added image.

EFI and 3D Image Capturing

High quality panorama capability can be coordinated with EFI and 3D image capturing as well. The ability to capture images across a wide area, show uneven surfaces in full focus, and produce 3D images means the data exceeds that of ordinary digital microscopes.

Improved Algorithms for Pattern Matching and Shading Correction

Wide angle, high-resolution and high-quality images are realized with optimized pattern matching and no misalignment.

Large Amounts of Data Automatically Captured

Programmed Recipes: Automatically Capture Independent Points

Through a convenient programmed recipe function, the DSX510i can automatically capture images of several registered points with autofocus.

Measurement

Advanced Technology Supports Precise Measurement

Accuracy and Repeatability Guaranteed

The DSX510i delivers precise and repeatable measurements, with accuracy traceable to national standards.

•To guarantee XY accuracy, calibration work must be undertaken by Olympus’s dedicated service staff

•To issue certificates, calibration work must be undertaken by Olympus’s dedicated service staff

*Differs according to national and regional statutes

The DSX510i Features Telecentric Optics

With telecentric optics, the image size is not altered with changes in focus. Since the image size does not change, precise measurement can be taken without concern for dispersion.

Point of focus Higher / Lower

Ample Measurement Functions

The Features Required to Ensure Optimal Measurement Results

All fundamental industrial microscope measurement capabilities are standard features of DSX510i software, making it easy to obtain optimal measurement results. Optional software is available for 3D measurement, caliper measurement, and particle analysis.

Area/Volume Caliper
Step Live measurement

Particle Line roughness Surface roughness Geometric

A Higher Level of Measurement and Analysis

Automate Measurement with a Simple Wizard Function

Using the simple wizard function of DSX510i, many measurement processes can be easily automated. Once an image is created, the measurement wizard can run automatically or the wizard can also be connected to an automatic image capturing wizard, further automating the imaging and measurement process.

*1 Cannot be used with the embedded standard LED. *2 Calibration by Olympus or dealer specialists necessary. *3 When the objective lens MPLFLN50XBDP or MPLAPON50X is attached, the stroke is 25 x 25 mm.

This site uses cookies to enhance performance, analyze traffic, and for ads measurement purposes. If you do not change your web settings, cookies will continue to be used on this website. To learn more about how we use cookies on this website, and how you can restrict our use of cookies, please review our Cookie Policy.