SS Magnet status (JBoehm) All 3 CC magnets are cold and full. All still in solenoid mode. Will put FC into flip and reverse SSU. The magnets have not run since they tripped last year. Should run and check for "damage" soon. Heater performance on FC is worse - pressure went up by 10 mbar - with the A/C down.

Ramp FC alone to 165 A(?) in flip mode. FC group to define this threshold.

Then test FC + SSU to confirm forces

Extended test of full channel to wait until ops period when we have shift cover

JBoehm has link box wiring documentation near-complete. Windows updates on QP system are now disabled, but need to remove one of the ones that sneaked itself on [PF].

Hydrogen status (VBayliss) Cooldown over Christmas: absorber reached 20K and cooler 12 K. 12 K would freeze LH2; don't have heater control circuits in R9. Will do test with Neon in the next couple of weeks to confirm that a liquid would condense and drip into absorber pot, reducing the temperature difference between them.

Issues arising from previous running A complete list of the issues encountered during last data-taking period needs to be built up. Need to make sure that they will actually be dealt with (or at least ignored consciously).

e.g. What specific actions led to the PPS search drop?

e.g. Why did miceraid5 run out of disk space, when mitigation actions were decided in Feb. 2016? Have they actually been implemented yet?

Need to check eLog/paper log/MOM notebook and find what else is lurking

Need the list for next week's meeting, as we are running out of time in which to make fixes and test PHodgson [as DC] will ask Melissa for copies of her lists.

Shutdown work in Jan and Feb The official version The Beamstop and Potterax door have both been serviced. HN thinks that, with both chillers active, it should be possible to run conventional magnets at full power. After some discussion, the present plan is:

What is status of neutron monitor débâcle? Ours is still with ISIS Health Physics - we should get an update at Monday's MICE-ISIS Safety Mtg. The temporary monitor is still running OK but we should get the calibration checked.