Trigno™ Analog Adapter

Integrate analog outputs from third-party equipment

Use our wireless sensor technology to seamlessly integrate analog outputs from third-party equipment with the Trigno™ System. A single Trigno™ adapter is capable of sampling and transmitting up to 4 analog channels, in accordance with its technical specifications. The adapter sensor is ideal for synchronization with dynamometers, as it supports both Biodex and HUMAC Norm (CSMi) systems to collect position, torque and velocity measurements. Simply connect the analog port from one of these systems to the Trigno Analog Adapter and begin acquiring data.

Applications

Useful Links

Specifications

Channel

DSUB Pin

Sampling Period

Resolution Depth

Low Pass Cutoff

1 EMG

6

27/52 ms

16 bits

50 Hz

2 ACCX

9

6.75 ms

10 bits

50 Hz

3 ACCY

7

6.75 ms

10 bits

50 Hz

4 ACCZ

8

6.75 ms

10 bits

50 Hz

common

1,2,3,4,5

–

–

–

The Trigno™ Wireless EMG Systems are battery-powered biofeedback devices that enable researchers and clinicians to acquire EMG and related signals from subjects for biofeedback purposes. They are intended for relaxation training and muscle reeducation. Interpretation of the EMG and supporting signals by a qualified individual is required.