Applications

Digital
Hardware Verification and Debug

Monitor & Measure Digital Hardware
Performance

Breakthrough
Solutions for Real-time Digital Systems Analysis

The TLA700 Series
offers a range of digitizing oscilloscope modules that provide the acquisition
capabilities of the world's best digitizing oscilloscopes, tightly integrated
with the TLA700 logic analyzers. The 2 and 4 channel digitizing oscilloscope
modules available in the TLA700 Series offer sample rates of up to 5 GS/s
and bandwidths of up to 1 GHz, with 15 Kb memory depth per channel at all
times.

The digitizing oscilloscope modules offer precise time correlation
and flexible cross-triggering with other installed modules. This enables you
to see the quality of critical signals time-correlated with the digital signals
represented by the logic analyzer modules.

Runt Pulse Trigger - Triggers
on a pulse that crosses one threshold but fails to cross a second threshold
before crossing the first again; settable from 2 ns to 1 s.Runt Pulse trigger.

Slew
Rate Trigger - Triggers on pulse edge rates that are either faster or
slower than a set rate, edges can be rising, falling, or either; settable
from 2 ns to 1 s.Slew
Rate trigger.

Logic Pattern Trigger - Triggers
when a logical combination (AND, OR) of all the input channels (Hi, Lo, Don't
Care) stays true or false for a specified period of time; settable from 2 ns
to 1 s.Logic
Pattern trigger.

Setup-and-hold Trigger - Triggers
on violations of both setup time and hold time between clock and data which
are on separate input channels; setup time settable from -100 ns to +100 ns
in 200 ps increments; hold time settable from -1 ns to +102 ns; minimum settable
window of setup time + hold time is 2 ns.Setup-and-hold trigger.