Rohde & Schwarz FSU26 Spectrum Analyzers

Refurbished Rohde & Schwarz FSU26

Rohde & Schwarz FSU26 Details

Rohde & Schwarz FSU26

Spectrum Analyzer 20 Hz - 26 GHz

The Spectrum Analyzer R&S FSU is another milestone. New circuit concepts, advanced RF components, A/D converters and ASIC technology, extensive experience gained from a variety of applications and customer requirements - all this combines to form a solid basis on which the R&S FSU was developed. Its unparalleled features enable the use of new test methods - to your advantage. The future-oriented concept combines unprecedented performance with continuity. The R&S FSU is compatible with the R&S FSE and R&S FSIQ, the industry standards to date. Test routines and sequences generated for the R&S FSE or R&S FSIQ can be used on the R&S FSU too. The R&S FSU family thus secures your investment.

The operating concept of the top analyzer R&S FSU is the same as that of the general purpose analyzer R&S FSP, so these instruments offer a uniform platform for a variety of applications.

Performance

Frequency range 20 Hz to 26.5GHz

Displayed average noise level -158 dBm (1 Hz)

Phase noise -123 dBc (1 Hz) at 10 kHz

TOI typ. +25 dBm

Resolution bandwidth 1 Hz to 50 MHz

RBW-characteristics

Digital filters from 10 Hz to 100 kHz (1,2,3,5 steps)

Analog filters from 300 kHz to 10 MHz (1,2,3,5 steps)

39 channel filters from 100 Hz to 5 MHz

RRC filters (NADC, TETRA, 3GPP)

EMI filters: 200 Hz, 9 kHz, 120 kHz FFT filters 1 Hz to 30 kHz

FFT filter mode adjustable

Rohde & Schwarz FSU26 datasheet & specifications

Fill out the form below for more information about the Rohde Schwarz FSU26