Lloyds EZ50 Advanced Universal Testing System

The EZ50 materials testing system combines high performance, flexibility and exceptional ease of use. It is ideal for testing applications up to 50kN (11,240lbf) in tension and direct compression. The EZ50 features a twin column, twin lead screw frame, and incorporates a crosshead guidance system to prevent side loading of the sample under test. A high resolution encoder is used to measure sample extension and also provide a high accuracy speed control. The crosshead is driven by a high precision servo motor and DC servo system to achieve a wide range of speeds throughout the full load range. The EZ20 features a multi-position control console with membrane multi-function keypad and easy to read LCD display. The machine may be programmed to perform up to 10 different tests and the statistics for each test are automatically calculated. Up to 600 results may be stored from any of the test setups. Tests may also be performed using state of the art NEXYGEN PLUS Software via the integral RS232 interface. The crosshead speed is programmable up to 0.1mm/min to 250 mm/min ( 0.004 in to 9.84 in/min ). Speed accuracy remains +/- 0.2% throughout the range. Maximum travel is 855mm ( 33.66 in ) and the width between the columns is 400mm ( 15.75 in ). Frame stiffness is better than 50kN/mm. The EZ20 load measuring system exceeds the requirements of BS EN ISO7500: 1999 Class 0.5, ASTM E4, and DIN51221. The machine performance can be maximized when combined with NEXYGEN PLUS. This Windows based software is easy to use and allow detailed force measurement and materials testing data to be stored and displayed in graphical and tabular formats. The software also lets you manipulate data to produce detailed analytical reports, plus offers the ability to integrate with popular Windows software packages including EXCEL, ACCESS, WORD, POWERPOINT, and OUTLOOK. You must choose at least one load cell to have a functional system. Ships via common carrier.

10 programmable test setups

4 line backlit display

2 key press operation

Works with Nexygen series software

Saves 600 test results

Easy set safety limits

Extension resolution less than 0.1 micron

Selectable weighing units

Multiple languages

Exceeds national and international standards

The EZ50 materials testing system combines high performance, flexibility and exceptional ease of use. It is ideal for testing applications up to 50kN (11,240lbf) in tension and direct compression. The EZ50 features a twin column, twin lead screw frame, and incorporates a crosshead guidance system to prevent side loading of the sample under test. A high resolution encoder is used to measure sample extension and also provide a high accuracy speed control. The crosshead is driven by a high precision servo motor and DC servo system to achieve a wide range of speeds throughout the full load range. The EZ20 features a multi-position control console with membrane multi-function keypad and easy to read LCD display. The machine may be programmed to perform up to 10 different tests and the statistics for each test are automatically calculated. Up to 600 results may be stored from any of the test setups. Tests may also be performed using state of the art NEXYGEN PLUS Software via the integral RS232 interface. The crosshead speed is programmable up to 0.1mm/min to 250 mm/min ( 0.004 in to 9.84 in/min ). Speed accuracy remains +/- 0.2% throughout the range. Maximum travel is 855mm ( 33.66 in ) and the width between the columns is 400mm ( 15.75 in ). Frame stiffness is better than 50kN/mm. The EZ20 load measuring system exceeds the requirements of BS EN ISO7500: 1999 Class 0.5, ASTM E4, and DIN51221. The machine performance can be maximized when combined with NEXYGEN PLUS. This Windows based software is easy to use and allow detailed force measurement and materials testing data to be stored and displayed in graphical and tabular formats. The software also lets you manipulate data to produce detailed analytical reports, plus offers the ability to integrate with popular Windows software packages including EXCEL, ACCESS, WORD, POWERPOINT, and OUTLOOK. You must choose at least one load cell to have a functional system. Ships via common carrier.