Please notice!
This website uses cookies to provide necessary site functionality and improve your online experience. By using this website, you agree to the use of cookies as outlined in the
Privacy Statement
.

As a global market leader with numerous subsidiaries and distributors, Miltenyi Biotec is
committed to providing our customers around the world with the highest quality products.
In addition to direct selling in more than 20 countries in North America, Europe and
Asia/Pacific, Miltenyi Biotec also provides support for our customers through an
extensive distributor network covering dozens of additional countries.

As a global market leader with numerous subsidiaries and distributors, Miltenyi Biotec is
committed to providing our customers around the world with the highest quality products.
In addition to direct selling in more than 20 countries in North America, Europe and
Asia/Pacific, Miltenyi Biotec also provides support for our customers through an
extensive distributor network covering dozens of additional countries.

1 Referred value indicates the average of multiple experiments and can differ for individual sample values.2 For counting performance, full 96-well plates were loaded with 200 μL/well of peripheral blood mononuclear cell (PBMC) suspension at a nominal concentrationof 5000 cells/μL. The uptake volume was set to 50 μL at medium flowrate.3 For carry-over, full 96-well plates were loaded with 200 μL/well of PBMC suspension at a nominal concentration of 5000 cells/μL in every other well (“SRC-wells”).Alternating wells are loaded with an equal volume of MACSQuant® Running Buffer (“CO-wells”). The uptake volume was set to 50 μL at medium flowrate.The carry-over is defined by sum(CO-singlet count)/sum(SRC-singlet count) ×100%.4 The measurement speed is determined by measuring the time between the first movement of the robotic arm into the first measured well and the firstmovement out of the last well. The measurements itself were carried out at the highest possible flow rate in fast mode measuring 5 μL per well.