This
is a high performance intensified CCD camera
system with gating times down to 3 ns. With its
12 bit dynamic range and a high resolution CCD image sensor it
features an excellent signal-to-noise-ratio and the ability of
single photon detection. The system is suited for applications
in environments with high electromagnetic disturbances. A high
speed serial fiber optic data link connects the system to the
PC. The camera can be triggered by light or electrical input.

unit

setpoint

dicam
pro VGA

dicam
pro SVGA

resolution
(hor x ver)

pixel

640
x 480

1280
x1024

pixel
size
(hor x ver)

µm2

9.9
x 9.9

6.7
x 6.7

peak
quantum efficiency

%

depends
on photocathode material

depends
on photocathode material

dynamic
range A/D

bit

12

12

readout
noise

e-
rms

@
12.5 MHz

13..14

7..8

imaging
frequency, frame rate

fps

@full
frame

30

8

exposure
time

s

3ns..1000s

3ns..1000s

binning
horizontal

1,
2, 4, 8

1,
2, 4, 8

binning
horizontal

1,
2 , 4, 8, 16, 32

1,
2 , 4, 8, 16, 32

optical
input

Nikon
F-mount, C-mount or special mounts

Nikon
F-mount, C-mount or special mounts

data
interface

PCI
local bus, Rev. 2.1

PCI
local bus, Rev. 2.1

cooled
CCD temperature

°C

-15

-12

mnimum
interframing time

ns

500

500

photocathode
material

S20.
S25, GaAs, GaAsP, others on request

S20.
S25, GaAs, GaAsP, others on request

phosphor
screen material

P43,
P46

P43,
P46

MCP diameter
depends on photocathode material. The new GaAs- and GaAsP-MCPs
are only available with a diameter of 18 mm while all others have
25 mm!