XYG series HF HV X ray machine, becoming the best choice for customers in high quality flaw detection, which is composed of XFG series oil insulation HF HV power supply, industrial metal ceramic X ray tube, and etc. Control type is analog control, working frequency take up to 40KHZ, control precision increased 1%.

Technical Parameters:

Model

XLT-225/21

XLT-225/22

XLT-225HP-11KB

XLT-225HP/11

Max tubevoltage

225KV

225KV

225KV

225KV

Max power

640w/1600w

640w/3000w

800w/1800w

800w/1800w

Focal size EN12543

1.0mm/3.0mm

1.0mm/5.5mm

0.4mm/1.0mm

0.4mm/1.0mm

Max filament current

4.1A/4.2A

4.1A/4.2A

4.1A/4.1A

4.1A/4.1A

Max filament voltage

4.2V/5.5V

3.0V/5.5V

2.9V/7.3V

2.9V/7.3V

Inherent filter

0.8mmBe

0.8mmBe

0.8mmBe

0.8mmBe

Target material

W

W

W

W

Target angle

22degree

22degree

11degree

11degree

Beam angle

40degree

40degree

90x25degree

40x30degree

Cooling medium

water

water

water

water

Mini cool medium flux

4L/min

4L/min

4L/min

4L/min

Max inner temperature

35

35

35

35

HV cable connector

R24

R24

R24

R24

Tube Technology

Tube Lab

Looking for ideal Ceramic X Ray Detect Tube Manufacturer & supplier ? We have a wide selection at great prices to help you get creative. All the Portable X Ray Tube are quality guaranteed. We are China Origin Factory of Real Time Radiography X Ray Tube. If you have any question, please feel free to contact us.