INMEL 51

Calibrator INMEL 51 integrate generation and measurement function of electric signals, used in automatic industry.
The main features of this devices is high accuracy and resolution of generating or measuring signals, friendly operation and large display.

Checking and calibration:

measuring transducers,

temperature meters,

measuring and control lines.

BASIC PARAMETERS:

The measurement and generation

tabela1

INPUT/OUTPUT RANGE

RESOLUTION

ACCURACY

-10mV~100mV

0.01mV

±0.025%+2digits)

Maximal input voltage?30V

Measure (input)/Simulate (output) Thermocouple Specification

tabela1

FUNCTION

RANGE

RESOLUTION

ACCURACY

REFERENCE ERROR

J

-200~1200°C / -328~2192°F

0.1°C/°F

±(0.3°C+10uV)

±0.3°C

K

-200~1370°C / -328~2498°F

0.1°C/°F

±(0.3°C+10uV)

±0.3°C

T

-200~400°C / -328 ~ 752°F

0.1°C/°F

±(0.3°C+10uV)

±0.3°C

E

-200~950°C / -328~1742°F

0.1°C/°F

±(0.3°C+10uV)

±0.3°C

R

-20~1750°C / -4~3182°F

0.1°C/°F

±(0.1°C+10uV)

±0.3°C

S

-20~1750°C/ -4~3182°F

0.1°C/°F

±(0.1°C+10uV)

±0.3°C

B

600~1800°C / 1112~3272°F

0.1°C/°F

±(0.1°C+10uV)

±0.3°C

N

-250~1300°C / -418~2372°F

0.1°C/°F

±(0.3°C+10uV)

±0.3°C

Maximal input vol

General Specifications

Maximum voltage applied between any jack and earth ground or between any two jack?30V