Surgical Microscope Cold Light

Surgical Microscope Cold Light is commonly used by Dentists for operating a dental surgery to the patient. It is lightweight and flexible, can also be used in microscopy and diagnosis of diseases. It has Filter with the Cold Light Source that makes the light get in to the target and the make a clear image of the effected part. Specifications

Microscope Section

Binocular Tube : 45° Inclined

Eye Pieces : 12.5x/ 10x /16mm Dipptric Adjustment

Magnification Changes : Built In Manual 3-Step Changer

Total Magnification : 7.5x, 12.5x, 20x

Field Of View : 37.8mm, 23.5mm, 14.5mm

Working Distance : 200mm

Interpupillary Distance Adjustment : 55 mm to 75 mm

Illumination Section

Type : By Optical Light Guide System

Light Source : 24V/250 Watts Halogen Lamp

Illumination Control : Continuous Adjustment

Field Of Illumination : 70mm

Filter : Cobalt Blue, Red Free, Heat Absorbing Filter

Stand Section

Type : Floor Stand

Vertical Range Of Counterbalanced Arm : 400 mm

General Information

Base Size : 600mm X 550mm

Weight : 50Kg.

Height : 1550 mm

Electrical Power : AC 220V, 240V, 50 / 60 Hz

Power Consumption : 220 VA / 110V is also Available

Optional Accessories

Beam Spliter With CCD Camera Attachment

CCD Camera

Table Clamp Attachment

Optional accessories are listed as separate items in "Accessories" section under products, there brought separately.