FI-8000 Optical gas indicator

This highly accurate Optical Gas Indicator is customizable for up to 8 gases and ranges, and uses optical interferometry to acuratley measure a gas concentration in a known base gas, such as air. Numorous gases are available to choose from, and if not listed on our current gas list, please ask, it is probably possible to include.

Because the principle operation is the comparison of the optical refractive index of a gas, there are no moving parts or problems with sensor drift. This gives a stable calibration, allowing for long periods between calibration, thus saving on maintenance cost.

The FI-8000 is the latest model from Riken in a long line of optical interferometers dating back to the 1930’s when Riken’s founder first developed the technology for Methane detection to help prevent explosion accidents on oil tankers and in coal mines.

In the 1970’s Weatherall developed with Riken the instrument for anaesthetic gases. In conjunction with Dr John F Nunn’s research, published in the British Journal of Anaesthesia, helped validate the principle of interference refractometry as a technique suitable for measuring anaesthetic gases such as Halothane, Isoflurane, Sevoflurane, Desflurane and Enflurane.

Chamber length: 5mm

Detection gas

Base gas

Range

Methyl bromide

Air

0~100vol%

Sulfur Hexafluoride

Air

0~100vol%

Sulfur Hexafluoride

Air

0~99.9%up

Sulfur Hexafluoride

N2

0~100vol%

Propane

Air

0~100vol%

Iso-butane

Air

0~100vol%

N-butane

Air

0~100vol%

Flon 22

Air

0~100vol%

Dimethyl Ether

Air

0~100vol%

Dimethyl Ether

N2

0~100vol%

Xenon

Air

0~100vol%

Ethylene

Air

0~100vol%

Chorine

Air

0~100vol%

Vinyl chloride

N2

0~100vol%

Chamber length: 5mm

Detection gas

Base gas

Range

Propane

Air

0~101.3 MJ/m3Gross 0°C

Butane

Air

0~134.2 MJ/m3Gross 0°C

Propane

Air

0~93.15 MJ/m3Net 0°C

Butane

Air

0~123.7 MJ/m3Net 0°C

Chamber length: 24mm

Detection gas

Base gas

Range

Halothane

O2

0~6vol%

Isoflurane

O2

0~8vol%

Sevoflurane

O2

0~10vol%

Desflurane

O2

0~20vol%

Halothane

Air

0~6vol%

Isoflurane

Air

0~8vol%

Sevoflurane

Air

0~10vol%

Desflurane

Air

0~20vol%

Enflurane *

O2

0~10vol%

Enflurane *

Air

0~10vol%

*Available on request. Select up to a maximum of 8 gas ranges per instrument.