RAE UltraRAE 3000

Benzene, Butadiene, or Halocarbon specific PID Monitor

The UltraRAE 3000 PID is an advanced compound-specific monitor. It has a range of 0.05–10,000 ppm in VOC mode or 50 ppb to 200 ppm in benzene mode. 3 second VOC response times; 60-second benzene response times.

RAE Systems’ UltraRAE 3000 photoionization detector (PID) is an advanced compound specific VOC monitor with both total VOC and compound specific modes. The UltraRAE 3000 can be configured with a 9.8 eV (standard), 10.6 eV, or 11.7 eV PID lamp. Its sensor detects VOCs at an extended range of 0.05–10,000 ppm with improved linearity when in VOC mode and 0.5–200 ppm in benzene specific mode. Integrated humidity and temperature sensors provide compensation and automatic adjustment of the PID. The UltraRAE 3000 features a fast 3 second response time in VOC mode or a 60 second response in a compound specific mode. The UltraRAE 3000 is ideal for a range of applications including hazardous material response, pre-entry screening during plant and refinery maintenance, refinery down-stream monitoring, marine spills, and more.

The UltraRAE 3000 compound specific portable VOC monitor is the suggested replacement for the UltraRAE.

The UltraRAE 3000 features an IP-65 design for use in harsh environments, high resistance to EMI/RFI interference, memory for 6 months of datalogging at 1 minute intervals, and a 95 dB alarm buzzer with bright flashing LED alarm indications. A new sampling probe design provides instant tube breakthrough visibility. In compound specific mode a 15 minute benzene STEL calculation with real time recording and measurement is possible. The UltraRAE 3000 comes with an integrated correction factor list of more than 200 compounds.

Other hardware features of the UltraRAE 3000 include a large backlit display with intuitive navigation and support for multiple languages. RAE Sep tubes save money and reduce cost of ownership as compared to typical gas detection tubes. The UltraRAE 3000 does not require annual factory calibration and is less expensive to own than a portable GC. Its rechargeable, field-swappable Lithium-ion battery pack provides up to 16 hours of continuous operation and an alkaline battery pack adapter provides additional runtime when needed.

UltraRAE 3000 Features

PID Technology

3 second response time when in VOC mode

60 second response time when in compound specific mode

Extended range of 0.5–10,000 ppm with improved linearity in VOC mode

Detects benzene at a range from 50 ppb to 200 ppm

Comes with 9.8 eV PID Lamp (10.6 eV or 11.7 eV optional)

Operates in compound specific or total VOC modes

RAE-Sep™ benzene tube enable highly specific readings

Real time 15 minute benzene STEL calculation

Integrated correction factors list featuring more than 200 compounds

Field swappable Lithium-ion battery packs provide up to 16 hours of operation

Field Supplies

Great Prices for Project Supplies

Did you know that Equipco also sells environmental field supplies?

Whether you need calibration standards, tubing, or other supplies like gloves and labels, we stock everything you need to finish the job. Our high quality field supplies, including NIST traceable calibration solutions and gases, come at significant savings over manufacturer brands.

Used Equipment

Great Discounts for Used Equipment

Did you know that Equipco also sells used equipment?

Whenever we have used equipment to sell, whether like-new equipment used only for demonstration or a former rental, well maintained and available for a great price, find what you’re looking for here and check back for updates.

RKI Eagle 2

Multiple Gas Monitor

RKI’s Eagle 2 is an advanced portable gas detector capable of monitoring up to six sensors simultaneously including a PID. 16 different sensors are available providing for hundreds of configuration possibilities.

Landtec GEM5000

Landfill Gas Analyzer

The GEM5000 is the latest in the line of Landtec's GEM landfill gas analyzers. Now lighter, smaller, more accurate, and with an improved user interface, the only instrument you need to take into the field just got even better.

Levelogger Edge

Water Level Datalogger

Advanced and durable, the Levelogger Edge is the latest water level datalogger from Solinst. An all new sensor design manages to achieve excellent accuracy while being more robust then previous models and remaining economical.

MultiRAE Series

Multiple Threat Monitor

The MultiRAE Series of handheld gas detectors includes three models for detection of up to six threats simultaneously. Choose from 25 toxic sensors, 3 PID sensors with a 9.8 eV lamp option, and even a gamma radiation sensor.