Note: If you use pre-IEEE488.2 command types, the measuring instrument will respond to the "OD" command by writing multiple lines including the "END" line, and returning them one by one as they are requested. Therefore after sending the "OD" command, lines must be read out repeatedly until the "END" string is received.

Please refer to the Communications Interface manual for your instrument, for more information about the format of returned data from the various commands.

Related Products & Solutions

The PZ4000 is designed for High Frequency Power Measurements as well as Transient Power Measurements. Its real-time acquisition system can perform cycle-by-cycle power measurements. One to Four input elements offer Basic Power Accuracy of 0.2% with 2 MHz Bandwidth and 5MS/sec digitizing rates.

The High-end WT1600 is designed for a wide range of applications, from energy-saving to large load applications. The WT1600 has a six element input capability to provide maximum flexibility, and Basic Power Accuracy of 0.1% with 1 MHz Bandwidth.

The WT2000 digital powermeter series has been designed with emphasis on basic performance (bandwidth, accuracy, response speed, and noise immunity) from the viewpoint of measurement of electrical quantities. The broad range of functions of these power analyzers enable them to be used in various fields of applications.

This site uses cookies to simplify and improve your usage and to provide you with the best experience of this website. Cookies are small text files stored on the device you are using to access this website. If you ignore this message and continue without changing your browser settings, we will assume that you are consenting to our use of cookies. For further information on our use of cookies, please see our Cookie Policy.