WARNING
These products can expose you to chemicals including Di(2-ethylhexyl)phthalate (DEHP),
which is known to the State of California to cause cancer and birth defects or other reproductive harm.
For more information go to www.P65Warnings.ca.gov

The choice of
hemodialysis professionals for over 30 years, these
compact instruments have been designed specifically to
test dialysate solutions. By measuring electrical
conductivity, they will quickly determine if dialysate
concentrations are correct.

To use these hand-held
instruments, the operator simply fills the cell cup with
dialysate solution and pushes a button. The needle
indicates the total concentrations of salts in the
solution.

Model D-1 has an expanded scale of
11-15 millimhos and is useful for testing final
dialysate. A unique feature of this model is the
adjustable green color band above the dial. This
provides an immediate visual confirmation of whether or
not the dialysate concentration is within the patient’s
“safe zone”.

Model D-2 features dual scales on
the dial and can test both acetate and bicarbonate
dialysate. .

Both instruments have built-in
cell cups and are completely self-contained. They are
powered by 9-volt batteries which last for at least
2,000 tests. A glow light serves as a battery reserve
indicator.

Thousands of Myron L® Dialysate Meters are
still in daily use around the world after years of reliable
service

FEATURES

Fast and accurate dialysate tests

Models for both acetate and
bicarbonate solutions

Chip and crack resistant
polyethylene cell cup

Rugged design

Simple to use

Automatic temperature
compensation

AUTOMATIC TEMPERATURE COMPENSATION:
WHY IT’S IMPORTANT

The graph below illustrates conductivity
readings of a 14.0 millimhos dialysate sample tested with a
non-temperature compensated conductivity meter. The same
sample APPEARS to have a higher level of dialysate
concentration when heated (and lower concentration when
chilled). Actually, only conductivity readings change with
temperature. Dialysate concentration does not change. Some
instruments ignore the importance of temperature. Others
require a separate thermometer for testing sample
temperature then manual adjustment of the meter. All Myron L®
instruments, however, feature fast and precise Automatic
Temperature Compensation. The result: a more accurate,
useful instrument.

Accessories
we recommend

Standard
Solutions Myron L® standard solutions are
precisely formulated and are used to recalibrate
Dialysate Meters for greatest accuracy. The 12.5, 13.4
and 14.0 millimhos Standards are the same used in Myron
L® factory calibration of Models D-1 and D-2.

Porta-Pak Carry
Case made of ABS plastic and foam lined, giving maximum
protection to either model Dialysate Meter. (Plunger
must be removed from cell cup when storing in case.)

Limited
WarrantyMyron L® Dialysate
Meters have a Two (2) Year Limited Warranty. If
an instrument fails to function normally, return
to factory prepaid. If, in the opinion of the
factory, failure was due to materials or
workmanship, repair or replacement will be made
without charge. A reasonable service charge will
be made for diagnosis or repairs due to normal
wear, abuse, or tampering. Warranty is limited
to the repair or replacement of the Myron L®
Dialysate Meter only. The MyronLCompany
assumes no other responsibility or liability.