Categories

Neurosign There are 7 products.

Ideal for Meticulous Tumor and Scar Dissections
Accurate to Approximately 0.5 mm, but, will Not Stimulate Through a Layer of Tissue Covering the Nerve (current must be increased)
Bipolar Stimulating Probe
Circumferential Anode Pole and an Inner Straight Cathode Pole

Ideal for Meticulous Tumor and Scar Dissections
Accurate to Approximately 0.5 mm, but,...

Ideal for Meticulous Tumor and Scar Dissections
Accurate to Approximately 0.5 mm, but, will Not Stimulate Through a Layer of Tissue Covering the Nerve (current must be...

Accurate Stimulation Through Small Amounts of Tissue
Current is Passed Between Two Probe Tips for Localized Current, but, Will Still Flow Through Adjacent Tissue
The Further the Tip Separation, The Greater the Current Penetration (But Less Accurate)
It Will Penetrate Through Bone and Tissue
3 mm Tip Separation
1 mm Exposed Insulation

Accurate Stimulation Through Small Amounts of Tissue
Current is Passed Between Two...

Accurate Stimulation Through Small Amounts of Tissue
Current is Passed Between Two Probe Tips for Localized Current, but, Will Still Flow Through Adjacent Tissue
The Further the...

Intended for Stimulating Across the Bulk of the Tumor to Determine Whether a Nerve is Lying Behind the Structure
Not Designed for Accurate Nerve Stimulation (Coarse Stimulation Only)
Improved Accuracy can be Achieved by Placing the Ground as Close as Possible to the Surgical Region and by Using the Minimum Current to Stimulate the Nerve

Intended for Stimulating Across the Bulk of the Tumor to Determine Whether a Nerve is...

Intended for Stimulating Across the Bulk of the Tumor to Determine Whether a Nerve is Lying Behind the Structure
Not Designed for Accurate Nerve Stimulation (Coarse Stimulation...

Tests Pedicle Integrity
Teflon Insulated Shaft, Terminating in a 3 mm Ball Tip
Lead Wire Ends with a Touch Proof Connector
Shaft May Be Bent for Use Under a Microscope
Requires a Separate Needle to Act as a Return for Current (Anode)
Single Use Only

Ideal for Nasopharyngeal Recordings, Stimulating Cranial Nerves, Peripheral Motor Nerves and Spinal Nerve Roots and Assessing Pedicle Screw Placement
Teflon Insulated Shaft, Terminating in a 2.3 mm Ball Tip
Lead Wire Ends with a Touch Proof Connector
Requires a Separate Needle to Act as a Return for Current (Anode)
Shaft May Be Bent for Use Under a Microscope