The Stability Platform measures balancing ability, which is essential to successfully performing many activities. The Model 16030 Stability Platform features fully integrated timing functions for test control and electronic angle measurement for unsurpassed accuracy. The platform control allows a wide range of user settable parameters including variable test times, selectable angle limits, and digital tilt angle readout. With the Stability Platform's rugged design and electronic capabilities, it will provide many years of reliable operation.

FEATURES

Electronic tilt angle measurements with digital angle readout

User selectable balance thresholds for left, right and center zones

Platform re-zero ability

Built-in timing functions for test and rest timing

Built-in repeat cycle counter for multiple trials

Built-in display for test data

Test Parameter storage

Analog output voltage proportional to tilt angle

Event marker to indicate test and rest intervals

Tone indicator to signal beginning and end of tests

Friction Free tilt action

Remote start

SPECIFICATIONS

ELECTRONIC SPECIFICATIONS

Angle Measurement Resolution: 1.0°

Measurement Limit Range: +/- 90°

Angle Limit Setting Resolution: 1.0°

Analog Output Voltage Range: 0-5 Volts

Analog Output Resolution: .028 V/°

Analog Output Rate: 25 samples/sec

Analog Output Time: 0.040 sec/sample

Test/Rest Time Limit: 999 sec

Test/Rest Time Resolution: 1.0 sec

Repeat Cycle Limit: 25

Test Timing Resolution: 0.001 sec

Timing Accuracy: .005%

Power Supply: 10 V external adapter

Event Mark Output: 0-5 V TTL

PLATFORM SPECIFICATIONS

Platform Dimensions: 42" x 25.5"

Overall Dimensions: 42" x 38" x 22"

Platform Material: 3/4" Plywood

Frame Material: 1/8" Aluminum

Email For Quotation!Listed are retail price only, for healthcare institutions and dealers, please email us for quotation