High-performance generators with excellent features over a wide frequency range

R&S®SMGU and R&S®SMHU are ideal for applications which the majority of signal generators cannot handle. In addition to out-of-channel measurements, they are for instance able to determine the spurious rejection of radiotelephone equipment up to 4 GHz as laid down by CEPT.

Brief Description

The frequency can be set with a resolution of 0.1 Hz over the entire range, and this is sufficient even for measurements on extremely narrowband DUTs. Both instruments supply frequencies down to 1 kHz.

Key Facts

R&S®SMGU and R&S®SMHU are ideal for applications which the majority of signal generators cannot handle. In addition to out-of-channel measurements, they are for instance able to determine the spurious rejection of radiotelephone equipment up to 4 GHz as laid down by CEPT.

High-performance generators with excellent features over a wide frequency range

R&S®SMGU: 100 kHz to 2160 MHz

R&S®SMHU: 100 kHz to 4320 MHz

Brief Description

Frequency

The frequency can be set with a resolution of 0.1 Hz over the entire range, and this is sufficient even for measurements on extremely narrowband DUTs. Both instruments supply frequencies down to 1 kHz. The frequency setting time is below 10 ms. In the fast mode up to 200 user-defined frequencies can be handled by means of a trigger signal or by memory sweep in less than 1 ms per setting.

Spectral purity

R&S®SMGU/SMHU fulfill requirements for selectivity measurements on top-class receivers. Signals of extremely high spectral purity afford critical adjacent-channel, in-channel and out-of-channel measurements with a wide tolerance margin. Phase noise remains low right up to the carrier. R&S®SMGU/SMHU are therefore ideal for LO applications or as a low-noise reference in noise measurement systems.

Frequency modulation

The FM modulation frequency range extends from DC to 1 MHz. In FM DC mode a high carrier-frequency accuracy is attained. The frequency offset occurring with FM DC selected is extremely small.

Amplitude modulation

The whole of the modulation frequency range can be used down to carrier frequencies of less than 100 kHz. The minimal phase shift at 30 Hz (AM DC) and a flat frequency response make for the precision amplitude modulation that is required for testing VOR/ILS navigation receivers.

Pulse modulation

Rise/fall times of 20 ns (typ. <10 ns for frequencies >200 MHz) and an on/off ratio of 80 dB open up a wide range of possibilities for testing telemetry, microwave link, radar and satellite communications systems.

Digital and analog sweep

In addition to the digital, step-by-step sweep with presettable start and stop frequency, span, step width and step time, an analog frequency and level sweep is also provided.

Phase offset

The phase of the RF output signal can be varied in steps of 1 using keyboard entry or the spinwheel. This makes it easier to adjust for phase quadrature during noise measurements and to investigate phase-critical components.

Thank you.

Your request was successfully sent!

We will get in touch with you soon.

Your Rohde & Schwarz Team

Get a Demo

Inquiry*

Please enter your inquiry.

First Name*

Please enter your first name.

Last Name*

Please enter your name.

Email*

Please enter a valid email address.

Company*

Please enter your company name.

City*

Please enter your city name.

ZIP*

Please enter your postal code.

Location*

Please enter your location.

Marketing permission

I want to receive information from Rohde & Schwarz via

Email

Post

What does this mean in detail?

I agree that Rohde & Schwarz GmbH & Co. KG and the Rohde & Schwarz entity or subsidiary company mentioned in the imprint of this website, may contact me via the chosen channel (email or postal mail) for marketing and advertising purposes (e.g. information on special offers and discount promotions) related to, but not limited to, products and solutions in the fields of test and measurement, secure communications, monitoring and network testing, broadcast and media, and cybersecurity.

Your rights

This declaration of consent may be withdrawn at any time by sending an email with the subject "Unsubscribe" to news@rohde-schwarz.com. Additionally, a link to unsubscribe from future email advertisements is contained in each email sent. Further details on the use of personal data and the withdrawal procedure are set out in the Statement of Privacy.

Please enter the characters shown in the image

The input was not correct.

Email confirmation

Do not enter data in this field

Thank you.

Your request was successfully sent!

We will get in touch with you soon.

Your Rohde & Schwarz Team

Get a Quote

Inquiry*

Please enter your inquiry.

First Name*

Please enter your first name.

Last Name*

Please enter your name.

Email*

Please enter a valid email address.

Company*

Please enter your company name.

City*

Please enter your city name.

ZIP*

Please enter your postal code.

Phone*

Please enter your phone number.

Location*

Please enter your location.

Marketing permission

I want to receive information from Rohde & Schwarz via

Email

Post

What does this mean in detail?

I agree that Rohde & Schwarz GmbH & Co. KG and the Rohde & Schwarz entity or subsidiary company mentioned in the imprint of this website, may contact me via the chosen channel (email or postal mail) for marketing and advertising purposes (e.g. information on special offers and discount promotions) related to, but not limited to, products and solutions in the fields of test and measurement, secure communications, monitoring and network testing, broadcast and media, and cybersecurity.

Your rights

This declaration of consent may be withdrawn at any time by sending an email with the subject "Unsubscribe" to news@rohde-schwarz.com. Additionally, a link to unsubscribe from future email advertisements is contained in each email sent. Further details on the use of personal data and the withdrawal procedure are set out in the Statement of Privacy.

Please enter the characters shown in the image

The input was not correct.

Email confirmation

Do not enter data in this field

Thank you.

Your request was successfully sent!

We will get in touch with you soon.

Your Rohde & Schwarz Team

Request information

Inquiry*

Please enter your inquiry.

First Name*

Please enter your first name.

Last Name*

Please enter your name.

Email*

Please enter a valid email address.

Company*

Please enter your company name.

City*

Please enter your city name.

ZIP*

Please enter your postal code.

Location*

Please enter your location.

Marketing permission

I want to receive information from Rohde & Schwarz via

Email

Post

What does this mean in detail?

I agree that Rohde & Schwarz GmbH & Co. KG and the Rohde & Schwarz entity or subsidiary company mentioned in the imprint of this website, may contact me via the chosen channel (email or postal mail) for marketing and advertising purposes (e.g. information on special offers and discount promotions) related to, but not limited to, products and solutions in the fields of test and measurement, secure communications, monitoring and network testing, broadcast and media, and cybersecurity.

Your rights

This declaration of consent may be withdrawn at any time by sending an email with the subject "Unsubscribe" to news@rohde-schwarz.com. Additionally, a link to unsubscribe from future email advertisements is contained in each email sent. Further details on the use of personal data and the withdrawal procedure are set out in the Statement of Privacy.