Product Info

High-speed mixer and thermal mixer for small and very small volumes in microplates and reaction tubes

The BioShake series allows for the first time high precise and efficient mixing in the microliter scale for a wide range of applications. Assays in microplates or reaction vessels can be realized fast and safe with using adjustable speed of 200 up to 3,000 rpm. The BioShake mixing-technology is obviously more robust, vibration free and needs less maintenance compared to classical mixers.

Programming the BioShake XP and BioShake iQ works via direct touch buttons. In addition two buttons for start and storage of time and mixing modes enable the instrument to run complex applications. This opens new points of view on the daily laboratory work and optimizes routine application enormously.

The short mix button allows short and fast mixing in between.

The two line LCD display guarantees simultaneous and safe reading of all programmed and measured parameters as time, mixing frequency and for BioShake iQ additional temperature.

The BioShake iQ is the high end thermal mixer of the BioShake series. In addition to the technical specifications of the model BioShake XP, the BioShake iQ comes with an very accurate heating technology. This allows highly reproducible results. The temperature range from RT to 99 °C is adjustable in 0.1 °C steps.

The temperature accuracy is ±0.1 °C, the temperature uniformity through all samples is better than ±1 °C.

The BioShake series is characterized by minor space requirement.

The BioShake series comes with a variety of standardized and specific adapter plates. The adapters allow an optimal fit for standard tubes, lysis tubes, microplates, glass vials and other sample vessels. An excellent temperature uniformity and homogeneity is guaranteed by using these adapters.

Customized adapters!You need an adapter plate for specially shaped microplates, tubes or vials?Please send us a sample and detailed information about manufacturer,description and article number.You will receive your special formed adapter plate!

Technical Data

Removable Adapter Plates

BioShake XP

BioShake iQ

Microplates

96-, 384- and 1536 well microplates, deep well plates and PCR plates

96-, 384- and 1536 well microplates, deep well plates and PCR plates

Tubes

0.2 up to 2.0 ml standard and lysis tubes (with skirt)

5 ml conical tubes

15 ml und 50 ml Falcon Tubes

0.2 up to 2.0 ml standard and lysis tubes (with skirt)

5 ml conical tubes

15 ml und 50 ml Falcon Tubes

Glass vials

2.0 and 4.0 ml glass vials

2.0 and 4.0 ml glass vials

Others

On request

On request

Tempering function

Temperature regulation range

-

Ambient to 99 °C

Temperature setting

-

0,1 °C increments, adjustable from 0 °C to 99 °C

Temperature regulation accuracy

-

± 0,1 °C

Temperature uniformity

-

±0.5 °C up to 45 °C±0.7 °C from 45°C to 75 °C±1.0 °C from 75°C to 95 °C