The DRG Series signal conditioner modules accept a wide variety of input signals such as
thermocouples, RTDs, strain gages, DC voltages/currents, AC voltages/currents, frequency and
potentiometers and produce a proportional conditioned process output.
The inputs and outputs are both field configurable and offer flexible wide ranging capability.
The slim housing mounts on a DIN rail and is ideal for high density installation.
All modules provide 1800 Vdc isolation between the input, output and power supply.
One advantage of the DRG series is the field configurable input and output ranges.
Each module can be set to a number of ranges by dip switch selection.
Wide ranging precision zero and span potentiometers provide even further adjustment.
The signal conditioners may be set for an almost limitless number of ranges.
Range adjustment requires the use of a calibrator or reference source.

DRG-SC-TC Thermocouple Signal Conditioner
The DRG-SC-TC is a DIN rail mount thermocouple
input signal conditioner. It can be field configured for
over 60 different thermocouple temperature ranges.
The output is linear to temperature and can be set for
either 0-5 V, 0-10 V, 0-1 mA, 0-20 mA or 4-20 mA. Zero
and span pots allow 50% adjustability of offset and
span turn down within each of the ranges. For example
the 500-1000°C range could be offset and turned down
to provide a 4-20 mA signal representing 750-1000°C.

The DRG-SC-DC DC Input Signal Conditioner
The DRG-SC-DC is a DIN rail mount DC voltage and
current input signal conditioning module. The input
can be field configured for any one 12 voltage ranges
from 10 mV to 100 V or 6 current ranges from 1 mA to
100 mA. The output is linear to the input and can be
set to either 0-5 V, 0-10 V, 0-1 mA, 0-20 mA or 4-
20 mA for the DRG-SC-DC-U (unipolar outputs) and
-5 V to +5 V or -10 V to +10 V for the DRG-SC-DC-B
(bipolar outputs). Zero and span pots allow 50%
adjustability of offset and span turn down within each
of the ranges. For example the 0-2 mA input range
could be turned down to 0-1 mA and provide a full
scale output signal (e.g. 4-20 mA).Input Ranges (UNIPOLAR AND BIPOLAR)
Voltages: 20 mV, 50 mV, 100 mV, 200 mV 500 mV,
1 V, 2 V, 5 V, 10 V, 25 V, 50 V, 100 V
Current: 2 mA, 5 mA, 10 mA, 20 mA, 50 mA,
100 mA

DRG-SC-RTD RTD Input Signal Conditioner
The DRG-SC-RTD is a DIN rail mount RTD input signal
conditioning module. It accepts a wide variety of RTDs
including 100, 500 and 1000 Ohm Platinum RTDs as
well as 10, 25 and 100 Ohm copper. It works with RTDs
with α = 0.00385Ω/Ω/°C or 0.00392Ω/Ω/°C
The input can be field configured for any one of up to
sixteen temperature ranges. The output is linear to
temperature and can be set for either 0-5 V, 0-10 V,
0-1 mA, 0-20 mA or 4-20 mA.Input Ranges (°C):
PT100, PT500 & PT1000: -200 to 600, -200 to 400,
-100 to 400, -200 to 260, -200 to 0, -200 to -100,
-100 to 260, -100 to 100, -50 to 50, -18 to 50,
-18 to 100, -18 to 260, -18 to 300, -18 to 400,
-18 to 500, -18 to 600
Cu10, Cu25 & Cu100: -200 to 260,-200 to 0,
-200 to -100, -100 to 260, -100 to 100, -50 to 50,
-18 to 50, -18 to 100, -18 to 260

DRG-SC-FR Frequency Input
Signal Conditioner
The DRG-SC-FR is a DIN rail mount frequency input
signal conditioning module. The field configurable input
and output offers flexible, wide ranging capability for a
variable frequency drives, magnetic pickups, turbine
meters and other pulse or frequency output
transducers. The output can be set for either 0-5 V,
0-10 V, 0-1 mA, 0-20 mA or 4-20 mA. The DRG-SC-FR
can be configured for virtually any frequency input to
DC signal output within the ranges specified.
Calibration utilizes technology where the user simply
applies minimum and maximum input frequencies,
touching a recessed button to configure the minimum
and maximum output range.Input Range: 2 Hz to 10,000 Hz, 50 mVp to 150 Vrms

DRG-SC-PT Potentiometer Input Signal Conditioner
The DRG-SC-PT is a DIN rail mount potentiometer
input signal conditioning module.
The input provides a constant voltage and is designed
to accept any three-wire potentiometer from 100 . to
100 K.. The field configurable output can be set for
either 0-5 V, 0-10 V, 0-1 mA, 0-20 mA or 4-20 mA.
Wide ranging, precision zero and span pots, used in
conjunction with DIP switches, allow 80% adjustability
of offset and gain to transmit a full scale output from
any 20% portion of the potentiometer input.Input Range:
100Ω to 100 KΩ

Place Order(Specify Model Number)

What Other People Bought: When you see this icon, click on it to expand a list of products that other people have bought when they purchased this model.

Most Popular Models!

Show Only Stocked Items

Part Number/ Desc.

Qty

RoHS

Description

RoHS

The part number is RoHS Compliant.

The part number is NOT RoHS Compliant.

RoHS Compliancy of the part number is undetermined.

May be ordered as RoHS compliant, however, that requirement must be specified at the time of ordering by adding "-rohs" to the end of the part number.