Abstract

Purpose:

IGRT has become an indispensable tool in modern radiotherapy with kV imaging used in many departments due to superior image quality and lower dose when compared to MV imaging. Many departments use manufacturer supplied protocols for imaging which are not always optimised between image quality and radiation dose (ALARA).

Methods:

Whole body phantom PBU-50 (Kyoto Kagaku ltd., Japan) for imaging in radiology has been imaged on Varian iX accelerator (Varian Medical Systems, USA) with OBI 1.5 system. Manufacturer's default protocols were adapted by modifying kV and mAs values when imaging different anatomical regions of the phantom (head, thorax, abdomen, pelvis, extremities). Images with different settings were independently reviewed by two persons and their suitability for IGRT set-up correction protocols were evaluated. The suitable images with the lowest mAs were then selected. The entrance surface dose (ESD) for manufacturer's default protocols and modified protocols were measured with RTI Black Piranha (RTI Group, Sweden) and compared. Image quality was also measured with kVQC phantom (Standard Imaging, USA) for different protocols. The modified protocols have been applied for clinical work.

Results:

For most cases optimized protocols reduced the ESD on average by a factor of 3(range 0.9–8.5). Further reduction in ESD has been observed by applying bow-tie filter designed for CBCT. The largest reduction in dose (12.2 times) was observed for Thorax lateral protocol. The dose was slightly increased (by 10%) for large pelvis AP protocol.

Conclusion:

Manufacturer's default IGRT protocols could be optimised to reduce the ESD to the patient without losing the necessary image quality for patient set-up correction. For patient set-up with planar kV imaging the bony anatomy is mostly used and optimization should focus on this aspect. Therefore, the current approach with anthropomorphic phantom is more advantageous in optimization over standard kV quality control phantoms and SNR metrics.