This wide-bandwidth oscilloscope kit is capable of measuring frequencies up to 350MHz sampled at up to 1GS/s in real time. The accessories are selected to allow measurements with input impedance of 10MΩ as well as measurements through 50Ω coaxial cable.

Using the fastest random interleaved sampling (RIS) mode (50GS/s), you can set the timebase as fast as 125ps/div.

Wide Bandwidth Oscilloscope Kit Features

M595 Oscilloscope

The innovative design of the M595 oscilloscope is indeed extraordinary in this category. It offers an extremely fast real-time sampling rate (1GS/s for single channel, 500MS/s for both channels). Thanks to the precise construction, the input amplifiers offer up to 350MHz bandwidth. The device is powered from a wall adapter, which is supplied with removable plugs for Europe, USA, UK, and Australia.

The device has a metal casing which can withstand rough manipulation and is fully recyclable.

The M595 offers precise electronic engineering and automated calibration processes during manufacturing. Each kit contains a traceable calibration certificate. The inputs withstand ±200V, which is quite rare in this category.

Since the M595 device uses USB 2.0 for communication (with automatic fallback to USB 1.1), it provides you with high refresh rates comparable to stand-alone devices.

Software

The basic package contains complex yet easy-to-use oscilloscope software (Scope). Scope allows you to control the oscilloscope, use advanced functions (such as FFT, persistent display mode, mask testing) or export the data. The Spectrum Analyzer software is the part of the basic package as well (with top frequency of 350MHz).

This kit also includes a license for Roll Mode software, which allows you to acquire data as fast as 5MS/s and store them directly to the hard drive. You can use either the viewer to analyze the data, or export tools to analyze data in third-party applications.

Channel A and Channel B on the whole display range. External fixed on about 1.5V

Slope selection

Leading or trailing edge independently on each source

Minimum trigger pulse period

3 ns

Minimum trigger pulse length

1.5 ns

Maximum voltage on external trigger input

–10V to +13V at 20 kHz or less

Adjustments

Digital filter with valid pulse length settable up to 32768*Ts for each level and counter of valid triggering events settable from 1 to 32767 for each level. HOLD-OFF settable up to 32768*Ts with selectable AUTO mode, to sample proper amount of data before trigger. (Ts – actual real-time sampling period.)

Data acquisition system

No. of divisions

10

No. of pixels per division

50

Mode of operation

Sampling before and after trigger with continual selection of the trigger position

Record length

Settable from 1024 to 32768 (32k) samples for each channel

Time base range in 1:1 mode

1 ns/d to 20 ms/d in 1-2-5 sequence

Time base range using different ZOOM modes

125 ps/d to 1.28 s/d

Time base accuracy

0.01% to 20ns/d, 0.5% for 10ns/d to 1ns/d

Real-time sampling frequency

2.5 kHz to 1 GHz single channel; 2.5 kHz to 500 MHz dual channel

Equivalent sampling frequency

2.5 kHz to 50 GHz

Display range with respect to trigger event

32768 samples before and 32768 samples after trig. event in length of 32768 samples

Wide Bandwidth Oscilloscope Kit Documentation

Optional Development API

The Development Kit is an API allowing you to add control functions for the scope in your own software applications. It works in Windows 98SE, Me, 2000, XP and Vista. It comes with descriptions and examples of all libraries written in Inprise Delphi, Inprise C++ Builder, Visual C++ 6.0, Visual Basic 6.0, C# .NET and Visual Basic .NET.