CustomAir Ramvac Compressor Model 1029T30

Theory of Operation: When power is applied, fresh air is taken in through air filters and compressed by the pump pistons. The compressed air is very warm and is cooled for use by the heat exchanger and fan. Air is then delivered to coalescing filter to remove most of the moisture. The coalescing filter is equipped with a automatic draining feature that allows the draining of the excess moisture. The cool air enters the desiccant style dryer assembly to complete the moisture removal by removing 99.7% of the water vapor from the air. Air is directed into the dryer cartridges, from one to the other, by electronically controlled solenoid valves. The timing for the solenoid valves is provided by the C1 Control. While the air is moving through one cartridge, the desiccant in the other cartridge is being dried by air directed through an orifice in the outlet shuttle valve. This valve cycle will be noticeable by a short burst of air from the purge valves. When the air has left the dryer assembly it goes through a particulate filter to remove any remaining particulates from the air. It is then collected in the tank where a pressure gauge displays the tank air pressure. The tank pressure is monitored by a pressure transducer in the C1 Control, which runs the motors on demand. Pressure is factory set to start the compressor at 80 psi and shut off at 100 psi. Pressure settings are not adjustable.

MAX

USERS

MOTOR

PER UNIT

HP

PER MOTOR

VOLTAGE

BREAKER

SIZE

POWER

TANK

Db

LEVELS

DIM

WEIGHT

9

3

1

230V

20 AMP

15.4 AMPS

30 GALLONS

78

38" X

24.5" X 29.5"

342 LBS

Ships Freight on Wooden Pallet

* IMAGE IS OF ACTUAL UNIT. ONCE FULLY REFURBISHED, UNIT MAY HAVE SOME VARIATIONS, SUCH AS COLOR, TUBING ETC