The DBB series of loadcells are an S beam, bi-directional load cell suitable for tension and compression applications. Loadcells are available with a bi-directional range of 50Kgf up to 10,000Kgf with a standard 2mV/V output and accuracy of 0.03%. The smaller DBBE is an encapsulated cell sealed to IP67, the DBBW is a fully welded cell sealed to IP68 and available as an ATEX certified loadcell, whilst the DBBF has a range of 10,000Kg. Units can be supplied with suitable inline amplifiers to provide a calibrated 4-20mA output.

The DDE series of load cells are an inline bi-directional load cell suitable for tension and compression applications with restricted mounting space. The DDE's small size and reduced weight, along with its low deflection results in a high stiffness assembly, makes this load cell suitable for rapid measurement changes and high frequency transients. Load cells are available with a bi-directional range of 100N up to 50kN with a standard 2mV/V output and accuracy of 0.3%. Units can be supplied with suitable inline amplifiers to provide a calibrated 4-20mA output.

The CCDDT load cells are dome top column load cells designed for the measurement of compressive forces. The CCDDT can be used for applications including hydraulic press calibration, pile force measurement, jack force measurement, centre of gravity measurement systems and structural proof testing. Ranges 2 to 1,000 tonnes. Standard output is 1.5mV/V with an accuracy of ±0.25% and options for inline amplifiers for standard 4-20mA or voltage outputs.

Price range : £0.00 - £0.00

CBES Low Profile - Compression Load Cell

The CBES load cell is a low profile load cell designed for the measurement of compressive forces. The CBES is equally suited for both force measurement and weighing applications alike. The compact design allows the cell to be fitted into the smallest of spaces can be retrofitted in weighing systems, the low height gives an added benefit in that it does not increase the headroom required for vessels and is simple to install. Ranges 50 to 50,000Kg. Standard output is 2mV/V with an accuracy of ±0.15% and options for inline amplifiers for standard 4-20mA or voltage outputs.