PLF SERIES

PLF 160/5
PLF series have been used as trusted professional
chamber furnaces for many years in laboratories.
Available with brick and fiber insulation elements,
with a wide variety of options, these models can be
optimally used for your general procedures.

Standard PLF series furnaces cover a range from
1400°C to 1600°C, all of which have front loading for
easy operation and double skin construction to
maintain a cooler outer case.

Important advantage of the model is that element
change is very easy and economic. Furnaces starting
from 1400°C use Silicon Carbide heating elements,
providing a very long service life.

The durability of the SiC rods in periodic use, in
combination with their high heating speed, make
these furnaces to all-rounders in the laboratory.
Typical applications for the product is melting,
thermal ageing, ceramics sintering, metal heat
treatment, chemical decomposition and thermal
shock testing.

High quality heating elements ensuring a long service life System operation with solid-state-relays Electrical protection Working Temperatures of up to 1600°C Dual skin housing for low external temperatures and high inner temperature stability Easy replacement of heating elements Intuitive controller user interface

Specifications

Model

Maximum Temperature (°C)

Continuous Operating Temperature (°C)

Volume (L)

Inside Measurements (HxWxD) (cm)

Outside Measurements (HxWxD) (cm)

Power (kW)

Phase

PLF 140/5

1400

1350

5.25

14x15x25

65x55x58

3.3

3

PLF 140/9

1400

1350

9.0

18x20x25

72x56x64

4.4

3

PLF 140/15

1400

1350

15.0

22x22x31

79x59x69

5.8

3

PLF 140/30

1400

1350

29.7

27x25x44

80x66x73

11.5

3

PLF 150/5

1500

1450

5.25

14x15x25

65x55x58

4.9

3

PLF 150/9

1500

1450

9.0

18x20x25

72x56x64

6.5

3

PLF 150/15

1500

1450

15.0

22x22x31

79x59x69

5.8

3

PLF 150/30

1500

1450

29.7

27x25x44

80x66x73

11.5

3

PLF 160/5

1600

1550

5.0

14x15x25

65x55x58

5.2

3

PLF 160/9

1600

1550

8.6

18x20x25

72x56x64

5.9

3

PLF 160/15

1600

1550

15.0

22x22x31

79x59x69

7.0

3

PLF 160/30

1600

1550

29.7

27x25x44

80x66x73

11.5

3

Model

Maximum Temperature (°C)

Continuous Operating Temperature (°C)

Volume (L)

PLF 140/5

1400

1350

5.25

PLF 140/9

1400

1350

9.0

PLF 140/15

1400

1350

15.0

PLF 140/30

1400

1350

29.7

PLF 150/5

1500

1450

5.25

PLF 150/9

1500

1450

9.0

PLF 150/15

1500

1450

15.0

PLF 150/30

1500

1450

29.7

PLF 160/5

1600

1550

5.0

PLF 160/9

1600

1550

8.6

PLF 160/15

1600

1550

15.0

PLF 160/30

1600

1550

29.7

Model

Inside Measurements (HxWxD) (cm)

Outside Measurements (HxWxD) (cm)

Power (kW)

Phase

PLF 140/5

14x15x25

65x55x58

3.3

3

PLF 140/9

18x20x25

72x56x64

4.4

3

PLF 140/15

22x22x31

79x59x69

5.8

3

PLF 140/30

27x25x44

80x66x73

11.5

3

PLF 150/5

14x15x25

65x55x58

4.9

3

PLF 150/9

18x20x25

72x56x64

6.5

3

PLF 150/15

22x22x31

79x59x69

5.8

3

PLF 150/30

27x25x44

80x66x73

11.5

3

PLF 160/5

14x15x25

65x55x58

5.2

3

PLF 160/9

18x20x25

72x56x64

5.9

3

PLF 160/15

22x22x31

79x59x69

7.0

3

PLF 160/30

27x25x44

80x66x73

11.5

3

Optional Features

Over-temperature limiter for thermal protection Adjustable air intake opening in the furnace door Exhaust air opening in the of the furnace roof Option of sideways opening door Observation hole enabling the operator observe the load during the operation Data logger with the software

RS422/485 communication Cooling Fan for faster cooling processes Jet Fan for faster exhaust of gases Movable bottom enabling the operator to remove the sample while the furnace is still hot Quartz element protection for heating elements Protective gas system and connection