The applications for ThermaLab™-System are not limited to aqueous systems. It can be used in organic liquids as well. The use in solid- and gas-phases are not restricted, nor they are tested. CAUTION!!! Heated electrodes can get very hot. At ambient air, we observed a glow at platinum wires, while heating electrodes.

The ThermaLab-System can perform a big variety of voltammetric analysis, as well as techniques as functions of time, such as open circuit potentiometry and amperometric detection. More information can be found here. In case of any question, please contact us.

As in every other electrochemical measurement it depends on the special applications. For ThermaLab-System itself it is not necessarily to use vented solutions. For more information, please contact us.

We have been partner in a WK-Potential-Project, and are still partner in the HiCare-Project and the OxIOSCR-Project. Furthermore we have our own SME-Project and we are going to be partners in different upcoming H2020-Projects.

The working electrodes are designed and tested to have an surface area up to 2 cm2. Nevertheless we think that there's an opportunity to achieve a bigger area, as long as the electrodes keeps a wired-design. Please contact us.

The temperature-precision depends of the used electrode and the system-settings. The resolution can be set from <0.1°C (@ ΔT about 10°C) up to 10°C (@ ΔT about 600°C) with an accuracy of about 10% of the temperature-range.

Yes, you can use heated electrodes in gas-phase-reaction. But be careful. Our electrodes are not designed and thought to be used in gas-phase-reactions. We observed a glow while heating platinum-electrodes at air.

Yes, you can use every electrode you want, even this electrode fits to the connector. For heating-purposes our system requires an working-electrode with two connectors. For more details, please contact us.