TEVLAR

Heating and moistening of the respiratory mixture coming to the patient from the artificial lung ventilation apparatus (ALVA), inhalation anesthesia (IA) device or oxygen inhaler.

Advantages

Automatic maintenance of the temperature and humidity parameters of the respiratory mixture selected by the doctor at the outlet of the supply hose (in the patient) in a wide range of minute ventilation

Autonomous channels for controlling the temperature and moisture content of the respiratory mixture

Indication on the digital display of the set and measured temperature at the outlet of the supply hose

Availability of a spiral duct made of a porous metal in the evaporation chamber (tank) reduces the degree of water heating (below 65 °С), increases the reliability and convenience of operation

Signaling about the overheating of the respiratory mixture and disturbances in operation

Simple maintenance, small size and weight make it possible to effectively apply this model together with any lung ventilation devices, IA and oxygen inhalers

Delivery set

№

Name

Amount, pcs.

1.

Breathing mixtures humidifier TEVLAR-“UOMZ”, including:

1

1.1

Control unit

1

1.2

Reusable chamber for adults

1

1.3

Sensor

1

1.4

Reusable chamber for infants and neonates

1

1.5

Neonatal circuit (Ø10mm)*

1

1.6

Pediatric circuit (Ø 16mm)*

1

1.7

Y-shaped connector Коннектор (15F)*

1

1.8

Straight connector (10/15mm)*

2

1.9

Heating wire adapter *

1

1.10

Straight connector with temperature port (10/15mm)*

1

1.11

Straight connector with temperature port (15/22mm)*

1

1.12

Cable fixator (15mm)*

3

1.13

Cable fixator (10mm)*

4

1.14

Bracket

1

2.

Spares

2.1

Sensor

1

3.

Documentation

Specifications

Overall dimensions (width ∙ length ∙ height)

145х175х195 mm

Weight

2.7 kg

Tank connection holes

22 mm

The maximum volume of water poured into the tank (included)

250 ml

Supply hose outlet temperature range

from 25 °С to 45 °С

Relative humidity

Either 75%;Or 100 %

Range of respiratory mixture flow

from 2 to 50 l/min

The duration of reaching a temperature of 33 °C and a relative humidity of 85% at the outlet of the supply hose (at an ambient temperature of at least 20 °C and a flow of the respiratory mixture to 30 l/min)