Available with and without high/low alarm settings. No recorder output, no sample pump.

The only one that can be used with either polarographic or galvanic sensors

Oxygen Analyzer with or without Hi-Lo Alarms

The 102 series analyzers are designed for use as a continuous monitor on anesthesia machines, ventilators and incubators. The instruments feature a large digital LCD display, low battery indicator, sensor disconnect indicator, automatic temperature compensation and portable, battery operation using a standard 9-volt transistor radio battery.

They are the only monitors that can use both the standard supplied polarographic sensor or optionally, galvanic sensor number 100580 with 102111 adapter cable. This permits the user to standardize and reduce the number of sensors in his inventory.

The compact size and economical price of the model 102500 makes this the ideal spot check analyzer for use in the hospital or homecare environment. It has no high - low alarms, but is otherwise identical to model 102000.

With or without high-low alarms

Ideal for checking anesthesia machines,

ventilators, oxygen concentrators

Uses always available 9V battery

Large, easy to read digital display

Oxygen Analyzer Specifications

Parameter

Specification

Range

0-100% O2

Display

LCD, 0.1% O2 increments

Linearity

1% of full scale

Accuracy

2% of full scale

Response Time

20 seconds to 90%

@ 2 LPM in TEE

_______

Operating Temperature Range

15 to 40 degrees C

Storage Temperature Range

0 to 50 degrees C

Humidity

0-85% RH

Warm-Up Time

Not required

Low Battery Indicator

Yes, will appear on display

Power Requirements

One 9-volt alkaline battery

Battery Life

Approximately 2000 hours

Battery Replacement Number

.03003

Dimensions

5-5/8" x 3-1/4" x1-1/2" inches

(Height x Width x Depth)

144 x 83 x 40 mm

Weight

10 oz (285g)

Sensor type

Galvanic or polarographic

Sensor Internal/External

External with hose fitting

Sensor Life

Expected life of one year in normal medical conditions

Shelf Life

Minimum 3 months w/o degradation of life

Alarms (Model
102000)

Adjustable High
& Low(Visual,
LED & audio)

Alarms (Model
102500)

None

Recorder Output

Not available

Cat No

Description

102000

Oxygen
analyzer/Monitor with alarms complete with disposable sensor and cable

102500

Oxygen Analyzer/Monitor without alarms but complete with disposable sensor and cable