Main features
Flashpoint according to ASTM D6450/SHT0768 (Flash-Point by Continuously Closed Cup) and ASTM D7094 (Flash-Point by Modified Continuously Closed Cup)
FLASHPOINT COMMITTEE STATEMENT: "There is no statistically significant bias observed between ASTM D7094 and ASTM D93 Procedure A."), equivalent EN ISO 2719, excellent correlation to ASTM D56, ISO13736, IP170.
MINIFLASH TOUCH is a uniquely designed flash point tester for the determination of flashpoints of liquids and solids, using the Grabner flash point detection method of measuring the instantaneous pressure increase inside the continuously closed chamber, triggered by an energy-controlled electric arc. Heating the test chamber from the top avoids condensation of high volatile compounds, significantly improving the test results.
Its continuously closed cup design and small sample volume of only 1-2 mL offers maximum safety, avoids offensive fumes in the testing area and reduces sample waste.
The fast thermoelectric cooling at the end of a test and the fast and easy handling of the instrument save expensive labor time.
Furthermore, test results are equivalent to other well known standards such as Abel and TAG, and thanks to its unique design MINIFLASH can be used for more difficult applications such as flash-point testing of contaminated samples like:
- diesel fuel in lube oil or gasoline
- diesel fuel
- biodiesel testing
The new MINIFLASH TOUCH analyzer also includes improved methods for simulation of EN ISO 3679 and EN ISO 3680.

Main features
MINIFLASH is a uniquely designed flash point series of instruments for the automatic determination of the flash point of liquids and solid samples according to ASTM D6450 and the advanced ASTM D7094 standard. This procedure describes the “Standard Test Method for Flash Point by Modified Continuously Closed Cup (MCCCFP)”, which offers the highest repeatability and reproducibility and an excellent correlation to the D93 Pensky-Martens method: "There is no statistically significant bias observed between ASTM D7094 and ASTM D93 Procedure A” as concluded during the last Round Robin. ASTM D7094 has been approved by the ASTM committee for the Diesel Fuel and Kerosene specifications ASTM D975, D396, D2880 and D3699.
Furthermore, test results are equivalent to other well known standards such as Abel and TAG, and thanks to its unique design, MINIFLASH can be used for more difficult applications such as flash-point testing of contaminated samples like:
- diesel fuel in lube oil or gasoline
- diesel fuel
- biodiesel testing
Its continuously closed cup design and small sample volume of only 1-2 mL offers maximum safety, avoids offensive fumes in the testing area and reduces sample waste.
The fast thermoelectric cooling at the end of a test and the fast and easy handling of the instrument, save expensive labor time.

Main features
The MINIFLASH is a uniquely designed flash point tester for the determination of flashpoints of liquids and solids, using the GRABNER INSTRUMENTS flash point detection method of measuring the instantaneous pressure increase inside the continuously closed chamber due to a hot flame. The MINIFLASH flashpoint analyzer series is a new approach to flash point testing.
It revolutionizes traditional methods, when it comes to safety, sample volume, speed and instrument size
Furthermore, test results are equivalent to other well known standards such as Abel and TAG, and thanks to its unique design MINIFLASH can be used for more difficult applications such as flash-point testing of contaminated samples like:
- diesel fuel in lube oil or gasoline
- in diesel fuel
- biodiesel testing