Purchase

Quantity

The selected item is out of stock, please select a different item or combination

Description / Spec

Model SCO5W water jacketed CO2 incubator and is ideal for cell culture applications. It provides a precisely controlled, accurate, pure environment throughout the culturing process. Shel Lab CO2 incubators are designed to provide the uniform temperatures and highly stable CO2 environments that are necessary to prevent shocks to cells. Made in the USA.

Heated door to ensure superior temperature uniformity and also reduce the potential for condensation to form on the inner glass door

Durable stainless steel chamber with coved corners for easy cleaning

Exclusive heated copper CO2 inlet not only promotes temperature uniformity, but also prevents moisture from being introduced into the chamber

Copper shelving option

Autoclavable shelving system

5 year warranty parts and labor in the USA, 7 years on the IR sensor

Specifications

Weight

310.00 lbs

Access Port

1.5 inches

Shelving

3 Supplied (16 Maximum)

Relative Humidity Range

up to 95%

Jacket Type

Water

CO2 Recovery Rate

< 5 minutes

CO2 Sensor

IR

CO2 Range

0-20%

Temperature Uniformity

+/-0.2°C at 37°C

Temperature Range

Ambient +8°C to 60°C

Exterior Dimensions WxDxH in. (cm)

26 x 25.5 x 40.3 (66.1 x 64.8 x 102.3)

Interior Dimensions WxDxH in. (cm)

19.2 x 19.7 x 23 (48.9 x 50.1 x 58.4)

Capacity

5 Cu.Ft. (143 L)

Overtemperature Alarm

Yes

CO2 Alarm

Yes

Inner Glass Viewing Door

Yes

Nominal Voltage 50/60HZ (V)

110-120/220-240

Maximum Energy Consumption (W)

600/840

Humidity Display

Yes

Data Logging Capabilities

Yes

All specifications are determined by using average values on standard equipment at an ambient temperature of 25°C (77°F) and line voltages within +/-10% of unit type. Temperature specifications follow DIN 12880 methodology. We reserve the right to change specifications at any time.