Specifications

With the input sensitivity set to 10 mV/div and above, the 454A has a vertical bandwidth of 150 MHz.
At 5 mV/div it is 100 MHz, and at 2 mV/div it is 50 MHz.

It is possible to cascade the channel-1 and channel-2 input amplifiers to obtain a sensitivity of 400 μV/div.
In that configuration, the vertical bandwidth is 33 MHz.

The 454 weighs 29 pounds and consumes about 130 watts. please add

Internals

The 454 has a 140 ns delay line in the signal path after the
input channel switching and before the main vertical amplifier.
There is a phase equalizer after the delay line to mitigate the
harm done to pulse fidelity by the delay line.