Manufacture: FlukeProduct Description:The compact ScopeMeter 120 Series is the rugged solution for industrial troubleshooting and installation applications. Its a truly integrated test tool, with oscilloscope, multimeter and paperless recorder in one affordable, easy-to-use instrument. Find fast answers to problems in machinery, instrumentation, control and power systems.In todays complex systems, a meter measurement just doesnt give enough detail to determine the cause of a fault. Signal anomalies, dropouts and glitches that might cause a machine to go down, are best displayed with an oscilloscope. The ScopeMeter 120 Series meet todays need of simultaneously measuring and checking waveforms. The unique Connect-and-View triggering automatically displays stable waveforms of virtually any signal. It really is as easy as one-two-three!ScopeMeter 120 Series combines a 40 or 20 MHz dual input digital storage oscillo-scope, two true-RMS digital multimeters and a dual input TrendPlot recorder all in a compact, battery powered instrument. Leave all other test tools behind, the ScopeMeter 120 Series is the only tool youll need.High frequency waveform-, meter-, capacitance- and resistance-measurements as well as continuity checks are all covered by the shielded test leads. No time wasted finding or swapping leads. The included accessories allow hook-up at test objects of every dimension.

10:1 Voltage Probe included with Fluke 124 and 125 for high frequency measurements

Up to 7 hours battery operation

600 V CAT III safety certified

Optically-isolated interface for PC connection

Rugged, compact case

Fluke 125 gives bus health and power measurements

Power Measurements for single phase and balanced 3-phase systems. The Fluke 125 can directly present you the Total Power (Watts), Apparent Power (VA), Reactive Power (VAR) and the Power Factor (PF), over a wide range of applied frequencies, including those seen with motor drives and inverters. As a result, you are able to easily see the effects on the various power measurements during start-up or under changing operational conditions. A current clamp is included as a standard.

Harmonics mode graphically displays harmonics up to the 33rd harmonic to assist in fault-finding, e.g. with large non-linear loads.

RPM and Hz reading for use with electrical and combustion engines.

Vac pwm for use on motor drive outputs, reading the true output voltage experienced by the motor itself.