Special order item. Please use the Send Request button on the right for pricing and availability.

Declaration:

This product is one of the base models, however most items can be customized to your needs (voltage, ranges, sizes, additional documents, etc.). Please Contact Us if any specific requirements are needed.

Description

Series DA-7035N Temperature Switches include the same time-proven switching mechanism used in our Series DA Pressure Switches. Bourdon tube assures high sensitivity and long life. Fully adjustable deadband makes units suitable for a wide range of control applications. Visible, calibrated dial and external adjustments make changing set points simple and fast.

Features

Adjustable deadband

Snap action switch standard

Inert gas activated Bourdon tube

No cross ambient temperature effects

No bulb elevation correction required

Visible dial calibrated in both °F and °C

Visible on/off indication

Specification

Service

Compatible liquids or gases

Wetted Materials

Bulb and connection: 304 SS

Temperature Limit

Ambient: 180°F (82°C)

Pressure Limit

300 psi (20.6 bar)

Enclosure Rating

General purpose

Repeatability

±1% of full scale

Switch Type

SPDT snap switch

Electrical Rating

10A @ 120/240/480 VAC

Electrical Connections

Screw terminal

Conduit Connection

7/8" (22.23 mm) hole for 1/2" (12.7 mm) conduit hub

Process Connection

3/4" male NPT

Mounting Orientation

Vertical and level

Set Point Adjustment

External knobs for set point and reset point

Weight

5 lb (2.3 kg)

Deadband

Adjustable from minimum in model chart to full range

Capillary

6 ft (1.8 m) standard

Set Point Scale

Indication in °F and °C

Send an Inquiry

Please make sure to include your shipping address. This will help us estimate the shipping cost and provide a full quote.