The MSE display solution is a high-contrast LCD with simple operation and easy user guidance.The new Cubis is the first user-configurable and customizable laboratory balance series designed for users who always need the most advanced weighing technology, but who also want to invest only in what is necessary for the task at hand. more ...

Features

The MSE display solution is a high-contrast LCD with simple operation and easy user guidance.

The new Cubis is the first user-configurable and customizable laboratory balance series designed for users who always need the most advanced weighing technology, but who also want to invest only in what is necessary for the task at hand.

The Cubis series is a modular system that offers a choice of display and control units, weighing modules and draft shields

With Cubis, you never have to invest in completely new equipment. This is what makes the Cubis modular design so special, with a number of unique advantages:

Optimal interchangeability or extension of modules

Shorter innovation cycles for individual modules

Faster development of additional modules

A full range of accessories that enable you to customize your balance even further

All Cubis controls are easy to use, clearly designed and logically structured. Their user guidance features prevent time-consuming and cost-intensive operating errors.

In addition, you can adjust the tilting angle of the display and control unit to ensure optimal viewing at all times. All Cubis units come with standard internal motorized calibration

With Cubis theres a choice of three display and control units. They cover the entire range of laboratory applications from simple weighing right on up to management of complex work sequences by means of defined tasks and the User/ Password Management menu.

Specifications

Power Requirements: 100-240V~, -15%/+10%, 50-60Hz, 1,0A

Input Voltage: 15 Vdc, ± 5%

Ambient Temperature, Operation +5 °C...+41°F - +104°F

Highest Relative Humidity80% for temperatures up to 31°C/88° F, decreasing linearly up to 50% relative humidy for 40°C/104°F

Safety of electrical equipmentAccording to EN 61010-1:2001: Safety requirements for electrical equipment for measurement, control, and laboratory use - Part 1: General requirements

Electromagnetic CompatibilityAccording to EN 61326-1:2006: Safety requirements for electrical equipment for measurement, control, and laboratory use - Part 1: General requirements

Interference ResistanceSuitable for use in industrial areas

Immunity to Interference: Class B: Suitable for use in residential areas and areas that are connected to a low voltage network that also supply residential buildings.

Model:

MSE5203S-000-DE

MSE5203P-000-DE

MSE3203S-000-DE

MSE2203S-000-DE

MSE2203P-000-DE

MSE1203S-000-DE

MSE623S-000-DE

MSE623P-000-DE

MSE323S-000-DE

Capacity x Readability

5200g x 1mg

1200g x 1mg2400g x 2mg5200g x 5mg

3200g x 1mg

2200g x 1mg

1010g x 1 mg2200g x 10mg

1200g x 1mg

620g x 1mg

150g x 1mg300g x 2mg620g x 5mg

320g x 1 mg

Tare Range (Subtractive)

- 5200g

- 5200g

- 3200g

-2200 g

-1200 g

-620 g

-320 g

Resolution (mg)

1

Average Stabilization Time (s)

< 1 second

< 0.8 seconds

Average Response Time (s)

< 2 seconds

< 1.5 seconds

< 1 second

Repeatability (std. deviation)(mg)

< ± 1

< ± 1 < ± 6

< ± 0.7

< ± 1 < ± 2< ± 4

< ± 0.7

Linearity (±)

0.005g

0.005g

0.005g

0.003g

0.005g

0.002g

0.002g

0.005g

0.002g

Corner Load (mg)*(Test Load [g])

2 (2,000)

2 (2,000)

2 (2,000)

2 (2,000)

3 (1,000)

2 (500)

2 (200)

4 (200)

2 (200)

Minimum Sample Weight** (g)

1.5

Sensitivity Drift Within +10...+30°C/ +50°F

± ppm/K 1

± ppm/K 1.5

±: ppm/K 2

External Calibration Weight (of at least accuracy class...)(g)

5000 (E2)

3000 (E2)

2000 (E2)

1000 (E2)

500 (E2)

500 (F1)

200 (E2)

Display Update Rate (Depends on The Filter Level Selected)

0.1-0.4 seconds

Weighing Pan (W x D) [mm/in]

140 x 140/5.5 x 5.5

Weighing Chamber Height (DE Draft Shield) (mm/in)

172/6.8

Draft Shield

Manual analytical draft shield chamber, with smooth-action doors that open
wide and provide unimpeded access to the weighing chamber without any interfering frame braces. For all models with a readability of 0.01 mg, 0.1 mg or 1 mg.

The YDP20-0CE data printer is designed to receive
weighing data from a connected weighing instrument
and print it on plain paper (roll width:
57.5 mm), with up to 24 characters per line.
The data printer is equipped with a real-time clock.
After the printer has been configured according to
the required settings, it is suitable for use as auxiliary
apparatus in legal metrology.

The YDP20-0CE data printer meets the quality standards
of the chemical and pharmaceutical industries, in
compliance with the requirements of GLP (Good Laboratory
Practice) and GMP (Good Manufacturing Practice).
In addition to recording individual weighing data, the
data printer can generate a statistical evaluation of
weighing data. All values transferred from an on-line
weighing instrument are stored. The following can be
printed out, depending on your printer configuration:
– Number of weighing operations
– Mean value
– Standard deviation
– Variation coefficient
– Lowest measured value
– Highest measured value
– Difference between lowest and highest measured values
– Date and time

Data is transferred either automatically or manually, at
the press of a key.
The following can be computed/printed out:

Number of weighing operations

Mean value

Standard deviation

Variation coefficient

Minimum value/weight

Maximum value/weight

Difference between the maximum and minimum value/weight

Data are transferred either automatically or manually by pressing a key.
Built-in real-time clock
Built-in statistics program. These programs can be individually selected to meet your needs.
Counter for consecutive numbering of transactions can be activated.
High contrast LCD that shows the date or time as well as the current transaction number.
Integrated rechargeable batteries with battery level indicator.
Ultrafast print head.
Runs on the balances power, no separate power supply necessary.
Easy to change paper roll and ink ribbon cassette.

Sartorius YDP30 thermal transfer printer works with Practum, Quintix, Secura, MA37 and MA160 balances. When you plug the Sartorius YDP30 printer in one of these balances the Sartorius balance automatically detects the YDP30 printer allowing you to generate printouts of the results on standard printer paper or on continuous labels.

The Practum, Quintix, Secura, MA37 or MA160 balance will guide you through the steps for obtaining GLP-compliant printouts. The novel GLP user guidance features will support you, especially for working with traceable sample identification.