SHV-20 and SHV-B

SHV-20 Specifications for Grounded Coaxial

SHV-20 feedthroughs are rated for 20,000 Volts DC and up to 15 Amps. These specially designed interfaces incorporate standard high voltage contacts, and are designed for increased performance in pulsed high voltage applications. SHV-B feedthroughs are designed for bakeout at higher temperatures than the other SHV series. The air-side cable connectors included with these feedthroughs utilize a ceramic dielectric material. SHV-B feedthroughs and connectors are rated for 7500 Volts DC and up to 3 Amps.

Operating Conditions

The electrical and thermal ratings specified are safe operating limits determined by various factors including material properties, mechanical design, and the intended operating environment. All electrical ratings are based on operation with one side in dry atmosphere and the other side in vacuum of 1 x 10-4 Torr maximum pressure. Temperature ratings for various mounting options may reduce the operating range of an assembly. All assemblies have a maximum thermal gradient of 25oC per minute and may be damaged if this limit is exceeded.

SHV-20 Grounded

Single/Double Ended:

Single

Electrical Rating:

Voltage: 20000 Volts DC Current: 15 Amps

Thermal Rating:

Feedthrough: -200 to 450oC Air-side Connector: -65 to 165oC

Conductor:

Nickel

Impedance:

Non-constant

Get an Accurate Quote in 48 Hours for Hard-to-Find Ceramic to Metal Electrical Feedthroughs, Connectors, Isolators and UHV Viewports

Get a custom quote for your unique application. Built to your specifications, thoroughly tested and delivered when you need it.