PROBIUS UNO - Laser control equipment for the inside diameter and straightness of pipes

PROBIUS system measures different parameters of smooth pipes:

inner diameter;

ovality;

out-of-roundness;

non-straightness (warpage);

The system is able to measure pipes with inner diameters from 100mm. Pipe length is up to 10 m.

Custom-made solutions are available.

PARAMETER ACCURACY

Non-straightness

±0.005mm

Diameter/ovality measurement

±0.01mm

Probe positioning

±1mm

SPECIFICATION The system contains:

Adjustable laser source;

Self-moving wireless probe;

Target for laser beam adjustment;

Software;

Connection cables;

Battery charger;

Calibration block;

Manual.

OPERATION In order to prepare for the measurement, the laser source is to be placed at the end of the pipe. The target for laser beam adjustment is to be placed at the opposite end of the pipe. The user should aim the laser beam in the center of the target using special adjustment screws and then remove the target.

PROBIUS is to be inserted into the pipe. All the rest processes are handled automatically after pushing “Start” button. Status of measurement is displayed on the laptop screen. The final report can be printed or stored in a database or on a memory card.

The probe provides to carry out measurements over the entire surface of the pipe excluding the areas (at the beginning and at the end) which do not exceed 60. Using a special pipe extender (option) allow to measure all pipe surface.

The probe moves along the pipe automatically. The probe motion control and data transaction are wireless.

Periodic calibration of this instrument is required for which it is necessary to use a special block (included), imitating non-straightness and having two to five graduation rings for diameter (ovality) channel calibration.

Operation speed: 2 m per minute.

Total weight of the system: 10 kg.

Weight of the probe: 1.5 kg.

Other applications The device can be manufactured in accordance with the customer's individual specifications. It is also possible to implement additional channels for pipe wear and tear monitoring, pipe surface profile control, and a video channel.