I will be posting blog posts frequently below that will detail some of the customer questions that we have had, and hopefully they will help you as well. I look forward to hearing from you, and remember, just “Ask Robin!”

Certification is a crucial step to using a Flow Sciences enclosure. All FSI Units should be certified BEFORE operation. Below you will find the FSI Certification Guidelines, a blog from our Corporate Customer Advocate, Robin Williams, and other useful tips.

Large Glovebox Workstation for work with Anti-Cancer Drug Conjugate processing and other sensitive applications. The unit is made of polypropylene and has acrylic or glass glove port panels, a pass thru entry lock and a Bag-in/Bag-out HEPA System with integral fan. The unit is provided with an 18″ X 36″ inlet HEPA that provides the interior with an ISO 5 Cleanliness or better airflow. An LED interior lighting system provides an illuminated area ideal for precision work. Ideal for work with sensitive balances and related analytical equipment.

Size:
98″ Exterior Width
30″ Exterior Depth
31″ Interior Height

FSI Category: 7-8

Get a Quote / Info

Name*

Company*

Email*

Phone

State / Province / Territory*

Country*

Application / Process / Other Comments

GBWS Videos

GBWS Series

GBWS Features

FEATURES & BENEFITS

FEATURES & BENEFITS

SAFETY. The Glovebox Workstation series provides containment for highly toxic applications using APIs that need more safety than an opened face enclosure. FSI engineering controls are built in to prevent loss of containment. Third Party testing has proven containment on these units to below 50 ng/m³ based on process and quantity.

DESIGN. Designed as the next step in containment above the FSI Hybrid Isolator, this unit is able to effectively contain using HEPA clean air inlet and HEPA air out. This creates laminar airflow across the interior. The Glovebox Workstation is equipped with dual 4” thick HEPA filters that can be recirculated into the lab or sent directly to house exhaust.

ISO 5 INTERIOR CLEANLINESS. The HEPA inlet provides the interior with a laminar airflow. This HEPA clean environment meets or exceeds ISO 5 for clean processing of work while protecting operators from exposure.

FRONT LIFT DOOR. The glove ports sit in a front lift door that opens vertically to effectively load and unload equipment. This door is hinged for ease of use.

PHENOLIC BASE. The chemically resistant phenolic base is dished to maintain spills and protects the work surface from harmful chemicals.

INTERNAL LED LIGHT. This enclosure features an internal LED light for improved visibility inside of the enclosure and can be adjusted directly on to the application. This light comes standard with every Glovebox Workstation Enclosure.

STANDARD SIZES. The Glovebox Workstation series is available in 4, 5, and 6 foot standard width options. Customs are available.

The HEPA inlet provides the interior with a laminar airflow. This HEPA clean environment meets or exceeds ISO 5 for clean processing of work while protecting operators from exposure.

The Bag-in/Bag-out Dual HEPA exhaust ensures safe recirculation of air into the room or connection to house exhaust.