Easy to clean ABS waterproof IP65 enclosure; resistant to harsh and corrosive environments (IP40 on version with thermal printer).

Dimensions: 278x125x186 mm.

Suitable for use on the table, column or wall.

Real time clock and permanent data storage.

Fitted alibi memory, integrated on the motherboard.

Functions available with DINITOOLS utility- Configuration, calibration, data entry and print formatting.- Recording and filing of all the information on PC.- Complete retrieval of the data saved for quick substitution.- Instrument calibration with numeric values.

Selection of the programme language directly in the set-up.

24-bit A/D converter, 4 channels, up to 3200 conv./sec. and up to 8 signal linearisation points.

4 independent analogue channels (up to 2 in the E-AF03 version, 1 in the E-AF09 version) for reading of the weight.

Connection with up to 16 analogue load cells of 350 Ohm (45 load cells of 1000 Ohm) and with the main digital load cells available on the market (mod. E-AF03).

Up to 10.000e OIML or 3 x 3000e @ 0,3 µV/d CE-M legal for trade.

Up to 1.000.000 displayable divisions for internal factory use, with internal resolution up to 3.000.000 points.

Indicator with AF01 software and external power supply. Built-in rechargeable battery, RJ45 connector, glands and cups not included.
TAKE NOTE: no option can be installed at the time of order neither any calibration or approval service. Options can be provided separately.