Process

WIKA
Part of your business

Solutions for pressure, temperature, force and level measurement, flow measurement, calibration and SF6 gas solutions from WIKA are an integral component of our customers' business processes.

This is why we consider ourselves to be not just suppliers of measurement components but rather more a competent partner that offers comprehensive solutions in close co-operation with our customers – customised and precisely designed products for the exact requirement.

Can a 3-wire transmitter also work in a 2-wire configuration?

Can each pressure transmitter measure any medium?

No, the materials stated in the data sheet are to be checked by the customer for resistance.

Can the connector be altered?

The electrical connection cannot be altered.

Can you switch between 4 ... 20 mA and 0 ... 10 V?

No, the output signal is defined in the order and cannot be changed.

What causes zero point drift?

Through the ageing of the electrical components, through temperature and load cycling.

What solutions are there against pressure spikes?

For this, we recommend a process connection with a reduced pressure port of 0.6 mm or 0.3 mm.

Will a pressure transmitter also work without a power supply?

No.

Will the pressure transmitter still work properly when the supply voltage is slightly below the minimum specified level?

Up to a certain pressure, the pressure transmitter will still provide a correct output signal. If the pressure increases beyond this point, however, the output signal will remain at the same value, since the power supply is not sufficient to drive the output fully.

What is the overpressure limit?

Pressures up to the overpressure limit will not cause any permanent damage to the measuring instrument, however, the error limits specified in the data sheet could be exceeded.

At what pressure could the pressure measuring instrument be destroyed?

Pressure values above the overpressure limit can lead to irreversible damage of the measuring instrument. It does not matter whether this pressure is present constantly or only for a short period of time. In either case, the complete destruction of the parts exposed to the pressure and the sudden escape of the pressure medium can be expected.

What is equipment for ventilation and air-conditioning used for?

Ventilation and air-conditioning equipment, also known as air-handling units, exchanges the used room air with fresh air from the outside. Rooms are thus adequately ventilated and the air quality is improved.

Why is it important to make V/AC equipment as energy-efficient as possible?

For reasons of environmental protection and cost, it is very important to make ventilation and air-conditioning equipment as energy-efficient as possible, since the largest energy consumption in residential, office and commercial real estate is spent on ventilation and air-conditioning. In addition, some EU standards now even regulate energy efficiency.

What is the function of ventilation and air-conditioning measuring instruments?

Measuring instruments for ventilation and air-conditioning can indicate any excessive pressure loss due to dirty filters, enable the control of the correct operating state and prevent frost damage to the heat exchangers. In addition, they can be used for monitoring since they are available with digital interfaces.