The
Kent / AMCO C700 oscillating piston type water meters are the
most advanced design positive displacement meters available. The complete
family of top-performing C700 meters, sizes 5/8" to 2", can be offered
with a variety of registers and material configurations.

The Kent
/ AMCO T3000 is a high velocity Class II Turbine Meter exceeding
AWWA Standard C-701. Performance and simplicity are
incorporated into the T3000 design. The T3000's throated inlet
increases low flow sensitivity, assuring high accuracy with low pressure
loss. This design also provides space between the rotor and chamber
wall to allow passage of the suspended solids commonly seen in large
meter installations. For higher accuracy, in-line straightening
vanes reduce turbulence. The T3000's undergear assembly reduces
the speed of the driving magnet, preventing magnetic breakaway during
acceleration, maximizing usage registration. The T3000 design
is modular, facilitating maintenance by allowing replacement of the
measuring element without removing the meter from service. High
quality materials, including a tungsten carbide rotor shaft, PTFE (Teflon)
radial bearings, ceramic jewel thrust bearings and stainless steel
body trim, combine to maximize service life. Available
in bronze from 1 1/2 - 8" with various pulser and register options.

The H4000, sizes 1½'' through 12'', is an inferential
cold-water meter with a rotor that measures accurately in either direction,
driving the rotor in direct proportion to the velocity of the water passing
through the meter. The H4000 is recommended for industrial monitoring of
high flows of cold water (less than 120° F) for direct read, remote
totalization, and batching and continuous processes for rate of flow and
bi-directional flows.

The R1000, sizes 2'' to 10'', is designed for measurement of high flows of cold
water, in-plant applications for dirty water, standard agriculture, river and
lake pumping applications as well as normal irrigation applications for water
resource management.

Unparalleled
accuracy in measuring a wide range of flows. The C-3000 compound
meter combines all the great features of the T-3000 turbine meters
and C-700 piston meters to form a true high performanceClass
II meter.

The COMBO meter is designed for the highest possible metering
accuracy in installations where externally large variations
in flow are typical such as schools, hospitals, shopping
centers, and other installations where flow demands vary
from a few drops to large peak-hour demands.
The C-3000 Saves Money. Because the COMBO is lighter and less bulky,
it does not require special equipment and additional labor to install.
All C-3000 parts and subassemblies are completely interchangeable
with the T-3000 turbine meter and the C-700 positive displacement
meter, thereby eliminating dual inventory and cutting maintenance
costs.

The
C-3000 features a unique changeover valve. This spring loaded poppet
style valve functions on differential pressure, and is a positive
action valve, which means that it is fully opened or fully closed
at all times. This superior design assures the highest accuracy
(100% ± 1/2%) and eliminates the loss of revenue at changeover
which is common with weighted check valves.

The
H-3200 Fire Hydrant Meter is ready to deliver high performance,
high revenue capabilities and convenience.

The
rugged H-3200 fire hydrant meter is a high performance Class II turbine
meter ideal for temporary metering of well pumping, irrigation, construction,
testing and other similar nonpermanent applications.

The
H-3200 has a standard flow restrictor plate, which limits maximum
output to 400 gpm to protect the meter from overspeeding. Or you
may select the optional internal check valve, installed in place
of the restrictor plate, that serves as a flow suppression device
as well as a nonreturn valve and adds no weight or length to the
meter. Another option is a gate valve which allows the operator to
regulate the flow from the hydrant.

Light
weight & easy to handle. The light weight aluminum body is fitted
with inlet and outlet 2 1/2" NST national fire standard hose couplings,
with a removable delrin plastic screen in the inlet coupling to prevent
rotor damage from large particles. Convenient removable steel carrying
handles act as a "roll bar" to protect the register and can
easily be replaced.

The
H-3200 is built tough to withstand rough usage. One of the many advantages
is the interchangeable parts and subassemblies used in the T-3000
turbine meter and C-3000 compound meter, guaranteeing spare parts
availability and low maintenance costs.

PVDF Turbines, sizes ½'' and 1'' are designed
for those applications where a more rugged meter is required. This unit
retains all of the capabilities of the 316 SS Turbine Meter, plus the
PVDF (Kynar) material and ceramic shaft allow use with approximately
100 more chemicals than stainless steel.

This line of 0.5% accuracy turbine meters is
appropriate for a wide varity of flow applications. Choose among
the connection and output variants to get the meter best suited to your
needs. Ideal for food and dairy processing, beverage manufacturing and
pharmacuetical production.

Available with NPT, ANSI flange or sanitary clamp ends.

The construction material offer compatibility with a
wide range of industrial process chemicals. Outputs include sine wave
frequency pulse, conditional square wave pulse or blind 4-20mA analogue
signal. As with other inferential turbine metering technologies, installation
requirements must be observed for the best accuracy.

Electronic Registers are available for all cold water Positive Displacement
and Turbine Meters. The battery powered Electronic Register is designed
to replace mechanical registers with LCD displays of rate indication and
totalization in the basic unit.

The EBC2C is an AC powered totalizer, rate indicator, and batch controller. It
has both single relay output to a standard solenoid and dual relay output for
use with dual valve batch control systems, dual stage solenoids, or bi-directional
solenoids with auxiliary limit switches.

The Model R10 Rate Indicator/Totalizer is an economical battery-powered up-counting
totalizer and a 1/Tau rate meter, with an 8-digit LCD display. Separately programmable
count/rate scalers and decimal points allows for display of Rate and Count in
various engineering units.

AquaMaster revolutionizes high volume electronic
water measurement. Available in sizes 5/8" to 24", the rugged AquaMaster
is designed to maximize revenue collection by offering a broad flow range
with high accuracy, retention of high accuracy without testing or parts
replacement, low-cost, foolproof installation, optional data logging
capability and battery or mains-powered models to accommodate site requirements.

The MagMaster is an electromagnetic water and waste meter operating on Faraday's
principle of induction. MagMaster covers the full flow range of most mechanical
meters with a high accuracy. MagMaster is available in sizes 2” to 12” with
industry standard 4-20 mA output of rate of flow data.