Specifications1. 210W high efficiency,55N/cm high torque eletric motor, low noise and vibration.2. The speed of motor: 300-50,000/rpm.3. Big screen can indentify all operation proceed quickly.4. 10 setting progrem.5. The electric motor can be high pressure and temperatuer sterilized.6. The new foot pedal meet the international waterproof standard IPX4, it can control the speed, program, output of supplying water, the direction of speed.

Maximum output power

120w

Power

AC 110/230V, 50/60 HZ

Torque

5.0-55N.cm (20:1)

BLDC Speed

0 rpm -50000 rpm

RPM

30-2500 rpm (20:1)

Infusion flow

0-100ml/min

Motor

Original import Surigical brushless motor

Volume

400*330*265mm

Net weight

7kg

Gross weight

8.2kg

OTY per master carton

1pc/ctn

Content:

Major control unit

1

Foot pedal

1

Electric wire

1

Motor with wire

1

Water supply bottle holder

1

Motor holder

1

Pipe clasp

8

Y type isocon

1

Internal waterspray nozzle

1

Contra angle for implant

1

Fuse

1

Prevent steam bolt

1

Water supply pipe

1

Nozzle for contra angle

1

Buyer Note

International Buyer - Please NoteImport duties, taxes and charges are not included in the item price or shipping charges. These charges are the buyer responsibility.Please check with your country customs office to determine what these additional costs will be prior to bidding/buying.

About us

Foshan Vimel Dental Equipment Co.,Ltd. is a professional enterprise that specializes in research and development, production and marketing of dental equipment.

"Superior quality, reasonable price and satisfactory service" is our principle. We are sincerely looking forward to cooperating with companies from the whole world for mutual benefit cooperation business with a brilliant future!

Q&A

★1. Q: Are you a manufacturer, or a trading company? A: We are a manufacturer.