Software leverages the accuracy of an 8.5 digit DMM, with the versatility of the PRS-300 Programmable Resistor to make the most revolutionary standard resistor ever manufactured

Resistance Range: 0.1 Ω to 20 MΩ

Resolution: 1 μΩ

Accuracies better than 20 ppm; source resistance with the same accuracy that your 8.5 Digit DMM can measure.

All you need to do is enter the target resistance. The software uses the DMM to measure the resistance being sourced by the PRS-300 and readjusts the source resistance as close as possible to the target resistance.

Software automatically provides the measurement uncertainties of every measurement