Description

Product Description

The Power-Turn Elite Mattress System combines two clinically effective therapies: Lateral Rotation and True Low Air Loss in a portable system. The mattress utilizes 18 independent 9" deep air cells to provide proper pressure redistribution for the prevention and treatment of Stages I-IV pressure ulcers. Lateral Rotation Therapy reduces the risk of pulmonary complications for patients who are immobile. A Full, Left or Right Turn mode is offered with either a 25º Small Turn or a 40º Big Turn. The mattress replacement is designed to fit all standard hospital beds and includes a high density foam base, head bolster and abduction wedge for additional patient safety. The Power-Turn Elite Lateral Rotation Mattress System supports patients weighing from 650 lbs. to 1000 lbs. The Power-Turn Elite meets California Technical Bulletin #117 for fire retardancy. An optional fire barrier is available to meet the most stringent fire safety requirements when requested for foam base.

Mattress Sizes

Features

Bladders: Ventilated low air loss cells have air holes throughout the mattress to help minimize skin maceration by reducing excess moisture under the patient. Air cells are constructed of TPU, thermal polyurethane material. This combination insures strength, durability and patient comfort.

CPR Valve: CPR Valve quickly deflates alternating pressure mattress system for medical emergencies. A simple twist releases the air immediately. This multipurpose CPR Valve can be used to inflate or deflate the mattress with our optional blower pump.

Blower Pump:The blower pump is controlled by a micro-processor which allows the caregiver total control of comfort and chosen therapy. Cycle time ranges from 5 to 95 minutes. The blower pump inflates the mattress within minutes. This reduces the waiting time for therapeutic treatment to begin.

Head Rest and Leg Abductor: Head rest and leg abductor wedge provide support during the turning cycles. Head support is constructed from Viscotec, a visco elastic foam. The leg abductor is constructed from PPF, Pressurized Polymer Foam. Both are covered with Vyvex-III.