Setaflash Series 3 ActiveCool Flash Point Tester, 33200-3

Setaflash Series 3 Active Cool provides fast and reliable flash point results and requires minimum operator expertise. It has been designed to rapidly and efficiently carry out Flash / No Flash tests to determine the flash point of liquids and semi-solids in the 10 to 135°C temperature range.

• Compliance
• Safety
• Product quality
• 2ml of sample and a result in less than 2 minutes

Setaflash Series 3 Active Cool has a very simple user interface with colour digital display and touch screen icons which guide the operator through a straightforward testing process.

For easy record keeping the instrument stores 100,000 test results which can be saved and transferred via a USB port.

The versatile range of Setaflash instruments is ideal for use in the laboratory, production line or for portable test applications.

Rapid Results in just 1 or 2 Minutes The Setaflash Series 3 Active Cool is an easy to use instrument that can complete a flash/no flash test in less than two minutes and in many cases in 60 seconds, or determine the flash point of a sample within a temperature range of 0°C to 135°C. The sample cup is heated/cooled by Peltier cells. This gives very accurate and stable sub-ambient test temperatures and the benefit of rapid temperature normalization between tests.

Ease of Operation All functions on the Series 3 Active Cool including test parameters, instrument status, and test results are accessed via the touchscreen display and push button. An audible prompt is sounded when a flash is achieved. An automatic flash detector reduces the chance of misinterpretation of the test result by the operator and enhances repeatability.

Innovative Design Gas for the test jet/shutter assembly is supplied from an integral gas tank, which is filled using a standard butane (lighter) refill cartridge. The Series 3 Active Cool Flash Point Tester is lightweight and suitable for mobile use when operated from the optional 12-volt power supply adaptor. Supplied with: 2ml syringe, gas ignitor, mains lead, and instruction manual.