More Views

Tektronix AFG31000 Series

Arbitrary Function Generator

The Tektronix AFG31000 Series is a high-performance AFG with built-in arbitrary waveform generation, real-time waveform monitoring, and the largest touchscreen on the market. Providing advanced waveform generation and programming capabilities, waveform verification, and a modern touch-screen interface, the new AFG31000 is sure to delight and simplify the job of every researcher and engineer.

Key Performance Specifications

1 or 2 channel models

Output amplitude range 1 mVP-P to 10 VP-P into 50 Ω loads

Basic (AFG) mode:

25 MHz, 50 MHz, 100 MHz, 150 MHz, or 250 MHz sine waveforms

250 MSa/s, 1 GSa/s or 2 GSa/s sample rates

14-bit vertical resolution

Built-in waveforms include sine, square, ramp, pulse, noise, and other frequently used waveforms

Sweep, Burst, and Modulation modes (AM, FM, PM, FSK, and PWM)

Advanced (Sequence) mode:

Continuous mode (optional Sequence, Triggered and Gated modes)

16 Mpts arbitrary waveform memory on each channel (128 Mpts optional)

Up to 256 steps in sequence mode with loop, jump and wait events

Variable sampling clock 1 μSa/s to 2 GSa/s

Key Features

Patented InstaView™ technology enables engineers to see the actual waveform at the Device Under Test (DUT) in real time, without the need of an oscilloscope and probe, eliminating the uncertainty caused by mismatched impedance

Sequencing option adds the ability to program long, complex waveforms with up to 256 steps

The 9-inch capacitive touch screen works like a smart phone and has short-cuts to frequently used settings

Built-in ArbBuilder lets you create and edit arbitrary waveforms on the instrument, eliminating the need to connect to a PC

Outputs are protected from over voltage and current to minimize potential instrument damage

Compatible with TekBench™ software to help students set up, control, and analyze test results in the lab