The product uses saturated steam to rapidly and efficiently sterilize
articles. It can be used in hospitals, public health centers, health
stations, clinics in factories and mines, scientific research
institutes, etc, to sterilize medical equipment, surgical dressings,
glass utensils, solutions, substrata. It can also be used as cooker in
plateau areas and be used to make high quality drinking water in
industrial and mining enterprises. It has hundreds of other uses as
well.

The product is heated by the 2KW immersed
electro-thermal tube. It gets hot very fast and with low power
consumption (saving about 50% of energy as compares with electric
stove).

The container is made of 0CrlgNi9 (stainless steel),
which is not only alkali & acid proof but also more than 3 times
more tensile and compressive than aluminum alloy, ensuring safety and
service life.

The lid is equipped with a safety valve and air
releasing valve. When the steam pressure exceeds 0.16MPa, the safety
valve can automatically release the pressure that is too high, having
double safety function and featuring safety and reliability in use. The
lid is also equipped with a pressure gauge with megabar and temperature
readings, which makes it convenient for the user.

Instructions for use:

1. Pour 3.5 liters of water into the container and
be ready for use. As sterilization will consume certain amount of
water, add water to the rated amount before using again~ otherwise the
electro-thermal tube will be damaged due to shortage of water.

2. Put the articles to be sterilized on the sieve
plate in the sterilizing drum, properly wrapped up, leaving certain
space between the items so as to let the steam penetrate, thus
guaranteeing sterilizing quality.

3. Put the sterilizing drum into the container and
then insert the hose (on the lid) into the half-round slot in the
sterilizing drum. Making the alignment of the lid and the ear slot on
the container and evenly tightening the handle to make the lid and
container closely fit each other.

4. Plug in the power to the unit. It displays the
beginning of heating, and at this time air releasing valve should be
opened to let cold air out. When the steam jets out from the
air-releasing valve, close the valve. The pointer of pressure gauge
will indicate steam pressure in the sterilizer.

5. When steam pressure reaches the requested
pressure of sterilization, begin to account the needed time of
sterilization. When steam pressure rises up to 0.14MPa, safety valve
starts to operate automatically so as to keep the constant pressure. If
needing the pressure below 0.14MPa to sterilize, you can control the
power switch on or off by hand to keep constant Pressure when the
pressure reached the requested range.

6. Once complete with the sterilization, cut off the
power. Wait for the readings of pressure gauge to return to zero, then
open the air releasing valve and the lid. Do not release steam or open
the lid immediately to avoid the boiling liquid to flow out the
container, which can cause injury. *Again, make sure the temperature
has cooled down full, before opening the lid.

ORDER ONLINE NOW

OR CALL 1-866-606-3991

Packing List:

Portable sterilizer unit

Drum

Sieve plate

Power cord (110v 60hz for North American market)

Operation manual

Everything you need is included. It will be ready to operate out of the box!