Fluke 368 True-rms Leakage Current Clamp Meter

Fluke 368 True-rms Leakage Current Clamp Meter

The Fluke 368 FC true-rms leakage current clamp meter helps users detect, document, record and compare leakage current readings over time as a means of preventing unplanned downtime, and identifying intermittent GFCI and RCD trips, all without taking equipment off line.
The Fluke 368 FC has a large 40 mm diameter jaw for work with large conductors. The clamp’s jaw is fully shielded to accurately capture very small leakage signals, and to minimize external electromagnetic interference.

Expected battery life, without using backlight and spotlight, is more than 150 hours

Automatic shut-down

Meter automatically shuts down after 15 minutes of inactivity

Safety and Environmental Specifications

General safety

IEC 61010-1: Pollution Degree 2

Measurement safety

IEC 61010-2-032: CAT III 600 V / CAT IV 300 V

Operating temperature

-10 °C to +50 °C

Storage temperature

-40 °C to +60 °C

Operating humidity

Non-condensing (<10 °C)

90% relative humidity (10 °C to 30 °C)

75% relative humidity (30 °C to 40 °C)

45% relative humidity (40 °C to 50 °C)

Protection class

IEC 60529: IP30 (jaw closed)

Operating altitude

2000 m

Storage altitude

12000 m

Current sensor action category

IEC 61557-13: Class 2, ≤30 A / m

Electromagnetic compatibility (EMC)

International

IEC 61326-1

Industrial Electromagnetic Environment

CISPR 11

Group 1 Class B

Group 1

Generated inside the equipment and / or use radio frequency energy associated with conducting the energy for the device’s own internal functions is essential.

Class B

Equipment for home appliances and residential buildings directly connected to a low voltage power supply network equipment. When this device is connected to a test object, it may exceed the emission levels produced by CISPR 11 requirements.
Korea (KCC): A type of equipment (radio and communication equipment industry)

Class A

This product meets the requirements of industrial electromagnetic equipment, vendors or users should be aware of this. This device is intended for use in a commercial environment, instead of the home environment.
USA (FCC): 47 CFR 15 B Subpart. In accordance with section 15.103 provides that the products are considered tax-free device.