Dantest Hardware Devices

HW5 Device

The HW5 device is the top hardware product offered by QHRV. This device can communicate with any computer using USB cable, Infra Red and Bluetooth.

HW5 has two data input channels: ECG and Pulse Wave signal (Plethismogram). Both channels are 16bit with available reading frequency of 100, 250, 360, 400 and 600 Hz. The strength of the signal is automatically amplified.

HW5 can operate in two separate modes:

Real Time: While connected to a PC, patient data transfers directly and in real time to the computer for instant evaluation and analysis.

Standalone: Data from both channels is able to be stored in the hardware memory for analysis without the aid of a computer. The hardware can store measurement from more than 50 patients at any given time. All patient data can be transferred later to a PC for further analysis and reporting.

HW6 Device

HW6 has two data input channels: ECG and Pulse Wave signal (Plethismogram). Both channels are 16bit with available reading frequency of 100, 250, 360, 400 and 600 Hz. The strength of the signal is automatically amplified.

HW6 can operate only while connected to a PC, patient data transfers directly and in real time to the computer for instant evaluation and analysis.

This device meets all USA and EU medical safety standards.

This device is FDA 510k cleared and meets all USA and EU medical safety standards.

HW6C Device

HW6C has 3 data input channels: ECG and two Pulse Wave signals (Plethismogram) with SpO2 capability. All channels are 16bit with available reading frequency of 100, 250, 360, 400 and 600 Hz. The strength of the signal is automatically amplified.

HW6C can operate only while connected to a PC, patient data transfers directly and in real time to the computer for instant evaluation and analysis.

This device meets all USA and EU medical safety standards.

This device is FDA 510k cleared and meets all USA and EU medical safety standards.

HW8 Device (Lite)

HW8 has one data input channels: Pulse Wave signal (Plethismogram). The channel data is 16bit with available reading frequency of 100, 250, 360, 400 and 600 Hz. The strength of the signal is automatically amplified.

HW8 can operate in two separate modes:

Real Time: While connected to a PC, patient data transfers directly and in real time to the computer for instant evaluation and analysis.

Standalone: Data from both channels is able to be stored in the hardware memory for analysis without the aid of a computer. The hardware can store measurement from more than 50 patients at any given time. All patient data can be transferred later to a PC for further analysis and reporting.