DL350 Portable ScopeCorder

The DL350 Portable ScopeCorder combines all the measurement and recording capabilities you need when you are away from your office or lab - all in one compact instrument.

Featuring a resistive touch screen, the DL350 provides the flexibility you need when you need it. Whether you are recording high-speed transients or long-term signals, using “quick and simple” setup or advanced features, the DL350 makes complete measurements completely portable.

Choose from 20 types of input modules and install up to 2 in the DL350 Portable ScopeCorder at a time to capture up to 32 channels plus 16 built-in channels of digital input. Input types include:

Voltage & Currents

Sensor Outputs

Temperature, Vibration/Acceleration, Strain, Frequency

Logic Signals & CAN / CAN FD / LIN and SENT

Use it like a data acquisition system or a long memory oscilloscope

Up to 20 Gpoints* of data per module can be recorded directly to an SD card. This means that the DL350 can be used for continuous recording for up to 50 days. For high speed signals, up to 100 M points per module of internal memory is available to capture fast transients. This is up to 10,000 times more than other portable oscilloscopes or test tools and thus signals can be captured with higher sample rates or for much longer periods.

* The firmware version 1.20 or later is required.

Verify power line quality using harmonic, power or FFT analysis

The power in single- and 3-phase systems can be evaluated. Additionally for fundamental waveforms of 50 or 60 Hz, up to 40 harmonic orders can be analyzed. Alternatively use the suite of FFT functions to perform full frequency analysis.

Harmonics analysis (bar graph)

FFT analysis

Environmental features for portability

Vibration resistant
Instruments used for in-vehicle driving tests or field maintenance must be able to make reliable measurements. The DL350 has an aluminum inner frame and an external rubber bumper and conforms to the Japanese JIS D1601 standard for resisting in-vehicle shock and vibration.

Wide Temperature Range
Even when used with the rechargeable battery, the DL350 will operate in temperatures from 0 to 45 degrees. The DL350 brings high-quality laboratory measurements into the harsh environments of the field.

A wealth of triggers for fault finding

The user has a choice of a simple level trigger or can use enhanced triggers such as pulse width, waveform period and triggers across multiple channels. For example, the wave window trigger is ideal for AC power line monitoring which enables voltage sags, surges, spikes, phase shifts or drop outs to be easily captured (available for 40 to 1,000 Hz waveforms).

Leave a DL350 unattended and automatically save the waveform to a file or send a notification email if and when it triggers.

Vehicle Edition

The ScopeCorder Vehicle Edition is designed for engineers working in the automotive and railway industry, providing a thorough insight into the dynamic behavior of the electromechanical system. This integrated approach results in considerable time saving compared to using multiple separate instruments.

In-vehicle Testing

CAN/CAN FD bus, LIN bus and SENT monitoring

Use the DL350 with /VE option and bus monitor module to decode CAN/CAN FD bus, LIN bus or SENT signals and display information such as engine temperature, vehicle speed and brake pedal position as trend waveforms and compare this with the analog data coming from the actual sensors. This enables automotive engineers to gain an insight into the dynamic behavior of the complete electromechanical system.

The DL350 offers a choice of operating modes, including recorder mode, which makes it easier than ever to set up long-term data recordings. Choose Easy Setup, and the setup wizard quickly guides the operator through the entire setup process. Just select the desired channels, recording duration, and sampling interval, and press Start!

Choosing Easy Setup opens a guided, three-step process to begin recording for up to 50 days.

Use the touchscreen during recording to scroll across the entire data set.

Long-Term Temperature Monitoring

Using the DL350 Memory Recorder Mode Easy Setup, combined with a 16-channel thermocouple input, it's easy to record hours, days, or weeks of data with a few simple settings.

Examples of Complex Measurements

Field

Application Purpose

Measurement Item

User Advantages

Slot 1

Slot 2

EV (electric vehicle)

Evaluation of battery
voltage fluctuation
while driving

Battery voltage

CAN/CAN FD communication data

Small size, battery drive,
synchronization with GPS*
position and time data

Power tool

Evaluation of
practical behavior

Battery voltage, motor rotation pulse

Tool vibration

Small size, battery drive,
complex measurement of
voltage and vibration

Logging GPS and Analog Data

Combining Motion and Analog Measurements

Capture data from motion accelerometers synchronized with other analog data. The DL350 series ScopeCorder features input modules and functions to make this possible.

Power Line Monitoring in Plants and Factories

By using a wave-window trigger, voltage sags, surges, spikes and dropouts can be detected and captured.

Multi-phase voltages up to 1 kVrms and 1.4 kV peak can be recorded using 720268 high-voltage isolation modules.

In the case of unattended operation, waveforms can be saved, or an e-mail sent, when the DL350 is triggered.

Industrial robot maintenance

It is possible to monitor and record the control signals to the servomotors and their speed and torque at the same time.

For condition monitoring, FFT analysis can be used on the vibration signals from accelerometers to help identify potential failures in machines or components.

Remote operation is available using the ‘assistant software’ or the input/output terminals making it potentially safer to use.

Additional Option License for DL350

This is a software license allowing users to add “vehicle edition”, a software option for the DL350 ScopeCorder, by themselves after purchasing the main unit.

PC data and setup file management

DL350 Assistant software
Data files or setup configuration files stored in the DL350 SD card can be backed up with the press of a button. Remote setting, start-stop control and setup file editing can also be easily done on the connected PC.

Input logic signals can be displayed as waveforms at 10 MS/s sampling resolution. Each port can measure 8 bits, and 2 ports of input are available on each module. Thus, a single module can observe 16-bit logic signals. Compatible with 700986, 700987, 702911 and 702912 logic probe.
Applicable to DL350 and DL850 series.

Use the CAN/LIN bus monitor module to decode the CAN or LIN-Bus signals and display information on physical data like engine temperature, vehicle speed and brake-pedal position as analog waveforms and compare this with the data coming from real sensors.
Applicable to DL850V & DL850EV, DL350 (/EV option) ScopeCorder Vehicle Edition only.

Use the SENT monitoring module to decode the SENT protocol signal and display information on physical quantities from SENT output sensors.
Applicable to DL850V & DL850EV, DL350 (/EV option) ScopeCorder Vehicle Edition only.

Strain Module, type NDIS (2 channels) to be used in combination with 701955 or 701956 NDIS bridge head.
Using the strain module and strain gauge the ScopeCorder can detect and display mechanical stress (strain). It works on the principle that the resistance value of a metal foil changes as it expands and contracts.
Applicable to DL350, DL850 series, DL750 series and SL1000.

Strain Module with Shunt Calibration, type DSUB (2 channels) to be used in combination with 701957 or 701958 DSUB bridge head.
Using the strain module and strain gauge the ScopeCorder can detect and display mechanical stress (strain). It works on the principle that the resistance value of a metal foil changes as it expands and contracts.
Applicable to DL350, DL850 series, DL750 series and SL1000.

Typically, a portable scope or test tool is used by an engineer to diagnose faults on a production line in real-time, or to carry out tests in a workshop. But how can faults be detected and diagnosed when they are intermittent and the engineer needs to be somewhere else? On the other hand, a portable data logger can capture lots of data, including the faults over a period of time, but requires the data to be searched and analysed later to make sense of it.

In this customer case story, Reiner Schultheiss; electrical engineer at Michelin tire manufacturing, reflects on his time with the company and explains how the portable Yokogawa DL350 ScopeCorder was welcomed to Michelin and how it ensures quality at the tire manufacturing process. In this customer case story, Reiner Schultheiss; electrical engineer at Michelin tire manufacturing, reflects on his time with the company and explains how the portable Yokogawa DL350 ScopeCorder was welcomed to Michelin and how it ensures quality at the tire manufacturing process.

Firmware

Product Overviews

In this video we review the major features of the DL350 showcasing its portability, functionality, and operability. This device features battery power, 18 signal conditioning input modules, and touchscreen access to enhanced triggers, math, and analysis.

Capture, display, record and analyze the widest variety of electrical and physical parameters in automotive, transportation, electronics, energy and mechatronics. In this video Lee Thomas, Yokogawa T&M Sales Manager in the UK showcases the Yokogawa DL350 ScopeCorder.

In this video we demonstrate the GPS data logging capability of the DL350 Portable ScopeCorder. Recording Position, Velocity, and Altitude simultaneously with accelerometers or other analog inputs is simple with the DL350's built-in features.

This site uses cookies to simplify and improve your usage and to provide you with the best experience of this website. Cookies are small text files stored on the device you are using to access this website. If you ignore this message and continue without changing your browser settings, we will assume that you are consenting to our use of cookies. For further information on our use of cookies, please see our Cookie Policy.