Field Mount F126-EG

Field Mount F126-EG

APLICACION

MODELO

TECNOLOGIA

MARCA

Displays

F126-EG

Fluidwell

FUNCION

CODIGO

SECCION

Computadores de Flujo.

F126-EG

22

The F126-EG flow computer has been developed to calculate corrected gas volume at normal conditions. The corrected volumetric flow is calculated by measuring the uncorrected volumetric flow, actual line temperature and pressure which are processed with the equations stored in the flow computer. A compressibility factor can be set to approach a real gas behavior. The reference conditions can be defined as desired (e.g. 15°C or 60°F and 1.013 bar).

The display shows the compensated flow rate, total, accumulated total and the actual line temperature and pressure. On-screen engineering units like Nm³ are easily configured from a comprehensive selection.

Advantages

Robust IP66, IP67 (NEMA4X) field enclosure. It is so rugged, you can even stand on it!