All you need to start using your flowcell. Do you want a flowcell exactly according to your wishes? Contact us to find out what the possibilities are for your custom chip

Product Code

FLOW_CELL_PLATFORM

Dimensions

128 mm x 85,4 mm x 20 mm

Material

Aluminium

Sealing mechanism

Load ‘n Seal

Maximum operation temperature

80°C

Max. operating pressure

10 bar

Sealing material

Perlast (FFKM)

Contents

Teflon tubing and 5 FFKM ferrules

Length

5m

Outer Diameter (OD)

1/16 inch

Inner Diameter (ID)

250 μm

Number of chips per pack

3

Distance between channel and top surface

depending on the chosen chip

Distance between channel and bottom surface

depending on the chosen chip

Channel location

No

Total chip thickness

depending on the chosen chip

Chip size

45 mm x 15 mm

Channel width

depending on the chosen chip

Channel height

depending on the chosen chip

Internal Volume

depending on the chosen chip

Number of Inlets

3

Number of outlets

3

Inlet/outlet hole sizes on top of the chip

1.70 mm

Inlet/outlet holes size at channel

0.60 mm

Optical properties

Optical clear view from all sides

Supplied in Fluidic slide?

Yes

Material chip

Borosilicate

Material black cartridge

Polypropylene

Filters

2 filters, 4 connectors (PEEK)

Compatible to tubing with diameter

1/16” OD (PTFE tubing)

Write Your Own Review

You're reviewing: Flow Cell Platform

How do you rate this product? *

1 star

2 stars

3 stars

4 stars

5 stars

Quality

*Nickname

*Summary of Your Review

*Review

There's no flow in my setup. How do I fix this?

Is your setup clogged with dirt or fully blocked?Chips can get clogged, for example when they’ve been used before and have not been cleaned the right way. If your chip is clogged with dirt, check our document on the prevention of clogging and/or read the answer to question number one on this page (regarding the cleaning of chips). Clogged chips often do let some flow through the chip, which makes it easier to identify the issue as clogging. If there is no flow at all it’s more likely that the flow path is not connected correctly.

Pinpoint the issueWe recommend walking through the full track of the fluid flows: check if your syringe or pump is not blocked, if your tubing is actually hollow and if your filters let fluids pass through. Then check if your chip has inlet and outlet holes and if these are positioned so that the fluids can go through the chip. The easiest way to do this is to disassemble the setup and check if fluids come out of the pump, then connect the tubing and check if fluids can come through. Keep on adding parts and checking if the fluids go through. This way you can pinpoint the problem.

If the problem remainsIf the problem cannot be solved after following the above-mentioned steps, then take some pictures or make a short movie of your setup and send this, together with the description or name of your chip, to Micronit. We will gladly assist you in identifying the issue.