EasyLog | EL-GFX-D2

Dual Channel Temp, RH & Dew Point Data Logger with External Probe

Price

$230.99

Overview

This dual channel data logger measures and stores over 250,000 temperature and relative humidity readings using up to two external temperature & humidity probes*. It has user-programmable, audible alarm thresholds with confidence/alarm LEDs.

Its high contrast, dot-matrix LCD with three face-buttons for navigation gives real-time trend analysis, data summaries and the ability to start, stop and restart the data logger without connecting to a PC. EasyLog USB software enables quick configuration and data download via the USB port. Supplied with two lithium batteries, which can typically allow logging for up to 1 year. It can also be powered from USB.

Logging rates between 10 seconds and 1 hour

Stores over 250,000 readings per channel for temperature and humidity

User triggered logging mode

Up to two channels supported (supplied with one probe)

*only one probe is included with the device.

PLEASE NOTE:You can add up to FOUR SPARE BATTERIES for every data logger ordered. If you wish to order more than this quantity, please contact your local sales office for special shipping information.

Specification

Temperature:

Measurement Range

-30 to +80ºC (-22 to +176ºF)

Accuracy (overall error)

±0.3ºC (±0.6ºF) typical

Resolution

0.1ºC (0.2ºF)

Relative Humidity:

Measurement Range

0 to 100%RH

Accuracy (overall error)

±2.05%RH typical (20 to 80%RH)

Resolution

±0.1%RH

Dew Point Accuracy

±1.1ºC (±2.0ºF) typical

Probe Cable Length

2m (only one probe supplied)

Battery Life

1 year

Application Info

The EL-GFX-D2 is particularly suitable for any application where you need to separate the logger from its sensor - perhaps for aesthetic reasons or logistical restrictions. Applications include: