• Environment temperature range: 15℃ ~ 60℃
• Relative humidity: Less than 85%R.H.
• Atmosphere: 86 ~ 106 Kpa
• Without strong vibration around
• Without strong electromagnetic field around the influence, no high concentration of dust and corrosive materials
• Avoid direct sunlight or other heat source direct radiation
• Without strong airflow around, when the surrounding air should be forced to flow, flow should not blows directly to the body
• Machine should be placed flat, maintain the level, because the upper water seal to prevent salt fog spillover
• Around the machine there shall be a certain distance, convenient maintenance operation

Collector: Nozzle have eruptive fall fog quantity to free-fall way down to 80 cm of the funnel cup inside, again by duct flow to measuring cup inside.

Shelf: The shelves of window of model steel is made, pool weight can't more than 2 kg, if place dispersedly samples, affordable 10 kg weight within the supporter and lined with two rows round hole, round hole is for placing buy object bar, vertical plane is divided into 15 degrees and 30 degrees.

Heating water tanker: This tank adherent to the laboratory bottom, used to hold water humidifying keep laboratory temperature smooth, have added wet and heat preservation function.

MACHINE EXTERNAL:

Measuring pail: It collects the spray quantity, the highest scale is 50ml.

Insulated tank: Use water principle, avoid salt fog were leaked.

Saturated air barrel: Placed in control bottom part, USES the SUS304 stainless steel plate is made, its function lies in the air after the barrel warming humidification, allowing air to saturated humidity to the nozzle spray.

Test cover: Roof type bevel 100 degrees perspective cover, and used to cover in the laboratory, combined as one body.

Water inlet: Supplementary laboratory and saturation bucket of water to normal usage.

CONTROL SYSTEM:

Temperature controller: Control laboratory temperature, in accordance with the standard temperature setting.

Saturated air barrel temperature controller: Control saturated air barrel temperature.

Timer: Adjustable 1 ~ 9999 hr can be arbitrary set test time, automatic stop, end with automatic zero setting.

Accumulation time device: Direct reading type 0 ~ 9999.9 hr to instruct a test of the accumulation of time.

Heating water tank thermostat: To protect the water heating temperature, the controller is fit laboratory temperature control this tester surrounding temperature difference too big when, must cooperate with adjustment, the setting temperature is 75 degrees in the control.

Saturated air barrel thermostat: Saturated air barrel temperature control, temperature safety set value than the actual high temperature 5 degrees.