No matter whether you work in the lab or in production - the Microwave Power Meter R&S NRP is always the right choice. The versatility of the novel Power Meter Series R&S NRP is due to the newly developed sensors. These sensors are intelligent standalone instruments that communicate with the base unit or a PC via a digital interface. The SMART SENSOR TECHNOLOGY™ now implemented for the first time, sets new standards in terms of universality and accuracy.

Some highlights of the NRP:

Dynamic range: up to 90 dB (sensor dependent)

Level range: -67 dBm ... +45 dBm (sensor dependent)

Frequency range: 9 kHz to 40 GHz (sensor dependent)

Measurement speed: 1500 measurements per second in the buffered mode

Linearity uncertainty: 1.5 % (20 to 25°C, f = 100 MHz to 4 GHz)

Precise average power measurements irrespective of modulation type and bandwidth

Flexible measurements on up to 128 time slots per power sensor (26 time slots with base unit)

Our temperature and static controlled Calibration Laboratory and Repair Facilites are staffed by fully credentialed professionals who are meticulous in their attention to detail, operates in full compliance with strict NATA (National Association of Testing Authorities) standards of measurement and reporting (ISO/IEC 17025:2005).

TMG provides a uniquely single point to send all of your RF, AC, DC, Optical and Video related test equipment for careful checking.