3. Depending on the different division for calibration
w Standard division (under 10000)
It is necessary to do the capacity setting calibration and weight calibration.
w High precision division (over 10000 ～ 150000 division)
It must do the linearity calibration, capacity setting calibration, weight calibration.

4. Option with RS232 output and 3 relay output

5. With 3 kinds of HOLD function

6. AC / DC power supply

7. LCD display

8. Auto-power cut off to assure the stability performance
( If the power is below the low battery warning , it will be cut off automatically. )

9. With built-in-backlight

ERROR MESSAGE

E1 ⇒ Zero range for turning on machine is over F.S 10%.
E2 ⇒ Zero range for turning on machine is under F.S 10%.
E4 ⇒ Zero range for turning on machine is unstable.
E6 ⇒ Zero range is too high out of its range during calibration mode.
E7 ⇒ Zero range is too low out of its range during calibration mode.

Press ∧ or TARE to select F2 function ⇒ the display appears
Press ↵
The display shows
the factory setting
Press ∧
to input the setting number
∧ = key for moving up (01~15)
Press ↵
-8. you can select the setting.
♦ Before calibrating. 05/07/19
F2 ⇒ NOISE FILTER SETTING
The range of stability is from 01~15.
♦ The bigger the figures you choose.
♦ If the surrounding is stable. depending on the environment. please reduce the setting value to “ 05 ”. SM2020010106
PDF created with pdfFactory Pro trial version www.com
. please adjust the mini jumper SWA1 to “ADJ” position.
♦ The original setting from the factory delivery is “05”. the setting value can be adjust up to “15”. the stable time is quicker but the performance of the
accuracy is not so sensitive as it is.
♦ After finished the calibration.
♦ If the surrounding factor is less stable.pdffactory. please re-adjust the mini jumper SWA1 to “LOCK” position.

SM2020010106
PDF created with pdfFactory Pro trial version www.20 . When the scale returns to 0 (the weight is less than 10 division). the scale will hold the value which is displaying on the screen.
the hold function mode will be released.
.com
. the scale will automatically hold
the maximum value displaying on the screen. the hold value indicating on screen will be print-out. 05/07/19
F8 ⇒ HOLD FUNCTION SETTING
4 In “Hold” mode. if press PRINT key.
Press any key except PRINT key to cancel the hold function.
= In the continuos changing weight value.
= When it is stable.
= When it is stable. please refer to F5 setting)
Press ∧ or TARE to select F8 function ⇒ the screen displays
Press ↵
Press ∧
Press ∧
Press ∧
Press ∧
Press ↵
= No hold function. the scale will hold the value which is displaying on the
screen.
Press any key except PRINT key to cancel the hold function.pdffactory.
(It is not related to F5 interface option but your should have to select proper baud rate
for the receiving transmit beforehand)
(to set the transmitting baud rate.

SM2020010106
PDF created with pdfFactory Pro trial version www.
♦ When the mini jumper SWA1 is in “ADJ” position on the hardware (PCB).
♦ The scale can be calibrated on if the mini jumper SWA1 is in “ADJ” position
when the software F9 is in “n” condition.pdffactory.
. the procedure is finished.22 . 05/07/19
F9 ⇒ SOFTWARE LOCK SETTING
Remarks
♦ When the mini jumper SWA1 is in “LOCK” position on the hardware (PCB).com
.
Press ∧ or TARE to select F9 function ⇒ the screen displays
Press ↵
= UNLOCK
Press ∧
Press ∧ = LOCK
Press ↵
Press ∧ key to select the one you need.
Press ↵ key.
♦ Before setting. please adjust the mini jumper SWA1 on the PC board to “ADJ” position.
the software F9 locking is effected.
the software F9 locking is invalid.