CLC Firmware Release Notes

Firmware Release Notes for CLC version 1.31.190 indicate revisions made to the CLC Firmware version since its initial release.

Compatibility

These firmware updates are compatible with all PhysioTel™ Digital Implants and hardware, Ponemah versions 5.10+, 6.xx and all associated service packs. However, Ponemah v5.20-SP9+ and v6.33+ need to use CLC firmware 1.29.206 or later.

Firmware version 1.30.80 or later is required when using the following implant models: L03, L04, M0G, M1G.

CLC Firmare v0.1.28 IS NOT compatible with Ponemah v5.10, v5.20, and all associated service packs, as well as Ponemah v6.32 and previous versions of Ponemah 6.xx.

Release History

Version

0.1.20*

0.1.21

0.1.22/23

0.1.24

0.1.25

0.1.26

0.1.27

0.1.28

1.29.206

1.30.80

Release Date

December 2012

January 2014

DSI Internal Releases

April 2014

February 2016

June 2016

October 2016

August 2017

September

2017

April 2018

*Original firmware released with CLC.

New features, enhancements, and fixes

Reference #

Release Version

Description

Disposition

42430

1.31.190

Added new commands to the Implant Diagnostics webpage to permit changing how the implant behaves when searching for a system to join to conserve battery life.

Enhancement

42426

1.31.190

The NTP Configuration file was updated to pass log information into the CLC system logs.

Enhancement

42349

1.31.190

The NTP Configuration file was updated reduce the polling time from 64 seconds to 8 seconds to more quickly react to time synchronization changes.

Enhancement

38257

1.30.80

Added support for large animal glucose implant channels.

Enhancement

39288

1.30.80

Disabled CLC debug logging to the CLC SD Card by default. When using Ponemah v6.41+ or Ponemah v5.20-SP11+ (i.e. DSI-Link2.0), these logs are saved to the acquisition computer.

Enhancement

39619

1.30.80

Fixed data synchronization gap (~15 samples) between implants on the same CLC after signal dropout.

Enhancement

39635

1.30.80

Added PhysioTel Digital Support for additional geographic regions.

Enhancement

39710

1.30.80

Added the GetStatus implant command to the Diagnositics webpage to permit Implant operation mode to be obtained while in Diagnostics.

Enhancement

40185

1.30.80

Support up to 500Hz hardware sample rate on some models (e.g. L03/L04).

Enhancement

40403

1.30.80

Implant data samples will be averaged over 1 seconds if channel sample rate is set to 1Hz in the software.

Enhancement

40887

1.30.80

Increased the frequency the Implant will attempt to communicate with a CLC for better responsiveness in obtaining status updates.

Enhancement

37840

1.29.206

Improving process for reconnecting with implants that were out of range, and have come back into range. Will result in a quicker re-connection.

Enhancement

38676

1.29.206

Change the CLC firmware revision scheme to include a unique build number. (Releases changing from 0.1.28, to 1.29.xxx)

Enhancement

39292

1.29.206

CLC will now blink the yellow TRX LED if there is an error with the TRX, or if there is a CRC error between the communication of the CLC and TRX.

Enhancement

39401

1.29.206

CLC can get stuck in a loop if implant is out of range while doing scheduled sampling. It would not be possible for the CLC to every acquire data from that implant again even if it was brought back into range.

Fix

39514

1.29.206

Fixes a bug when using new CLC FW versions with the older (SP8, SP6) configuration or wizard process. It was not possible to proceed with the wizard if changing the CLC frequency.

Fix

39555

1.29.206

TRX Power can be adjusted by EU & CH Users

Enhancement

37813

0.1.28

Fix a bug within CLC to show the correct status of the implant in Ponemah 6.x

Fix

37512

0.1.28

Fix issue of the CLC reporting an impossibly high battery voltage to DSILink

Fix

36961

0.1.28

Improve the shutdown/reboot process of the CLC to prevent lockups

Fix

37023

0.1.28

CLC Logs can now be obtained by a button on the System Status page of the CLC Webpage

Enhancement

38303,38305

0.1.28

Decreasing amount of data written to SD card, and adding ability to stop writing data to SD card by checkbox on CLC Settings Page

Fix

37941,37940

0.1.28

Fixes a data loss bug within the CLC dealing with reading/writing data from memory

Fix

36810

0.1.28

CLC will no longer support Listen Before Talk in EU/Japan while in Assessement mode. This will prevent the TRX reporting an error of an unrecognizable command

Fix

36738

0.1.28

Update to support the Frame Beacon Lock to prevent implants from listening to beacons from adjacent systems

Enhancement

36812

0.1.28

CLC will notify DSILink of an implant reset as a reset instead of an error. CLC will also record what caused the reset in its logs, and send to DSILink as well

Fix

35660

0.1.28

TRX Power can be adjusted by EU only Users

Enhancement

34885

0.1.28

PhysioTel Digital China support

Enhancement

34887,34888,35279,

35156,38306,38322

0.1.28

Updates to support 6 implants per CLC

Enhancement

386677, 38381, 37195

0.1.28

Updates to improve user experience with DSILink 1.9

Enhancement

34470

0.1.28

Fix for uploading TRX firmware

Fix

35270

0.1.27

Fix for implant channel signal misalignment when sampled using different sampling rates.

Fix

34804

0.1.26

Fixed an issue that caused Ponemah v5.xx and v6.00 to crash when launching the PhysioTel Digital configuration if a CLC with firmware 0.1.25 was on the same network as the acquisition computer.

Fix

34355, 34470, 33515

0.1.26

Fixes to reduce the amount of data loss seen by the system.

Fix

34083

0.1.26

Improved rare issue that prevented the CLC from being able to boot up properly after upgrading CLC firmware.