Problem savannah SNF 2010-11-28 18:16:53: Chamber is flaking heavily

i vacuumed the chamber and gave it a thorough wipe down to reduce (and
hopefully for the meantime) eliminate the flakes. the chamber will be bead
blasted to remove film build up over the christmas holidays.
to clean the system you can always wipe it down with clean-wipes (be careful
it is hot), or vacuum it the same way you would with innotec (clean wipes
covering the nozzle of the house vacuum).
j
On Sun, Nov 28, 2010 at 6:16 PM, <jennyhu at snf.stanford.edu> wrote:
> I'm not sure how to clean it, but the chamber was flaking very heavily
> before my run. I'm worried that these particles may be falling onto samples.
>>-------------- next part --------------
An HTML attachment was scrubbed...
URL: <http://snf.stanford.edu/pipermail/savannah/attachments/20101129/07458359/attachment.html>