Functions available with DINITOOLS utility- Configuration, data entry and print formatting.- Recording and filing of all the information on PC.- Complete retrieval of the configuration.- Instrument calibration with numeric values.- Customization of the screens.

24-bit A/D converter, 4 channels, up to 3200 conv./sec. and up to 8 signal linearisation points.

4 independent analogue channels (up to 2 in the E-AF03 version) for reading of the weight.

Connection with up to 16 analogue load cells of 350 Ohm (45 load cells of 1000 Ohm) and with the main digital load cells available on the market (mod. E-AF03).

Up to 10.000e OIML or 3 x 3000e @ 0,3 µV/d CE-M legal for trade.

Up to 1.000.000 displayable divisions for internal factory use, with internal resolution up to 3.000.000 points.

Connectable to printers and labellers with free programmation of the printout formats.

PC software for the quick compiling of the databases. For 3590Exx and CPWExx instruments.

Software for the customization of the TOUCH screen

Code

Description

DDT

Software for the quick customization of the touch screen display. For 3590ET and 3590EGT series indicators.

External battery

Code

Description

BP6V12A

IP67 stainless steel external battery pack, 6V 12Ah. TAKE NOTE: this battery can be directly recharged from the indicator's motherboard, when the indicator is connected to the main line (230Vac)(recharge time about 12h with turned off indicator). Otherwise, it's possible to recharge the battery by using the BP6CBE recharger (recharge time about 8h).

Built-in USB port kit for data storing on USB key. To be coupled with 5,7 inches touch screen weight indicators. It does not occupy serial ports. Fitted with USB key for PC. TAKE NOTE: this interface is not available on its own.

(S1) Thermal printer for mounting attached to 3590EGT indicator. IP65 stainless steel case and a cover door for protection from sprays and dust, for harsh or strict hygiene environments. Power supply from weight indicator.

(S1) Built-in analog output, 0-4/20 mA, 0-10VdC (cable gland excluded). For DFW, 3590 and CPW indicators. NOTE: For 3590, only with IOEXPB optional board. TAKE NOTE: this interface is not available on its own.

(S1) Built-in board with 12 OUT and 6 IN (mounting and cable gland excluded). It includes expansion circuit of the power supply range of the indicator (8-40Vdc). NOT COMPATIBLE WITH 3590E4CH AND TPRXT. With UL marked PCB, Terminal block, plastic box and labels (if expected).