SF6 LEAKCHECK P1:p

Based on the globally successful SF6 GASCHECK P1, the new SF6 LEAKCHECK P1:p was developed for a more mobile and convenient on-site leak detection. The portable version of the Allround Leak Detector is characterised by a high degree of reliability and measurement certainty, while at the same time being easy to use. The dimensions of the leak detector have been designed from the outset to comply with current hand luggage regulations.

For regular measuring operation, the only thing you need to do is switch the instrument on and off. Since the operator primarily does not need to measure a leak, but first to search and find it, P1:p automatically starts up in "search mode". In this mode, when the sniffer probe approaches a leak, the audio signal and optional vibration alarm as well as the analogue pointer all indicate full scale, signaling very clearly that an unknown leak has been found. Once spotted, the operator switches to "measurement mode" by pressing a key on the handset. The leak rate is now displayed in analogue and digital form in the desired unit of measurement, and a clear pass/fail indication based on the selected alarm threshold is given.

SmartSensorAll parts of the system subject to wear and tear are integrated in the plug-in SmartSensor. It can be replaced within a few seconds. As a result, maintenance intervals are longer and the availability of the P1:p is significantly increased. The 3 sensors included in the scope of delivery ensure that the P1:p is permanently ready for use.

Peli Protector Case For outstanding ruggedness, the instrument is built into a waterproof and unbreakable Peli Protector Case. The case is equipped with an automatic ventilation valve to accomodate to air pressure changes, e.g. during air transport. Its surface is resistant to corrosion, dents and scratches.

Unlimited OperationThe P1:p is suitable for worldwide use due to its wide range power supply from 100 to 240V - 50/60 Hz. The device also works for approx. 10 hours in mains independent battery operation. The charging process does not require monitoring, and the GASCHECK P1:p may by design remain connected to mains permanently. Where applicable, a standby mode can be selected to partially power down the P1:p when the handset is deposited for an adjustable period of time, saving both SmartSensor and battery life time. Return to normal operation takes place automatically when the hand unit is picked up again.

SF6 LEAKCHECK P1:p - Features

Portable version in waterproof and unbreakable Peli Protector Case

Maximum availability due to exchangeable SmartSensors

Highly sensitive 1.0E-8 mbar l/s and 0.1 ppm for SF6

Very short response and recovery times

No warm-up or run-in time required

High operating comfort

Immune against 100% SF6 concentration

Parameterization protected by passwords in several levels

Convenient input of parameters via LCD touch screen

World's top-selling SF6 leak detector

Technical Data Overview

Detector

NIC

Sensitivity

1.0E-7 ml/s - 1.0 ppm / Optional HIGHsens 1.0E-8 ml/s - 0.1 ppm

Response time t90

approx. 0.5 s

Recovery time t10

approx. 0.5 s

Alarms

Sound output, status LEDs, vibration alarm in the handset

Display

Handset analogue, base unit digital, plain text display

Self-diagnostics

Intake volume, sensor life, battery status, hardware error

Battery

Integrated rechargeable battery with automatic charge control

Storage

-10°C.... 60°C

Operating temperature

0°C.... 50°C

Operating voltage

100-240V 50/60Hz

Dimensions

Console 486 x 398 x 194 mm

(max. flight hand luggage 550 x 400 x 230 mm)

Weight

Console 12,5 kg

Hand Unit 563 g

SF6 LEAKCHECK P1:p - Applications

Mobile, highly sensitive leak detection and measurement of SF6 insulated switchgear in the field

Leak-tightness measurement and leak detection of SF6 filled components during production