Uses:Portable Tensile Tester combines high-low temperature tester with
peel strength tester wonderfully. It can restrain the error brought
by changing environment between testing process. Setting up
different fixtures can test 180&90 degree peeling strength,
separating force, ect. for the adhesive material in the high-low
temperature environment. This machine can be changed into
programmable temperature&humidity tester according to your
needs.

Character:1. The structure is made of the paint-coated aluminium blanking
plate. The interior is used the high-accuracy, low-resistance and
zero clearance of two ball screw and oriented pole which improve
loading efficience and structure rigidity.2. Use Panasonic servo motor which ensure high efficiency, steady
transmission and low noise. The accuracy of speed can be controlled
in 2%.3. That Using the business computer as the main control machine
plus the special testing software of our company can conduct all of
testing parameter, work state, collecting data&analysis, result
display and printing output.4. Have steady performance, high accuracy, powerful software
function and easy operation.5. Use double insulated glass, user-friendly observing test; Adopt
stainless steel frame and bladder, not only beauty but corrosion
resistance.

Specifications:

Capacity choice

2,5,10,20,50,100,200,500,1000,2000,5000kg

Choosing capacity

3 to 10 times of the most value about the specimen damage

Operation method

Windows operation

Resolution

1/2500,000

Accuracy

≤0.5%

Testing speed

1~500mm/min

Stroke

400,500,600mm(Height can be customized)

Test width

40cm(can be changed by customer needs)

Test space

(L×W×H)40×40×70cm

Motor

Panasonic servo motor

Kind of drive

high-precision ball screw

Dimension

(W×D×H)1200×600×1500mm

Weight

200kg

Power

1φ,220V/50Hz

Rate of work

0.4KW(depend on motor model)

Clamp

customize by customer needs(tensile and compressive clamp)

Temperature control system:

Temperature range

-40℃～150℃

Temperature method

computer P.I.D. automatic control

Humidity range

20%RH~98%RH

Stability of control

±2.5%RH,±1℃

Accuracy of distribution

±3%RH,±1℃

Temperature&humidity control

Korea TEMI880

Circulatory System

Wind forced circulation

Heating System

SUS#304, stainless steel heater

Humidifying System

Surface evaporation, a transparent structure.

Cooling system

Fan cooling mode, high efficiency of all sealed compressor

Dehumidifying System

Freezing latent heat dehumidification mode

Internal&external material

SUS#304

Inner dimension:

(L×W×H)40×40×70cm

Safety Device:

Stroke protection

Machinery, computer double protection, prevent over preset

Force protection

system setting

Emergency stop device

Handling emergencies

Software Functions:1. Use windows working platform, set all parameter with dialog
forms and operate easy;2. Using a single screen operation, do not need to change the
screen;3. Have simplified Chinese, traditional Chinese and English three
languages,switch conveniently;4. Plan test sheet mode freely;5. Test data can be directly appeared in the screen;6. Compare multiple curve data through translation or contrast
ways;6. With many units of measurement, the metric system and british
system can switch;7. Have automatic calibration function;8. Have user-defined test method function9. Have test data arithmetic analysis function11. Have the function of automatic magnification, to achieve the
most appropriate size of graphics;12. Can test tensile, compression, peel, fatigue and so on.