Go to pcfs terminate the e remote control and restart if cannot connect.

End message, if problem exists, please mention/describe the error.

Handover Issues

R4892:

Handover note created without experiment name.

“Experiment started” information not provided.

Data volume at beginning not specified.

CRD102:

Ho: 1307UT - fmset puts the field system time 6.2 seconds ahead of the Mark5 and Computer time, this seems to result in scans ending (and starting) ~5 seconds ahead of schedule (when going off the log time). We are in the last 50 mins of the experiment so I am going to leave as is rather than spend 25 mins restarting machines and end up losing more data than would be (potentially) saved. (Tiege)

on going issue. Jamie will solve it tomorrow. Get the 26m to log the time difference

R1894:

Yg: pcfsyg disk filled up 15 minutes before experiment started. Needed to remove some old epi/png files from the auto_corr_test directory. (Tiege)

Cronjob to clean up directory everyday

New machine coming in a few months with new system.

Might still need to use the old system.

AOV035:

Ho: 1725UT - Cryo monitoring system not working at Ho (ERROR ma -4 Device rx timed-out on response). I have replaced the cryomon and rxmon procedures with nops in EXPERIMENT procedure library for the time being, because otherwise it waits for each rx measurement to time out. I have also commented out the rxmon command in the midob procedure in STATION library, as that is also waiting for timeout, and screwing up scan timing. (Patrick)

Fixed by resetting TTY distributor with “reset” switch on old S/X MarkIV rack. Serial communications from Hobart to s/x receiver are still patched through this rack. (Warren and Brett)