The R&S®NRP18S three-path diode power sensor is a self-contained, fully characterized instrument.

Specifications

NRPxxS/SN sensors

Frequency range

R&S®NRP8S/SN, N (m)

10 MHz to 8 GHz

R&S®NRP18S/SN, N (m)

10 MHz to 18 GHz

R&S®NRP33S/SN, 3.5mm (m)

10 MHz to 33 GHz

R&S®NRP40S/SN, 2.92mm (m)

50 MHz to 40 GHz

R&S®NRP50S/SN, 2.4mm (m)

50 MHz to 50 GHz

Key facts

10 000 triggered measurements/s

More than 50 000 readings/s

Unprecedented measurement speed and accuracy even at low levels

Ideal for universal applications

Detachable cables for flexible operation

93 dB dynamic range thanks to innovative three-path concept

Built-in trigger I/O port

Brief description

The R&S®NRP18S three-path diode power sensor is a self-contained, fully characterized instrument.

They can be operated with the R&S®NRP2 base unit, with a laptop/ PC via USB, and with many Rohde & Schwarz instruments (e.g. signal generators, signal and spectrum analyzers, network analyzers).

Specifications

Three-path diode power sensors

Frequency range

Powermeasurementrange

Impedance matching (SWR)

Rise timevideo bandwidth

Uncertainty forpower measure-ments at+20 °C to +25 °Cabsolute (in dB)

Uncertainty forpower measure-ments at+20 °C to +25 °Crelative (in dB)

R&S®NRP8S/SN, N (m)

10 MHz to8 GHz

100 pW to200 mW(–70 dBm to+23 dBm)

10 MHz to2.4 GHz:

< 1.13

< 5 µs > 100 kHz

0.053 to 0.065

0.022 to 0.050

>2.4 GHz to8.0 GHz:

< 1.20

R&S®NRP18S/SN, N (m)

10 MHz to18 GHz

100 pW to200 mW (–70 dBm to+23 dBm)

10 MHz to2.4 GHz:

< 1.13

< 5 µs > 100 kHz

0.053 to 0.094

0.022 to 0.069

>2.4 GHz to8.0 GHz:

< 1.20

> 8.0 GHz to18.0 GHz:

< 1.25

R&S®NRP33S/SN, 3.5 mm

10 MHz to33 GHz

100 pW to200 mW (–70 dBm to+23 dBm)

10 MHz to2.4 GHz:

< 1.13

< 5 µs > 100 kHz

0.053 to 0.134

0.022 to 0.136

>2.4 GHz to8.0 GHz:

< 1.20

> 8.0 GHz to18.0 GHz:

< 1.25

> 18.0 GHz to26.5 GHz:

< 1.30

> 26.5 GHz to33.0 GHz:

< 1.35

R&S®NRP40S/SN, 2.92 mm

50 MHz to40 GHz

100 pW to100 mW (–70 dBm to+20 dBm)

50 MHz to2.4 GHz:

< 1.13

< 5 µs > 100 kHz

0.073 to 0.138

0.028 to 0.142

>2.4 GHz to8.0 GHz:

< 1.20

> 8.0 GHz to18.0 GHz:

< 1.25

> 18.0 GHz to26.5 GHz:

< 1.30

> 26.5 GHz to33.0 GHz:

< 1.35

> 33 GHz to40.0 GHz:

< 1.37

R&S®NRP50S/SN, 2.4 mm

50 MHz to50 GHz

100 pW to100 mW (–70 dBm to+20 dBm)

50 MHz to2.4 GHz:

< 1.13

< 5 µs > 100 kHz

0.073 to 0.138

0.028 to 0.184

>2.4 GHz to8.0 GHz:

< 1.20

> 8.0 GHz to18.0 GHz:

< 1.25

> 18.0 GHz to26.5 GHz:

< 1.30

> 26.5 GHz to33.0 GHz:

< 1.35

> 33 GHz to40.0 GHz:

< 1.37

> 40 GHz to50.0 GHz:

< 1.40

High-power three-path diode power sensor

Frequency range

Powermeasurementrange

Impedance matching (SWR)

Rise timevideo bandwidth

Uncertainty forpower measure-ments at+20 °C to +25 °Cabsolute (in dB)

Uncertainty forpower measure-ments at+20 °C to +25 °Crelative (in dB)

R&S®NRP18S-10, N (m)

10 MHz to18 GHz

1 nW to2 W(–60 dBm to+33 dBm)

10 MHz to2.4 GHz:

< 1.14

< 5 µs > 100 kHz

0.083 to 0.198

0.022 to 0.087

>2.4 GHz to8.0 GHz:

< 1.20

> 8.0 GHz to12.4 GHz:

< 1.25

> 12.4 GHz to18.0 GHz:

< 1.30

R&S®NRP18S-20, N (m)

10 MHz to18 GHz

10 nW to15 W (–50 dBm to+42 dBm)

10 MHz to2.4 GHz:

< 1.14

< 5 µs > 100 kHz

0.083 to 0.198

0.022 to 0.087

>2.4 GHz to8.0 GHz:

< 1.25

> 8.0 GHz to12.4 GHz:

< 1.30

> 12.4 GHz to18.0 GHz:

< 1.41

R&S®NRP18S-25, N (m)

10 MHz to18 GHz

30 nW to30 W (–45 dBm to+45 dBm)

10 MHz to2.4 GHz:

< 1.14

< 5 µs > 100 kHz

0.083 to 0.198

0.022 to 0.087

>2.4 GHz to8.0 GHz:

< 1.25

> 8.0 GHz to12.4 GHz:

< 1.30

> 12.4 GHz to18.0 GHz:

< 1.41

TVAC-compliant three-path diode power sensor

Frequency range

Powermeasurementrange

Impedance matching (SWR)

Rise timevideo bandwidth

Uncertainty forpower measure-ments at+20 °C to +25 °Cabsolute (in dB)

Uncertainty forpower measure-ments at+20 °C to +25 °Crelative (in dB)

R&S®NRP33SN-V, 3.5 mm

10 MHz to33 GHz

100 pW to200 mW (–70 dBm to+23 dBm)

10 MHz to2.4 GHz:

< 1.13

< 5 µs > 100 kHz

0.053 to 0.134

0.022 to 0.136

>2.4 GHz to8.0 GHz:

< 1.20

> 8.0 GHz to18.0 GHz:

< 1.25

> 18.0 GHz to26.5 GHz:

< 1.30

> 26.5 GHz to33.0 GHz:

< 1.35

Thank you.

Your request was successfully sent!

We will get in touch with you soon.

Your Rohde & Schwarz Team

Get a Demo

Inquiry*

Please enter your inquiry.

First Name*

Please enter your first name.

Last Name*

Please enter your name.

Email*

Please enter a valid email address.

Company*

Please enter your company name.

City*

Please enter your city name.

ZIP*

Please enter your postal code.

Location*

Please enter your location.

Marketing permission

I want to receive information from Rohde & Schwarz via

Email

Post

What does this mean in detail?

I agree that Rohde & Schwarz GmbH & Co. KG and the Rohde & Schwarz entity or subsidiary company mentioned in the imprint of this website, may contact me via the chosen channel (email or postal mail) for marketing and advertising purposes (e.g. information on special offers and discount promotions) related to, but not limited to, products and solutions in the fields of test and measurement, secure communications, monitoring and network testing, broadcast and media, and cybersecurity.

Your rights

This declaration of consent may be withdrawn at any time by sending an email with the subject "Unsubscribe" to news@rohde-schwarz.com. Additionally, a link to unsubscribe from future email advertisements is contained in each email sent. Further details on the use of personal data and the withdrawal procedure are set out in the Statement of Privacy.

Please enter the characters shown in the image

The input was not correct.

Email confirmation

Do not enter data in this field

Thank you.

Your request was successfully sent!

We will get in touch with you soon.

Your Rohde & Schwarz Team

Get a Quote

Inquiry*

Please enter your inquiry.

First Name*

Please enter your first name.

Last Name*

Please enter your name.

Email*

Please enter a valid email address.

Company*

Please enter your company name.

City*

Please enter your city name.

ZIP*

Please enter your postal code.

Phone*

Please enter your phone number.

Location*

Please enter your location.

Marketing permission

I want to receive information from Rohde & Schwarz via

Email

Post

What does this mean in detail?

I agree that Rohde & Schwarz GmbH & Co. KG and the Rohde & Schwarz entity or subsidiary company mentioned in the imprint of this website, may contact me via the chosen channel (email or postal mail) for marketing and advertising purposes (e.g. information on special offers and discount promotions) related to, but not limited to, products and solutions in the fields of test and measurement, secure communications, monitoring and network testing, broadcast and media, and cybersecurity.

Your rights

This declaration of consent may be withdrawn at any time by sending an email with the subject "Unsubscribe" to news@rohde-schwarz.com. Additionally, a link to unsubscribe from future email advertisements is contained in each email sent. Further details on the use of personal data and the withdrawal procedure are set out in the Statement of Privacy.

Please enter the characters shown in the image

The input was not correct.

Email confirmation

Do not enter data in this field

Thank you.

Your request was successfully sent!

We will get in touch with you soon.

Your Rohde & Schwarz Team

Request information

Inquiry*

Please enter your inquiry.

First Name*

Please enter your first name.

Last Name*

Please enter your name.

Email*

Please enter a valid email address.

Company*

Please enter your company name.

City*

Please enter your city name.

ZIP*

Please enter your postal code.

Location*

Please enter your location.

Marketing permission

I want to receive information from Rohde & Schwarz via

Email

Post

What does this mean in detail?

I agree that Rohde & Schwarz GmbH & Co. KG and the Rohde & Schwarz entity or subsidiary company mentioned in the imprint of this website, may contact me via the chosen channel (email or postal mail) for marketing and advertising purposes (e.g. information on special offers and discount promotions) related to, but not limited to, products and solutions in the fields of test and measurement, secure communications, monitoring and network testing, broadcast and media, and cybersecurity.

Your rights

This declaration of consent may be withdrawn at any time by sending an email with the subject "Unsubscribe" to news@rohde-schwarz.com. Additionally, a link to unsubscribe from future email advertisements is contained in each email sent. Further details on the use of personal data and the withdrawal procedure are set out in the Statement of Privacy.