3 kN Flexure Fixture, Fatigue 3 Point

Catalog no.2810-600/605

Suitable for both low force dynamic and static testing along with dental standards ISO 9917, ISO 6872, ISO 4049 or ISO 20795 the low-force 3-Point bend fixture is designed to be compatible with ElectroPuls E1000. The specimen is supported on two lower anvils and the load is applied in the center of the specimen by a single upper anvil. The t-slot base allows adjustment of the lower span from 15 - 80mm (0.59 - 3.15in) allowing for variable specimen lengths and is submersible to be compatible with an optional fluid bath and recirculator.

Optional Accessories

4-Point Bend Fixture Conversion Kit (2810-605): Used to convert 2810-600 from a 3-point to a 4-point bend fixture, the 4-point bend conversion kit has a variable span of 5 - 40mm (0.19-1.57in) and is suitable for both static and dynamic testing. The test specimen is supported on two lower anvils while the load is evenly applied by two upper anvils.

The 3-Point bend fixture is designed to easily provide the capability of a 3-point flexure test; it is suitable for both Dynamic & Static testing. The specimen is supported on two lower anvils and the load is applied in the centre of the specimen by a single upper anvil. The fixture has a variable span from 15 – 80 mm (0.59 – 3.15 in).

Please choose a subscription optionOne or more required fields are empty, complete the form to continue.One or more fields contain formatting errors, fix the errors to continue.You must agree to Instron's Privacy Policy in order to continue.

“It is our objective at Instron to provide our customers the best ownership experience by delivering the highest quality products, expert support and world-class service.”