Product overview: Fluke 787B ProcessMeter™

The Fluke 787B ProcessMeter™ doubles troubleshooting capabilities by combining the power of a safety rated digital multimeter and mA loop calibrator into a single, compact test tool. Based on the trusted measurement capabilities of the Fluke 87 DMM, the 787B adds the ability to measure, source and simulate mA with the accuracy and resolution you would expect from a Fluke mA loop calibrator, giving you the ideal tool for troubleshooting and calibrating current loop applications. With Fluke Connect® mobile app and desktop software compatibility technicians can wirelessly monitor, log, and share data from the field with their team anytime, from anywhere*.

*Requires Fluke IR3000FC module (not included). Not all models are available in all countries. Check with your local Fluke representative.

Specifications: Fluke 787B ProcessMeter™

Measurement function

Range and resolution

Best accuracy (% of reading + LSD)

V dc

400.0 mV, 4.000 V, 40.00 V, 400.0 V, 1000 V

0.1% + 1

V ac (true-rms)

400.0 mV, 4.000 V, 40.00 V, 400.0 V, 1000 V

0.7% + 2

mA dc

30.000 mA

.05% + 2

A dc

1.000 A (0.440 A continuous)

0.2% + 2

A ac

1.000 A (0.440 A continuous)

1% + 2

Resistance

400.0 Ohms, 4.000 k, 40.00 k, 400.0 k, 4.0 M, 40 M

0.2% + 1

Frequency (0.5 Hz to 20 kHz)

199.99 Hz, 1999.9 Hz, 19.999 kHz

.005% + 1

Diode test

2.000 V (shows diode voltage drop)

2% + 1

Continuity

Beeps for resistance < approx. 100 ohms

Output function

Range and resolution

Drive capability

Accuracy (% of span)

DC current output (Internal battery operation)

0.000 to 20.000 mA or 4.000 to 20.000 mA, (selectable at power-up) Over-range to 24.000 mA

24 V compliance or, 1,200 ohms, @ 20 mA

0.05%

DC current simulate (Ext. 15 V to 48 V loop supply)

0.000 to 20.000 mA or 4.000 to 20.000 mA, (selectable at power-up) Over-range to 24.000 mA