FD-A1/A10/A10AR/A50/A50AR/A100A/100AR: ±0.5 % of F.S. or lessFD-A250/A600: ±1 % of F.S. or less(at the operating ambient temperature of 25 °C)

Rating

Power voltage

12 to 24 VDC, Ripple (P-P) 10 % or less

Environmental resistance

Ambient temperature

0 to +50 °C 32 to 122 °F (No freezing)

Relative humidity

35 to 85 % RH (No condensation)

Vibration resistance

10 to 55 Hz, Double amplitude 1.5 mm 0.06", 3 hours in each of the X, Y, and Z axis

Material

Housing material: Polycarbonate, Key top: Silicon rubber

Accessories

Mounting bracket, Power cable, Instruction manual

Weight

Approx. 85 g (including power cable)

*1 The amplifiers, whose model names have "A" or "AP" in the end, are upgraded ones along with the supplementation of a new head. The old type of amplifiers, whose model names do not have "A" or "AP", cannot be used with the sensor heads other than FD-A10/A50/A100.