pressure sensor. The in-built process circuit transform mV signal to standard voltage and current output, which is suitable for using with computer, control equipment and display unit. The high stable pressure sensor and high performance circuit ensure PT-VP can transmit signal from remote distance.

Features

■OEM service

■All stainless steel integrated design with easy installation features, suitable for long term use in hazardous environment

■Various electrical connectors and thread types for selection

■Wide measurement range: -100kPa…0~35kPa…100MPa

■Anti-vibration, anti-impact, and anti-corrosive

■High interference with long term stability

■Zero and span field calibration

Main Specifications

▼Pressure connection

M20 x 1.5

G1/2”

M14x1.5

G1/4”(male)

G1/4”(female)

1/2”NPT

1/4”NPT

1/4”PT

Material: SS304

▼Electrical connector

Hirschmann connector /

Hirschmann with cable outlet /

Cable outlet / Packard /

BP3 aluminum case /

Aviation plug

▼Sealing O-ring

NBR

▼Pressure range

-100kPa…0~10kPa…100MPa

▼Pressure type

Gauge pressure /

Sealed pressure /

Absolute pressure /

Compound pressure

▼Applicable fluid

Liquid and gas that compatible with SS304

▼Ambient temperature

-20 to 85℃

▼Storage temperature

-40 to 125℃

▼Vibration resistance test

20g (20-5000Hz)

▼Impact resistance test

100g (11ms)

▼Protection level

IP 54 (Hirschmann connector)

IP 65 (Packard, aluminum case)

IP 67 (Aviation plug, cable outlet)

▼Explosion-proof class (option)

Exia II CT6

Other Specifications

▼Case

Stainless steel

▼Power supply

24VDC / 12VDC

▼Output signal

4~20mA

1~5V

0~5V

0~10V

0.5~4.5V

▼Measuring elements

Diffusion silicone chip

Transducer

Material: SS316L

▼Zero & span drift

±1.5% F.S. max (@-10~70℃)

▼Accuracy

0.5% F.S. (standard) /

0.3% F.S. (option) /

0.1% F.S (option)

▼Stability

±0.2% F.S. /year

▼Response time

≤1ms

(pressure increase to 90%F.S.)

▼Insulation resistance

100MΩ, 250VDC

▼Temperature compensation

-10 to 70℃ /

-20 to 85℃ (option)

▼Overpressure

1.5*F.S.

▼EMC surge

(IEC61000-4-5) 2KV

▼EMC discharge

(IEC61000-4-2)

Contact discharge 8KV

Air discharge 15KV

Supplementary Product Illustration:

PT-VP.pdf

Competitive Advantages:

Well and High Quality Control,Prompt Delivery,New Product,Customer's Design and Logo are Welcome,Competitive Prices,Earth-friendly Products,In a Variety of Design,Small Order Acceptable,ODM Accepted,OEM Accepted