The ACCU-flo/NP is a combination precision nozzle and multi-point, self-averaging Pitot airflow measurement station. When combined with an ultra high accuracy transmitter, such as Air Monitor’s VELTRON II< MASS-tron II or VELTRON DPT-plus, the ACCU-flo/NP provides high accuracy (±0.5%) flow measurement for process monitoring and control applications. The combined effect of the integral flow straightener

This all-stainless steel combination air equalizer/straightener with self-averaging Pitot tube traverse station provides highly accurate measurement of low air volumes ranging from 2-2000 CFM. It is capable of operating up to 1000°F in a variety of applications, including applications with corrosive gases, within the limitations of the 326 stainless steel construction.

The VOLU-probe/SS Stainless Steel Airflow Traverse Probe is ideal for clean or harsh and particulate laden applications operating at temperatures ranging from -20 – 900°F. It consists of multiple Pitot total and static pressure sensing ports positioned along the length of each probe to traverse the duct cross-section. They average the sensed pressures in their