Unique Features of Cyclops 390L range of thermometers

The unique features of the Cyclops 390L portable non-contact thermometer make it the ideal instrument for accurate non contact temperature measurements in hydrocarbon-fuelled furnaces. Operating at a 3.9µm (a transmissive waveband), for hot targets, water vapour and CO2are invisible.

Details

The Land Cyclops L family of high quality portable non-contact thermometers provides precision spot temperature measurement with unmatched accuracy and reliability, for a wide range of industrial applications. Features include precise view of target spot with simultaneous digital display of temperature in the viewfinder, choice of operating and calculating modes, digital output and out of range alarms, Bluetooth® serial communications and optional data logger software on Mobile Phone, PC/Laptop/Tablet.

The Cyclops 390L provides both a USB port and convenient Bluetooth® wireless technology. The measured temperature is displayed in four simultaneous modes: continuous, peak, average and valley. Accurate sighting is ensured by the clear, wide angle (9°) field of view and small, clearly defined (1/3°) square measurement area. Focusing is variable from 1m/3ft to infinity. The operating waveband (3.9µm) has been carefully chosen as a narrow spectral response centred at 3.9µm eliminates errors caused by absorption/emission bands present in combustion gases

Options

Includes new Data Logger software for both PC and Mobile Devices including the unique Route Management utility

UKAS Certified instrument calibration

Data Output Modes

Single mode: logs a measurement for each trigger release

Latch mode: logs continuous, average, peak and valley readings from the first trigger release until the second

Burst mode: logs a stream of measurements whilst the trigger is held pressed - approximately 30 readings / sec, up to a maximum of 999 readings per burst.

Route mode: This mode is ideal for users who need to take a number of readings on a regular tour of their plant or process. The pre-loaded locations help to guide the user along each route

Applications:

Steel Applications

Stock temperatures in the preheat and heating zones of reheat furnaces.

Unique Features of Cyclops 390L range of thermometers

The unique features of the Cyclops 390L portable non-contact thermometer make it the ideal instrument for accurate non contact temperature measurements in hydrocarbon-fuelled furnaces. Operating at a 3.9µm (a transmissive waveband), for hot targets, water vapour and CO2are invisible.

Details

The Land Cyclops L family of high quality portable non-contact thermometers provides precision spot temperature measurement with unmatched accuracy and reliability, for a wide range of industrial applications. Features include precise view of target spot with simultaneous digital display of temperature in the viewfinder, choice of operating and calculating modes, digital output and out of range alarms, Bluetooth® serial communications and optional data logger software on Mobile Phone, PC/Laptop/Tablet.

The Cyclops 390L provides both a USB port and convenient Bluetooth® wireless technology. The measured temperature is displayed in four simultaneous modes: continuous, peak, average and valley. Accurate sighting is ensured by the clear, wide angle (9°) field of view and small, clearly defined (1/3°) square measurement area. Focusing is variable from 1m/3ft to infinity. The operating waveband (3.9µm) has been carefully chosen as a narrow spectral response centred at 3.9µm eliminates errors caused by absorption/emission bands present in combustion gases

Options

Includes new Data Logger software for both PC and Mobile Devices including the unique Route Management utility

UKAS Certified instrument calibration

Data Output Modes

Single mode: logs a measurement for each trigger release

Latch mode: logs continuous, average, peak and valley readings from the first trigger release until the second

Burst mode: logs a stream of measurements whilst the trigger is held pressed - approximately 30 readings / sec, up to a maximum of 999 readings per burst.

Route mode: This mode is ideal for users who need to take a number of readings on a regular tour of their plant or process. The pre-loaded locations help to guide the user along each route

Applications:

Steel Applications

Stock temperatures in the preheat and heating zones of reheat furnaces.

Petrochem Applications

Tube temperatures in tube furnaces (pyrolysis, reformer, cracking).

Temperatures

Low Temperature (C):

450

High Temperature (C):

1400

Low Temperature (F):

842

High Temperature (F):

2552

Technical Details

Measurement range:

450 to 1400°C/842 to 2532°F (Operating)

Indication:

4-digit LCD in viewfinder; external backlit LCD display

Measuring modes:

Continuous, Average, Peak, Valley

Data Storage

Internal storage of up to 9999 readings, including date & time stamp. Also 4 ‘routes’ of up to 99 readings per route

The Bluetooth word mark and logos are owned by the Bluetooth SIG, Inc. and any use of such marks by LAND is under license. Other trademarks and trade names are those of their respective owners.

Accessories for this instrument listed below

The thermometers and thermometer systems listed here are specially designed to solve temperature measurement problems in specific industrial applications. LAND are also willing to consider making 'customized' non-standard products, or even to develop a particular system from scratch to answer individual requirements. Talk to LAND about your own particular needs.