Please read this manual carefully before using the equipment and understand the precautions. The user assumes all risk in operating this device.

This device, is operated on batteries (DC 4.5V. 3 x AAA) as well as on regular voltage (AC 110-240 volt).

This electro sex stimulator sends electrical impulses to the electro parts/toys which can be connected to this power box and placed on any body part as long as the shortest distance between electrodes does not cross the heart region. This stimulation has a sexual stimulating origin but can also be used as a medical device to fight, incontinency, erection problems and muscle weakness in the erogenous zones. Please ask and consult your doctor first before you use this device for medical treatments.

The basic of Electro sex stimulation is to create uncontrollable contractions of the muscles in the intimate areas by controllable electrical current.This current will stimulate the nerves in the treated areas and this will lead to the muscle contractions. The stronger the current is, the tighter the contraction will be. Rimba created and developed a nice variety of different accessories to use on all Rimba Power Boxes. The electrodes have all different sizes and shapes to please both men and women.

This electro sex power box # 7890 has 4 channels which can be individually controlled over 15 different levels. Just use the + and – button. Each channel is bi-polar, so 4 different bipolar accessories can be used at the same time, each on the requested power level.

This # 7890 power box unit has 5 different stimulation /therapy programs. Your selection will be shown on the digital display by a matching graph. This program and will run on all 4 channels. If you change the mode, it will show on the display and run again for all 4 channels.

Electro Sex Stimulation is not for children and not a children’s toy! Use always your common sense and read this manual carefully.

- Do not use if you wear cardiac-pacemaker.- Do not use if you are pregnant.- Do not use on eyelids.- Do not use on broken skin.- Avoid abrupt changes in stimulation control settings.- Keep out of the reach of children.- Turn the powerbox off before applying, removing or changing the electrodes.- Stimulation of areas where bones are near the skin (knee, forehead or chin) can be painful for some patients.- Stimulation controls, especially the level of intensity should be adjusted slowly.- Do not immerse the stimulation-powerbox in water or other liquids.- Do not use the power adapter when broken or damaged- Keep out of the reach of children.

Side effects:

Skin irritation may occur beneath or around the electrode sites. If skin irritation persists discontinue use immediately.

Turn on the requested channel by pressing the “+” button to the First level. If wanted, please press “+” button again to go to the next level. There are 15 levels per channel. The current level is shown per channel individually on the display’s diagram.

It is possible to change the MODE while running therapy. The graph will change on the display.

In case of an emergency, use the ON/OFF button to stop therapy

After you have turned off the power box, the electrodes or pad scan be disconnected and taken from the body.

TIPS:

Clean the body part and the area around that you want to treat with electro therapy. The self adhesive pads will hold better and last much longer.

Emergency: press the ON/OFF button to stop treatments immediately.

Take the batteries out, when the power box is not used.

The better the contact the better the therapy. Contact gel will help you out.

Repeat all these actions before every use!

The key pads: (press buttons)The power box is equipped with following buttons:

ON / OFF button: to switch on and off.

Time + and - button: the requested time can be adjusted up and down in steps of 5 minutes. The count down will start at once. The maximum timer level is 30 minutes. If the count down reaches zero, the power box automatically switches off.

‘Mode’ buttons: The selected mode is indicated by a blue LED light.

Intensity buttons + and -. Each channel can be adjusted separately with the + and – buttons over the total of 15 levels.

LCD screen:The LCD screen will indicate which channel is active and on what level. Each bar is one level. The graph shows how the therapy runs and the timer shows how much time remains in the therapy session.

Safety:This power box is no children’s toy. If you are being monitored by or under treatment with a medic / doctor, please do consult your doctor for possible health risks before you start using this power box. If you suffer from tumors, infection, heart rhythm troubles, or skin irritations, please do consider this as risk.Do not use this digital sex power box in combination with other medical equipment or other medical supporting devices such as:

Pacemaker or similar cardio devices.

Prostheses and/or metal pins, screws, plates in your body.

All electronic devices needed and used to monitor, control or protect your body and health.

Never use electro sex accessories near heart, eyes, mouth or irritated/damaged skin.Do not use the power box if you are or trying to get pregnant.

Cleaning and Maintaining:Clean the body of the power box with a damp cloth and a small amount of soft soap. Gently rub the stains away. Never soak the device or use petroleum products or alcohol to clean.Clean electrodes (gel-pads) carefully with a damp cloth and dry them slowly. Keep them away from direct sunlight.Keep the power box and accessories in a safe and dry environment. Avoid extreme heath, cold, fire and liquids. Do not repair this power box device by yourself.Do not use any other power cord than the one enclosed.Take out the batteries when the power box unit is not used!

Product specifications :

Power supply

4.5V, 3 x AAA

Frequency

1- 100HZ

Power Consumption

Max. 540mw

Output Power

Min. 25uA, 1.25mW, Max. 100uA , 20 mW (into 2K 0hm load)

Puls Widthe

20-220 µS

Trouble shooting: Causes and Solutions:

Status of the product

Cause

Solutions

No LCD display after the batteries were installed.

Battery contacts may need cleaning.

Check and remove the batteries.

Batteries are used or polarity incorrect

Check the cause and remove or re-install.

The contact is bad between the batteries and the spring

Expand the anode spring by using a suitable tool.

No LCD display after power cord was connected

The power is not connected

Check again if plugs have been inserted correctly, if necessary use the travel adapter

There is no power

Power cord might be broken or damaged, please try battery operation

You are using a non original power cord

Please use the original supplied power cord

No feeling but the LCD display is in working

Plug inserted in the jack firmly?

Check and correct connection between electrode and chord, chord and jack.