Thermcore IR113 Thermal Imaging Module

Thermcore IR113Thermal Imaging Module

The IR113 thermal imaging module series are designed for original equipment manufacturer (OEM). They can be easily integrated into infrared systems or used with its standard interface. They produce 384 x 288 pixels high quality thermal images.

The thermal imaging module has option to use several lenses. With lightweight, small size and low power consumption design, it is an ideal choice for clients to make secondary development and system integration thanks to our flexible customization. Additionally it has plug-and-play OEM module.

Specifications of the Thermal Imaging Module

Thermcore IR113

Detector

Detector type

UFPA microbolometer, Asi

Pixels

384 × 288

Pitch

25μm

Spectral range

8~14μm

NETD

<80mk - 30℃

Thermal response time

7ms

Fill Factor

>80%

Bad Pixels

<1%

Image Presentation

Video output

PAL (50Hz) NTSC (60Hz)

Settings

Auto/Manual brightness & contrast adjustment

Digital zoom

2×, 4× interpolating

2× interpolating

2×, 4× interpolating

2×, 4× interpolating

Polarity

Black and white, black and white inverse and pseudo color optional

Black and white, black and white inverse

Black and white, black and white inverse

Black and white, black and white inverse

Video recording

N/A

N/A

Live video recording on board (optional)

N/A

Storage media

N/A

N/A

2GB SD card, extended to 4GB optional

N/A

Storage contents

N/A

N/A

Images and videos optional

N/A

Interfaces

Command and Control

RS232/RS485/RS422

Video output

RCA/BNC alternative

RCA/BNC alternative

RCA/BNC alternative

RCA/BNC alternative

USB2.0

Transfer digital live video to PC optional

N/A

transfer saved images and videos to PC optional

N/A

Power System

Power supply

6VDC- 12VDC (110-220VAC adapter)

7~10V DC (110- 220VAC adapter)

7~12V DC (110/220VAC adapter)

7~12V DC (110- 220VAC adapter)

Power Dissipation

<3.5W

≤3.5W

≤3.4W

≤2.5W

Environmental Specification

Operating temperature

-20℃ to 60℃ (-40℃ to 60℃ optional)

Storage temperature

-40℃ to 60℃

Physical Characteristics

Weight

75 g

65 g

90 g

70 g

Size

47.2 mm x 42.1 mm x 41 mm

65 mm x 37 mm x 9 mm

39 mm x 39 mm x 45 mm

39 mm x 39 mm x 40 mm

Specifications are subject to change without prior notice.
Described company name and model are brand name or registration of trademark.