AQ6319 Optical Spectrum Analyzer (DISCONTINUED)

The AQ6319 packs the power of cutting-edge technologies with many years of research and experience. It holds the best gratings based monochromator optical performance on the market and provides customers with a user-friendly interface for faster and more accurate measurement. In fact, the AQ6319 raises the standards in the optical testing industry.

With an unprecedented wavelength accuracy of ±10 pm (1520-1580 nm), the AQ6319 OSA has become an outstanding instrument in both general and specialized R&D labs for high accuracy wavelength measurement. Yokogawa keeps the "free-space" design between the incident beam and the monochrometer; this proven technology which is inherited from the world-renowned AQ6317 series OSAs offers much greater facility for researchers using specialty fibers, especially with large core diameters. The new AQ6319 offers a much faster sweep speed than any other models. It also features a LAN connection for faster and easier data transfer, as well as remote control capabilities. In addition, many new analysis functions have been included.

If you have products currently in China that require calibration or repair, you may either return them to your local Yokogawa service center or the Yokogawa China Service & Repair Center. Please be advised that ...

Level Accuracy:Level Accuracy is a tolerance to the true value of measured value, when a standard level is measured with a standard wavelength. Level Linearity:Level Linearity is the width of error dispersion ...

Wavelength accuracy: Wavelength accuracy is a tolerance to the true value of a measured value when the standard wavelength is measured. Wavelength repeatability:Wavelength repeatability is the stability of ...

The warning "142 WL calibration failed" is displayed when a level of light source is not enough at wavelength calibration or calibration cannot be executed because wavelength difference is out of calibration ...

If you are performing wavelength calibration using the instrument's built-in light source, the warning message is an indication that the monochromator needs to be repaired.
Please visit our Contact site to locate ...

The Chopped Light mode is used:
To increase measurement level sensitivity by cutting the high frequency noise
To detect only the light that the LS emitted, effective for free space measurement. In free space ...