FISCHERSCOPE UMP150

The UMP 150 is a high-precision ultrasound device designed for ultra-fine, non-destructive wall thickness measurement. It uses single-element probes and a high-frequency square wave generator (30 MHz), which together ensure an excellent resolution of 1 μm. Besides its wide measuring range of 0.1 to 500 mm, it also offers a variety of modes for highly sensitive measurements. The device can be upgraded simply and at any time: many software options and measurement modes can be enabled directly via the keypad.

Fischer’s ultrasonic line is ideal for measuring wall thicknesses in all aspects of corrosion protection.

Features:

Comes equipped with a 10 MHz probe with delay line; other probes optionally available

Measuring range from 0.1 - 500 mm

Bandwidth 0.5 - 30 MHz (-3 dB)

Settings for up to 30 different applications can be stored and retrieved

Zero-point compensation for reliable measurements even after frequent use and/or at fluctuating temperatures