Hydrogen sulphide levels (H2S) in landfill gas have increased and Geotech has responded to changes reported by landfill site operators by now offering analysers measuring H2S up to 10,000 ppm. Rapidly rising H2S levels are thought to be caused by waste-to-landfill content changing largely as a result of success in recycling and other initiatives to avoid sending waste to landfill. Previously, H2S was measured at 0-500ppm – still typical for post-scrubber H2S measurement – but higher level monitoring is now essential on many UK landfill sites.

Currently the two main methods of landfill gas measurement are either with portable or fixed-position gas analysers. Geotech reports strong interest in high-range H2S analysis at 0-10,000 ppm which is an option on its newly launched portable landfill analyser, the GA5000. This option is also available on the Geotech fixed-position, automated landfill and biogas gas analysers, the GA3000 and the Geotech AEMS system.

PurgeAs well as presenting a significant health risk to operators, H2S can also ‘poison’ the electrochemical cells which provide the most cost-effective solution for H2S monitoring. Higher H2S levels present a greater challenge. The cells must be regularly purged with clean air. Geotech’s portable analysers are easily purged manually after taking each reading and Geotech fixed position gas analysers purge automatically. Without purging, the life of the H2S cell is dramatically reduced and accuracy compromised. Some analyser systems dilute biogas with air in order to monitor H2S continuously but there is a significant danger of adding error with this approach and it may well not be enough to cope with new, high H2S levels.

DamageFor the many landfill operators using landfill gas to produce electricity from CHP engines, accurate H2S measurement is essential. To protect CHP engines, inbuilt gas content and quality sensors may cause auto-shut down to protect engines from corrosive H2S damage and CHP engine manufacturers usually specify an upper H2S limit above which warranties can be invalidated. Consequently, landfill operators often prefer to analyse the gas used in CHPs. If the gas is not suitable for CHP use and without an H2S scrubber, likely to cause damage with excessive H2S, they may divert it to flare or use it elsewhere.

ScrubbingHigh levels of H2S are often reduced with scrubbing systems and gas analysers can measure levels before and after scrubbing. This monitors scrubber performance as well as avoiding high levels of corrosive H2S reaching and damaging CHPs.

Good adviceGeotech advises that all landfill gas analysers must be used and managed properly, including appropriate purging, to give operators the reliable gas readings essential for gas field management. As H2S levels rise, so does the need for accurate, dependable landfill gas analysis. Geotech works closely with landfill operators and other stakeholders to ensure equipment is in-line with changing conditions as well as being reliable and well-supported.

Related news

Hydrogen sulfide, H2S, is a very poisonous, corrosive, and flammable gas. As such, it needs to be carefully monitored and controlled in industrial processes where it is generated or used. At relatively low concentrations, it has an odour of rotten eggs which effectively works as a warning, but at higher concentrations where it becomes acutely toxic and even lethal, it can no longer be smelled and the awareness of danger disappears.
It is important to be able to monitor concentrations of hydrogen sulfide, both...

As focus around the globe is increasingly shifting towards renewable energy, anaerobic digestion (AD) has become an economically viable energy source that capitalizes on reusing our wasted materials. It is a relatively easy and natural means of turning a broad variety of complex waste into a simple fuel gas.It’s no surprise therefore, that the number of AD plants across the UK has risen in recent years. These plants have a significant role to play in helping the country meet the Government’s target of...

Following the update of health recommendations and the management measures of the High Council of Public Health (HCSP, France), the Regional Health Agency and GWAD’AIR (Approved Association for the Monitoring of Air Quality) have deployed a network to measure hydrogen sulfide (H2S) and ammonia (NH3). This network aims to continuously measure the emanation of these gases from the decomposition of Sargassum algae.
The monitoring system currently in place will be expanded starting in August with the addition...

Following the update of health recommendations and the management measures of the High Council of Public Health (HCSP, France), the Regional Health Agency and GWAD’AIR (Approved Association for the Monitoring of Air Quality) have deployed a network to measure hydrogen sulfide (H2S) and ammonia (NH3). This network aims to continuously measure the emanation of these gases from the decomposition of Sargassum algae.
The monitoring system currently in place will be expanded starting in August with the addition...

New rules for flare stack combustion from the US Environmental Protection Agency (EPA) set to take effect in early 2019 include additional monitoring and analysis requirements that make it necessary for refinery operators to quickly and accurately determine the heat values of flare stack gases.
Determining the heating values of vent gas is important because sufficient combustible material must be continuously present to achieve high combustion efficiencies. To assist refiners in making that determination, AMETEK...

Customer comments

No comments were found for Landfill gas H2S increase measured with high-range analysers. Be the first to comment!

Add your comment

Great! comment successfully added!

Contact supplier

Your message:

Your email

Your first name

Your last name

I would like to receive periodic email updates and special offers from select suppliers.