Rocking Stage

About

The Rocking Stage adds an additional degree of freedom to your SEM's or FIB/SEM's sample stage. This axis is a rotation that is perpendicular to the microscope stage's rotation axis.

Advantages

The most important use case for the Rocking Stage is to minimize curtaining effects during cross-sectional FIB-milling. Of course, the stage can also be used for any other task that requires flipping a sample on its side.

Specifications

Travel: 360° unlimited

Resolution: 0.1°

Torque: 0.01 mNm

Temperature range: 273 K to 353 K

Lowest pressure: 10-7 mbar

All technical specifications are approximate. Due to continuous development, we reserve the right to change specifications without notice.