Model 6510e/6530 Power Analyzers

The Magtrol 6510e Single-Phase and 6530 Three-Phase Power Analyzers are
easy-to-use instruments ideal for numerous power measurement
applications. From DC to 100 kHz AC, the 6510e and 6530 measure volts, amps, watts, volt-amps, frequency,
crest factor, Vpeak, Apeak and power factor in one convenient
display. They may be used either as stand-alone instruments
or in conjunction with any Magtrol Hysteresis, Eddy-Current
or Powder Brake Dynamometer;
any Magtrol Dynamometer
Controller and M-TEST Software for more demanding motor test applications.

Features

Single/Three-Phase
Capabilities: Equipped to handle
single (6510e)
or three-phase (6530) power measurements

The Analog Output is a plug-in module designed specifically
for use with the 6510eSingle-Phase Power Analyzer and 6530 Three-Phase Power Analyzer.
It provides up to 12 channels (3 for the 6510e,
12 for the 6530) of analog output corresponding to volts, amps
and watts. Each output is capable of, and calibrated to, ± 10
volts. With the analog output option, the user may apply a scale
factor to all outputs. Amps scaling is applied to all amp channels,
volts scaling is applied to all volt channels and watts scaling
is applied to all watt channels.

The Analog Output can be used along with a 6510e or 6530
Power Analyzer to output information to a strip recorder or
interface with a data acquisition system. The Analog Output
is also used as a safety feature to help trigger alarm events.

The HA Series shunt, designed for power cable hook-up utilizing 3/8-16 screws, has an insulating base and can be mounted on conductive surfaces. The LAB Series shunt also comes with an insulating base, along with thumb screws for the power leads and knurled nuts on the sensing terminals. The FL Series is a relatively small bus, bar-mounted shunt with large end blocks. The comparatively large end blocks and short spacing aid in cooling and allow the shunt to operate in more extreme environments. All shunts are calibrated on equipment with current certifications traceable to N.I.S.T.

Specifications

HA Series

LAB Series

FL Series

Rated Current

50 A to 500 A

750 A to 1000 A

2000 A to 3000 A

Output

50 mV

Bandwidth

DC to 60 Hz

Accuracy

± 0.25% (± 0.1% accuracy with calibration certificate
is available for an additional cost)

Operating Temperature

For optimum accuracy, temperature of shunt
should be 30 °C to 70 °C

Available Models

Type

P/N

Amps

HA

004640

50

HA

004641

100

HA

004642

150

HA

004643

200

HA

004644

250

HA

004645

300

HA

004646

400

HA

004647

500

LAB

004648

750

LAB

004649

1000

FL

005214

2000

FL

005286

3000

Sensor Substitution Wiring Connection

NOTE: For the 6530 Three-Phase Analyzer, any (or all) of the Amps inputs may be replaced with a shunt for any of the wiring modes.