From design to production, the Huvitz name is one you can trust. Now, with the addition of the HS-5000 2X, you can enjoy a quality slit lamp featuring world-class optics at an outstanding price. This Haag-Streit Style Tower Illumination system features a Galilean Converging Binocular Optical System and a two-position, drum-style magnification changer that offers great field of view and a crisp clear image every time. Every movement with the HS-5000 2X is designed to be smooth and precise. Plus, an omnidirectional joystick with trigger button and five built-in filters are among the many “extras” you’ll enjoy when selecting the quality designed slit lamp. The HS-5000 2x is a smart choice for the budget-conscious practice.

Design
The HS-5000 2X features a Haag-Streit Style Tower Illumination System that sits above the observation system, allowing the illumination column to be decoupled in the vertical meridian. This allows greater flexibility when minimizing unwanted reflections during gonioscopy and increases the peripheral field of view during indirect ophthalmoscopy.

Microscope
HS-5000 2X features a Galilean Converging Binocular optical system that delivers crisp images and a wide field of view. Operators can select from total magnification ratios equivalent to 10x (∅22.3mm) and 16x (∅15.2mm).

Magnification Control System
The uniquely designed Huvitz optic system eliminates image distortion at all magnification levels, enabling you to achieve a more precise diagnosis. The unit’s two-position drum-style magnification changer provides easy access to magnification from 10x to 16x with the 12.5x eyepiece

Five Filters, Including Yellow
Five built-in Illumination Pathway Filters can be effortlessly inserted into the illumination pathway using the selection lever. Options include a heat absorption filter that aids in patient comfort during lengthy exams, a red-free filter that enhances structural contrast, a cobalt blue filter for use in fluorescein assessment, a grey filter, and a yellow filter to absorb blue light and enhance contrast.

Integrated Control
The omnidirectional style joystick is simple to control. An integrated trigger button allows for convenient image and video capture. When the slit lamp is connected to imaging devices, images and videos can be stored simultaneously.

Ergonomicaly Designed Illumination Control
The illumination lever is conveniently located next to the joystick. The brightness control switch can be set to High or Normal, making diagnostic procedures even easier.

High Quality Standards
From design to production, the Huvitz name is one you can trust.

Durability is Assured
All Huvitz slit lamps are made to be durable, even in extreme conditions. Joints and moving parts undergo tens of thousands of repeated operations to ensure long lifetime use.

Smooth Precision Movement
The easy operation of the joystick and knobs factor into making diagnostic procedures even easier. Every movement with the HS-5000 2X is designed to be smooth and precise.

International Quality Assurance
All Huvitz slit lamps are certified by the international quality assurance system.

Slit Illumination

Slit Length(mm)

0.3~12

Slit Width (mm)

0~12 continuous

Slit Projection

1x

Aperture Diaphragms

0.02 / 0.04 / 0.11 / 0.19 / 0.4 / 0.47in (0.3 / 1 / 3 / 5 / 9 / 12mm)

Filters

Cobalt Blue, Red-Free, Gray, Heat absorption and Yellow

Slit Rotation

0°~180°continuous

Patient's Eye / Prism

Angle of Incidence

0°, 5°, 10°, 15°, 20°

Surface Working Distance

3.14in (80mm)

Microscope

Type

Galilean Converging Binocular

Magnification

2 Position Rotating Drum

Eyepieces

12.5x

Total Magnifications

10x, 16x

Real Fields of View (mm)

38.5,22.2, 15.2, 10.5, 6.7

Inter-Pupillary Adjustment

2.16in~3.14in (55mm~80mm)

Base

Vertical Movement

1.10in (28mm)

Longitudinal Movement

3.07in (78mm)

Lateral Movement

3.85in (98mm)

Fine Base

0.4in (10mm)

Voltage Frequency

50/60Hz

Power Consumption

70VA

Instrument Voltage

12V DC

Halogen Bulb

12V 30W

Fixation Point Bulb

6V 0.2A

OPTION: PC, Table Designs and details can be changed without prior notice for the purposes of improvement

From design to production, the Huvitz name is one you can trust. Now, with the addition of the HS-5000 2X, you can enjoy a quality slit lamp featuring world-class optics at an outstanding price. This Haag-Streit Style Tower Illumination system features a Galilean Converging Binocular Optical System and a two-position, drum-style magnification changer that offers great field of view and a crisp clear image every time. Every movement with the HS-5000 2X is designed to be smooth and precise. Plus, an omnidirectional joystick with trigger button and five built-in filters are among the many “extras” you’ll enjoy when selecting the quality designed slit lamp. The HS-5000 2x is a smart choice for the budget-conscious practice.

Design
The HS-5000 2X features a Haag-Streit Style Tower Illumination System that sits above the observation system, allowing the illumination column to be decoupled in the vertical meridian. This allows greater flexibility when minimizing unwanted reflections during gonioscopy and increases the peripheral field of view during indirect ophthalmoscopy.

Microscope
HS-5000 2X features a Galilean Converging Binocular optical system that delivers crisp images and a wide field of view. Operators can select from total magnification ratios equivalent to 10x (∅22.3mm) and 16x (∅15.2mm).

Magnification Control System
The uniquely designed Huvitz optic system eliminates image distortion at all magnification levels, enabling you to achieve a more precise diagnosis. The unit’s two-position drum-style magnification changer provides easy access to magnification from 10x to 16x with the 12.5x eyepiece

Five Filters, Including Yellow
Five built-in Illumination Pathway Filters can be effortlessly inserted into the illumination pathway using the selection lever. Options include a heat absorption filter that aids in patient comfort during lengthy exams, a red-free filter that enhances structural contrast, a cobalt blue filter for use in fluorescein assessment, a grey filter, and a yellow filter to absorb blue light and enhance contrast.

Integrated Control
The omnidirectional style joystick is simple to control. An integrated trigger button allows for convenient image and video capture. When the slit lamp is connected to imaging devices, images and videos can be stored simultaneously.

Ergonomicaly Designed Illumination Control
The illumination lever is conveniently located next to the joystick. The brightness control switch can be set to High or Normal, making diagnostic procedures even easier.

High Quality Standards
From design to production, the Huvitz name is one you can trust.

Durability is Assured
All Huvitz slit lamps are made to be durable, even in extreme conditions. Joints and moving parts undergo tens of thousands of repeated operations to ensure long lifetime use.

Smooth Precision Movement
The easy operation of the joystick and knobs factor into making diagnostic procedures even easier. Every movement with the HS-5000 2X is designed to be smooth and precise.

International Quality Assurance
All Huvitz slit lamps are certified by the international quality assurance system.