We use cookies on this website to improve your shopping experience. We use cookies to remember log-in details and provide secure log-in, collect statistics to optimize site functionality, and deliver content tailored to your interests.Click accept to give your consent to accept cookies and go directly to the site or click on more information to see detailed descriptions of the types of cookies we store.
Click Here To Learn About Cookie Settings...

Part Number: MSR145

Measures Temperature, Humidity, Light, Pressure and Acceleration Simultaneously and 2 or 4 Analog Inputs Also Available

Stores up to 2 Million (Internal Memory) or 1 Billion (SD Card) Total Readings

Rechargeable 900 mAh Lithium Polymer Battery

Data from any number of loggers can be merged synchronously into a single Data Record

Accelerometer Range of ±15g with -20°C to 65°C Temperature Range

Made in Switzerland, Swiss Made

Includes Software CD, USB Cable and 4GB microSD Card

Click on the "Configurator" Tab Below to Choose the Ordering Options

MSR Data Loggers are Built to Order and Not Eligible for Return. All Sales Are Final.

Special order - please allow approx. 2 to 3 weeks for this item to ship.

Configurator

MSR145: The revolutionary mini Data Logger

This thumb-sized Swiss Quality MSR145 data logger records over 2 000 000 measured values - with an optional microSD card, even over a billion measured values - and is ideal for long-term measurements. It is capable of simultaneously measuring and recording a variety of measures such as temperature, humidity, pressure, 3-axis acceleration and light. All measured values can be rapidly transferred to a PC or laptop via the USB interface or microSD card for data analysis purposes.

Versatile universal data logger

You have the choice between different casings, battery capacities and sensors. Additionally, you can equip your MSR logger with analogue inputs (selectable input configuration). The analogue inputs allow you to connect individual, external sensors for different measuring tasks (e.g., CO2, conductivity, pH, etc.). In addition to your MSR145 data logger you will also receive the user-friendly software for data acquisition and analysis free of charge.

With a few clicks you are able to select and send us your order or request a no-obligation quotation of your desired MSR145.

2. Select the sensors.

Please select the sensors for your data logger here.
For the measured parameters temperature, humidity and fluid pressure you can now choose whether you wish to provide your MSR145 data logger with more than one sensor of the same measured parameter.
Please note the following restrictions for this option:

Only one measured parameter can be equipped with multiple sensors.

The slot for a microSD card cannot be combined with this option.

The light sensor (L2), the temperature sensor (FT) and the port for a thermal element of Type K (T3, T4) cannot be selected.

Can your desired sensor combination not be selected? Contact us for a customized fit.

4. Would you like to expand the memory capacity of your MSR145 to over 1 billion measured values?

Please order the expansion slot for plugging-in a standard microSD card here.

Item no.

Expansion slot for microSD card (available only for design housing B7 and B8)

Please note that:

this expansion slot with plugged-in microSD card only meets the protection category IP60,even with the design housing B8,

the expansion slot is not available in combination with light or thermal element,

the expansion slot is not available in combination with a multiple selection of sensors of the same measurand,

one formatted and ready for use microSD card (≥4GB) is included.

N10006Error

5. General technical data on the MSR145.

Memory capacity:

Over 2 000 000 measured values – expand­able to over 1bn.

Key:

Set bookmark or start and stop recording.

LED:

blue:
red:
yellow:

Record indicator
Alarm indicator
Battery charge indicator

Power supply:

Lithium-polymer battery, 230 mAh or 900 mAh.

The battery is charged via the USB connection.

One battery charge is sufficient for operation of the unit for up to two years, depending on the battery and the measurement rate.

Options:

Charging station (USB hub) for 7 MSR145 (item number B47002)

Interface:

USB

PC software:

Free Setup, Reader, Viewer and Online software (Windows XP/Vista/7/8/10) to record and evaluate data. All measurement parameters can be rapidly transferred to a PC via USB connection both during the measurement operation and once the measurement task has been completed. Thanks to the integrated clock (RTC), data from as many MSR145 units as required can be synchronised and merged into a single data record.

Leave a Reply

Ask Your Question

Name*

Email*

Question*

*Please type the letters below

* Required Fields

An authorized stocking distributor of over 30 manufacturers of data logging, data acquisition and weather station products. We address the data logging and recording needs of the transportation, agricultural, pharmaceutical, HVAC/R, environmental, educational, scientific and industrial communities.

video title

When you visit any website, it may store or retrieve information on your browser, usually in the form of cookies. This information does not usually identify you, but it does help companies to learn how their users are interacting with the site.

We respect your right to privacy, so you can choose not to accept some of these cookies. Choose from the different category headers to find out more and change your default settings.

Strictly Necessary Cookies Always Active

These cookies are essential for the website to function and they cannot be turned off. They are usually only set in response to actions made by you on the MicroDAQ.com website: Logging In, Adding Items to Your Cart or Filling in Forms. If you browse our website, you accept these cookies.

Cookies are Used By

Website and Store

Performance Cookies

These cookies allow us to keep track of how many people have visited our website, how they discovered us, and how they interact with the site. All the information used is aggregated, and completely anonymous. If you do not allow these cookies we won’t know you have visited us.

Cookies are Used By

Google Analytics

Google Remarketing

Google Adwords

Bing

Facebook Pixel

Targeting Cookies

These cookies are placed on our site by our trusted third-party providers. They help us to personalise our adverts and provide services to our customers such as live chat.

Cookies are Used By

Live Chat

Google/Double Click

Functional Cookies

These cookies allow the provision of enhance functionality and personalization, such as videos and live chats. They may be set by us or by third party providers whose services we have added to our pages. If you do not allow these cookies, then some or all of these functionalities may not function properly.