We use cookies to remember your preference and provide you with the very best experience on our website. By using our website, you consent to our cookie policy. Find out more by reading our privacy policy.

In observation of the Christmas holiday, APS will be closed on Monday, December 24th and will resume normal operations on Wednesday, December 26th.

DESCRIPTION

Probe

Batteries

2 AA Alkaline; Battery Life of approximately 200 hours Battery manufacturer’s battery operating temperature range -4 to 130°F, -20 to 54°C Note: Battery may not provide adequate power if ambient temperature is extremely low or high.

Enclosure Material

Polypropylene Blend, Stainless Steel and Aluminum.

Temperature

Resolution 0.01

Range

-40 to +400°F
-40 to +204°C

Calibrated Accuracy

±0.2°F from 32 to 200°F
±0.5°F from 200 to 400°F
±0.1°C from 0 to 100°C
±0.3°C from 100 to 200°C
Long-term drift not to exceed 0.05%/year Meets API requirements

The TP7-D employs the proven RTD design that has been used in the TP 7 and TP 8 for many years. A sealed industrial quality overlay provides a user interface that is easy to use with gloves. A polypropylene blend enclosure protects the circuit board and large LCD from penetration by impact, water and reactive liquids. The new TP7-D circuit board is an evolutionary step up from our highly accurate, reliable and successful TL1 laboratory thermometer. The Power Button’s primary function powers the instrument for intervals of about 20 minutes since the last button was accessed. The Power Button can also be used to conserve power and clear the memory, or to make adjustment while in calibration mode. A simple menu operation is displayed by holding the Function Button, and alternately functions to allow adjustments in the calibration mode. Arrows on the left side of the display show the direction of the temperature reading and whether stability has been reached. At the user’s discretion stabilized temperatures can be logged at numerous liquid levels for a running average and later displayed for the user’s documenting purposes. But this feature never interferes with simply getting an accurate temperature reading. To endure the environment and be intrinsically safe, the TP7-D is manufactured of materials, which are both immune to petrochemicals and are non-sparking. The enclosure is made of a polypropylene blend. The probe assembly is constructed using non-stick cable and stainless steel sensor components.

Filter

Deadman Control Systems

JOIN OUR NEWSLETTER

Sign up for latest industry news and exclusive offers delivered to your inbox.

Please enter a message

American Petroleum Services, LLC is a distributor/supplier of aviation refueling equipment. We handle quality control equipment that must test for jet fuel, which includes fuel receipts from the barges and pipelines.