The Multilogger is a battery operated, hand-held datalogger capable of measuring and recording up to 4 inputs. The measured values can be viewed on a web browser, as well as stored and downloaded for later analysis. Features large non-volatile memory, wide adjustable logging rate, alarms and easy-to-read LCD display and keypad for local monitoring and control.

The M1300 variant can accept up to 4 voltage or 4-20mA inputs, 2 of which can be a digital inputs and 1 input can count pulses.

It is possible to set two independent alarm limits for each channel (ie. measured or calculated value) which can be configured either as an upper and lower limit or who limits exceeding in a consistent direction. Alarm signalization can be acoustic (built-in beeper), optical (3 LEDs), alarm output or sending an e-mail alert.

Email

Warning email are sent when measured value exceed selected limits. Emails are also send when values returns back into safe range. SMTP authentication is supported, but SSL not. Domain name for SMTP server address is supported.

XML

XML protocol for actual measured values reading. This protocol is suitable for device integration into 3rd party SCADA systems.

SNMP

SNMP version 1 protocol for IT infrastructure. Using SNMP protocol you can read actual measured values, alarm status and alarm parameters. MIB tables with OID description are available.

MIN/MAX memory

Memory for minimum and maximum values. Memory is independent on values in history memory. Minimum and maximum values can be cleared according user requirements.

Vision Software

The Vision Software is designed for communication with Multiloggers, read out the configuration of the device and setting up the parameters.It also allows to read out the data from the device in order to get the measured values.

Data can be viewed and printed as table or graph. It it also possible to export the data to DBF or TXT files. You can change the axis and select any time range to view or print out the data. It is allowed to zoom the graph, modify the drawing width and the its colour. It is also possible to separate the quantities from each other and print some of them or all of them separately.