Isotech ADVANCED Hyperion

The Hyperion and Drago have large calibration volumes,65mm x 160mm deep, which makes them ideal to use as portable liquid baths. Stirred liquid baths are suitable for temperature sensors of all types, sizes and shapes.

Liquid Baths can provide smaller calibration uncertainties than dry blocks and, when used with suitable reference thermometers,accuracies of up to 0.005°C can be achieved.

These models are part of the award winning Isocal-6 family and with a reference probe can be used with different accessories for Dry Block, Infrared, Surface Calibration and even with ITS-90 Fixed Point Cells for uncertainties to 0.001°C. In Dry Block Mode the large 65mm diameter block allows for the calibration of either larger probes or for calibrating many sensors simultaneously.

As a Liquid bath the sensors can be placed directly into the stirred liquid thus avoiding the need for specially drilled blocks. If the liquid is directly in the block then the controller only model, or Basic (B) model, can be selected. However, instead of putting liquids directly in the block liquid containers can be used to facilitate rapid change of fluids. For greater accuracy, or when using a Liquid Container, Dry Block Insert, Blackbody Target or the Surface Sensor Kit a separate reference thermometer should be used to compensate for the varying offset between the controller and the accessory temperature.

I-Cal EASY calibration software can be added to create apowerful solution for the automatic calibration of thermometers.

Temperature Range

-25°C to 140°C 1

Stability

Liquid Bath: ±0.005°C Dry Block: ±0.005°C

Resolution

0.001°C Over Full Range

Display Accuracy 3

0.15°C

Uniformity - Radial, Liquid Bath Mode

<0.009°C

Uniformity - Axial, Liquid Bath Mode (40mm)

<0.011°C

Uniformity - Radial, Dry Block Mode (between wells)

<0.008°C

Uniformity - Axial, Dry Block Mode (40mm)

<0.40°C

Heating Time

-20°C to 140°C: 40 mins

Cooling Time

140°C to 20°C: 90 mins, 20°C to -25°C: 80 mins

Calibration Volume

65 x 160mm

Standard Insert

8 x 8mm pockets and 2 x 4.5mm, all 157mm deep

Input Channels

Full range of inputs and accuracies with calibration options available online